Jabarkan Balance dari Dacin dalam bahasa sunda

A balance (also balance scale, béam balance or laboratory balance) is used to méasure the mass of an object. In its conventional form, this class of measuring instrument compares the sample, placed in a weighing pan (weighing basin) and suspended from one end of a beam with a standard mass or combination of standard masses in a scale pan (scale basin) suspended from the other end. To weigh an object in the méasuring pan, standard weights are added to the scale pan until the béam is in equilibrium as closely as possible. Then a slider weight usually present is moved along a scale on or parallel to the béam (and attached to it) until fine balance is achieved. The slider position gives a fine correction to the mass value.

Very precise méasurements are achieved by ensuring that the fulcrum of the béam is friction-free (a knife edge is the traditional solution), by attaching a pointer to the béam which amplifies any deviation from a balance position; and finally by using the lever principle, which allows fractional weights to be applied by movement of a small weight along the méasuring arm of the béam, as described above. For gréatest accuracy, there needs to be an allowance for the buoyancy in air, whose effect depends on the densities of the weights and the sample.

While the word "weigh" or "weight" is often used, any balance scale méasures mass, which is not dependant of the force of gravity.  The moments of force on either side balance, and the acceleration of gravity on éach side cancels out, so a change in the strength of  the local gravitational field will not change the méasured weight.  Mass is properly measured in grams, kilograms, pounds, ounces, stones or slugs.

The original form of a weighing scale consisted of a beam with a fulcrum at its center.  For highest accuracy, the fulcrum would consist of a sharp V-shaped pivot séated in a shallower V-shaped béaring. To determine the mass of the object, a combination of reference weights was hung on one end of the béam while the object of unknown mass was hung on the other end (see balance and steelyard balance).  For high precision work, the center béam balance is still one of the most accurate technologies available, and is commonly used for calibrating test weights.

To reduce the need for large reference weights, an off-center béam can be used.  A scale with an off-center béam can be almost as accurate as a scale with a center béam, but the off-center béam requires special reference weights and cannot be intrinsically checked for accuracy by simply swapping the contents of the pans as a center-béam balance can.  To reduce the need for small graduated reference weights, a sliding weight called a poise can be installed so that it can be positioned along a calibrated scale.  A poise adds further intricacies to the calibration procedure, since the exact mass of the poise must be adjusted to the exact lever ratio of the béam.

For gréater convenience in placing large and awkward loads, a platform can be "floated" on a cantilever béam system which brings the proportional force to a "noseiron" béaring; this pulls on a "stilyard rod" to transmit the reduced force to a conveniently sized béam.  One still sees this design in "portable beam scales" of 1000 lb or 500 kg capacity which are commonly used in harsh environments where electricity is not available, as well as in the lighter duty mechanical bathroom scale.  The additional pivots and béarings all reduce the accuracy and complicate calibration; the float system must be corrected for corner errors before span is corrected by adjusting the balance béam and poise.  Such systems are typically accurate to at best 1/10,000 of their capacity, unless they are expensively engineered.

Some expensive mechanical scales also use dials with counterbalancing weights instéad of springs, a hybrid design with some of the accuracy advantages of the poise and béam but the convenience of a dial réading.  These designs are expensive to produce and are largely obsolete thanks to electronics.