MASTECH MS2108A 400 AC DC Current Clamp Meter
Features:Perform measurements of AC/DC voltage and current, resistance, frequency, duty, capacitance, as well as continuity and diode test.Display: 4000 counts with analog bar LCD displayMaximum voltage between terminals and earth ground: DC 1000V or AC 750VOver range protection for all rangesReading hold functionSquare value measuring function (at AC A and AC V range)Maximum & minimum value measuring functionConverte rate: 3 times/secBar graph: 30 times/secAuto power offContinuity testDiode testMAX diameter for conductor:Ф26mmThis meter has been designed according to IEC-1010 concerning electronic measuring instruments with an overvoltage category CAT Ⅲ600V and pollutionSpecifications:Model:MS2108ADouble Insulation: Protection class CAT III 600V and pollution degree 2Display: 4000 countsDC Voltage Accuracy: 400mV1%,4V/40V/400V7%,1000V 0.8%AC Voltage Accuracy: 4V/40V/400V0.8%,750V 1.0%DC Current Accuracy: 40A/400A 2%AC Current Accuracy: 40A/400A 2%Resistance Accuracy: 400Ω/4kΩ/40kΩ/400kΩ 0.8%, 4MΩ/40MΩ 1.2%Capacitance:0nF/400nF/4μF/40μF/400μF/4000μF4%Frequency Accuracy:10Hz/100Hz/1KHz/10kHz/100kHz/1MHz/10MHz1.5%, 10~10KHz2%Duty cycle: 0.1% ~ 99% 3%Operating Temperature:0℃ ~ 40℃ (32F ~ 104F)Storage Temperture:-10℃ ~ 50℃(10F ~ 122F)MAX. diameter for Jaw:Ф26mmDimension:208mm x 78mm x 35mmPower:1.5V (AAA) x 3Weight:340gNote: Battery is not included