Your Experience on LCR Meter DER EE DE-5000 / Hantek 1833C - Page 1
Forum section: electronics / testgear

author: andymok
author_rank: Contributor
timestamp: March 21, 2022, 08:37:31 pm
content: I am considering getting a LCR Meter (being very new on this) and have been doing research on DE-5000 and 8813C. While we all know DE-5000 is a legend, I realize people talk so little about Hantek 1833C, most of them are open boxes. I see comments saying 1833C is fancier, having extras frequencies, extended low capacitance range and different measuring levels, are they any useful at all?I would like to understand more about your real life experience on these machines, anything like or dislike, grateful if you can share. Thanks!

author: Grandchuck
author_rank: Super Contributor
timestamp: March 22, 2022, 11:57:59 am
content: Have you watched this?https://www.bing.com/videos/search?q=Hantek+1833C&view=detail&mid=082CBCE315F23D591578082CBCE315F23D591578&FORM=VIRE

author: hpw
author_rank: Frequent Contributor
timestamp: March 23, 2022, 10:16:15 am
content: Quote from: andymok on March 21, 2022, 08:37:31 pm1. Hantek 1833C requires the latest FW!!!2. using the above 1833C requires for SMD additional tools, while crocodile will not work, so China & ALi is near3. Have a clear requirements about low SMD cap's (1pF) and inductor's as RF (BF for 1Mhz...xxxMHz) one will fail..just my 2 centsHp

author: dietermontanez
author_rank: Newbie
timestamp: May 10, 2022, 07:48:58 am
content: Does anybody know how to make the Windows app (Ver. 1.07) work with Windows 10.The video says it requires some fideling but does not elaborate.What do I need to do?

author: coromonadalix
author_rank: Super Contributor
timestamp: May 10, 2022, 04:53:34 pm
content: try the software compatibility modes in the file properties, im running very old aps with this mode

author: rvalente
author_rank: Frequent Contributor
timestamp: May 11, 2022, 08:14:04 pm
content: Quote from: Grandchuck on March 22, 2022, 11:57:59 amI once asked myself: who uses bing? haha, now I've found.

author: David Aurora
author_rank: Frequent Contributor
timestamp: May 11, 2022, 11:01:15 pm
content: I've had the DE-5000 for a while now, I'm still on the fence about it for my needs. For detailed measurement/sorting/etc it's quite good. For troubleshooting in a repair shop it's painfully slow. About once a week I hit a point where I want to throw it in the bin and go back to using my old Silicon Chip/Bob Parker ESR meter instead.The main issue is that it boots into auto mode by default which locks out half the buttons, so you need to cycle through a bunch of different buttons to get where you want to be every time you turn it on. Then it times out and turns off if you stop to scratch your head for 13 nanoseconds and you have to do it all again. It would be so nice if you could store your own default settings or tell it to boot into the last used settings or something, or change the auto off time, etc.I've also never managed to get it to successfully calibrate to a regular set of probes so I'm stuck using Kelvin clips clumsily for every single reading. Again, not the end of the world, but clunky in practice. Measurements can be pretty slow too, a lot of times if I'm trying to work fast I end up just grabbing something else.So yeah, I think it really depends on what exactly you intend to use it for.

author: TimFox
author_rank: Super Contributor
timestamp: May 11, 2022, 11:11:48 pm
content: I have no experience with the Hantek.With the DE-5000, I've learned to live with the boot-up idiosyncrasies mentioned above.My calibrations, though, are very successful. Every time I measure a lab-standard capacitor, my results are quite accurate for such a reasonably-priced unit.For low-accuracy quick results, I use the supplied alligators on short leads (4 terminal connection up to the clips) or (worse) a pair of 18" long leads to grabbers from the DE-5000 banana jacks.For higher accuracy, I bought some extra alligator fixtures and replaced the wiring to BNCs or binding posts or Kelvin clips.My only real problem, though, is sometimes the circuit I am measuring that looks like a capacitor (or inductor) is actually inductive (or capacitive) at the test frequency, but the meter indicates positive capacitance (or inductance).I would prefer that it show negative capacitance, since I can easily deal with that. In that case, I would prefer a display in terms of resistance/reactance or admittance/susceptance, like some expensive bench meters have.

author: MathWizard
author_rank: Super Contributor
timestamp: May 12, 2022, 06:06:19 am
content: I got 1 of the cheap Mastech ones, or a knock off of one. And I can't complain, since it sure beats trying to measure it all myself on the scope. They are similar to the DE5000, but bigger, but the screen is huge tho, so I guess.Maybe next year I will get a better, more compact model. They get expensive fast tho.

author: philwong5176
author_rank: Unknown
timestamp: May 31, 2022, 05:15:52 pm
content: I recently received my Hantek 1833c. I need a Kelvin Clip but found nothing in their website.Does anybody know which Kelvin Clip models works with this meter?Also:I am aware I could plug this meter into a PC, but what software could I use to interface with the meter?Yes I emailed the company, heard NOTHING back.

author: Marko56
author_rank: Newbie
timestamp: August 23, 2022, 04:26:30 am
content: AliExpress has UNI-T Kelvin Clip sets with clip and tweezer models. I paid ~$45 CDN for the clip version ( UTR-L100K-H) IIRC the tweezers were $65CDN.

author: coromonadalix
author_rank: Super Contributor
timestamp: August 23, 2022, 06:50:54 am
content: Quote from: philwong5176 on May 31, 2022, 05:15:52 pmThey have an +/- 200 mb software download ??http://www.hantek.com/products/detail/16180oh their download speed is sloooooooooooooooooooooow pff loll

author: NoMoreMagicSmoke
author_rank: Regular Contributor
timestamp: May 18, 2023, 01:18:44 am
content: Can anyone who has the Hantek measure a resistor at the 100KHz frequency? I have an ET342 that measures significantly low, and since the Hantek appears to share a very similar design, I am curious if it has this same issue.

author: bdunham7
author_rank: Super Contributor
timestamp: May 18, 2023, 02:29:14 am
content: Quote from: TimFox on May 11, 2022, 11:11:48 pmDoes it not have the option of simply displaying Z and Ø ? This is where I go with my LCR meter when results are unstable or unexpected. If the readings don't make sense and Z is very high or very low you then know to try a different frequency, level or mode.

author: hexreader
author_rank: Frequent Contributor
timestamp: May 18, 2023, 03:11:35 pm
content: 1 Megohm resistor shows as about 500k at 100kHz serial mode"X" shows as about 450k - maybe you need to add the two together? - wild guess - I have no understandingParallel mode shows about 900k Ohms with 100kHz100Hz gives probably correct value of 980k Ohms in any modeHantek 1833C

author: mawyatt
author_rank: Super Contributor
timestamp: May 18, 2023, 05:43:42 pm
content: The DE-5000 (bracket value from Hioki IM3536) for a 1/8w leaded 1% 1M metal film shows:DCR of 1.0015M (1.00188M)Rp @ 100KHz 1.0016M (1.00102M)Best,

author: TimFox
author_rank: Super Contributor
timestamp: May 18, 2023, 05:46:41 pm
content: The high-frequency behavior of resistors depends greatly on the internal construction.For old-fashioned A-B carbon composition resistors, the talc grains mixed with the carbon particles "short out" the carbon capacitively at high frequency, so that the equivalent parallel resistance is surprisingly low.

author: David Hess
author_rank: Super Contributor
timestamp: May 18, 2023, 06:07:00 pm
content: One thing I like about the DE-5000 is that the accessories are readily available, so for instance another set of the short Kelvin sensed clips can be purchased and then converted into a long set with coaxial cables.I would not mind having a low frequency VNA for component measurement but until then, the DR-5000 has been outstanding for me.

author: dreamcat4
author_rank: Frequent Contributor
timestamp: May 18, 2023, 06:10:11 pm
content: Quote from: David Hess on May 18, 2023, 06:07:00 pmsounds like a fantastic idea... not sure how to do though? any guidance or help on this matter very much appreciatededit:or can we make from scratch? for example with a pcb to insert into the slot at riight angle. then solder to that perhaps with a 2nd joining board at 90 degrees. maybe things like this should be possible? although idk if it is really worth it instead versus modifying an existing tl-21 like you sayalso debating whether or not to get a 2nd nother tl-21 initially. or just sacrifice the one it came with.... and then if ever change mind can order another one to get that stock version back againso really the question is do i make by purchaing other existing coal kalvin clips pre made. or make from pieces of coax. to then solder some probe handles which is more custom or something than an off the shelf pair of kalvin leadsor you know, to make some bnc termination onto as an adapter making from the existing tl-21. that can plug in unmodified kalvin probes of other brands of esr meters

author: David Hess
author_rank: Super Contributor
timestamp: May 18, 2023, 06:41:06 pm
content: Quote from: dreamcat4 on May 18, 2023, 06:10:11 pmAfter I bought my DE-5000, I picked up a second TL-21 just so I would have the hardware to make a second set of more useful Kelvin probes. The TL-21 provides access to the 2 force connections, the 2 sense connections, and 1 ground/shield connection.Moving interrupted this project so I have not finished it, but my plan is to run 2 thin coaxial cables to each clip, with 1 for force, 1 for sense, and the shields connected to the ground/shield connection.

author: mawyatt
author_rank: Super Contributor
timestamp: May 18, 2023, 06:46:08 pm
content: Here's something that folks might find interesting.https://www.eevblog.com/forum/testgear/de-5000-lcr-tl-21-mod/msg4754858/#msg4754858If you deal with SMDs, the modded TL-21 with BNC connectors and a SMD fixture like shown works really well.Best,

author: NoMoreMagicSmoke
author_rank: Regular Contributor
timestamp: May 20, 2023, 02:33:08 am
content: Quote from: TimFox on May 18, 2023, 05:46:41 pmGood point. I should have specified the type of resistors. I tested my ET432 with a precision metal film resistor, which measures perfect on a DE-5000 at 100KHz, but is 1.8% low at 100KHz on the ET432.

author: NoMoreMagicSmoke
author_rank: Regular Contributor
timestamp: May 20, 2023, 02:35:05 am
content: Quote from: hexreader on May 18, 2023, 03:11:35 pmWhat type of resistor did you test? As others have said, a carbon resistor dropping is expected, but a metal film should not have an appreciable drop at 100KHz.My ET432 has all sorts of problems in the MOhm range as well. I suspect this is because the highest range resistor internally is only 100K, whereas a meter like the DE-5000 includes a 1MOhm range resistor.

author: hexreader
author_rank: Frequent Contributor
timestamp: May 20, 2023, 07:11:54 am
content: Quote from: NoMoreMagicSmoke on May 20, 2023, 02:35:05 amCheap nasty 0.25 Watt 5% resistor from a bundle offer many years agoI have no idea what the construction would beI have no other high value resistors to tryI have low expectations of the Hantek LCR meter and the resistor. Both are good enough for hobby use

author: NoMoreMagicSmoke
author_rank: Regular Contributor
timestamp: May 20, 2023, 05:18:05 pm
content: Quote from: hexreader on May 20, 2023, 07:11:54 amSo probably a carbon resistor. Ok. A dropoff in resistance is probably normal for that resistor then.Thanks for testing though!!!

author: ozkarah
author_rank: Regular Contributor
timestamp: May 20, 2023, 09:51:43 pm
content: Quote from: dreamcat4 on May 18, 2023, 06:10:11 pmIf you have a 3D printer you can also build it yourself :https://www.eevblog.com/forum/testgear/de-5000-lcr-tl-21-mod/msg4753598/#msg4753598

author: Finderbinder
author_rank: Regular Contributor
timestamp: January 21, 2024, 06:43:17 pm
content: Quote from: David Aurora on May 11, 2022, 11:01:15 pmJust was considering DE-5000 as a cheep but reasonably good meter. But your real life experience with it ruined my expectationsThose day-to-day use flaws would also annoy me.

author: audiotubes
author_rank: Regular Contributor
timestamp: January 21, 2024, 06:54:44 pm
content: There's also a new Uni-T LCR meter and I can't remember the name, but there's a bench LCR meter for around 240 euros on Banggood that people have reviewed very favorably.

author: Finderbinder
author_rank: Regular Contributor
timestamp: January 21, 2024, 07:12:42 pm
content: Quote from: audiotubes on January 21, 2024, 06:54:44 pmI can only find 10kHz in mentioned price range.

author: dreamcat4
author_rank: Frequent Contributor
timestamp: January 21, 2024, 07:22:02 pm
content: Quote from: audiotubes on January 21, 2024, 06:54:44 pmut612 perhaps this one is it?https://meters.uni-trend.com/product/ut612/

author: csuhi17
author_rank: Frequent Contributor
timestamp: January 21, 2024, 08:27:35 pm
content: Uni-T 622E/C/A

author: Finderbinder
author_rank: Regular Contributor
timestamp: January 21, 2024, 09:39:15 pm
content: Quote from: csuhi17 on January 21, 2024, 08:27:35 pmUni-t UT622E looks promising, will dig deeper...unless someone recommends even better alternativep.s.: I used a dmm uni-t ut71e, really solid build, hit the ground many times with no bad consequences.

author: Finderbinder
author_rank: Regular Contributor
timestamp: January 30, 2024, 09:13:52 am
content: ...also considering Keysight U1733C, BUT:1) not 4 wire?2) no C and ESR at once?3) what else NOT?

author: The Electrician
author_rank: Frequent Contributor
timestamp: January 30, 2024, 03:46:33 pm
content: Tonghui TH2822C and Applent AT826 both have 4 wire capability and display C and ESR at the same time, and cost less than the Keysight U1733C