page_title: Fabry–Pérot interferometer
section_title: Applications
hierarchical_section_title: Fabry–Pérot interferometer / Applications
caption_attribution_description: Français&#160;: Interféromètre de Perot-Fabry. Modèle de la société Spectra-Physics (model 470 confocal). Ancien équipement utilisé au Laboratoire de Physique de l'Université de Bourgogne (désormais Laboratoire Interdisciplinaire CARNOT de Bourgogne), équipe Solitons Lasers et Communications Optiques. Le dispositif est désormais utilisé dans le cadre des travaux pratiques du département de Physique de l'UFR Sciences et Techniques.
context_page_description: In optics, a Fabry–Pérot interferometer or etalon is an optical cavity made from two parallel reflecting surfaces. Optical waves can pass through the optical cavity only when they are in resonance with it. It is named after Charles Fabry and Alfred Perot, who developed the instrument in 1899. Etalon is from the French étalon, meaning "measuring gauge" or "standard".
  Etalons are widely used in telecommunications, lasers and spectroscopy to control and measure the wavelengths of light. Recent advances in fabrication technique allow the creation of very precise tunable Fabry–Pérot interferometers. The device is called an interferometer when the distance between the two surfaces can be changed, and an etalon when the distance is fixed.
context_section_description: Telecommunications networks employing wavelength division multiplexing have add-drop multiplexers with banks of miniature tuned fused silica or diamond etalons. These are small iridescent cubes about 2 mm on a side, mounted in small high-precision racks.  The materials are chosen to maintain stable mirror-to-mirror distances, and to keep stable frequencies even when the temperature varies.  Diamond is preferred because it has greater heat conduction and still has a low coefficient of expansion. In 2005, some telecommunications equipment companies began using solid etalons that are themselves optical fibers.  This eliminates most mounting, alignment and cooling difficulties.
  Dichroic filters are made by depositing a series of etalonic layers on an optical surface by vapor deposition.  These optical filters usually have more exact reflective and pass bands than absorptive filters.  When properly designed, they run cooler than absorptive filters because they can reflect unwanted wavelengths.  Dichroic filters are widely used in optical equipment such as light sources, cameras, astronomical equipment, and laser systems.
  Optical wavemeters and some optical spectrum analyzers use Fabry–Pérot interferometers with different free spectral ranges to determine the wavelength of light with great precision.
  Laser resonators are often described as Fabry–Pérot resonators, although for many types of laser the reflectivity of one mirror is close to 100%, making it more similar to a Gires–Tournois interferometer. Semiconductor diode lasers sometimes use a true Fabry–Pérot geometry, due to the difficulty of coating the end facets of the chip. Quantum cascade lasers often employ Fabry-Pérot cavities to sustain lasing without the need for any facet coatings, due to the high gain of the active region.
  Etalons are often placed inside the laser resonator when constructing single-mode lasers. Without an etalon, a laser will generally produce light over a wavelength range corresponding to a number of cavity modes, which are similar to Fabry–Pérot modes. Inserting an etalon into the laser cavity, with well-chosen finesse and free-spectral range, can suppress all cavity modes except for one, thus changing the operation of the laser from multi-mode to single-mode.
  Fabry–Pérot etalons can be used to prolong the interaction length in laser absorption spectrometry, particularly cavity ring-down, techniques.
  A Fabry–Pérot etalon can be used to make a spectrometer capable of observing the Zeeman effect, where the spectral lines are far too close together to distinguish with a normal spectrometer.
  In astronomy an etalon is used to select a single atomic transition for imaging. The most common is the H-alpha line of the sun. The Ca-K line from the sun is also commonly imaged using etalons.
  The methane sensor for Mars (MSM) aboard India's Mangalyaan is an example of a Fabry-Perot instrument. It was the first Fabry Perot instrument in space when Mangalyaan Launched. As it did not distinguish radiation absorbed by methane from radiation absorbed by carbon dioxide and other gases, it was later called an albedo mapper.
  In gravitational wave detection, a Fabry–Pérot cavity is used to store photons for almost a millisecond while they bounce up and down between the mirrors. This increases the time a gravitational wave can interact with the light, which results in a better sensitivity at low frequencies. This principle is used by detectors such as LIGO and Virgo, which consist of a Michelson interferometer with a Fabry–Pérot cavity with a length of several kilometers in both arms. Smaller cavities, usually called mode cleaners, are used for spatial filtering and frequency stabilization of the main laser.
caption: A commercial Fabry-Perot device
exif: {}