<b><br>Specification: </b> <br><br>E--G: <br>Lead Mode:5 Leads ( R, L, F, N, C or RA, LA, LL, RL,V) <br>Lead selection:I, II, III, avR, avL, avF, V, <br>Waveform:2 ch <br>Lead mode:3 Leads ( R, L, F or RA, LA, LL) <br>Lead selection:I, II, III, <br>Waveform:1 ch <br>Gain:´2.5mm/mV, ´5.0mm/mV, ´10mm/mV, ´20mm/mV, auto <br>HR and Alarm <br>Range: <br>Adult:15 ~ 300 bpm <br>Neo/Ped:15 ~ 350 bpm <br>Accuracy:± 1% or ± 1bpm,which great <br>Resolution:1 bpm <br>Sensitivity:> 200 (uV P-P) <br>Differential Input Impedance:> 5 MΩ<br> <b><br>RESPARATION: <br> <br>Method:Impedance between R-F(RA-LL) <br>Differential Input Impedance:>2.5 MΩ <br>Measuring Impedance Range:0.3~5.0Ω <br>Base line Impedance Range:0 – 2.5 KΩ <br>Bandwidth:0.3 ~ 2.5 Hz <br>R-esp. Rate <br>Measuring and Alarm Range <br>Adult:0 ~ 120 rpm <br>Neo/Ped:0 ~ 150 rpm <br>Resolution:1 rpm <br>Accuracy:±2 rpm <br>Apean Alarm:10 ~ 40 S<br> <b><br> Pulse Rate: <br> <br>Measuring and Alarm Range:0~254bpm <br>Resolution:1bpm <br>Accuracy:±2bpm <br>I.5TEMP measurement: <br>Channel:1 <br>Measuring and Alarm Range:0 ~ 50 °C <br>Resolution:0.1°C <br>Accuracy:±0.1°C <br>Actualization interval: about 1 Sec. <br>Average Time Constant: < 10 Sec. <b><br>Sp(0)2: <br> <br>Measuring Range:0 ~ 100 % <br>Alarm Range:0 ~ 100 % <br>Resolution:1 % <br>Accuracy: 70% ~ 100% ±2 %; 0% ~ 69% unspecified <br>Actualization interval:about 1 Sec. <br>Alarm Delay:10 Sec. <br> <b><br>About N(1)BP: <br> <br>Method:Oscillometric <br>Mode:Manual, Auto, STAT <br>Measuring Interval in AUTO Mode: 1, 2, 3, 4, 5, 10, 15, 30, 60, 90, 120, 180, 240,480 (Min) <br>Measuring Period in STAT Mode:5 Min <br>Pulse Rate Range:40 ~ 240 bpm <br>Alarm Type:SYS, DIA, MEAN<br>