<b>Features:</b><br/> Measure AC/DC current (minimum 1mA by jaws)<br/> Measure microamps μA (minumum 0.1μA by terminals)<br/> Measure AC/DC voltage<br/> V-alert measurement, live-wire check<br/> Measure frequency, temperature, resistance, continuity and capacitance<br/> Low impedance voltage measurement<br/> Clamp on frequency function<br/> Auto range, Auto power off, data hold, low battery indication with backlight<br/> Conform to the IEC-61010-1, IEC-61010-2-030, IEC-61010-2-032, IEC-61010-2-033 standards<br/> Over-voltage protection class CAT III 300V, CAT II 600V, pollution degree II<br/><br/> <b>Specifications:</b><br/> Display: 6000 counts<br/> DC current: 6A/60A/100A 0.001A/0.01A/0.1A ±(3.0%+5counts)<br/> DC current(μA): 0-200μA 0.1μA ±(0.8%+5counts)<br/> AC current: 6A/60A/100A 0.001A/0.01A/0.1A ±(2.5%+5counts)<br/> DC voltage: 6V/60V/600V 0.001V/0.01V/0.1V ±(1.0%+3counts)<br/> AC voltage: 6V/60V/600V 0.001V/0.01V/0.1V ±(1.0%+3counts)<br/> Low impedance LOZ: Max 600V Resolution 0.1V<br/> Resistance: <br/> 600Ω/6kΩ/60kΩ/600kΩ/6MΩ 0.1Ω/0.001KΩ/0.01KΩ/0.1KΩ/0.001MΩ ±(1.0%+3counts)<br/> 60MΩ 0.01MΩ ±(1.2%+20counts)<br/> Capacitance: 600µF/6000µF 0.1µF/1µF ±(4.0%+3counts)<br/> Frequency: 60Hz/1000Hz 0.1Hz/1Hz ±(1.0%+5counts)<br/> Frequency(by jaws): 60Hz/1000Hz 0.1Hz/1Hz ±(1.0%+5counts)<br/> Buzzer: If the resistance is less than 30Ω, the continuity beeper sounds<br/> Temperature: -20℃~500℃ 1℃/2℉ ±(2.0%+2counts)<br/> Jaw Capacity: 20mm<br/> Power: 1.5V (AAA) x 2 (not included)<br/> Size: 179 x 67 x 36 mm<br/> Weight: 137g<br/><br/> <b>Package Includes:</b><br/> 1 x Digital Clamp Meter<br/> 1 x Probes<br/> 1 x Thermocouple<br/> 1 x Carrying Case<br/> 1 x Screwdriver<br/> 1 x User Manual<br/><br/> <b>Note:</b><br/> The batteries are not included in the package due to Aviation Security Check