If the signal does exist, but sometimes the oscilloscope can catch it, sometimes it cannot, it may have something to do with the settings of the oscilloscope. Usually, if you can set the trigger mode of the oscilloscope to Normal, set the trigger condition to edge trigger, adjust the trigger level to an appropriate value, and then set the sweep mode to single mode, if this method does not work, usually the instrument may fail problem.