血浆
The measurement cuvette with the blood plasma sample is placed inside the water thermostat. Clotting starts when activator with immobilized TF is immersed into the cuvette. The clot then propagates from the activating surface into the bulk of plasma. Image of growing clot is registered via the CCD camera using a time-lapse microscopy mode in scattered light and then parameters of coagulation are calculated on the computer. Thrombodynamics analyser T-2 device also supports measurement of spatial dynamics of thrombin propagation during the process of clot growth via usage of the fluorogenic substrate for thrombin. Blood plasma sample is periodically irradiated with the excitation light and the emission of the fluorophore is registered by CCD camera.