Philips PM3055 50 MHz, some questions - Page 1
Forum section: electronics / testgear

author: Ridocar
author_rank: Newbie
timestamp: December 18, 2017, 10:26:14 am
content: Hi,Here's a new PM3055 owner. Although, i've got the 50 MHz version. It works well, but needs calibration.Luckily i've found some manuals (both user and maintanance), so this should be easy. But i've still got some questions.Does anyone know the technical difference between the 60 and 50 MHz? I couldn't find this info anywhere. The manuals are all about the 60 MHz, and i don't want to screw things up because of differences between those scopes.Some parts are very brittle, they break off by even looking at them. Can i get some spare case-parts anywhere?Thanks.

author: oldway
author_rank: Super Contributor
timestamp: December 18, 2017, 06:47:30 pm
content: Hello and welcome to the forum.To calibrate a device, you need precision instruments. Have you got some ?An analog oscilloscope is not a high precision measuring device, it is very rare that it requires calibration.If you are not a specialist, do not touch the adjustments, you risk doing worse than it is.If it works, use it as it is.If something is totally wrong, it's probably a failure, not a calibration problem.For plastic parts of Philips oscilloscopes, it is true that they become brittle like glass .... No solutions for that because these parts are not manufactured for many years ....You can always try to ask for parts on the Dutch forum :https://www.circuitsonline.net/Philips oscilloscopes were manufactured in Holland, (Eindhoven) and it is then easier to find parts in this country.

author: bd139
author_rank: Super Contributor
timestamp: December 18, 2017, 07:01:05 pm
content: Bit of preventative maintenance: replace the RIFA X2 (gold coloured) caps in the power supply before it blows up. It is 100% certain that it will blow up and will make a pretty bad mess of the scope and fill your room with smoke which takes hours to go away. Every bit of Philips kit I’ve had from that era has either blown those caps or they are cracked and running hot.

author: oldway
author_rank: Super Contributor
timestamp: December 18, 2017, 07:11:34 pm
content: If you are 100% sure your scope need a calibration, ask for help with somebody who knows what he is doing....In Holland, see here : (he is member of this forum)http://schneiderelectronicsrepair.nl/

author: HighVoltage
author_rank: Super Contributor
timestamp: December 18, 2017, 07:29:29 pm
content: I have a couple of these scopes in the field with customers and every year I need to do a required calibration to fulfill ISO requirements for this client.However, I never had to do any adjustments on a PM3055.After they are warmed up, they are usually spot on, even after years of service.

author: Ridocar
author_rank: Newbie
timestamp: December 19, 2017, 01:07:37 pm
content: Quote from: oldway on December 18, 2017, 06:47:30 pmThanks.QuoteNope, no precision instruments, just a bunch of multimeters. (none of them are calibrated)QuoteGot it. It's not that the measurements are rally off, i just want to know if it's worth finetuning the scope. But i'll leave it just as it is. it's working fine at this point.QuoteI'll take a look over there. maybe it's worth replacing the front panel...if i can find one.

author: Ridocar
author_rank: Newbie
timestamp: December 19, 2017, 01:11:17 pm
content: Quote from: bd139 on December 18, 2017, 07:01:05 pmI saw another topic about those caps, took a peek inside my scope and yep, there are some cracks in them. They need replacement, that's for sure. Good reminder.