SOLD Rohde & Schwarz FSEA30 3.5GHz Spectrum Analyzer w/ OCXO, Tracking Generator - Page 1
Forum section: electronics / general

author: ci11
author_rank: Regular Contributor
timestamp: May 05, 2023, 10:44:33 pm
content: Here are the details:This refurbished low-usage, well-equipped FSEA30 includes the following factory installed and licensed options:1.B4 OCXO Low Phase Noise2. B5 FFT Bandwidth 1Hz to 1kHz3. B7 Vector Signal Analysis4.B9 Tracking Generator with I/Q Modulator5. B12 Switchable Attentuator for Tracking Generator6. B15 Computer Function with Windows NT4.0 and 2nd VGA Display connection7. B16 Ethernet Variant 2 – AUI connection8. B17 2nd IEEE-488 Bus Interface9. K10 GSM Mobile Station Analyzer Firmware10. K11 GSM Base Station Analyzer FirmwareSerialized factory option stickers can be clearly seen on the rear panel.Recent refurbishment includes:1.TFT LED backlight installation2. Settings Memory/RTCLithium Battery replacement3. Diskette Drive replacement4. Internal airflow clean-upPerformanceThe analyzer has been power-on cycled and tested for 4 weeks. Its OCXO oven has been powered up and stabilized during this time. The built-in full CAL routine has been run 3 times and passed all tests 100%. The DANL specification of -150dBm according to the datasheet has been confirmed. Also confirmed is functionality of the Ethernet adapter which included accessing the World Wide Web using a Firefox browser using an attached keyboard and mouse which are not included in this auction. Actual attenuator counts attest to the analyzer’s usage history.Listing photos show the external appearance with actual screenshots. Since these photographs do not show the real, vibrant color of the TFT, additional screenshot photos without ambient lighting have been added which reflect the actual display color more accurately.CosmeticsThe front and rear panels of the analyzer are in excellent shape with only minor blemishes. All pushbuttons and the encoder work well. The AF output connector has been tested and works correctly. The Probe/Code and Probe Power connectors have not been tested and condition is unknown.Enclosure panels show minor scratches from install and uninstall activity, and a number of rubber bumpers are missing. The 2 carry straps function well with minor rubber cracks on one and one end cap is also missing.

author: VanBudd
author_rank: Contributor
timestamp: May 06, 2023, 09:48:27 am
content: Hi,Would you ship to Germany?BR

author: ci11
author_rank: Regular Contributor
timestamp: May 06, 2023, 12:28:19 pm
content: Quote from: VanBudd on May 06, 2023, 09:48:27 amHello,I am happy to ship to Germany via UPS if you pay the shipping cost. It maybe about US$300 depending on your location based on the shipping box being 60cm x 60cm x 30cm and weighs in at about 33kg. Payment would be via a wire transfer to my bank.

author: N_basil
author_rank: Newbie
timestamp: September 21, 2023, 07:38:12 am
content: Hi!How do you take screenshots using FDD?Unfortunately, we used to have them done, but those skillets are now lost (. The user manual couldn't help (I would like to thank you very much in advance!

author: ci11
author_rank: Regular Contributor
timestamp: September 21, 2023, 12:46:26 pm
content: It's on page 4.73 of the R&S FSE Operating Manual 1065.6016.12. Good luck.

author: N_basil
author_rank: Newbie
timestamp: September 21, 2023, 02:41:43 pm
content: Thank you!I know it...But...Two years ago the device was under repair.And now only the following files are written to FDD:They can't be opened by any image editor, and I'm also not sure if they are actually screenshots (

author: ci11
author_rank: Regular Contributor
timestamp: September 21, 2023, 02:51:19 pm
content: I am not sure how anyone can help if your machine had been "repaired" and its behavior now deviates from the manual. Sorry.

author: N_basil
author_rank: Newbie
timestamp: September 22, 2023, 07:08:39 am
content: Thank you anyway!Sorry, but I remembered, this is done (saving a screenshot) from the menu "hardcopy"!Good luck.