Bside 1mA DC/AC Clamp Meter True RMS 6000 Counts Temperature Low Impedance Voltage Amp V-Alert Live Wire Check Tester with Back Clip
Brand: BSIDE
Color: Gray
Features:
Measure AC/DC current (minimum 1mA by jaws)
Measure microamps μA (minumum 0.1μA by terminals)
Measure AC/DC voltage
V-alert measurement, live-wire check
Measure frequency, temperature, resistance, continuity and capacitance
Low impedance voltage measurement
Clamp on frequency function
Auto range, Auto power off, data hold, low battery indication with backlight
Conform to the IEC-61010-1, IEC-61010-2-030, IEC-61010-2-032, IEC-61010-2-033 standards
Over-voltage protection class CAT III 300V, CAT II 600V, pollution degree II
Specifications:
Display: 6000 counts
DC current: 6A/60A/100A 0.001A/0.01A/0.1A ±(3.0%+5counts)
DC current(μA): 0-200μA 0.1μA ±(0.8%+5counts)
AC current: 6A/60A/100A 0.001A/0.01A/0.1A ±(2.5%+5counts)
DC voltage: 6V/60V/600V 0.001V/0.01V/0.1V ±(1.0%+3counts)
AC voltage: 6V/60V/600V 0.001V/0.01V/0.1V ±(1.0%+3counts)
Low impedance LOZ: Max 600V Resolution 0.1V
Resistance:
600Ω/6kΩ/60kΩ/600kΩ/6MΩ 0.1Ω/0.001KΩ/0.01KΩ/0.1KΩ/0.001MΩ ±(1.0%+3counts)
60MΩ 0.01MΩ ±(1.2%+20counts)
Capacitance: 600µF/6000µF 0.1µF/1µF ±(4.0%+3counts)
Frequency: 60Hz/1000Hz 0.1Hz/1Hz ±(1.0%+5counts)
Frequency(by jaws): 60Hz/1000Hz 0.1Hz/1Hz ±(1.0%+5counts)
Buzzer: If the resistance is less than 30Ω, the continuity beeper sounds
Temperature: -20℃~500℃ 1℃/2℉ ±(2.0%+2counts)
Jaw Capacity: 20mm
Power: 1.5V (AAA) x 2 (not included)
Size: 179 x 67 x 36 mm
Weight: 137g
Package Includes:
1 x Digital Clamp Meter
1 x Probes
1 x Thermocouple
1 x Carrying Case
1 x Screwdriver
1 x User Manual
Note:
The batteries are not included in the package due to Aviation Security Check
[1mA Sensitivity]: 1mA sensitivity when measures AC/DC current by clamp.
[Dual Line Display]: Dual line display can show both AC voltage/current and frequency or Celsius and Fahrenheit on the screen.
[Low Impedance Mode]: Low Impedance (LoZ) mode for identifying and eliminating ghost or stray voltages.
[Activate V-Alert at Any Function]: Activate V-Alert (non-contact voltage detection) function under any function design makes non-contact voltage testing more convenient.
[Back Clip Design]: Back clip design make it more portable, hang on your belt or pocket.