B&K Precision 879B Dual Display Handheld Deluxe Universal LCR Meter with Backlit Display
40000 Count handheld LCR meters measures inductance, capacitance, and resistance quickly and precisely. The meters also includes handy functions such as data hold, Minimum/Maximum/Average recording, tolerance sorting, and relative mode. Measurement data can continuously transfer to a PC via the meter's mini USB interface, using either the provided data logging software or SCPI commands sent from a custom program. Fast auto range design for rapid, easy component measurements. USB (Virtual COM) interface. Software for datalogging and front panel emulation available as free download. Selectable auto-power-off options. Applications: Passive component trouble shooting. Electronic assembly. Quality control (component sorting). Measurement parameters: L/C/R/Z/D/Q/Theta/ESR. Test frequency accuracy is 0.02 percent of actual test frequency: 100 Hz, 120 Hz, 1 KHz, 10 kHz (test setting), 100 Hz, 120.048 Hz, 1 kHz, 10 kHz (actual frequency). Tolerance mode: 1 percent, 5 percent, 10 percent, 20 percent. Backlit display: yes Test signal level: almost equal to 0.6 Vrms. Measuring circuit mode: Series mode/Parallel mode. Basic accuracy: 0.5 percent. Ranging mode: Auto. Measuring terminals: 3 terminals with sockets. Measurement rate: 1.5 reading /second (range auto search not included). Response time: almost equal to 680 ms/DUT. Auto power-off: 5, 15, 30, 60 minutes, None. Operation temperature: 32 degree F to 104 degree F; 0-70 percent R.H. Storage temperature: -4 degree F to 122 degree F (-20 to +50 degree C); 0-80 percent R.H. Low battery indication: almost equal to 6.8 V. Battery life: almost equal to 16 hours using Alkaline Battery at 1 kHz with 100 ohm DUT, with backlight off. Power requirements: 1) DC 9V Battery, 2) Ext. DC Adaptor: DC 12 Vmin -15 V Maximum (Load 50 mA Minimum.). Measures 7-1/2" length by 3-1/2" width by 1-19/32" height. 1 year warranty.