Publication: Magyar Közlöny
Issue: MK-2007-70 (Year: 2007, Number: 70)
Era: 2004-2010
Section: Melléklet a 2007. évi XLVI. törvényhez
Paragraph Index: 6548

b) Sample temperature: the normal mode of operation is for analysis of the sample in its (untreated) “wet” condition. This requires that the sample cell and all other components in contact with the sample in this sub-system be maintained at a temperature of not less than 50°C, with a stability of ±2°C. The option to measure CO and CO2 on a dry basis (with suitable water traps) is allowed, in which case unheated analysers are permissible and the interference limits for H2O vapour removed, and subsequent correction for inlet water vapour and water of combustion is required.

Source: https://magyarkozlony.hu/hivatalos-lapok/4c6310a937d14bac566ee9c9d944896656c292dd/dokumentumok/5a6ac3c6db12e692ae41096677ad5f072d2cc9f3/letoltes