Testo 0563 5505 RSA Basic Refrigeration System Analyzer Kit with Clamp Probe, -58 to 302 Degree F Range
Testo refrigeration system analyzers (RSA) are high performance instruments that accurately perform temperature and pressure measurements in seconds. Includes clamp probe, batteries and certificate. These instruments provide fast, real-time superheat and subcooling readings for 39+ profiles in real-time. Built rugged for the field, Testo RSAs give you the accuracy and repeatability you've come to expect from Testo. The testo 550 RSA is ideal for servicing, maintaining and commissioning, refrigeration and heat pump systems including the newest 16 SEER systems. Compact design, hardened housing and extremely convenient operation for ease-of-use, transport and setup. Measurement of low and high pressure as well as calculation of temperatures, fast and accurate. Performance, value and speed cool. Real-time superheat and subcooling readings. Simultaneous digital readout of temperatures and pressures. Pipe clamp temperature probe (0.2 - 2.5" diameter). Built-in magnifying sight glass. Two NTC temperature probe inputs. Fast response time. Compact design for easeof-use transport and setup. Large easy to read back-lit display. Hardened housing that protects the analyzer. Ambient temperature and barometric pressure compensated. Rugged case. Aluminum valve block. Operating temperature: 14 to 122 degree F. Storage temperature: -4 to 140 degree F. Battery life: approximate 150 hr (w/o illumination). Pressure media: CFC, HFC, N, H2O, CO2. Pressure measuring range: -14.7 to 725 psi. Pressure accuracy (at 72 degree F): +/-0.75 percent fs (+/-1 digit). Pressure resolution: 0.1 psi/0.01 bar. Measuring range: -58 to 302 degree F. Temp. accuracy (at 72 degree F): +/-1 degree F (+/- 1 digit). Temperature resolution: 0.1 degree F , 0.1 degree C. Probe connections: 2x plug-in (NTC). 39 refrigerant profiles built in for automatic PT calculations. Measures 4-1/2" length by 2-1/2" width by 8" height.