Intended Use:<br> 1.Use with metene TD-4116 blood glucose monitor to quantitatively measure glucose in fresh capillary whole blood from the finger, palm, forearm, and upper arm. 2. Intended for single-patient use (lay-users at home) and should not be shared. It should not be used for the diagnosis of or screening for diabetes, not for use on neonates.<br> Limitation:<br> 1.Hematocrit: limited to 20% ~ 60%.<br> 2.Do not test blood glucose during or soon after the xylose absorption test. Xylose in the blood will elevate the results.<br> 3.The device is not intended for use in health care or auxiliary Settings, such as hospitals, physician offices or long-term care facilities.<br> Storage and Use:<br> 1.Expiration: 6 months after first opening.<br> 2.Storage conditions: 39.2℉ ~ 104℉, 10% ~ 85% R.H, keep dry and cold.<br> 3.Make sure the surface of strip is free from moisture or other contamination and avoid touching the blood sampling area of the strip.<br> 4.Close the bottle after taking out the strip. It is best to use the strip within 3 minutes after taking it out.<br> Read Your Result:<br> 1.Your blood glucose readings deliver plasma equivalent results and are displayed in milligrams of glucose per deciliter of blood (mg/ml) or millimoles per liter of blood (mmol/L).<br> 2.Normal plasma glucose range for people without diabetes:<br> Fasting and before meals: ≤100 mg/ml (5.6 mmol/L)<br> 2 hours after a meal: ≤140 mg/ml (7.8 mmol/L)<br> Performance Characteristics:<br> 1.Blood size: 0.7 µl<br> 2.Reaction time: 7 seconds<br> 3.Measurement range: 20~600 mg/dL(1.1~33.3 mmol/L)<br> 4.Hematocrit range: 30% ~ 55%<br>