Horizontal tank: choose a position inside the tank bottom support and far away from the liquid inlet Installation of Ultrasonic Tank Level Sensor in Horizontal Tank Spherical tank: For new tanks, if the manhole flange is not planned to be opened in recent years, and there is few liquid deposits, it is best to install it on the manhole flange. Because of the surface level, the signal is strongest. Installation of Ultrasonic Tank Level Sensor in spherical tank When opening the inlet flange, remove the liquid level sensor first. After the inlet flange is installed and fixed, the liquid level sensor can be reinstalled. If the deposits are not cleaned for a long time, a large amount of deposits may accumulate on the inner surface of the flange. In this case, the sensor installation position that can work stably for a long time is a circle around the manhole flange. Here, the level is small, the sediment is small, and there is no rapid liquid flow interference. Although the signal for the new tank without sinking is not strong on the flange, it works stably for a long time. Vertical tank: Generally, choose the installation groove far from the liquid inlet and outlet (more than 90°). Installation of Ultrasonic Tank Level Sensor in Vertical TankInstallation of Ultrasonic Tank Level Sensor in Vertical Tank with a bypass pipe