Testo 0563 5572 Refrigeration System Analyzer Kit with 2 Clamp Probe and Hard Carry Case, -58 to 302 Degree F Range
Testo refrigeration system analyzers (RSA) are high performance instruments that accurately perform temperature and pressure measurements in seconds. These instruments provide fast, real-time superheat and subcooling readings for 39+ profiles in real-time. Built rugged for the field, testo RSAs give you the accuracy and repeatability you’ve come to expect from testo. Performance, value and speed cool. Real-time superheat and subcooling readings. Simultaneous digital readout of temperatures and pressures. Pipe clamp temperature probe (0.2 - 2.5" diameter). Built-in magnifying sight glass. Two NTC temperature probe inputs. Fast response time. Compact design for easeof-use transport and setup. Large easy to read back-lit display. Hardened housing that protects the analyzer. Ambient temperature and barometric pressure compensated. Rugged case. Aluminum valve block. Temperature compensated tightness test. Surface temperature compensation factor. Heat pump mode. Two temperature probe inputs for simultaneous real-time calculation of superheat and subcooling. Compact design for ease of use, transport, and setup. 40 refrigerant profiles stored eliminating the use of PT charts. Operating temperature: 14 to 122 degree F. Storage temperature: -4 to 140 degree F. Battery life: approximate 150 hr (w/o illumination). Measures 11" length by 5.5" width by 3" height. Pressure media: CFC, HFC, N, H2O, CO2. Pressure measuring range: -14.7 to 725 psi. Pressure accuracy (at 72 degree F): +/-0.75 percent fs (+/-1 digit). Pressure resolution: 0.1 psi/0.01 bar. Measuring range: -58 to 302 degree F. Temp. accuracy (at 72 degree F): +/-1 degree F (+/- 1 digit). Temperature resolution: 0.1 degree F , 0.1 degree C. Probe connections: 2x plug-in (NTC). Refrigerants: 40. Measures 5-1/2" length by 3" width by 11" height.