Non-Invasive Blood Pressure Monitor Accuracy in Neonatal Subjects

Study Overview
=================
Brief Summary
-----------------
This study is being conducted to validate the accuracy of non-invasive blood pressure (NIBP) measurement technology from Mortara Instrument, Inc. (Milwaukee, Wisconsin, USA) and Zoe Medical, Inc. (Topsfield, Massachusetts, USA) in a neonatal, infants and children up to 3 years old patient population. The study is defined according to the International Standard ANSI/AAMI/ISO 81060-2:2013, Non-invasive sphygmomanometers - Part 2: Clinical investigation of automated measurement type.

Detailed Description
-----------------
The device under test is Mortara Instrument's Surveyor S12/S19 patient monitors (Surveyor) incorporating neonatal NIBP technology. Specifically, NIBP measurements will be taken using a cuff on the patient's limb (exact location will be determined by clinician) using a variety of cuffs sized for neonates and pediatrics. The device utilizes oscillometric technology to obtain quickly and conveniently non-invasive blood pressure measurements. Results will be compared to invasive blood pressure measurements from patients already monitored with such sensors. The Device under Test will be calibrated and verified by a Mortara representative on site.

Official Title
-----------------
Clinical Investigation for Non-invasive Blood Pressure Monitor Accuracy Validation Performed on Neonatal, Infants and Children up to 3 Years Old

Conditions
-----------------
Non-invasive Blood Pressure

Intervention / Treatment
-----------------
* Device: Surveyor S12 and S19 patient monitor


Participation Criteria
=================
Eligibility Criteria
-----------------
Patient inclusion criteria Before proceeding with the blood pressure non-invasive measurements detection, patients shall be examined and met the following stated criteria: Patients shall have already an invasive blood pressure monitoring (with catheter), this criteria applies only for Method 1. It shall be possible to perform multiple measurements of Non Invasive Blood pressure on one or more limbs with inflatable cuff. Patient with weight and age range according to the protocol requirements previously defined. Once the patient is identify a pre-screening activities is performed in order to assess any differences in the blood pressure. If the differences are minimal, according to the criteria described in the next section Exclusion criteria, the patient can be included in the study group. Exclusion criteria Patients with unstable clinical or cardiovascular conditions; Patients with circulatory problems which can results in different blood pressure values in the limbs; patients with heart disease which can results in blood pressure wave morphology abnormalities; patients where the difference of systolic blood pressure between two limbs is greater than 15 mm Hg or the difference of lateral diastolic blood pressure is greater than 10mmHg; Patients with arrhythmias.

Ages Eligible for Study
-----------------
Minimum Age: 29 Days
Maximum Age: 3 Years

Sexes Eligible for Study
-----------------
All

Accepts Healthy Volunteers
-----------------
Accepts Healthy Volunteers

Study Plan
=================
How is the study designed?
-----------------

Arms and Interventions

| Intervention/Treatment |
| --- |
|Device: Surveyor S12 and S19 patient monitor|Non-invasive blood pressure measurement|

What is the study measuring?
-----------------
Primary Outcome Measures

| Outcome Measure | Measure Description | Time Frame |
| --- | --- | --- |
| To study is the non-invasive blood pressure measurements accuracy validation performed with the Surveyor S12 / S19 monitor in pediatric, infants and children up to 3 years patients per ISO 81060-2: 2013 requirements. | To study the accuracy of non-invasive blood pressure measurements in neonatal patients | 4 hours per patient | 

 Terms related to the study
=================
Keywords Provided by Centre Hospitalier Valida
-----------------
NiBP