Sigrok-Support for APPA 150/208/506 / BENNING MM 12 / CM 12 / Sefram 7352B / etc - Page 1
Forum section: electronics / testgear

author: Cymaphore
author_rank: Regular Contributor
timestamp: November 02, 2020, 02:06:53 pm
content: Hi,I created a driver for lots of APPA-based devices for sigrok this weekend. This includes Multimeters and Clamp Multimeters with USB/Serial and Bluetooth LE.Currently the driver supports the BENNING MM 12, BENNING CM 12, Sefram 7352B, APPA 506(B), 208(B), 155B, 156B, 157B, 158B and a couple of others under all kinds of brands.I implemented and tested it with a BENNING MM 12 (APPA 506B). Functions for the other meters with the same communication protocol are implemented, but obviously untested, since I don't have the hardware.Currently all live reading functions for USB/Serial/BLE are implemented, but log download and calibration will follow when I have the time.Sigrok-Wikipage for the meter I used, including some pictures:https://sigrok.org/wiki/BENNING_MM_12So if anyone here has one of those APPA 150/208/506 based meters, regardless of the brand, I would be happy about any tests with the new sigrok driver.Here's my github-fork of libsigrok containing the driver "appa-b":https://github.com/Cymaphore/libsigrokThe driver is currently under review and will be included with mainline sigrok once it passed.Thanks and best regards,Martin

author: Cymaphore
author_rank: Regular Contributor
timestamp: November 04, 2020, 11:24:52 am
content: Number of supported models has just increased:Code:Supported Models:ID | Model | Brand Name | OPT | BLE | State------|-----------|-----------------------|:---:|:---:|---------------0x01 | 150B | APPA 155B | | X | untested0x01 | 150B | APPA 156B | | X | untested0x01 | 150B | APPA 157B | | X | untested0x01 | 150B | APPA 158B | | X | untested0x01 | 150B | BENNING CM 12 | | X | untested0x0c | 172 | APPA 172B | | X | untested0x0d | 173 | APPA 173B | | X | untested0x0d | 173 | BENNING CM 10 | | X | untested0x0e | 175 | APPA 175B | | X | untested0x0f | 177 | APPA 177B | | X | untested0x0f | 177 | BENNING CM 10-PV | | X | untested0x14 | 179 | APPA 179B | | X | untested0x03 | 208 | APPA 208 | X | | untested0x04 | 208B | APPA 208B | X | X | untested0x07 | 501 | APPA 501 | X | | untested0x08 | 502 | APPA 502 | X | | untested0x15 | 503 | APPA 503 | X | | untested0x15 | 503 | CMT 3503 | X | | untested0x15 | 503 | ISO-TECH IDM503 | X | | untested0x15 | 503 | Voltcraft VC-930 | X | | experimental0x16 | 505 | APPA 505 | X | | untested0x16 | 505 | RS PRO IDM505 | X | | untested0x16 | 505 | Sefram 7355 | X | | untested0x16 | 505 | Voltcraft VC-950 | X | | experimental0x05 | 506 | APPA 506 | X | | ok0x05 | 506 | Sefram 7352 | X | | ok0x06 | 506B | APPA 506B | X | X | ok0x06 | 506B | BENNING MM 12 | X | X | ok0x06 | 506B | Sefram 7352B | X | X | okN/A | 507 | APPA 507 | X | | experimental0x12 | A17N | APPA A17N | | X | untested0x13 | S0 | APPA S0 | | X | untested0x09 | S1 | APPA S1 | | X | untested0x09 | S1 | RS PRO S1 | | X | untested0x0a | S2 | APPA S2 | | X | untested0x0a | S2 | BENNING MM 10 | | X | untested0x0a | S2 | RS PRO S2 | | X | untested0x0b | S3 | APPA S3 | | X | untested0x0b | S3 | BENNING MM 10-PV | | X | untested0x0b | S3 | RS PRO S3 | | X | untested0x10 | SFLEX_10A | APPA sFlex-10A | | X | untested0x11 | SFLEX_18A | APPA sFlex-18A | | X | untestedSee also:APPA Multimeters - sigrokIf anyone has any test results with sigrok on one of those meters, please just let me know.

author: ratmole
author_rank: Contributor
timestamp: March 24, 2021, 10:54:50 pm
content: hello, looking for APPA 700 Communication Protocol, do you have one available?i'm trying to implement, expand your appa-dmm driver...