Reed R5001 LCR Meter, 20 MicroHenry to 2000 Henry Range, 0.001 MicroHenry Resolution, +/-0.2% Accuracy
Measure inductance/capacitance/resistance with secondary parameters including dissipation factor (D), quality factor (Q), phase angle, equivalent series/parallel resistance (ESR or Rp). Autoranging operation for AC impedance and DE resistance measurement. Auto LCR smart check and measurement. Series/parallel modes. Ls/Lp/Cp with D/Q/ESR parameters. Support DCR mode 200 to 200M ohm. High accuracy with 4 or 5 terminal Kelvin measurement. Testing frequency is up to 10kHz and output frequency is up to 100kHz. Dual display with backlight. Data hold function. Low battery indication with auto power off. Dust and water proof to IP67. Includes test leads, batteries and carrying case. Inductance: 20µH to 2000H range, 0.001µH resolution, +/-(0.2 percent +5 digit). Capacitance: 20pF to 20mF range, 0.001pF resolution, +/-(0.2 percent +3 digit.). 100/120/1k/10k/100k Hz Test frequencies. 0.5 VAC rms typical test signal level. 6 x 1.5V AA batteries power supply. Approx. 30 hours battery life. Measures 96mm width by 220mm height by 60mm depth.