Fluke 179 Batt error - Page 1
Forum section: electronics / repair

author: mattltm
author_rank: Newbie
timestamp: August 24, 2016, 04:01:55 pm
content: Hi all.My hard worked Fluke 179 has finally given up and is displaying "Batt" whenever it is turned on. I've tried 3 new batteries and the batt message remains.Has anyone seen this happen before?Thanks.Matt

author: rsjsouza
author_rank: Super Contributor
timestamp: August 24, 2016, 04:24:12 pm
content: I found a few threads around the internet that mention certain Fluke issues that show "Batt" at random times. A common denominator is that apparently the issue is with the main board and only repairable through Fluke.One of the hits:https://www.eevblog.com/forum/testgear/fluke-28-ii-battery-life/

author: retiredcaps
author_rank: Super Contributor
timestamp: August 24, 2016, 05:39:52 pm
content: Questions/Comments/Suggestions.Q1) Are you the original owner of the Fluke 179 and did you purchase this from an authorised Fluke dealer?If yes, seehttp://www.antiqueradios.com/forums/viewtopic.php?f=8&t=264761Q2) If you are not the original owner, check the serial number to see if it has been recalled as perhttps://www.eevblog.com/forum/repair/multimeter-repair-time-again!-fluke-177-reads-8-ohms/?allQ3) If you are not the original owner and your 179 is not being recalled, then I have other tests/suggestions, but will await to see how you answer Q1 and Q2 first.

author: rsjsouza
author_rank: Super Contributor
timestamp: August 24, 2016, 06:21:46 pm
content: retiredcaps, coincidentally I sawthis other threadaround the internet where you state your annoyance at folks that state a problem, get a ton of help and simply vanish.

author: retiredcaps
author_rank: Super Contributor
timestamp: August 24, 2016, 06:31:40 pm
content: Quote from: rsjsouza on August 24, 2016, 06:21:46 pmYes, it happens enough that it changes the time and effort I put into responding now (as I started with this thread).Over the last 3 months on eevblog, a few multimeter repair threads just go dead with the OP not responding.

author: Robomeds
author_rank: Frequent Contributor
timestamp: August 27, 2016, 04:13:12 am
content: That really is too bad. As one who has benefited from your knowledge and the knowledge of others here I hate to see it go without recognition.

author: retiredcaps
author_rank: Super Contributor
timestamp: August 31, 2016, 05:04:51 pm
content: Quote from: rsjsouza on August 24, 2016, 06:21:46 pmAnd 6 days later, no response from OP. Yep, good thing I didn't write anymore than. No thank you, nothing.

author: rsjsouza
author_rank: Super Contributor
timestamp: September 09, 2016, 04:55:22 pm
content: Quote from: retiredcaps on August 31, 2016, 05:04:51 pmI feel your pain with past experiences...A somewhat related beef: the other day I gave some suggestions on another thread (the oscilloscope as a prop thread) but it was so sidetracked that they were completely ignored. Interestingly a few posts later others started suggesting very similar things... Oh well, at least the OP didn't disappear.

author: HeloRN
author_rank: Newbie
timestamp: December 03, 2019, 04:06:46 am
content: Perhaps you had just changed the battery or a fuse when this started?Not sure if anyone will read this, but this same exact problem was happening with me after I had changed a fuse and I found something that fixed it. Not sure if it would work for you, but it's at least worth checking.When I opened the back of my meter to change the fuse, a rotary plate that is supposed to couple with the rotary switch got bumped out of place. To check and see if this is your problem, follow this:-remove the 4 screws and remove the back of your fluke 179-remove the single screw under the 440 mma fuse that holds the board in place (mine was a #1 philips)-The rotary switch coupled with another rotary part on the board, and under the board was a small, white plate that was coupled to the intermediate rotary part. It is about 5/8" in diameter. It's the part that rotates with the rotary switch and makes the contacts for whichever function you are using. It has two tiny springs on it that make contact with the board.-That little rotary plate has a notch in the edge that fits into a key that holds it in place. Mine had 'jumped' out of it's resting place with the notch in the key. I simply put it back in place, put it back together, and it was running like new.Hope it helps! Peace!