Digital Salinity & Temperature Meter with ATC, Handheld Pen Type Water Quality Measurement Monitor Tester, 0~70 PPT, for Saltwater, Hydroponics, Pool, Aquarium, Fish Pond, Spa, Monitoring & Testing
Brand: Gain Express
Color: Salinity Meter(70.0ppt)
Dual meter is designed to measure the salt content in liquid without the need for conversions. It measures both temperature (ºC/°F) and salinity (ppt) and display simultaneously on its clear LCD. It has an IP65 Rating, waterproof and dustproof housing and waterproof membrane keypad.

Applications & Usage:

It quickly and accurately measure salt concentration in koi ponds. . Ideal for checking salt in fish hospital tanks and tropical aquariums, water monitoring, seawater-prepared food solutions, breeding ponds, aquariums, holding tanks and more!

Features:
Built-in NaCL Temperature coefficient
Conductivity to TDS conversion factor
Multi point calibration guarantees full range accuracy
Allow factory default resetting
With over & below the limit value indicator
Low battery indicator

Specifications:
Measuring Range:
Salinity: 0.00~10.0 ppt (NaCL) and 10.1~70.0ppt (NaCL)
Temperature Compensation: 0~50°C
Accuracy:
Salinity: 1% Full Scale ± 1 digit or 2% Full Scale ± 1 digit (10.1-70.0ppt)
Temperature: ±0.5°C
Resolution:
Salinity: 0.01ppt or 0.1ppt
Temperature: 0.1°C/°F
TDS Factor: Built-in NaCL conductivity to TDS conversion factor
Built-in NaCL Temperature coefficient
Normalization Temperature: Fixed at 25°C
Operating Temperature: 0~50°C
Power Supply: 4 x 1.5V button batteries (last up to 80hrs continuous work)
Auto power off if idle or no operation for 20 minutes
Package Weight: 0.14kg
Package Size: 19cm x 8cm x 5cm

Set Includes:
1 x Salinity and Temperature Meter
4 x 1.5V Button Batteries (FREE)
English Quick Start Guide
Standard packaging box
MEASURING RANGE - Salinity: 0.00~10.0 ppt (NaCL) and 10.1~70.0ppt (NaCL) ; Temperature Compensation: 0~50°C
HIGH ACCURACY - Multi point calibration guarantees full range accuracy
FUNCTIONAL DESIGN - IP65 Waterproof housing with built-in NaCL temperature coefficient, conductivity to TDS conversion factor and data holding
USER FRIENDLY - Allow factory default resetting, with over & below the limit value and low battery indicator
QUALIFIED - With CE Marking, tested according to European international guidelines and standards