The output sensitivity of the load cell is based on its rated capacity so with a 10Vdc excitation applied you would get an output of 2mV * 10V = 20mV with the cell’s full rated capacity applied. This then varies linearly with the applied load so, if we assume a 50N rated capacity as an example, with 25N applied your output would be 10mV. If your system requires a higher-level output signal then we can provide a signal conditioner to give signals such as 0-5Vdc, 0-10Vdc or 4-20mA, as well as bi-polar voltage signals including ±5Vdc and ±10Vdc. Here are some links to suitable signal conditioners: https://appmeas.co.uk/products/instrumentation/load-cell-amplifier-sga/ https://appmeas.co.uk/products/instrumentation/miniature-load-cell-amplifier-ica/