Endotracheal Tube Holder and Bite Guard Research

Study Overview
=================
Brief Summary
-----------------
This study aims to test the effectiveness of a novel combined endotracheal tube holder and bite guard when compared with the current standard of care, tape and an oral airway.

Detailed Description
-----------------
Endotracheal tubes (ETT) are secured during surgery and ICU care so that they are not accidentally removed from the airway. The position of the tip of the ETT is important and must not vary much within the trachea. The standard technique for securing an ETT is to use tape to attach it to the upper lip and face. The technique is cheap and simple, but suffers from certain drawbacks. First, the upper lip is mobile and thus even when well secured, the tube can move within the trachea due to this mobility. Second, conditions on the face such as sweating, drooling, facial hair, skin flaking or breakdown, all interfere with the ability of the tape to adhere, thus reducing its efficacy. Furthermore, the tape itself can irritate and or injure the facial skin. The adhesive irritates skin and certain patients are allergic to these substances. Also, removing the tape can physically damage the skin, especially if the skin is vulnerable to sheer forces (such as elderly patients or patients on systemic steroids). Patients that are intubated often need a bite block or bite guard inserted into their mouths. This serves two main purposes: 1) to prevent the patient from biting the ETT and occluding airflow, 2) to prevent the patient from biting their tongue, cheek, or lips causing injury. Furthermore, endoscopic procedures are often performed on intubated patients and thus a bite block is needed to protect the delicate instrument from being bitten and damaged. For these reasons, many providers will insert bite blocks into the mouths of intubated patients. Many such devices exist on the market, but the most commonly used is the Guedel oral airway. This device is actually designed to increase airway patency during mask ventilation, and long term use of this device as a bite block has been linked to numerous complications including: 1)tongue swelling, often compromising the airway patency, 2) tooth damage, and 3)lip injury. Because of these shortcomings, specific ETT bite blocks have been developed, and some of these even double as an ETT securing device. However, these are usually made of hard, stiff materials that in and of themselves can injure the delicate oral structures. In summation, no single device that both serves as an ergonomic ETT holder and bite block has been successfully developed and marketed, and therefore the standard of care remains tape plus an oral airway. The Haider airway is a combination ETT holder-securing device and bite block. Made of soft silicone rubber, the device was engineered from the ground up to serve those purposes with safety and comfort in mind. The device is FDA approved and in a pre-marketing trials phase to determine the efficacy of the product in specific clinical scenarios. Our institution has been tasked with testing the device in clinical scenarios where tape + oral airway has been historically problematic. We plan to compare this new device to tape + oral airway, the current standard of care in neurosurgical patients in the supine position. We hypothesize the device will be superior to the standard of care in both efficacy and patient comfort.

Official Title
-----------------
Testing a Novel Combined Endotracheal Tube Holder and Bite Guard in Neurosurgical Patients, Thoracic Surgery Patients, and Orthopedic and Neurosurgical Spine Surgery Patients.

Conditions
-----------------
Surgery, Anesthesia

Intervention / Treatment
-----------------
* Device: Haider ETT Tube Holder and Bite Guard
* Procedure: Tape


Participation Criteria
=================
Eligibility Criteria
-----------------
Inclusion Criteria: Age ≥ 18 years General anesthesia requiring oral endotracheal intubation in the prone position for the duration of surgery (head secured in a foam head holder, horseshoe frame, or Mayfield pins). General anesthesia for neurosurgery requiring oral endotracheal intubation, supine or slight lateral bump position for the duration of surgery. (Special emphasis will be placed on including patients who will undergo motor evoked potentials, and thus will not be paralyzed during the surgery). General anesthesia requiring oral endotracheal intubation with a left sided double-lumen endotracheal tube for lung isolation, lateral position for the duration of surgery. Exclusion Criteria: Dentition issues (loose teeth, missing teeth) Temporomandibular joint disease Maxillofacial abnormalities (deformities of the jaw, lips, tongue) Surgical procedures involving the teeth, lips, jaw History of asthma or bronchospasm Immunosuppression.

Ages Eligible for Study
-----------------
Minimum Age: 18 Years

Sexes Eligible for Study
-----------------
All

Accepts Healthy Volunteers
-----------------
No

Study Plan
=================
How is the study designed?
-----------------

Design Details

Primary Purpose: Treatment
Intervention Model: Single Group Assignment
Masking: None (Open Label)


Arms and Interventions

| Participant Group/Arm | Intervention/Treatment |
| --- | --- |
| Experimental: ETT holder and bite guard<br>Participants will have endotracheal tubes secured with tape prior to having surgery, which is the current standard of care. Following the traction test and measurement of displacement, participants will have endotracheal tubes secured with a combined Haider ETT Tube Holder and Bite Guard. | Device: Haider ETT Tube Holder and Bite Guard<br>* Surgery patients will have endotracheal tubes secured with a novel combined endotracheal tube holder and bite guard change in ETT tip position caused by traction up to 15N was measured prior to surgery.<br>* Other names: ETT Tube Holder and Bite Block;Procedure: Tape<br>* Surgery patients will have endotracheal tubes secured with adhesive tape, as the current clinical procedure; change in ETT tip position caused by traction up to 15N measured prior to re-securing the ETT with the Haider device.<br>|

What is the study measuring?
-----------------
Primary Outcome Measures

| Outcome Measure | Measure Description | Time Frame |
| --- | --- | --- |
| Change in ETT Position | The ability of the securing device to resist endotracheal tube dislodgement under axial strain for patients in the supine position was measured. The endotracheal tube, once secured with either tape or the Haider airway device, encountered an axial force to simulate the endotracheal tube being pulled from the mouth. The force increased over approximately 5 seconds until the target of 15 N was reached or until the principal investigator deemed that the force be aborted to prevent possible tracheal extubation. The change in position, measured in cm, was recorded. | 5 minutes after intubation (which occurs at the beginning of the anesthesia about 2-5 minutes after the patient goes to sleep). | 

Secondary Outcome Measures

| Outcome Measure | Measure Description | Time Frame |
| --- | --- | --- |
| Participants With Irritation or Minor Injury to the Face and Oral Structures Likely Attributable to the Study Device | The patients were examined after surgery (both immediately after surgery and at the end of the recovery room period) to determine if the patient suffered any irritation or minor injury to the face and oral structures. The patients were asked to fill out a questionnaire after recovery with questions about their overall experience with specific relation to any irritation and/or minor trauma to their face, oral structures, throat, jaw, and temporomandibular joint. | Immediately after surgery and 1-3 days following surgery, before discharge. | 
| Clinically Significant Movement | Number of participants experiencing ETT movement >1cm under each fixation technique | 5 minutes after intubation (which occurs at the very beginning of the anesthesia about 2-5 minutes after the patient goes to sleep) | 
| Potentially High Extubation Risk | Number of participants experiencing ETT movement >4cm under each fixation technique | 5 minutes after intubation (which occurs at the very beginning of the anesthesia about 2-5 minutes after the patient goes to sleep) | 

 Terms related to the study
=================
Keywords Provided by Centre Hospitalier Valida
-----------------
ETT Holder, bite guard, airway, endotracheal tube