The standard WIKA measuring insert allows use up to 3g (amplitude). This corresponds to a loading of 6g, peak to peak, per DIN EN 60751 (58.86 m/s^2). In EN 60751 only 20-30 m/s^2 peak-to-peak is specified (1 g = 9.81 m/s^2). The vibration-resistant design is suitable for up to 20g peak-to-peak. Special designs, on request, up to 50g are possible. (The values given above always apply to the vibration load directly at the measuring resistor.)