Video Agilent 53131A Universal Counter Upgrade OPT-030 Analysis and Experiments - Page 1
Forum section: electronics / testgear

author: Hugoneus
author_rank: Frequent Contributor
timestamp: July 13, 2014, 01:26:55 am
content: In this episode, Shahriar upgrades an Agilent 53131A Universal Counter with the OPT-030 which extends its frequency range from 225MHz to 3GHz. The upgrade kit is a replica PCB intended to emulate the behavior of the original Agilent branded option.The PCB is examined carefully with attention to microwave layout techniques along the signal path. The datasheet of all the parts are reviewed and the reverse-engineered block diagram of the PCB is presented. The expected behavior of the PCB is then measured in both small-signal and large-signal operation using an active 3GHz probe. The PCB is then installed inside the unit and the functionality of the instrument is verified within its specifications. All the documents presented in the video can be found at The Signal Path Blog.Watch the video here: [51 Minutes]http://youtu.be/21_9Dspzu_wMore videos at:http://www.TheSignalPath.com

author: berk98
author_rank: Newbie
timestamp: July 13, 2014, 05:11:57 am
content: Excellent video! Always happy to see new ones.I have a GPS based disciplined oscillator. It's a bit annoying to use, and it's actually currently on loan to someone else. But if you don't find a better option in a while (and you have a good place for an antenna), I'd be glad to ship it over.There will be no question over accuracy, too, although next to an OCXO, a rubidium standard will be fine.Keep up the great videos.

author: vaualbus
author_rank: Frequent Contributor
timestamp: July 13, 2014, 01:53:15 pm
content: I see on the agilent site that there is an option to go to 12ghz. Would be the circuit different? It could be custom amplifier and custom dividr chip or the circuit would be different?

author: Hugoneus
author_rank: Frequent Contributor
timestamp: July 14, 2014, 01:09:12 am
content: Quote from: vaualbus on July 13, 2014, 01:53:15 pmThe principle of operation should be the same: amplify, divide, count. Jus the division ratio has to be higher.

author: Lukas
author_rank: Frequent Contributor
timestamp: July 14, 2014, 04:55:35 am
content: I'm wondering how it detects the absence of an input signal (around 46:20) since there's no kind of RF detector or power meter on the option board.

author: pmcouto
author_rank: Unknown
timestamp: July 14, 2014, 12:12:08 pm
content: Quote from: Hugoneus on July 14, 2014, 01:09:12 amHi,Unfortunately, building a 12 GHz prescaler for the HP 53131A is not that simple…Have a look at this thread:https://www.eevblog.com/forum/testgear/hp-53131a53132a53181a-high-stability-timebase/msg300677/#msg300677

author: Hugoneus
author_rank: Frequent Contributor
timestamp: July 14, 2014, 04:53:16 pm
content: Quote from: pmcouto on July 14, 2014, 12:12:08 pmThank you. Yes it is more difficult to build a presaler at 12GHz... But it is ultimately doing the same thing.I should take apart my 20GHz frequency counter and have a look.

author: Hugoneus
author_rank: Frequent Contributor
timestamp: July 14, 2014, 04:54:46 pm
content: Quote from: Lukas on July 14, 2014, 04:55:35 amEither it looks for activity on the divided output or there is a peak detector on the board. There is an extra diode after the amplifier which I didn't talk about. I'll have to look if they are using that as a peak detector.

author: HighVoltage
author_rank: Super Contributor
timestamp: July 14, 2014, 05:09:11 pm
content: What a great video.Thank you

author: Hugoneus
author_rank: Frequent Contributor
timestamp: July 15, 2014, 06:12:23 pm
content: Quote from: HighVoltage on July 14, 2014, 05:09:11 pmThanks! I am working on a new video... This one needs a lot of preparation.

author: Fsck
author_rank: Super Contributor
timestamp: July 15, 2014, 09:35:58 pm
content: Excellent video as always, really wish you had more time in your life to dedicate to these videos!