NEW UPGRADED Vector Network Antenna Analyzer 50KHz-900MHz MF HF VHF UHF with Shell<br/><br/><b>Specification:</b> <br/>- PCB: 54mm x 85.5mm x 11mm (without the size of connectors, switches)<br/>- Measurement Frequency: 50KHz ~ 300MHz (50KHz -900MHz, enable extended firmware)<br/>- RF Output: -13dbm (maximum -9dbm)<br/>- Frequency Accuracy: ï¼œ0.5ppm<br/>- Measurement Range: 70dB (50kHz-300MHz), 50dB (300M-600MHz), 40dB (600M-900MHz) enable extended firmware)<br/>- Port SWR: < 1.1<br/>- Display: 2.8 inch TFT (320 x240)<br/>- USB Interface: USB Type-C; Communications Mode: CDC (Serial)<br/>- Power Supply: USB 5V 120mA, built-in 450mAh electricity, maximum charging current 0.8A<br/>- Number of Calibration Points: 101 (Fixed)<br/>- Number of Scanning Points: 101 (Fixed)<br/>- Display Tracking: 4, Marking: 4, Save Setting : 5<br/>- Measuring S parameters, voltage standing wave ratio, phase, delay, Smith chart and the like<br/><br/><br/><b>Package Included:</b><br/><br/>1X NanoVNA Host (2.8-inch touch screen and 450mAh battery are included)<br/>2X 20cm SMA Male to Male RG174 RF Cable <br/>2X SMA Female to Female Connector <br/>1X SMA Simple Calibration Kit SHORT <br/>1X SMA Simple Calibration kit OPEN <br/>1X SMA Simple Calibration kit LOAD <br/>1X USB Type-C Data Cable <br/>1X Type-c to Type-c Line<br/>1X Touch Paddle   <br/>