Theoretically, you can engrave an infinite number of FBGs on a single fiber, as long as you keep different Bragg wavelengths on different FBGs. But practically speaking, the wavelength band is limited due to the limitation of the broadband light source and the receiving and detecting module of the fiber grating interrogator. At the same time, since the Bragg wavelength changes with the influence of temperature and strain. Therefore, it must be ensured that the gap distance between the Bragg wavelengths of the fiber grating does not overlap. In this way, the interrogator can distinguish different fiber gratings according to different reflection wavelengths. Another characteristic that affects the number of fiber gratings is the power available to receive reflections from the Bragg gratings. This depends on the transmit power of the interrogator, the losses along the way (including bend loss, connector loss, splice loss, fiber length loss), and the reflectivity of the Bragg grating. Therefore, there are many situations that will affect the number of FBGs. Generally, in the commonly used C-band, it is recommended to use no more than 13 or 14 fiber gratings on a single fiber grating sensor.