page_title: Heliometer
hierarchical_section_title: Heliometer
caption_attribution_description: English: EB1911 - Heliometer - Fig. 11, Repsolds completed a new heliometer for Yale College, New Haven, USA. The object-glass is of 6 inch aperture and 98 inch focal length. The mounting, the tube, objective-cell, slides, &amp;c., are all of steel. —from EB1911
context_page_description: A heliometer is an instrument originally designed for measuring the variation of the sun's diameter at different seasons of the year, but applied now to the modern form of the instrument which is capable of much wider use.
  The basic concept is to introduce a split element into a telescope's optical path so as to produce a double image. If one element is moved using a screw micrometer, precise angle measurements can be made. The simplest arrangement is to split the object lens in half, with one half fixed and the other attached to the micrometer screw and slid along the cut diameter. To measure the diameter of the sun, for example, the micrometer is first adjusted so that the two images of the solar disk coincide. The micrometer is then adjusted so that diametrically opposite sides of the two images of the solar disk just touch each other. The difference in the two micrometer readings so obtained is the diameter of the sun. Similarly, a precise measurement of the apparent separation between two nearby stars, A and B, is made by first superimposing the two images of the stars and then adjusting the double image so that star A in one image coincides with star B in the other.
context_section_description: A heliometer (from Greek ἥλιος hḗlios "sun" and measure) is an instrument originally designed for measuring the variation of the sun's diameter at different seasons of the year, but applied now to the modern form of the instrument which is capable of much wider use.
  The basic concept is to introduce a split element into a telescope's optical path so as to produce a double image. If one element is moved using a screw micrometer, precise angle measurements can be made. The simplest arrangement is to split the object lens in half, with one half fixed and the other attached to the micrometer screw and slid along the cut diameter. To measure the diameter of the sun, for example, the micrometer is first adjusted so that the two images of the solar disk coincide (the "zero" position where the split elements form essentially a single element). The micrometer is then adjusted so that diametrically opposite sides of the two images of the solar disk just touch each other. The difference in the two micrometer readings so obtained is the (angular) diameter of the sun. Similarly, a precise measurement of the apparent separation between two nearby stars, A and B, is made by first superimposing the two images of the stars and then adjusting the double image so that star A in one image coincides with star B in the other. The difference in the two micrometer readings so obtained is the apparent separation or angular distance between the two stars.
  The first application of the divided object-glass and the employment of double images in astronomical measures is due to Servington Savary from Exeter in 1743. Pierre Bouguer, in 1748, originated the true conception of measurement by double image without the auxiliary aid of a filar micrometer, that is by changing the distance between two object-glasses of equal focus. John Dollond, in 1754, combined Savary's idea of the divided object-glass with Bouguer's method of measurement, resulting in the construction of the first really practical heliometers. As far as we can ascertain, Joseph von Fraunhofer, some time not long before 1820, constructed the first heliometer with an achromatic divided object-glass, i.e. the first heliometer of the modern type.
  The first successful measurements of stellar parallax (to determine the distance to a star) were made by Friedrich Bessel in 1838 for the star 61 Cygni using a Fraunhofer heliometer. This was the 6.2-inch (157.5 mm) aperture Fraunhofer heliometer at Königsberg Observatory built by Joseph von Fraunhofer's firm, though he did not live to see it delivered to Bessel. Although the heliometer was difficult to use, it had certain advantages for Bessel including a wider field of view compared to other great refractors of the period, and overcame atmospheric turbulence in measurements compared to a filar micrometer.
caption: Heliometer at Yale College, New Haven, US: c. 1910, built by Repsolds
exif: {}