Uni-T UTD1050DL oscilloscope firmware dump needed - Page 1
Forum section: electronics / testgear

author: Finderbinder
author_rank: Regular Contributor
timestamp: September 10, 2021, 03:51:44 pm
content: Hello,I got this handheld oscilloscope new for cheep. When powering on, the SHIFT button turns green and display goes to a red Uni-T logo on black screen and then nothing happens/changes. I can power it off. When I connect it to a PC there is no reaction from PC side.I suspect factory defect, but can't guarantee. Would want to reflash a NAND chip. Or/and SPI Flash. Measurements in some control points would be welcome also.I may share photos of internals of the device.

author: Tappaja
author_rank: Newbie
timestamp: November 03, 2021, 07:52:59 am
content: Hi,did you moved forward?ThanksRoman

author: Finderbinder
author_rank: Regular Contributor
timestamp: December 28, 2022, 10:02:29 pm
content: Hello,without changes. Dump still needed.

author: rlohmann
author_rank: Contributor
timestamp: June 06, 2023, 03:43:13 pm
content: Hi,I'm in a similare situaltion:UTD1050DL no start,only one button light up green,not even the display gets on, in my case,Poked around and found a uart on the 12 pin-connector "9J2":Pin 9: 3V3Pin10: GNDPin11: TXPin12: RX (I guess)With a FTDI adaptor (115.2kb) I can read:CCCCCCCCU-Boot SPL 2011.09 (Feb 27 2020 - 10:56:49)Texas Instruments Revision detection unimplementedU-Boot 2011.09 (Mar 20 2020 - 13:08:02)I2C:Checked I2C flash "8U6" (GD25Q80) but it's empty (all „FF“).Maybe someone could help my with a dump of "8U6".

author: darkspr1te
author_rank: Frequent Contributor
timestamp: June 07, 2023, 07:19:55 am
content: Looking at this board/layout I suspect 8U6 is calibration data or similar. The large flash next to it is where i suspect the main firmware lies.As you have u-boot access see if a serial terminal will allow you to type "printenv" no quotes and see if it dumps the uboot environment.you can also dump the flash to serial port this way too.depending on the uboot build you may even be able to write the firmware via serial tooI also suspect that the two row connector next to 9LED2 (pic 3, bottom right ) is JTAG and with a device that runs uboot then it's no doubts for me.Also if this is built like of uni-t scopes then it will have a DMM chip too in some way and they wont finish bootup until that DMM chip spits out it's initial values, I have tested this with a faulty uni-t 81B and a working one and the broken unit had no serial output from the dmm chip but the dmm chip still functioned (i could connect lcd to it's lcd output pins and that showed it working and displaying correct values)I dont know what under the pads in the board pics but i will guess you will find a DMM Single chip solution that they use for basic value reading (non scope mode)darkspr1te

author: rlohmann
author_rank: Contributor
timestamp: June 08, 2023, 07:57:48 am
content: Hi darkspr1te,thank for your feedback. I'll give "printenv" a try and keep you updated.Maybe you could provide some instructions on how to dump the flash via serial.Yes, I assume the DMM in in between the two DSO front-ends.Will also check if I can find some kind of serial communication there.Will also try to peel-off the thermal pads and get some IDs from the devices underneath.Regards,

author: pindrei
author_rank: Newbie
timestamp: May 28, 2025, 07:47:31 pm
content: Hello,i have the same problem. But my output looks like this:Terminal readyCCCCCCCCU-Boot SPL 2011.09 (Feb 27 2020 - 10:56:49)Texas Instruments Revision detection unimplementedU-Boot 2011.09 (Mar 20 2020 - 13:08:02)I2C: readyDRAM: 256 MiBWARNING: Caches not enabledFound a daughter card connectedSKU#01Configurou interrupcaozipCnt:1930unzipCnt:76800Configurou LCDsizeof(image1) :307232NAND: HW ECC Hamming Code selected128 MiBMMC: OMAP SD/MMC: 0, OMAP SD/MMC: 1My display shown only a "loading" text. No Logo.Maybe it is helpfull or somebody have a idea to find a way do verify my problem....

author: Pyati
author_rank: Newbie
timestamp: July 04, 2025, 03:52:10 pm
content: Hello. I encountered this problem on the utd1050cl. It didn't start every time. I counted 50 times and saved it on my computer. Then I flashed the new memory one by one with each saved firmware using the Xgecu t56 programmer. On the fifth firmware, the oscilloscope started working. However, it's unclear how accurate it is. If anyone needs it, I can share it.

author: Pyati
author_rank: Newbie
timestamp: July 04, 2025, 03:55:28 pm
content: maybe it will be useful to someone

author: r011
author_rank: Newbie
timestamp: July 10, 2025, 11:03:01 am
content: It would be nice if you can send me bin fileThanks

author: SeagateAL
author_rank: Newbie
timestamp: October 12, 2025, 03:59:24 pm
content: Hello, could you send me your oscilloscope firmware dump?