Proster LCR Meter LCR Multimeter Tester for Capacitance Resistance Inductance Measuring Meter with LCD Over-range Display
Brand: Proster
Color: Yellow
Description:
Proster inductance, capacitance and resistance measuring meter,it is a Specialdigital instrument which is easy to be operated, the reading accuracydegree is higher with liquid crystal display 3 1/2.
It adopted double integralA/D converter core and a large-scale integrated circuits,these make the meter anexcellent performance instrument for the users.

Fetures:
1.Display: LCD
2.Biggest display value: 1999 (3 1/2).
3.Measuring method: Double integral A/D conversion.
4. Sampling rate: 3 times per second
5. Overrange display: "1"at the higgest digit
6.Auto power off and Data Hold function
7.Working temperature: 0℃ to 40℃,
8.Storage temperature: -10℃ to 50℃
9.Power: a 9V laminated battery(6F22)

Specifications:
Accuracy:±(%reading+digits)
Environment temperature: 23℃±5℃,Relativehumidity: <75%

Capacitance:
Range:200pF;Accuracy:±(2.5%+5);Resolution:0.1pF
Range:2nF;Accuracy:±(2.5%+5);Resolution:1pF
Range:20nF;Accuracy:±(2.5%+5);Resolution:10pF
Range:200nF;Accuracy:±(2.5%+5);Resolution:100pF
Range:2uF;Accuracy:±(2.5%+5);Resolution:1nF
Range:20uF:Accuracy:±(2.5%+5);Resolution:10nF
Range:200uF;Accuracy;±(2.5%+5);Resolution:100nF
Range:2000uF;Accuracy:±(5%+5);Resolution;1uF
PF=10-12F;nF=10-9F;μF=10-6F;mF=10-3F

Inductance:
Range:200uH;Accuracy:±(3%+5);Resolution:0.1uH
Range:2mH;Accuracy:±(2%+5);Resolution:1uH
Range:20mH;Accuracy:±(2%+5);Resolution:10uH
Range:200mH;Accuracy:±(2%+5);Resolution:100uH
Range:2H;Accuracy:±(2%+5);Resolution:1mH
Range:20H;Accuracy:±(5%+5);Resolution:10mH
μH=10-6H ;mH=10-3H

Forward voltage drop of diode:
Displaty forward voltage drop of diode;
Forward voltage of DC:Approx.1mA
Reverse DC voltage:Approx.2.8V

Package incluedes:
1*LCR meter
1*pair of crocodile clip
1*9V battery
1*user manual
1*Portable bag
1*pair of mini Alligator Clip
1*pair of Test Loads1*screwdriver
Proster LCR Meter: inductance, capacitance and resistance measuring meter,it is a Specialdigital instrument which is easy to be operated, the reading accuracydegree is higher with liquid crystal display 3 1/2.
High Accuracy : (Inductance) Measuring accuracy up to ±(2%+5); (Capacitance) Measuring accuracy up to ±(2%+5)Forward voltage drop of diode:Displaty forward voltage drop of diode;Forward voltage of DC:Approx.1mA;Reverse DC voltage:Approx.2.8V
Rotable LCD Display : Multi angle adjustment for easy reading.There is no need to hold tester while measuring, more convenient operation.
Multi-function: Auto power off; Data Hold function; Low Power Indication; ZERO ADJ for Capacitance.
New Version Test Leads: Comes with crocodile clip/ Test Loads/ mini Alligator Clip; The black pen is connected to the negative pole and the red test lead is connected to the positive pole