The measured value registration is made through the liquid-filled measuring system which consists of a temperature probe, capillary and Bourdon tube. All three components combine to form a closed tube system. The internal pressure in the system changes with the adjacent temperature. This causes the pointer axis connected to the spring to turn and the temperature value to be displayed on the scale. The capillary, with lengths between 500 and 10,000 mm also enables measurements to be made on remote measuring points. The scale ranges are between -40 and +400 °C with accuracy classes 1 and 2 in accordance with EN 13190.