LEAGY VICTOR VC60B+ Digital Insulation Resistance Tester Megohm Meter DC250/500/1000V AC750V Orange with Black
Brand: LEAGY
Color: Black
ntroduction: The whole instrument is stylish and superior grade in appearance and highly stable in performance. It can be operated by both hands through using a shoulder strap. It is most suitable for the needs of insulation resistance check, such as electric, cable, electrical and mechanical equipment, telecommunication equipment and electrical facilities.

Specification:
Model: VC60B+
Measuring Voltage: 50V±10% 500V±10% 1000V±10%
Insulation Range: 0.1MΩ~20 MΩ 0.1MΩ~50 MΩ 0.1MΩ~100 MΩ 20MΩ~500 MΩ 50MΩ~1000 MΩ 100MΩ~2000 MΩ
Accuracy: ±(4%+2)
Current of Short Circuit: 1.8mA 1.8mA 1.6mA
Median Resistance: 2MΩ 2MΩ 5MΩ
Terminal Position: L & E
Voltage Test(AC750V) Accuracy: ±(1.0% of reading + 6 digits)
Input Impendance: 1MΩ
Frequency Response: (50~200) Hz
Terminal Position: ACV & G
Alarm Function Red indicator light is illuminated
Low Battery Indication: Yes
Over Load Indication: Yes
Unit Symbol Display: Yes
Auto Power Off: Yes
Max. Display: 1999
Over Range Display: “1”
Operating Temperature: 0℃~40℃
Relative Humidity: 30%RH~85%RH
Dimension: 105×164×70mm
Weight: 900g
Power: 9V,6 x AA/1.5V battery (Included)

Package Include:
1 x Digital Insulation Tester
1 x 10A Test Leads
2 x Alligator Clip
1 x Shoulder Strap
1 x User Manual
6 x AA/1.5V battery
Measure Range up to 2000MΩ 1000V with Accuracy of ±(4% of reading +2 digits)
Large LCD display (84.8*59.8mm), with backlight, max display "1999"
Data hold and auto power off
Over Range indication: display "1"
Low power consumption: Less than 300mw at unload measurement