Abstract:
The invention concerns an inertial sensor or an actuator based on diamagnetic levitation, said inertial sensor or actuator comprising support means serving as main support body for an inertial sensor or for an actuator, a two dimensional array of permanent magnets and a diamagnetic element facing the said array characterized in that said diamagnetic material constitutes the inertial mass or the moving part of the actuator.

Description:
FIELD OF THE INVENTION  
       [0001]     The present invention relates to non-contact systems based on diamagnetic levitation.  
       BACKGROUND OF THE INVENTION  
       [0002]     Current inertial sensing systems such as accelerometers, gravimeters and inclinometers are based on the relative displacement between an inertial mass and the base of the instrument when the said base is subject to an external perturbation (vibration, modification of the “g” level, angle); and gyroscopes, which are another kind of inertial sensing system, are made of an inertial mass which is rotated about one of its axis of inertia and the measurement principle relies on the relative movement between the said axis and the base of the instrument, or on the force generated by the said axis on the base of the instrument, when the said base is subject to an external movement.  
         [0003]     The limitation of all those inertial sensing systems is mainly due to friction between the inertial mass and the base of the instrument.  
         [0004]     Indeed this friction is responsible of imprecision in the measurement, of wear between the mechanical parts in contact and it might also lead to failure due to mechanical fatigue.  
         [0005]     In addition inertial sensing systems, such as seismometers in seismology or inclinometers in civil engineering, are often placed to monitor structures or machines and the power consumption of such systems is sometimes a critical factor.  
         [0006]     Hence there is a need to make non-contact (contact less) inertial sensing systems with little energy consumption.  
         [0007]     One way to answer this problem is to use diamagnetic levitation, which is the only stable passive (no energy input) levitation at room temperature:diamagnetic materials are repelled by magnetic fields. If the magnetic field is created by permanent magnets, a piece of diamagnetic material can thus be passively and stably levitated.  
         [0008]     In the U.S. Pat. No. 3,831,287 a tiltmeter is designed using diamagnetic levitation but without axial contact-less stabilization of the diamagnetic inertial mass. The diamagnetic force exerted over the inertial mass is created by a 1D arrangement of large horseshoe magnets resulting in an unstable levitation in the axial direction.  
         [0009]     In the U.S. Pat. No. 5,396,136 an array of permanent magnets is levitated by magnetic interaction with a diamagnetic material (pyrolitic graphite).  
         [0010]     In such a configuration magnets are heavier than graphite for the same volume of material, and diamagnetic materials (such as pyrolitic graphite) is much more expensive than magnets for the same volume of material (or for the same weight). In addition such a stabilisation, using a bowl shape diamagnetic material, is not active and would not, if used as a sensor, behave with the high sensitivity of a force balance inertial sensing system such as the ones of the present invention.  
         [0011]     Other prior art references are listed below: 
    R. Moser, J. Sandtner, H. Bleuler, 
 
 Diamagnetic Suspension System for Small Rotors, Journal of Micromechatronics, Vol. 1, No. 2, 2001. 
    R. Moser, J. Sandtner, H. Bleuler, Diamagnetic Suspension Assisted by Active Electrostatic Actuators, 6 th  International Symposium on Magnetic Suspension Technology, Oct. 6 th  2001.     R. Moser, Y-J. Regamey, H. Bleuler, Passive Diamagnetic Levitation for Flywheels, ISMB, Sep. 24 th  2002.     R. Moser, F. Barrot, H. Bleuler, 
 
 Optimization of Two-Dimensionnal Permanent Magnet Arrays for Diamagnetic Levitation, MAGLEV, Sep. 9 th  2002. 
    Science Toys, Levitating Pyrolitic Graphite: http://www.scitoys.com/scitoys/scitoys/magnets/pyrolitic graphite.html July 2002.    
 
       SUMMARY OF THE INVENTION  
       [0017]     The present invention relates to a sensor as defined in claim  1  and to a bi-directional actuator as defined in claim  11 .  
         [0018]     Preferred embodiments are defined in the dependent claims.  
         [0019]     Advantageously, the present invention combines the use of diamagnetic levitation and electrostatic actuators to create highly sensitive non-contact inertial sensing systems based on the magnetic levitation of a diamagnetic body over a two-dimensional permanent magnet array.  
         [0020]     In the invention a diamagnetic element is facing a two dimensional planar array of permanent magnets  
         [0000]     and thus, when the bi-dimensional array of magnets is placed horizontally, the diamagnetic element floats above the 2D arrangement of magnets.  
         [0021]     The magnets are arranged in such a way (for instance: opposite polarities for neighbouring magnets) that the diamagnetic force exerted by the array of magnets overcomes the weight of the diamagnetic element.  
         [0022]     Preferably the relative horizontal position between the inertial mass and said array of magnets is sensed with one or several non-contact position sensors; This position information is then used to maintain or to move, with the use of non-contact electrostatic actuators, the diamagnetic element at a precise position above the array of magnets.  
         [0023]     To move the diamagnetic element, electrostatic forces are created by at least 3 electrostatic actuators and the diamagnetic material is part of an electrode that is common to all said electrostatic actuators. The other electrode of each electrostatic actuator is made of a non-ferromagnetic material.  
         [0024]     As for the said common electrode, if, for instance, we take a disc shaped diamagnetic material, the diamagnetic material will be inserted inside a non-ferromagnetic ring shaped metal or a ring shaped electret (that can be pre-charged by electrostatic charges). The association of the diamagnetic disc and the ring shaped metal (or ring shaped electret) constitutes the common electrode of the said electrostatic actuators and also constitutes the inertial mass used in this invention.  
         [0025]     Of course the shape of the diamagnetic material and the corresponding non-ferromagnetic metallic surface (or electret surface) that will transform the said diamagnetic mass into said common electrode, can be freely chosen.  
         [0026]     Of course the invention comprises a mechanical base to hold together the previously mentioned components.  
         [0027]     In a preferred embodiment the invention comprises a feed-back loop incorporating the electrostatic actuators, the non contact position sensors, a signal conditioning unit for the sensors, a High voltage power supply, and a controller which computes the amount of voltage to apply to the independent electrodes of the said electrostatic actuators in order to maintain the inertial mass at a predefined position. Moreover the invention may comprise a signal processing unit that can be the same unit used for the controller or a separate unit.  
         [0028]     When a relative movement of the base occurs (due to an acceleration, a tilt, etc. . . . ) the controller apply to the electrodes a voltage that is proportional to the disturbance (acceleration, angle).  
         [0029]     It is preferable to measure differentially the position of said inertial mass. This can be done using a unique sensor with several sensing unit or using two identical non-contact position sensors facing two opposite sides of the inertial mass.  
         [0030]     In addition, if a disc (or cylindrical) shape diamagnetic element is used with a ring (or cylindrical) shape electret (the electret can be pre-charged), and if each of the three (at least) electrodes facing the electret is made of, at least, three alternating comb electrodes, then a motor function can be implemented in order to spin the inertial mass about its main inertial axis.  
         [0031]     When the relative displacement between the inertial mass and the array of magnets is due to an external perturbation (such as a vibration, a tilt, a variation in the level of g) then the voltage applied to the electrostatic actuators in order to keep in place the inertial mass is proportional to the intensity (acceleration, angle) of the applied perturbation.  
         [0032]     Two-dimensional accelerometers (or seismometers), two dimensional tiltmeters (inclinometer) or gravimeters can be designed on such a concept.  
         [0033]     Hence various small inertial sensing systems made of low cost components can be designed on this same basic embodiment (implementation) requiring mainly changes in the signal processing part.  
         [0034]     In addition such an invention can also be used as a small X-Y actuator to move light objects with very small (less than a 1000 micrometer) and precise displacement (less than 50 nanometer depending on the sensitivity of the non contact distance sensors used). To do so, one only needs to give a varying order to the position controller instead of giving it a fixed position order.  
         [0035]     Besides a feedback loop along a direction Z, orthogonal to the X-Y plan of the magnet array, incorporating a Z actuator, as well a Z position sensor, can be added to the system.  
         [0036]     The total absence of contact is the main advantage of the present invention since it allows high sensitivity and high accuracy measurements.  
         [0037]     Furthermore, in such an approach, friction problems are overcome without spending much energy since the inertial mass is both: 
        levitated passively (without energy input) over the base of the instrument using diamagnetic levitation     and maintained at a precise position, adopting the force balance concept for inertial sensing systems, with the use of electrostatic actuators.       
 
     
    
     BRIEF DESCRIPTION OF THE DRAWINGS  
       [0040]     For a better understanding of the nature and objects of the invention, reference should be made to the following detailed description and the accompanying drawings, in which:  
         [0041]      FIG. 1  shows a cut side view of the inertial sensor system with: 
         1 ) Diamagnetic disk      2 ) 2D arrangement of permanents magnets      3 ) Aluminum crown      4 ) Copper electrode      5 ) Optical sensor      6 ) Infrared LED      7 ) Infrared beam passing through a hole in the diamagnetic disc and a hole in the array of magnets      8 ) Isolating layer      9 ) Structural support      10 ) Magnet with its polarity          
         [0052]      FIG. 2  explains the position control of the diamagnetic disc with: 
         11 ) Electrostatic actuator (Copper Electrode+aluminum crown)      12 ) Controller          
         [0055]      FIG. 3  shows details of the position control of the diamagnetic disc in differential mode (for V y  the procedure is the same as for V x ) with: 
         13 ) Electrostatic force exerted on the diamagnetic disc by the electrostatic actuators      14 ) Voltage amplifier          
         [0058]      FIG. 4  explains the signal conditioning of the 4 segments sensor that can be used in the inertial sensing system. i A , i B , i C , i D  are the currents which are proportionals to the illuminated surface and to the intensity of the spot. With: 
         15 ) Infrared spot          
         [0060]      FIG. 5  shows a linear drive made of a diamagnetic material over a 1D Halbach array of magnets with: 
         17 ) Diamagnetic material          
         [0062]      FIG. 6  shows diamagnetic material in a 6 degrees of freedom stable levitation over a 2D Halbach array of permanent magnets.  
         [0063]      FIG. 7  shows an unstable contact-less suspension with active electrostatic actuators (diamagnetic disc with its aluminum crown and its electrostatic electrodes over a 2D Halbach array of permanent magnets).  
         [0064]      FIG. 8  shows a side view of a diamagnetic material in a 5 degrees of freedom levitation over a 2D Halbach array of permanent magnets, with magnetic field lines with: 
         18 ) Magnetic field lines          
         [0066]      FIG. 9  shows a side view of a diamagnetic material in a three degrees of freedom levitation over a 2D Halbach array of permanent magnets, with magnetic field lines.  
         [0067]      FIG. 10  shows the implementation of the inertial sensor with 2 pairs of non contact distance sensors with: 
         16 ) Displacement sensor          
         [0069]      FIG. 11  shows a sketch of the electrode used to create a 3 phases alternating electric field in the gyroscope application with: 
        a) Phase 1 electrode     b) Phase 2 electrode     c) Phase 3 electrode      20 ) Front view of the 3 phases electrode      21 ) Back view of the 3 phases electrode          
         [0075]      FIG. 12  shows a top view of an inertial disc shaped mass with two orthogonal pairs of double electrodes with: 
         4 ′) Inner copper electrode      4 ″) Outer copper electrode          
         [0078]      FIG. 13  Possible magnets arrays for diamagnetic levitation, a) Opposite2D b) Opposite 1D, c) Repulsive 2D, d) repulsive 1D, e) Halbach 1D f) Halbach 2D g) reference magnet. 
     
    
     DETAILED DESCRIPTION OF THE INVENTION  
       [0079]     Several kinds of precision instruments can be designed on the principle of a diamagnetic material levitated over a 2D array of permanent magnets and kept in a precise position with electrostatic actuation.  
         [0080]     We will describe a preferred embodiment (cf  FIG. 1 ,  FIG. 2 ,  FIG. 3 ,  FIG. 4 ,  FIG. 7 ) that can be used as: 
        A bidirectional acceleration sensor or a bi-directional seismometer     A bi-directional tiltmeter (inclinometer)     A gyroscope with the addition of an alternating voltage applied on comb electrodes  20  and  21  ( FIG. 11 ) instead of the plain electrodes  4  (or  4 ′ and  4 ″).     A bidirectional actuator.     A gravimeter.        
 
         [0086]     In the preferred embodiment described in  FIG. 1 ,  FIG. 2 ,  FIG. 3 ,  FIG. 4 , and  FIG. 7 ,  
         [0087]     a diamagnetic disc  1  is surrounded with a cylindrical aluminum crown  3  and is levitated over a Halbach-2D array  2  (cf.  FIG. 13 . f ) of permanent magnets  10 .  
         [0088]     The position of the disc  1  is controlled in a feed-back loop.  
         [0089]     This feed-back loop is made of: 
        2 pairs of electrodes  11  made by the association of the aluminum crown  3  and copper electrodes  4 . The four electrostatic actuators are diametrically disposed in pairs along two orthogonal axis. One of the electrode of such actuators is the cylindrical aluminum crown  3  and the other electrode of each actuator, has a cylindrical shape as seen on  FIG. 3 ,  FIG. 7  and  FIG. 10 .     a 4 segments optical sensor  5  spotted by an infrared LED  6 - 7 - 15  through the diamagnetic disc  1  and through 2.     and a digital controller  5  that generates the required voltages ( FIG. 2  and  FIG. 3 ) to the 2 pairs of electrodes  11  (or  4 + 3 ) in order to maintain (thanks to the generated electrostatic forces  13 ) the diamagnetic disc  1  at a predefined position when it is subjected to motion (due to shaking in case of a seismic sensor, or due to angular displacement in case of an X-Y tiltmeter, or due to a variation of g in the case of a gravimeter).        
 
         [0093]     In the case of, for instance, the seismometer application, the forces  13  generated by the control  12  (or the voltages applied to the electrodes  4 ) are proportional to the soil accelerations.  
         [0094]     As we can see in  FIG. 4  the measurements (X and Y) with the four segments optical sensor  5  are differential.  
         [0095]     The advantage of such a differential measurement is that it cancels out the effect of temperature (or pressure or humidity . . . ) variations.  
         [0096]     Of course we can also use 2 pairs of position sensors  16  ( FIG. 10 ) (facing each others around the aluminum disc  1 , cf  FIG. 10 ) and we would also have differential measurements: subtracting the output signal of two diametrically opposite sensors  16  gives the position information of the inertial mass  1 + 3  along the direction defined by the two said diametrically opposite sensors  16 .  
         [0097]     The sensors  16  are non-contact position sensors; they can be optical reflection sensors, eddy current sensors, capacitive sensors (with a comb structure for instance) or interferometric sensors.  
         [0098]     Of course, a signal-conditioning unit is needed for the non-contact position sensor(s)  5  or  16  and also a high voltage power supply or a high voltage amplifier  14 , with at least two inputs and four outputs, is needed to apply high voltage (with very low currents) to the electrodes.  
         [0099]     If a feedback loop along a direction Z, orthogonal to the X-Y plan of the magnet array, incorporating a Z electrostatic actuator, as well as, at least, one Z position sensor (facing one of the face of the inertial disc), is added to the inertial sensing system embodiment described previously, then the whole system becomes an X-Y-Z actuator and the high voltage amplifier needs an additional input and an additional output. The Z actuator consists of at least one electrostatic actuator made of the diamagnetic disc  1  and the array of magnets  2  on which a high voltage is applied in order to attract the diamagnetic disc.  
         [0100]     Such an actuator can be used as the scanning module for an Atomic Force Microscope probe. The AFM probe is fixed in the centre of the diamagnetic disc  1  and points downward, towards the array of magnets  2 . Just under the AFM probe, a magnet of the magnet array  2  has been removed from the magnet array  2  and the element to be scanned is positioned inside the hole left by the missing magnet;  
         [0101]     Such a triaxial actuator (X-Y-Z) can also be used as a precise positioning unit that can, for instance, be incorporated in a larger X-Y table.  
         [0102]     The preferred arrangement for inertial sensing systems, or for bidirectional or tri-directional actuators, is a 2D (such as Halbach 2D or opposite 2D) array of magnets in the configuration of  FIG. 9 .  
         [0103]     Indeed, depending on both the shape of the inertial mass  1 + 3  and the shape of the 2D magnet array  
         [0104]      FIG. 5 ,  FIG. 6 ,  FIG. 7 ,  FIG. 8 ,  FIG. 9 , several kinds of stabilisation of the inertial mass can be achieved: 
        In  FIG. 5  the diamagnetic material  17  has a rectangular shape and is stably floating over a Halbach 1D. Such a configuration is especially well adapted to design a linear diamagnetic drive that can, for instance be used to convey small spare parts in a microfactory.     In  FIG. 6  the diamagnetic material  17  has a square shape and is in a six degree of freedom stable and passive levitation over a Halbach 2D array of permanent magnets. This configuration is illustrated in  FIG. 8 .          
         [0107]     In  FIG. 7  the diamagnetic disc  1  with an aluminium crown  3  constitutes an unstable contact-less suspension which is radially stabilized with active electrostatic actuators  3 + 4 . This configuration corresponds to the configuration illustrated in  FIG. 9 . Such a configuration is well adapted to the design of inertial sensing system as described in the present invention.  
         [0108]     As for the 2D magnet array it does not have to be a Halbach 2D array (2 or  FIG. 13 - f ), it can be any of the possible magnets arrays shown in  FIG. 13 . The Halbach 2D array ( 2  or  FIG. 13 - f ), is the one which produces the largest diamagnetic force among all those arrays; but, for instance, an opposite 2D array  FIG. 13 - a  of magnets can also be used and it is auto-stable (that is to say: no glue is needed for the assembly of this array) by opposition to the Halbach 2D array ( 2  or  FIG. 13 - f ).  
         [0109]     Except for the use of this inertial sensor as a gyroscope, the shape of the inertial mass  1 + 3  does not have to be a disc nor a cylinder; It can be a parallelepiped, a triangle or a square for instance.  
         [0110]     To be used as a gyroscope, some small changes have to be applied to the preferred embodiment: 
        The aluminium crown  3  is replaced by an electret crown that can be pre-charged or not.     The plain copper electrodes  4  are replaced by comb shaped electrodes  20 - 21  with at least three phases as described in  FIG. 11 . The inertial mass  1 + 3  rotates when a rotating electric field is applied to such electrodes     Some additional plain electrodes  4 ′ (cf. FIG. 12 ), pointing towards the inner surface of the electret crown can be added to better stabilize the axial movements of the inertial mass.        
 
         [0114]     Of course the place of the plain electrodes  4 ′ and the comb shaped electrodes  20 - 21  can be inverted; that is to say the plain electrodes  4 ′ (cf. FIG. 12 ), points towards the outer surface of the electret crown and the comb shaped electrodes  20 - 21  point towards the inner surface of the electret crown.  
         [0115]     The electrodes  4  do not have to be made of copper but they should be made of a non-ferromagnetic metal. Indeed, the electrodes  4  being placed next to the array of permanent magnets  2 , the flux line  18  of the later  2  would be severely modified if the electrodes  4  where made of a ferromagnetic material. In addition, an insulation layer  8  is needed on electrodes  4 - 4 ′- 4 ″ or on the aluminum crown  3  in order to avoid short circuits when the inertial mass  3 + 1  is in contact with more than one electrode  4 .  
         [0116]     The number of electrostatic electrodes needed to conceive inertial sensing systems as described on this invention, could be reduced to 3; the control would be a little bit more complicated than with 4 electrodes but the high voltage power supply stage would only need 3 independent outputs whereas it needs 4 independent outputs in the 4 electrodes embodiment.  
         [0117]     Moreover the electrostatic force exerted on the inertial mass can be doubled by using double electrodes as illustrated in  FIG. 12 . In such a configuration the voltage applied to an inner copper electrode  4 ″ is the same as the voltage of the diametrically opposed outer electrode  4 ′.  
         [0118]     As for the crown  3 , it does not have to be made of aluminum, it can be made of an electret (pre-charged or not) or a non-ferromagnetic metal.  
         [0119]     Indeed, the crown  3  cannot be made of ferromagnetic metal because, otherwise, it would be attracted towards the magnet array and it would cancel the diamagnetic force as well as the diamagnetic levitation.  
         [0120]     The main parameters of this inertial sensing system, such as the equivalent rigidity and damping of the electrostatic actuators, can be varied by changing numerical values in the digital controller. Hence the bandwidth of the sensor is user selectable. In addition some pre-processing or processing tasks can be implemented directly in the digital controller and the results can be saved in a data storage component.  
         [0121]     In addition with the adjunction of a compensating magnet a bigger body can be diamagnetically levitated and other apparatus such as flywheel can be designed or larger inertial masses can be used for the inertial sensor of the present invention.  
         [0122]     Everything that has been previously said about the inertial sensing system application can also be applied for the bi-directional X-Y and tri-directional X-Y-Z actuator.  
         [0123]     The precision of the measurements will mainly depend on the electronics used (resolution and sensitivity of the sensor, number of bits of the AD converters, number of bits used in the controller, etc. . . . );  
         [0000]     and if we apply the void inside the system, we can also enhance the precision of the measurements.