<b>Specifications  </b><br> Safety/Compliances: IEC61010 CAT. II 600V over voltage and double insulation<br> Display: 2.2 inch LCD display; Maximum display 1999<br> Range selection: Auto ranging<br> Overload protection: resistance measurement with PTC to protect circuit<br> Dimensions: 147mm*71mm*45mm (L*W*H)<br> Battery test: ✔<br> AC/DC Current test: ✔<br> AC/DC Voltage test: ✔<br> Resistance test: ✔<br> Diode test: ✔<br> Continuity test: ✔<br> NCV: ✔<br> Data Hold: ✔<br> LCD Backlight: ✔<br> Flashlight: ✔<br> Auto Power 0ff: ✔<br> Continuity Buzzer: ✔<br><br> <b>Battery Test:</b><br> - 1.5V Battery: Resolution: 0.001V; Built-in load: 1.8KΩ<br> - 6V Battery: Resolution: 0.01V; Built-in load: 150Ω<br> - 12V Battery: Resolution: 0.01V; Built-in load: 62Ω<br><br> <b>DC Voltage:</b><br> - Range and Accuracy: 200mV ± (0.5%+2); 2/20/200V ± (0.5%+2); 600V ± (0.8%+2)<br> - Input Impedance: 10 MΩ<br> - Overload Protection: 600V DC (RMS)<br> <b>AC Voltage (50Hz - 400Hz):</b><br> - Range and Accuracy: 2/20/200V ± (1%+3), 600V± (1.2%+3)<br> - Input Impedance: 10 MΩ<br> - Response impedance: Average (sine wave RMS)<br> - Overload Protection: 600V AC (RMS)<br> <b>Resistance:</b><br> - Range and Accuracy: 200Ω ± (0.8%+2); 2/20/200KΩ ± (0.8%+2); 2MΩ ± (0.8%+2), 20MΩ ± (1%+2)<br><br><br> <b>DC Current :</b><br> - Range and Accuracy: 200/2000μA ± (1.5%+3); 20/200mA ± (1.5%+3), 10A ± (2.0%+5)<br><br> <b>AC Current (50Hz - 400Hz):</b><br> - Range and Accuracy: 200/2000μA ± (1.5%+4); 20/200mA ± (1.5%+4), 10A ± (2.5%+5)<br> - Overload Protection: μA/mA Range: F250mA (Fast blows), 10A Range: F10A Fuse (Fast blows).<br> - Maximum Input Current: mA jack: 250mA, 10A jack: 10A<br><br> <b>Package Includes</b><br> 1x Tacklife DM10 Digital Multimeter<br> 1x Screwdriver<br> 3x 1.5V AAA Batteries<br> 1x User's Manual <br> 1x Test Leads with Alligator clip (Pair)<br> 1x 24-month Service Card<br>