Coating Thickness Gauge DPM-816 Pro | Paint Meter for Car | Fast & Accurate | Range 0-137 mils | Ferrous, Non-Ferrous | LCD with Backlight | Precision Probe (Black)
Brand: CARSYS
Color: Black
The device DPM-816 is used to measure thickness of various non-metallic coatings (paint, primer etc) over metallic base (steel, iron, alloys, aluminum etc).
An example of case for use would be to measure paint layer thickness of car body panels.

The device has 5 exclusive modes and has a wide range of uses:

Auto:

Fast&Easy mode, automatically determines the measurement method. Perfect for simple checking the car body. No additional settings required.

Manufacturing:

Automatically determines the measurement method. Higher resolution for thick coatings. Great for factories, paint shops and workshops.

Program:

Automatically determines the measurement method. Used for custom calibration at 6 points. For special cases when the default calibration is not suitable. Advanced mode.

Fe - Magnetic induction method:

Created for the most accurate measurement of non-magnetic coatings (paint, polymer coatings, powder coatings, electroplating) on magnetic metals (steel, iron). Has the function of continuous measurement. * Manual selection of measurement method

nFe - Eddy current method:

Created specifically for the most accurate measurement of dielectric coatings (paint, powder paint, polymer, plastic) on non-magnetic (non-ferrous) metals (aluminum, zinc, copper). Has the function of continuous measurement. * Manual selection of measurement method.

- Measurement range: 0-137 mils)

- Resolution: 0.01-0.1-0.5 mils.

- Accuracy (0-28 mils): ± 1% + 0.04 mils.

- Accuracy (28-99 mils): ± 1% + 0.4 mils.

- Measurement units: mm / microns / mil.

- Wide operating temperature range -13 to 104 F.

This set includes:
- Digital coating thickness gauge (the device).
- Aluminum and Steel calibration base.
- Calibration plates 500µm and 1000µm thick.
- User manual.

Order online right now and enjoy the most accurate Coating Thickness Gauge ever!
Wide range; Accurate Fe/nFe sensor; Resolution up to 0.01 mils; Accuracy up to 1%+0.04 mils
Auto recognition for ferrous and non-ferrous substrates (distinct sound signals for each)
Fast measurements in Single (up 40 times per min) or Continuous mode (4 times per sec)
Easily switch measuring units between mils/μm/mm; Calibrate your device in just 30 seconds
Automatically switches to Winter Mode in cold weather and Adjusts to the Current Environment