NOYITO USB Logic Analyzer Device with USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug
Brand: NOYITO
Description
Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz;
The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions;
8-channel;
Supports the latest version: Version 1.1.15

Voltage range:
Input voltage range: -0.5V to 5.25V
Input Low Voltage: -0.5V to 0.8V
Input High Voltage: 2.0V to 5.25V
Works with 5V, 3.3V, 2.5V, 2.0V systems. May work with 1.8V but not recommended.
Input Impedance: 1Mohm || 10pF (typical, approximate)
Crystal: ±20ppm, 24MHz
Error/Accuracy: pulse-width measurement: ±42ns (at 24MHz).

Packing Included:
1xLogic Analyser
1xUSB Cable
1xDupont Cable

Download Address:
https://www.amazon.com/clouddrive/share/NFh0JRc8M2JpCevtrtNEl2WQ2Bblp5Uo3HnLFdsSjEA
The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions; 8-channel.<br>
Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz;<br>
The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions;<br>
Input voltage range: -0.5V to 5.25V; Input Low Voltage: -0.5V to 0.8V; Input High Voltage: 2.0V to 5.25V<br>
Input Impedance: 1Mohm || 10pF (typical, approximate); Crystal: ±20ppm, 24MHz.<br>