Digital Oscilloscope RIGOL DS2072A
RIGOL DS2072A digital oscilloscope is a new high-quality dual-channel oscilloscope (70 MHz). This compact lightweight digital oscilloscope provides exceptionally accurate waveform viewing and measurements. RIGOL DS2072A provides up to 50 000 wfms/s waveform acquisition rate.RIGOL DS2072A digital oscilloscope is ideal for production tests, field service, research, design and all other applications involving analog/digital circuit tests and troubleshooting, as well as education and training.Technical SpecificationsBandwidth 	70 MHzChannels 	2Real time sample rate 	2 GSa/s (single-channel), 1 GSa/s (dual-channel)Memory depth 	14 Mpts (Std.), 56 Mpts (Opt.)Waveform acquisition rate 	50,000 wfms/s (Max. )Vertical sensitivity range 	500 μV/div - 10 V/divOffset range 	500 μV/div - 50 mV/div ( 2 V)51 mV/div - 200 mV/div ( 10 V)205 mV/div - 2 V/div ( 50 V)2.05 V/div - 10 V/div ( 100 V)Vertical resolution 	8 bitsMath 	Add, Subtract, Multiply, FFT, editableOutput voltage 	~ 3 V, peak-peakFrequency 	1 kHzGeneral SpecificationsTrigger modes 	Auto, Normal, SinglePower 	50 W (max.)Power voltage 	100-240 V, 45-440 HzTemperature range 	Operation: 0C to +50CNon-operation: -20C to +70CCooling method 	FanStandard Ports 	USB host (support USB-GPIB), USB device, LAN, Aux (TrigOut/PassFail)Display 	8" WVGA color LCD (800480)Package ContentsRIGOL DS2072A digital oscilloscope (1 pc.)Passive probes (2 pcs.)Power cable (1 pc.)USB data cable (1 pc.)Quick guide (1 pc.)CD (Software and User's guide) (1 pc.)