Wirbelstrom
The principle of Lorentz force velocimetry is based on measurements of the Lorentz force that occurs due to the flow of a conductive fluid under the influence of a variable magnetic field. According to Faraday's law, when a metal or conductive fluid moves through a magnetic field, eddy currents generate there by electromotive force in zones of maximal magnetic field gradient (in the present case in the inlet and outlet zones). Eddy current in its turn creates induced magnetic field according to Ampère's law. The interaction between eddy currents and total magnetic field gives rise to Lorentz force that breaks the flow. By virtue of Newton's third law "actio=reactio" a force with the same magnitude but opposite direction acts upon its source - permanent magnet. Direct measurement of the magnet's reaction force allows to determine fluid's velocity, since this force is proportional to flow rate. The Lorentz force used in LFV has nothing to do with magnetic attraction or repulsion. It is only due to the eddy currents whose strength depends on the electrical conductivity, the relative velocity between the liquid and the permanent magnet as well as the magnitude of the magnetic field.