A weighing system is all the more precise when the weighed structure is free of friction. When there are pipes, before anchoring the pipe to the weighed structure, check that it is close to and in line with the nozzle to which it will be attached.
Friction can be limited by using flexible hoses and elastic joints or free inlets with rubber protection, such as of the bellows type.
Alternatively, position the first anchor bracket in the horizontal section as far away as possible from the structure being weighed, at least 40 times the diameter of the pipe.
Share this FAQ