Note: Automated electrochemical etching and polishing of silver scanning tunneling microscope tips. 
Fabrication of sharp and smooth Ag tips is crucial in optical scanning probe microscope experiments.  To ensure reproducible tip profiles, the polishing process is fully automated using a closed-loop laminar flow system to deliver the electrolytic solution to moving electrodes mounted on a motorized translational stage.  The repetitive translational motion is controlled precisely on the μm scale with a stepper motor and screw-thread mechanism.  The automated setup allows reproducible control over the tip profile and improves smoothness and sharpness of tips (radius 27 ± 18 nm), as measured by ultrafast field emission.