QPX600DP - "Incorrect Configuration" (Updated with bounty offer) - Page 1
Forum section: electronics / repair

author: Mars
author_rank: Contributor
timestamp: August 27, 2023, 01:28:58 pm
content: I have a LXI QPX600DP with no strings attached where, after turn on, the LCD display saysQuoteAre there any obvious known ways forward here?

author: voltsandjolts
author_rank: Unknown
timestamp: August 27, 2023, 01:32:46 pm
content: Maybe some onboard parameters are corrupt, but I would start with a very thorough visual inspection and unplug/re-plug of everything pluggable.

author: Mars
author_rank: Contributor
timestamp: August 27, 2023, 01:37:30 pm
content: Thanks. I've not positively exhausted every possible oxidized connection, but I've gone over most of the reachable stuff with Deoxit etc. I've also done a superficial visual inspection and it looks in pristine condition (which is why I'd like to get it working again).

author: voltsandjolts
author_rank: Unknown
timestamp: August 27, 2023, 02:04:24 pm
content: IME TTi / Thurlby Thandar stuff is well built and repairable with off the shelf parts - no custom asics etc. There is firmware to consider though. Please post some pictures of the inside for all us TTi fans

author: Mars
author_rank: Contributor
timestamp: September 02, 2023, 05:00:42 pm
content: Do you think this "Incorrect Configuration" could be caused by a hardware error though? Have you seen anything like that on other similar TTi gear? I'm not sure what to do with this thing, but I'm reluctant to throwing it out without being confident there's no reasonably easy fix here. A more thorough disassembly is next on my list, but it would be nice if someone had seen this error before and could give me the encouragement I need to bother with this.I'll get back to you with pictures later.

author: glitchcatcher
author_rank: Contributor
timestamp: November 15, 2023, 10:00:55 pm
content: Have you made any progress in this?Unfortunately I cannot help here, but I was thinking about purchasing a unit with such an error.Pictures of the inside would indeed be nice.

author: Mars
author_rank: Contributor
timestamp: November 25, 2023, 01:01:32 pm
content: I have not, no. But I have not tried either. I actually have two of these where the other one is a different model, but otherwise in similar good condition etc, so while I'm not actively trying to find a solution here, I do have some incentive to do so. But it would require encouragement in some form or other, which at the moment I don't have. Anyway, here is a picture. I started on a block diagram a while back that I wanted to post at the same time as the pictures, but got bored and never finished and now I can't even find it, so this will have to do for now.

author: glitchcatcher
author_rank: Contributor
timestamp: November 25, 2023, 04:54:22 pm
content: Thanks for the picture. I could only find a QPX1200 service manual which contains circuit diagrams but no block diagram, although it has circuit descriptions :-)

author: Mars
author_rank: Contributor
timestamp: November 26, 2023, 12:35:24 am
content: The other model that I mentioned is the QPX1200SP (who doesn't need 20A@60V, right?), which has the same error.In a short moment of inspiration I tried to re-flash the unit over USB with the TTi flash utility and a QPX600_ALL_209.hex file I found on their website. FYI, the "LAN reset" button on the back has to be pressed in while you power on the unit to get it into this mode. But even after installing the usb drivers for windows, the utility just sat there seemingly stuck at doing nothing so I gave up. Else it responds over Ethernet and I'm able to access the built in web server where you can read out info about the instrument and a few other things. So overall it feels like there should be some kind of fairly simple fix here, but I don't know what it is.

author: Mars
author_rank: Contributor
timestamp: February 10, 2024, 10:04:47 pm
content: If someone can find a solution that fixes both these units I'll give you the QPX1200SP for free and ship it to you (or someone else if you want to sell it) if you pay the shipping. I'll provide on-the-ground support and will help with whatever needs to be done, but I have limited time so I'm not gonna sit nights on end trying to debug this thing. I can provide pictures if you insist, but they are both spotless appearance wise if that helps your motivation...

author: CLOLUS
author_rank: Newbie
timestamp: April 24, 2024, 09:04:27 pm
content: Hi,I have a QPX600D with the same problem of message "Incorrect Configuration". I find on the web the shematic diagram of this power supply :https://everist.org/NobLog/pics/20151112/sch_ps/HP_54120B_power_supply.htmThe fuse 15AT was broken and I changed two diode in short circuit (US1J 600V 1A) ref D36 D37. The power supply is on now without problem with the Fuse, but i have this message of "Incorrect configuration".I don't know how to solve this problem. I checked the documentation without success...

author: CLOLUS
author_rank: Newbie
timestamp: May 11, 2024, 05:18:29 pm
content: Hello,I find the solution of the message "Incorrect configuration". It's not a problem of software but a electrical problem on the main board. I changed many components and now it's ok. I loaded the ultimate firmware 2.09 without error message. I find a shematic of the QPX1200 but I would like to have the manual service of the QPX600D.