page_title: Fourier-transform infrared spectroscopy
section_title: Michelson interferometer
hierarchical_section_title: Fourier-transform infrared spectroscopy / Michelson interferometer
caption_attribution_description: English: Interferometer for FTIR
context_page_description: Fourier-transform infrared spectroscopy is a technique used to obtain an infrared spectrum of absorption or emission of a solid, liquid or gas. An FTIR spectrometer simultaneously collects high-spectral-resolution data over a wide spectral range. This confers a significant advantage over a dispersive spectrometer, which measures intensity over a narrow range of wavelengths at a time.
  The term Fourier-transform infrared spectroscopy originates from the fact that a Fourier transform is required to convert the raw data into the actual spectrum.
context_section_description: In a Michelson interferometer adapted for FTIR, light from the polychromatic infrared source, approximately a black-body radiator, is collimated and directed to a beam splitter. Ideally 50% of the light is refracted towards the fixed mirror and 50% is transmitted towards the moving mirror. Light is reflected from the two mirrors back to the beam splitter and some fraction of the original light passes into the sample compartment. There, the light is focused on the sample. On leaving the sample compartment the light is refocused on to the detector. The difference in optical path length between the two arms to the interferometer is known as the retardation or optical path difference (OPD). An interferogram is obtained by varying the retardation and recording the signal from the detector for various values of the retardation. The form of the interferogram when no sample is present depends on factors such as the variation of source intensity and splitter efficiency with wavelength. This results in a maximum at zero retardation, when there is constructive interference at all wavelengths, followed by series of "wiggles". The position of zero retardation is determined accurately by finding the point of maximum intensity in the interferogram. When a sample is present the background interferogram is modulated by the presence of absorption bands in the sample.
  Commercial spectrometers use Michelson interferometers with a variety of scanning mechanisms to generate the path difference. Common to all these arrangements is the need to ensure that the two beams recombine exactly as the system scans. The simplest systems have a plane mirror that moves linearly to vary the path of one beam. In this arrangement the moving mirror must not tilt or wobble as this would affect how the beams overlap as they recombine. Some systems incorporate a compensating mechanism that automatically adjusts the orientation of one mirror to maintain the alignment. Arrangements that avoid this problem include using cube corner reflectors instead of plane mirrors as these have the property of returning any incident beam in a parallel direction regardless of orientation.
  Systems where the path difference is generated by a rotary movement have proved very successful. One common system incorporates a pair of parallel mirrors in one beam that can be rotated to vary the path without displacing the returning beam. Another is the double pendulum design where the path in one arm of the interferometer increases as the path in the other decreases.
  A quite different approach involves moving a wedge of an IR-transparent material such as KBr into one of the beams. Increasing the thickness of KBr in the beam increases the optical path because the refractive index is higher than that of air. One limitation of this approach is that the variation of refractive index over the wavelength range limits the accuracy of the wavelength calibration.
caption: Schematic diagram of a Michelson interferometer, configured for FTIR
exif: {}