Siglent SDS1052DL+ / SDS1102CML+ hack - Page 1
Forum section: electronics / testgear

author: miyagi
author_rank: Contributor
timestamp: September 24, 2019, 03:09:20 am
content: Hello，I have been searching around, but didn't find any info about Siglent SDS1052DL+ or SDS1102CML+ hack. Is there anyone know about this? Thank you very much!

author: tautech
author_rank: Super Contributor
timestamp: September 24, 2019, 09:54:06 pm
content: Quote from: miyagi on September 24, 2019, 03:09:20 amThe DL+ is only available in a 50 MHz model these days so I don't know if the internal SW supports higher bandwidth.However the CML+ models do as they come in 50, 100 or 150 MHz models.Sometime back I installed a later firmware onto a new SDS1102CML+ and for some reason it flashed it to SDS1152CML+ and my customer got a nice surprise !

author: rf-loop
author_rank: Super Contributor
timestamp: September 25, 2019, 08:02:30 am
content: Quote from: tautech on September 24, 2019, 09:54:06 pmSDS1000DL+/CML+ domestic markets sisters (A)带宽 50 MHz 、70 MHz 、100 MHz、150 MHz实时采样率SDS1052A：500 MSa/s <<=== This 1052 is exception in this serie with 500MSa/s max ADC system *)SDS1072A：1 GSa/sSDS1102A：1 GSa/sSDS1152A：1 GSa/s通道数 2 + EXT存储深度SDS1052A：32 KptsSDS1072A：2 MptsSDS1102A：2 MptsSDS1152A：2 Mpts*) and because this, imho, it is difficult to think possible SW change for 1GSa/s models (different HW)But what I have seen 52DL+ analog front end BW is really far over 50MHz. (perhaps front end like 1072A but different memory and ADC system)72A, 102A, 152A have same 40k short and 2M long memory and max 1GSa/s52A (as also 52DL+ outside of china) have 32k short memory alone and max 500MSa/sAlso inside these series systems there are not finnish guy Linus Thorwalds secretions and emissions.Only what I remember that I have seen somewhere in this forum is that someone have added 2M memory to SDS1000CNL so that is works like SDS1000CML after mod.(and CNL or CNL+ are 1GSa/s and are not same as DL or DL+ what have 500MSa/s)Previously there have been25 - 200MHz models in DL series (pdf)so hardware mod for rise analog front end bandwidth is perhaps possible because I think analog front end have not changed so much.

author: miyagi
author_rank: Contributor
timestamp: October 10, 2019, 01:51:45 am
content: Thank you Tautech and rf-loop. This confirms my assumption, so me and my friend did some work. The result is pleasing.This is verified on SDS1102CML+ with FW version 6.01.01.21R2 and SDS1052DL+ with FW version 6.02.01.10R2.The trick is the same.1) Input below command in a Notepad file and rename the file as startup.sh#! /bin/shtelnetdAnd then put the startup.sh file in a USB disk root directory, and insert the USB disk to scope. The telnet will be available.2) Connect scope to local network via LAN, set IP by DHCP.3) Open cmd window on PC, input command ' telnet xx.xx.xx.xxx '(means IP address of the scope)4) Enter the user name 'root', enter the password 'ding123'5) mount -o remount,rw,sync /usr/bin/siglent/firmdata06) rm /usr/bin/siglent/firmdata0/bandwidth.txt7) Restart oscilloscopeNow go to System Status page, the model name is changed. Tested with a generator, the bandwidth is indeed upgraded.

author: tv84
author_rank: Super Contributor
timestamp: October 13, 2019, 09:38:35 am
content: @miyagi, could you show me the output of this SCPI command on both scopes (CML+ and DL+)?MD5_PR?

author: miyagi
author_rank: Contributor
timestamp: October 15, 2019, 09:00:44 am
content: @tv84,It's SDS1000A

author: yippy
author_rank: Newbie
timestamp: October 26, 2019, 03:52:30 am
content: Quote from: miyagi on October 10, 2019, 01:51:45 amWould this work with the SDS1072CML+ ?

author: tv84
author_rank: Super Contributor
timestamp: October 26, 2019, 08:06:16 am
content: Quote from: yippy on October 26, 2019, 03:52:30 amMost probably. I sent you a pm.

author: tv84
author_rank: Super Contributor
timestamp: October 26, 2019, 10:15:12 am
content: BTW, The SDS CML+ has some key combinations in the FW. If anyone wants to try them, be my guest:Code:key_combination.13722 - education_mode_key0x0E, 0x2B, 0x2B, 0x2B, 0x2B, 0x2B, 0x2B, 0x24, 0x2B, 0x2Bkey_combination.13711 - record_pkg_ctl0x32, 0x27, 0x24, 0x32, 0x27, 0x24key_combination.13257 - set_update_cfg_flag0x32, 0x0E, 0x27, 0x09, 0x1E, 0x19, 0x26, 0x26, 0x26, 0x260x09 - Channel 10x0E - Channel 00x14 - AutoSet0x15 - Horiz Menu0x18 - Ref0x19 - Save/Recall0x1C - Trig0x1D - Math0x1E - Default Setup0x21 - Print0x23 - Single0x24 - Utility0x26 - Menu (2nd Menu)0x27 - Display0x28 - Run/Stop0x2B - Menu (2nd Menu)0x2D - Cursors0x32 - MeasureI'm not 100% sure of the keys decoding but I did my best. If anyone tries, please report.

author: yippy
author_rank: Newbie
timestamp: October 27, 2019, 01:47:08 am
content: Quote from: tv84 on October 26, 2019, 08:06:16 amJust did the SCPI command from your pm, rebooted and it's now showing as a SDS1152CML+. Thanks for all your help

author: Labrat101
author_rank: Unknown
timestamp: November 20, 2019, 11:39:23 am
content: Hi ,I saw your Hack . I have the BK 2190E which has the Siglent M/Board SDY8.007.107D CML 160700 .. some how I got screwed.The original bios was 6.01.01.19 so I got the update Bios from Siglent 6.01.01.21R2 .It excepted the update with no problem . But it did not solve my problem which is if I put 10Mhz same signal in on both channels .set it to maths - (minus) I should get a flat line. some how channel 2 is out of phase by 30degs .I am not sure if its a Bandwidth problem or software.I also noticed that in: Save/Recall/ Setups / Factory / Uptate BW , options at the moment it has the code "CCZ24QW9JKUCT6BZ" . which can be edited.Can someone verify this code ? or give me some help or suggestion on this. I have a Feeling that BK have rebranded this model and just added thereflash start up screen over Siglent'sAnd the strange difference between CH1 & CH2. I am using military grade cables 50 ohms of coarse. .Thanks

author: Labrat101
author_rank: Unknown
timestamp: November 22, 2019, 02:26:57 pm
content: Hi First i want to say thanks to the PM I got from a forum member who wishes to be anonymous ..Who walk me though the process of getting my Rebranded PK2190E back to being a real Scope again .And Now its showing its true color SDS1152 @ 150MHz Not as on the Box !00mhzA word of warning to anyone who is thinking of buying a PK2190E .. Stop .. Its a rebranded Siglent SDS 1000A series .. get an original Siglent they are cheaper..OK I got screwed . But my Guardian Angle came to my rescue . and debunked my Scope .O Thanks you .

author: tautech
author_rank: Super Contributor
timestamp: November 22, 2019, 06:22:28 pm
content: Quote from: Labrat101 on November 22, 2019, 02:26:57 pmNice, enjoy.QuoteYep, somewhat cheaper.QuoteNope, you just didn't do enough homework.There are 4 or 5 Siglent rebrands that I know of.Not all models are, and other manufacturers just pick from Siglents range what they need to help complete their lineup.Siglents most well known partnership is with LeCroy.

author: Labrat101
author_rank: Unknown
timestamp: November 22, 2019, 07:03:30 pm
content: Hi, I did not do a lot of home work.I just looked at what I had got and research that.If I had got the LeCroy. in side I would have been a very happy camperI needed my scope to finish some tests to make a customer happy.and had already spent 2 days chasing the dogs tail as they say looking for a problem that was not there.And the calibration function was not working correctly. Now it works and all problems solved.and my bench is clear for another week. and It can compares two channels .Thanks .I still say companies should tell the public that its some one else's engine inside. special big well known companies like PK. Maybe asking too much . I am Happy to have it as a Siglent .I look at things old school if I bought a BMW and found fiat engine under the bonnet.. Ferrari i'd except.No Hard feeling . All is good .

author: croma641
author_rank: Regular Contributor
timestamp: November 25, 2019, 02:41:13 am
content: Hi to all,I've got years ago an Atten ADS1062CAL. Upgraded at that times with the last Siglent firmware (perfectly working) :Software Version 5.01.02.32Hardware Version 6-41.2.1Product Type SDS106XThe result is a "Siglent" 60 Mhz version.My question is: can I upgrade it to 100 Mhz ? (Or more?)thanks in advance !P.S.:1) this oscilloscope have only USB and serial port...2) I've bought the RAM in order to upgrade this "Hybrid" to 2 Mpoints, now it has only 40Kpoints

author: DSE_Kid
author_rank: Newbie
timestamp: May 04, 2020, 06:09:49 am
content: Hi guys, I only recently replaced my 40 year old CRO with a new Siglent SDS1072CML+ and I saw that someone has successfully upgraded theirs from 70MHz to 150MHz by telnet-ing into it?Would anyone still have the info to do this that they could PM me?Many thanks

author: DSE_Kid
author_rank: Newbie
timestamp: May 06, 2020, 04:10:08 am
content: Ok so I've progressed through the instructions and updated my DSO to the latest firmware however I seem to be stuck and unable to TELNET into it because the 5025 ports are closed on my SDS 1072CML+How did you guys open those TCP/UDP ports up to allow telnet access?

author: miyagi
author_rank: Contributor
timestamp: May 06, 2020, 06:16:45 am
content: try port 23.

author: DSE_Kid
author_rank: Newbie
timestamp: May 06, 2020, 10:22:05 am
content: Thanks tv84 will do

author: Chris Roubis
author_rank: Contributor
timestamp: May 10, 2020, 04:31:24 am
content: Quote from: yippy on October 27, 2019, 01:47:08 amWOW great job guys.You mean all I need to do is punch in some scpi commands to my SDS1072CML+ to change it to a SDS1152CML+ ?

author: Labrat101
author_rank: Unknown
timestamp: May 10, 2020, 07:48:12 am
content: Yes ..But Remember that having a faster scope with low grade Probes .Will give the same results .A Good scope is only as Good as its input Connections ..

author: Chris Roubis
author_rank: Contributor
timestamp: May 10, 2020, 09:43:31 am
content: Just searched the internet for some instructions to tweak my SDS1072CML+Is this the same principle others are using?

author: tv84
author_rank: Super Contributor
timestamp: May 10, 2020, 10:26:47 am
content: Quote from: lordzeus on May 10, 2020, 09:43:31 amTry it. No harm can be done. If you don't succeed, maybe I can help.

author: Chris Roubis
author_rank: Contributor
timestamp: May 15, 2020, 04:51:20 am
content: Is it possible to use the EasyScopeX app that comes with the Siglent scope CD?I have the SDS1072CML+ already connected to my desktop win7 via USBTMC.Attached a screenshot of software and the commands as the poster Miyagi posted above.Nothing changed on my scope.Wonder what im doing wrong?

author: tv84
author_rank: Super Contributor
timestamp: May 15, 2020, 08:47:53 am
content: It's simple: you are mixing SCPI commands with linux commands!The only commands that you can send in that interface are SCPI commands like the *IDN?.

author: Chris Roubis
author_rank: Contributor
timestamp: May 16, 2020, 12:07:03 am
content: thanks to tv84 my scope is now a SDS1152CML+Blessings to you for your valuable know how.

author: allways60
author_rank: Newbie
timestamp: June 01, 2020, 07:26:29 am
content: Hi, I just ordered an SDS1072CML+ and would also like to convert it into an SDS1152CML+.Could you kindly send me a PM on how you did this.Thanks in advance.

author: yippy
author_rank: Newbie
timestamp: June 04, 2020, 08:10:29 am
content: Quote from: allways60 on June 01, 2020, 07:26:29 amGive tv84 a PM, he's helped a few of us upgrade our SDS1072CML+.

author: allways60
author_rank: Newbie
timestamp: June 04, 2020, 02:49:56 pm
content: tv84 is a legend! The hack worked for me too.Thank you all!

author: StillTrying
author_rank: Super Contributor
timestamp: June 05, 2020, 12:58:40 am
content: Quote from: allways60 on June 01, 2020, 07:26:29 amSpend some time testing all of its functions and modes that you're likely to use, before you "fix" it.

author: rf1911
author_rank: Newbie
timestamp: June 09, 2020, 01:17:33 pm
content: Hi,first of all thanks for all the work done.I'm trying to run the hack on SDS1052DL+, i updated to 6.02.01.10R2, i added the needed file to a fat32 formatted usb stick (which is recognized fine), but telnet port is not opened.TV84, what am i doing wrong?Thanks.

author: rf1911
author_rank: Newbie
timestamp: June 09, 2020, 01:46:57 pm
content: startup.sh to enable telnet, so to run the required commands.

author: tv84
author_rank: Super Contributor
timestamp: June 09, 2020, 04:33:39 pm
content: Quote from: rf1911 on June 09, 2020, 01:46:57 pmWhere did you read that such idea worked on a CML+??

author: rf1911
author_rank: Newbie
timestamp: June 09, 2020, 06:46:42 pm
content: I didnt say it would work on CML+, it's written on post #4 of this thread. I have SDS1052DL+. User Miyagi wrote it.Not sure at this point if i'm asking the correct question.

author: tv84
author_rank: Super Contributor
timestamp: June 09, 2020, 08:15:05 pm
content: Quote from: rf1911 on June 09, 2020, 06:46:42 pmSorry, I totally forgot such methods...Show me the output of this SCPI command:MD5_PR?

author: rf1911
author_rank: Newbie
timestamp: June 09, 2020, 09:32:00 pm
content: eheheh np, thanks.. this is the result:1: Write Operation (MD5_PR?)Return Count: 7 bytes2: Read OperationReturn Count: 9 bytesSDS1000A\n

author: AdrianFu
author_rank: Contributor
timestamp: June 26, 2020, 10:29:24 pm
content: This is still to be tested but.....Yesterday I updated the firmware on my SDS 1052DL+ to latest from Siglent after flashing the oscilloscope now reports it’s a SDS1202DL+Only reporting my experience may not work with other scopesAdrian

author: josempaz79
author_rank: Newbie
timestamp: July 21, 2020, 09:54:33 pm
content: Thank you very much TV84 for your guide to make a successful hack.

author: PiXEL8
author_rank: Contributor
timestamp: August 30, 2020, 04:52:45 pm
content: I recently acquired an SDS1202F+, it's a china only version of the SDS1202DL+The language is forced to chinese which can't be changed, but if I flash the latest fw or the previous 6.02.01.07 the language is changed to english until pressing the language button or having the unit off for some time.I was able to enable telnetd with the previous 6.02.01.07 fw only and dloaded the Siglent_CML.cfg. Is it possible to convert this unit to an SDS1202DL+ or set the default language to english?lastly is it possible to enable 1GSa/s and uncap the 200MHz bandwidth?Thank you for any all help with this.-PiXEL8

author: tv84
author_rank: Super Contributor
timestamp: August 30, 2020, 05:30:21 pm
content: Quote from: PiXEL8 on August 30, 2020, 04:52:45 pmShow me this SCPI result:MD5_PR?Nonetheless, why are you trying to achieve 200 MHz if you already bought a 200 MHz machine? ?

author: PiXEL8
author_rank: Contributor
timestamp: August 30, 2020, 06:04:24 pm
content: QuoteNot sure how to run SCPI commands yet. I'm unable to connect using EasyScopeX. Port scanning shows only 111 and 719 open (and 23/22 now that I have telnetd and dropbear running). No port 5024 or 5025 to telnet to.I was seeing if this scope could achieve more than 200MHz. I've seen some threads where removing a cap in the analog section would uncap the bandwidth.I found the SCDP utility. It connects and the title bar shows " Siglent,SDS1202F+,SDS1MCB4165473,6.02.01.07". Not sure if that's what you need or helps. The SCDP app throws a System.NullReferenceException error after connecting and when trying refresh.

author: tv84
author_rank: Super Contributor
timestamp: August 30, 2020, 06:19:46 pm
content: Quote from: PiXEL8 on August 30, 2020, 06:04:24 pmWon't happen. Another model...

author: tautech
author_rank: Super Contributor
timestamp: August 31, 2020, 12:21:21 am
content: Quote from: PiXEL8 on August 30, 2020, 04:52:45 pmI don't agree.F+ models have 1 GSa/s and mostly 1 Mpts memory while the 200 MHz version has just 32 Kpts which makes these some Chinese hybrid that we've never seen in the west.They don't match up with:SDS1kCNLSDS1kCMLSDS1kCFLSDS1kDLOr any of the later Plus versions.

author: rf-loop
author_rank: Super Contributor
timestamp: August 31, 2020, 02:48:38 am
content: Quote from: tautech on August 31, 2020, 12:21:21 amSDS1202F+ 200MHz 2通道500MSa/s（Max.） 32Kpts 7英寸（800*480）彩色显示屏But it is really bit confusing here due to also some "web shop" versions like example X-C models (aka Cheap E models) Or like top model SDS 6000 ProEven they specs looks quite same. 1202F+ is "web shop version" asGoogle bit fun translate it: "E-commerce SHOP Monopoly"But SDS1202DL+ is normal Siglent distribution model.This is whole mess here.If look Siglent factory official web sidesSDS1000CNL+/DL+SDS1102CNL+ 100 MHz 2 + EXT 1 GSa/s - 40 Kpts E-commerce MonopolySDS1102DL+ 100 MHz 2 + EXT 500 MSa/s - 32 Kpts E-commerce MonopolySDS1202DL+ 200 MHz 2 + EXT 500 MSa/s - 32 Kpts ¥3,180 (siglent side 0 VAT prices)------------------------------------SDS1000E+/F+SDS1202F+ 200 MHz 2 + EXT 500 MSa/s - 32 Kpts E-commerce SHOP MonopolySDS1122E+ 120 MHz 2 + EXT 1 GSa/s - 1 Mpts E-commerce SHOP MonopolyTaobao/TMALL both models same seller and it is not more but "siglent Dingyang flagship store"SDS1202F+ ¥ 2328SDS1202DL+ ¥ 3380Who believe they are same model and only other is chinese version and other not. Difficult to believe when look price difference. Same seller sell these bot just to domestic mainland market area.ETA: After reading both data sheets (bit tricky to get from Siglent sides documents side due to 404 if try F+ data sheet, but it is there when use other route via model info side...)I can not see any difference in data sheet between 1202DL+ and 1202F+exceptF+ can select only Chinese and DL+ have whole set of languages.Of course also F+ do not have normal overseas warranty. Language is expensive........If look example SDS1000X-E and X-CX-C is clearly more cheap "web shop" version. Not even priced in official sides.They are roughly same functionally. But C version wfm/s and Sequence mode speed is roughly half.Perhaps also there can find some differences in specs between 200MHz DL+ and F+.1000E+ is 1GSa/s model. with 1M memory but there is 120MHz version. It is priced in TMALL same seller as ¥ 1780.00But fun is that SDS1102X-C price is ¥ 1588.00and guess what... it is just like half wfm/s speed SDS1102X-E (this is not in overseas markets) with 1Gsa/s and 14M memory and perhaps it is.... we know overseas model SDS1202X-E... and 1202X-C is ¥ 2760.00 if compare to 1102X-C only diff is BW MHz but is is soft or HW made... try guess.1104X-C is ¥3780 and then 1104X-E is ¥ 5280 wfm speed difference and language (including chinese taxes)And I can imagine wfm/s speed is not half with all mem depths and time scales. Perhaps with some speed difference is less. What is internal reason for slower speed I do not want even guess.But even if it is not possible and 200MHz is not absolute go - no go question then compared to SDS1202F+ other ways this is light year ahead if this old generation F+. They are like night and day.And in some images it even looks like this X-C have also international languages but advertisement images are some times as they are...Perhaps some have used X-E images. Data sheet tell UI is only Chinese language.But as we know in this series peoples have done many nice things with serial terminal.If I am buying in China which one I select SDS1202F+ or SDS1102X-C. No need think even one second ifout from box200MHz BW is not absolute game end factor.Modifying DL+/F+ models are perhaps bit more or much more tricky than more modern models due to OS and HW.

author: tautech
author_rank: Super Contributor
timestamp: August 31, 2020, 03:36:43 am
content: Quote from: rf-loop on August 31, 2020, 02:48:38 amBut the 6k Pro will make it to the west but in LeCroy clothes.Later we may see it in an 8 bit version when 5kX requires a face lift.

author: rf-loop
author_rank: Super Contributor
timestamp: August 31, 2020, 03:39:07 am
content: Quote from: tautech on August 31, 2020, 03:36:43 amYepAlso previous msg edited about these low end models

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 04:37:48 am
content: QuoteHere is an image showing English language after flashing fw.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 04:41:15 am
content: Quote from: tautech on August 31, 2020, 12:21:21 amThis model uses the SDS1000DL+ firmware and manuals.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 04:45:10 am
content: Here is the Siglent support page for this scopehttps://www.siglent.com/products-overview/sds1000e/

author: rf-loop
author_rank: Super Contributor
timestamp: August 31, 2020, 05:21:54 am
content: Quote from: PiXEL8 on August 31, 2020, 04:45:10 am'My previous msg was just based to Siglent China sides because I am here. And also based to other knowledge about Siglent producs over around 10 years.Problem is that Im not chinese and I do not undesrtand language at all and google translation is not always available and if is result is many times nothing but fun. At this time I only need live here because due to Covid have not been possible to return my homeland.This your image is bit blurry so least I can not clear see text after FW update.After some image handling / filtering etc yes I can see it is english menu.Do you mean you just update it simply with DL+ FW using normal USB memory stick method without nothing else and after it there is full language set available.If this is case, how they can keep this price difference between these models example in web shops.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 03:20:27 pm
content: SDS1202F+ Firmware from China site:https://www.siglent.com/upload_file/zip/firmware/Oscilloscope/V6.02.01.10R2.zipSDS1000DL+ Firmware from US site:https://int.siglent.com/upload_file/zip/firmware/Oscilloscope/SDS1000DL+_6.02.01.10R2_EN.zipBoth firmware ADS files are identical CDC32: 0FD4EC95QuoteAs stated previously flashing any firmware version defaults the unit to English, but if you press the language setting it changes to Chinese with no other options. Also, if you leave the unit powered off for some time (less than an hr) it will boot with the Chinese language.I'm really hoping to find a way to have English the default or fix the crippled language menu. Another option could be changing the model number.

author: tv84
author_rank: Super Contributor
timestamp: August 31, 2020, 03:24:04 pm
content: Quote from: PiXEL8 on August 31, 2020, 04:37:48 amIf you reboot, it goes back to chinese?

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 03:26:01 pm
content: Quote from: tv84 on August 31, 2020, 03:24:04 pmyes

author: tv84
author_rank: Super Contributor
timestamp: August 31, 2020, 03:27:33 pm
content: Quote from: PiXEL8 on August 31, 2020, 03:20:27 pmThat I find very strange...If you reboot immediately, it comes back in english.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 03:28:39 pm
content: Would replacing the Siglent_CML.cfg file from a SDS1202DL+ unit change the model number the fw uses? This could be the easiest way to fix this.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 03:29:37 pm
content: Quote from: tv84 on August 31, 2020, 03:27:33 pmYes, I tried this several times and it stays English language.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 03:37:15 pm
content: Here is the Siglent_CML.cfg from this unit. I added .hex so it could be uploaded here.

author: tv84
author_rank: Super Contributor
timestamp: August 31, 2020, 03:42:06 pm
content: Thanks. Let me have a look at this file later.I had never seen such a .CFG file but it may be the solution. Didn't know DL+/CML+/ etc. also had .CFG files.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 05:59:04 pm
content: Quote>>>MD5_PR?SDS1000A

author: tv84
author_rank: Super Contributor
timestamp: August 31, 2020, 07:35:39 pm
content: Try this .CFG file and report.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 07:49:45 pm
content: Quote from: tv84 on August 31, 2020, 07:35:39 pmThere are two locations for this file. Replace both or a certain one?/usr/bin/siglent/usr/config_file/Siglent_CML.cfg/usr/bin/siglent/firmdata0/Siglent_CML.cfg

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 08:28:02 pm
content: Quote from: tv84 on August 31, 2020, 07:53:09 pmReplaced the one in firmdata0 and reboot. No change. Replaced the other under ../usr/config_file. Reboot and stuck at Siglent boot logo. Power cycle instead of reboot the same.Found hold 'Math' button to unbrick. Screen flashed at power cycle then came back to normal.Language menu has every choice now and it booted in English. Under system status the model entry is missing and there is a '200M' to the far right of the sw version.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 08:32:52 pm
content: SCPI model now shows SDS120X

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 08:45:03 pm
content: Quote from: tv84 on August 31, 2020, 08:30:06 pmBoth files are identical.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 08:58:57 pm
content: Quote from: tv84 on August 31, 2020, 08:52:10 pmYes both of the original cfg files were identical.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 09:03:38 pm
content: Quote from: tv84 on August 31, 2020, 09:00:21 pmUnder firmware load choose the modified .cfg file?

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 09:10:24 pm
content: Quote from: tv84 on August 31, 2020, 09:06:31 pm"The selected file format is incorrect" message when trying to load.

author: tv84
author_rank: Super Contributor
timestamp: August 31, 2020, 09:16:22 pm
content: Do like this (attach).(Taken from themanual.)

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 09:30:43 pm
content: It doesn't have that choice. Under Save/Rec and Factory there is only Depth Recover. and no load cfg under firmware update, it just goes to a file select menu.

author: tv84
author_rank: Super Contributor
timestamp: August 31, 2020, 09:37:54 pm
content: Quote from: PiXEL8 on August 31, 2020, 09:30:43 pmThat's interesting. Please attach a photo of that screen.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 09:57:01 pm
content: Quote from: tv84 on August 31, 2020, 09:37:54 pm502 Bad gateways on this forum are killing me...

author: tv84
author_rank: Super Contributor
timestamp: August 31, 2020, 09:57:46 pm
content: Try this version of the .CFG file.First try the update via menu. If that isn't yet possible, try replacing both files by hand.

author: PiXEL8
author_rank: Contributor
timestamp: August 31, 2020, 10:27:06 pm
content: Quote from: tv84 on August 31, 2020, 09:57:46 pmWent to copy the cfg in both places and the ../usr/config_file directory is empty. Copied new config to ../firmwaredata0 only. Reboot no change I could see. Did a hold "Math" button at startup to reset things. Still no change.Copied new cfg to empty ../usr/config_file directory. Reboot and language was Chinese again. Also, scope is frozen with no response from any button press. Did the hold "Math" reset and it's back to the previous cfg and English language. The ../usr/config_file directory is now empty again.

author: tv84
author_rank: Super Contributor
timestamp: September 01, 2020, 10:49:45 am
content: Your case is an excellent case study that can bring benefits to many here!Next steps:1. take a photo (and share) of the language menu option with the scope as it is. So that we can see all the lang options.2. Rollback the .CFG file to the original ones and see if you get again the chinese-only option and recover the model name, etc.Just some infos:- loosing the model info to "SDS120X" means the scope is not assuming the .CFG config and returns to default settings that are embedded in the app code.- we need it to, once again, start considering what is in the .CFG files so that we can make some experiments.- if that goes well, I think we can activate the "BW change" option, among other things.

author: PiXEL8
author_rank: Contributor
timestamp: September 01, 2020, 03:56:25 pm
content: Quote from: tv84 on September 01, 2020, 10:49:45 amGlad to help and also looking forward to this being unlocked and in English.QuoteMenu only shows 3 languages at a time so there are 6 screen shots;Copying the original config back to both locations brought the model number back and forced Chinese language.Here is an interesting description for the usr dir in a /root/version.txt file;~ # cat version.txtVersion 1.0Author:Bookwore.PengData:2014.05.041.Added dir: usr/bin/siglent/usr for user data dirusr/bin/siglent/config for manufacture dir2.Added more "portmap" and "pdump" app for surpport vxi11The contents of the config dir;/usr/bin/siglent/config # ll-rwxrwxrwx 1 1000 ding 256 Oct 23 2015 NSP_config_upgrade_info.xml-rwxrwxrwx 1 1000 ding 47 Jul 28 2015 version.txt

author: tv84
author_rank: Super Contributor
timestamp: September 01, 2020, 04:33:25 pm
content: So, basically, the language menu is as attached.Quote from: PiXEL8 on September 01, 2020, 03:56:25 pmThat is great news!Later, I'll share a few more .CFG to see their behavior.

author: tv84
author_rank: Super Contributor
timestamp: September 01, 2020, 06:53:41 pm
content: Test this one, please.

author: PiXEL8
author_rank: Contributor
timestamp: September 01, 2020, 07:16:33 pm
content: Quote from: tv84 on September 01, 2020, 06:53:41 pmNo change, still model SDS1202F+ and forced Chinese language.

author: Chris Roubis
author_rank: Contributor
timestamp: October 05, 2020, 01:53:27 am
content: Don't worry... Tv84 will eventually sort it out for you.

author: Lama12
author_rank: Newbie
timestamp: October 06, 2020, 10:52:23 am
content: Hi, I just ordered an SDS1072CML +. Will anyone still have information on how to convert it to SDS1152CML +.Could you please kindly send me a PM how you did it.Thank you in advance.

author: tv84
author_rank: Super Contributor
timestamp: December 30, 2020, 09:05:53 pm
content: Quote from: miyagi on October 10, 2019, 01:51:45 am

author: quinty
author_rank: Newbie
timestamp: January 28, 2021, 01:44:50 pm
content: SDS1072CML+ with FW version 6.01.01.21R2 confirmed, works like a treat.Is anybody has problems with file coding then use file from " Reply #80".Lama12, użyj załączonego pliczku, jak sam robiłem plik to nie działało. Telnet na porcie 23.

author: JTRACKER
author_rank: Newbie
timestamp: February 02, 2021, 10:04:20 pm
content: I know it's been a while, but can I ask what SCPI command was executed, to get the 1152 from the 1052?Thanks

author: tv84
author_rank: Super Contributor
timestamp: February 03, 2021, 10:02:30 am
content: It's the usual command to change BW:MCBD <BW_lic>

author: Raima010
author_rank: Contributor
timestamp: March 18, 2021, 05:49:57 pm
content: Wondering if someone have experience hacking SDS1102X+ oscilloscopes. All backdoors are closed to read memdump. Scope has ID xxxx.yyyy.zzzz so missing last 4 numbers. Tried some code calculators from forum which give some unlock codes with only 12 digits input but none of them work.Some way to read memdump with disassembly? Other ideas?

author: oz2cpu
author_rank: Frequent Contributor
timestamp: November 28, 2021, 06:02:00 pm
content: sorry to kick this thread alive again..but I find no other threads about SDS1102CML hacknew trick : i was not able to connect via putty, correct ip number, tribble checked it was corrent,none of the mentionened ports, SSH or telnet, no way. also checked with nmap where they could be..The solution : get siglent software EasyScopeXthis will connect to the scope just fine,its command interface also accept the hack code, and you can up the bandwidth that way

author: Toolworker
author_rank: Contributor
timestamp: December 18, 2021, 02:31:45 am
content: I can connect to my SDS1102CML+ on the LAN with EasyScopeX, and ping it, but can not Telnet to it even after rebooting with startup.sh on the USB drive, or bring up the IP address in a browser.The firmware version is 6.01.01.22, the 8/6/21 update.Any suggestions?

author: tautech
author_rank: Super Contributor
timestamp: December 18, 2021, 03:01:48 am
content: Quote from: Toolworker on December 18, 2021, 02:31:45 amCheck item #4 on P13 of the programming guide:https://siglentna.com/wp-content/uploads/dlm_uploads/2021/01/SDS1000-SeriesSDS2000XSDS2000X-E_ProgrammingGuide_PG01-E02D.pdf

author: Toolworker
author_rank: Contributor
timestamp: December 18, 2021, 04:43:23 am
content: Quote from: tautech on December 18, 2021, 03:01:48 amThanks. But that just opens an SCPI command processor, which can also be done with EasyScopeX through a LAN or USB connection.After trying a couple of USB drives with startup.sh, I was finally able to telnet to a Linux login prompt with one of the drives.

author: tautech
author_rank: Super Contributor
timestamp: December 18, 2021, 06:23:13 am
content: Oh, sorry. I knew some instruments need to have a port specified for access and believed 5024 was open for CML+. Then I found this:https://siglentna.com/operating-tip/instrument-socket-and-telnet-port-information/

author: razer.zt
author_rank: Newbie
timestamp: February 02, 2022, 02:00:44 am
content: HelloI need help.My device 2190e (sds1152cml+) was fully erased.Attention !!! Don't run "run mmc-all" from uboot. This command contains "run chip.erase"!!! This will fully erase nand .Factory partitions with calibration data was erased too.Who can share partitions from working device.For now the device is not usable.Thank you for help.

author: makersLabs
author_rank: Newbie
timestamp: July 17, 2022, 11:52:48 am
content: Hi,I have a SDS1122E+ from SIGLENTMy version was in Chinese only, I would appreciate changing to EnglishI try to DL from their website here :https://int.siglent.com/upload_file/zip/firmware/Oscilloscope/SDS1000CML+_6.01.01.21R2_EN.zipI finally succeed to flash BUT I cannot change to anything else than Chinese :-(How to change this and access to language options other than only Chinese?Rgds

author: DarkTherapy
author_rank: Newbie
timestamp: April 03, 2025, 10:08:23 am
content: I know this is an old thread, but i recently purchased an SDS1052DL+ and wanted to try the upgrade hack.I couldnt get it working with the startup.sh script no matter what USB drive I used. But then i remembered that windows likes to add hidden characters in text files for a newline character so i converted it to a unix style document and it worked perfectly, a quick port scan showed me telnet was available."To convert a Windows Notepad file (with Windows-style line endings, CR LF) to a Unix-style file (with Unix-style line endings, LF), you can use the dos2unix command-line utility on a Unix-like system or Notepad++ on Windows"Using Notepad++ (on Windows):Open the file: Open the Windows Notepad file in Notepad++.Perform EOL Conversion: Go to the "Edit" menu, then select "EOL Conversion" and choose "Unix/OSX Format".Save the file: Save the file, and it will now have Unix-style line endingsI hope this helps someone.

author: tautech
author_rank: Super Contributor
timestamp: April 03, 2025, 10:39:21 am
content: Quote from: DarkTherapy on April 03, 2025, 10:08:23 amWelcome to the forum.What model did you convert to ?In the west we only have the 50 MHz DL+ whereas in the east a SDS1202DL+ is available.....

author: DarkTherapy
author_rank: Newbie
timestamp: April 03, 2025, 12:08:34 pm
content: Thank you for the welcome, my SDS1052DL+ now reports as a SDS1202DL+ in the System Status page, but I'm not sure how to verify if it's now 200mhz capable.