LOW FREQUENCY SIGNALGENERATOR General description RAG-101 low frequency signal generator can generate 10Hz～1MHz test signal with low distortion, It attenuate gradually as 10dB from 0dB to 60dB, It is popular with manufactures、repair staffs and electronic enthusiast. Synchronization signal terminals control the consuming signal with a tiny control signal. Basic characteristics Frequency range: 10Hz～1MHz 5 test rangePrecision: Full scope 5%Output impedance: 600ΩOutput control: 0 dB、-10 dB、-20 dB、-30 dB、-40 dB、-50 dB adjustable。  Sine wave output1、Output voltage：5～6V virtual value2、distortion ratio:500 Hz～50 kHz  ≤0.05%`1' 50 Hz～500 kHz 0.5%3、Output smoothed line：11.5 dB ( Reference ) Waveform output 1、 Scope：10 Hz～1 MHz2、  Output voltage：10 VP-P3、   Rise time : 0.5 μs Synchronization characteristics 1、Range: Frequency can be changed 1% as 1V voltage output 2、Input impedance: 10 KΩPower supply: AC 110 V / 220 V (50 / 60 Hz), 8 W