HTI-Instrument HT-19 (320 × 240) - Page 1
Forum section: electronics / general

author: Lord of nothing
author_rank: Super Contributor
timestamp: August 31, 2020, 10:42:21 pm
content: https://hti-instrument.com/products/ht-19-thermal-imagerI received it some days ago. I payed US $472 who was around 224€ when I remember right.And additional 111.06€ for Tax, Import Fee, DHL Import,...Sometime I win sometime the Gov...Here some Images: (Please know I adjust the size by 2x.)Guessed what Brand that is.Just my Accesspoint...My Tritium Light dont produce any heat.The lights on the Ceiling.

author: Fraser
author_rank: Super Contributor
timestamp: August 31, 2020, 11:04:14 pm
content: Looks to be working OK. That appears to be the imagery of the standard Seek Pro J3-603-320 OEM core.Check the temperature measurement accuracy...... it is an area that can be an issue with the J3-603 core.Fraser

author: Lord of nothing
author_rank: Super Contributor
timestamp: August 31, 2020, 11:12:35 pm
content: HiI forgotten to mention that the Cam have an replaceable Accu to. The is in the Handle. The "only" Downside is the use some special Connector and maybe is not that easy to get.I just use them as expensive Toy. How should I measure it?

author: Fraser
author_rank: Super Contributor
timestamp: September 01, 2020, 11:27:58 am
content: Basic thermal camera measurement accuracy tests.....Set the Emissivity to 0.96 for Ice water and Ambient waterMeasure water with lots of melting ice in it. Result should be close to 0CMeasure boiling water in a pan or kettle. Result should be close to 100CMeasure a non reflective surface with a thermometer and then measure it with the camera. Compare results. Emissivity does need to be considered for the imaged surface.Fraser

author: ebclr
author_rank: Super Contributor
timestamp: September 01, 2020, 11:49:34 pm
content: I'm curious from where do you know that Emissivity of watter is 0.96

author: Fraser
author_rank: Super Contributor
timestamp: September 02, 2020, 08:46:45 am
content: Emissivity tables

author: sir
author_rank: Newbie
timestamp: November 15, 2020, 04:06:33 pm
content: Hello. Excuse my bad english.in my country I cannot find the answer to my question. I have an Hti-HT19 and Hti HT-A2 thermal imager.Very good appliances. But I miss setting the temperature range. For example, from + 5 to + 15C.Is there a possibility of a new firmware of the device?

author: Lord of nothing
author_rank: Super Contributor
timestamp: November 15, 2020, 04:13:32 pm
content: I dont see any Updates. I dont know if its possible for us "end user" to do so.Maybe there are some hidden Pins unprotected against a Hack.

author: hulk69
author_rank: Contributor
timestamp: November 16, 2020, 06:37:42 am
content: I got the Hti HT-A2 and I recommend it to anyone who can afford it!I managed to ask the seller on aliexpress to do a 50$ invoice => No import feesMy version of firmware is 2.1.8

author: radioorizont
author_rank: Regular Contributor
timestamp: June 11, 2021, 02:22:03 pm
content: I don't recommend the HTI-A2I have an HTI-A2 and also own a few FLIR and Hikkvision thermal cameras.The Hikvision DS-2TD2136 is a cctv module which i used with the built in adjustable focus lens (if you open it up) as an electronics camera, an absolute beast at 50Hz refresh rate 384*288 resolution and very good sensitivity, never seen anything like that, but anyways...The HTI-A2 is basically a Seek Pro core, so pretty noisy and thermal sensitivity isn't greatAlso it only saves JPEG , not radiometric JPEG, just standard jpeg as a full screenshot where you also see the battery level, emisivity, everything that was on screen when the image was taken.Now for the painful part: HTI don't EVER reply to your emails after you bought the product, EVER.I left a 1 star review for their product/service and that was never added to their official reviews page on their site, pretty crooked if you ask me, i did that twice, "thanks for your review" message, days and weeks later no review was published.Don't ever think about customer support or firmware upgrades for their products.You buy cheap because it is cheap as an overall experience ! (compared with other cameras of same resolution offcourse, not that it's cheap by any means)

author: Lord of nothing
author_rank: Super Contributor
timestamp: June 11, 2021, 03:20:21 pm
content: ServusI like mine its good (enough) for Home and "light" Office use. I can see where some Chips,... are faulty. In comparission with other Brands I can order truth my IT Distributor mine was WAY cheaper from Ali Express.

author: Antonio90
author_rank: Frequent Contributor
timestamp: June 28, 2021, 08:28:21 pm
content: Quote from: radioorizont on June 11, 2021, 02:22:03 pmI might be able to get a Hikvision DS-2TD2136-15 for a good price. Are they easy to setup as a PCB inspection camera? Or is there a lot of DIY stuff to do first? If you could share some details about your setup I would really appreciate it.I am OK with opening it and adjusting the lens, but I don't really know much about programming, if that is needed. sorry if it's somewhat off-topic, but I haven't been able to find a post with that kind of use for such camera.

author: radioorizont
author_rank: Regular Contributor
timestamp: August 31, 2021, 06:25:00 pm
content: The Hikvision only needs 12V and an ethernet connection to your router or laptopYou connect to the default IP then do all the configuration from the browser window, just like any other NVR CCTV systemYou can also use a POE ethernet source which will also provide the power, harder to get those , and more expensive.My setup is a pair of TP-LINK powerlines which allow me to install the setup anywhere in the houseI either connect with ethernet directly to the laptop or if i have the powerline i just use the Android app, i have the camera set to a fixed IP and it's relatively easy to use Hikvision's app and web interface.I use mine handheld but sometimes would just have it on an adapted tripoid facing down to the work surface (PCB, ETC)They are very expensive new, over $3k , if you can get one cheap it's worth purchasing it.All the best

author: downtodo
author_rank: Contributor
timestamp: September 06, 2021, 09:36:07 am
content: DS-2TD2136 2016https://www.hikvision.com/my/products/Thermal-Products/Security-thermal-cameras/Bullet-series/DS-2TD2136-15-V1/Temperature Accuracy± 8 °C (±14.4 °F)Not a good choiceThe correct model should be DS-2TD2136T 2018https://www.hikvision.com/europe/products/Thermal-Products/Thermography-thermal-cameras/Bullet-series/DS-2TD2136T-15/Temperature AccuracyMax (± 2 °C, ± 2%)Not a new product, this model has been discontinued

author: Lord of nothing
author_rank: Super Contributor
timestamp: September 07, 2021, 06:17:48 pm
content: I still think about a Cam from the Company to detect if something with a Heat Signature move in our Garden. Sadly with all the other Sensors on the Market the can be triggert by some movement from a Tree, Busch, Public Bus...

author: radioorizont
author_rank: Regular Contributor
timestamp: September 09, 2021, 06:00:25 pm
content: If you care about temperature accuracy and you are actually looking to measure reliably the temperature on a cctv camera you can get the +/-2% 2018 versionI care about value for money ( i think they are about $2.5K new, got mine cheaper) and netd both models 40mk so sensitivity wise and image quality they both display the same thermal image.

author: Sanochi
author_rank: Newbie
timestamp: February 07, 2022, 09:00:46 pm
content: Hi, I am a bit late to this post but I'm looking for the software of this thermal camera. I can't find a download link anywhere.The software I'm talking about is this:

author: metrologist
author_rank: Super Contributor
timestamp: May 25, 2022, 05:24:34 pm
content: So I was battling insomnia as I recover from vaccinations and it appears that I bought the HT-19 at 3am last night. It was overpriced too. I do not know what happened because I was sure I was going to buy the UNI-T UTI260B.How bad have I messed up?

author: zrq
author_rank: Frequent Contributor
timestamp: May 25, 2022, 07:05:30 pm
content: Look at the bright side, you are getting 1.5 times more pixels and a more sensitive FPA.

author: metrologist
author_rank: Super Contributor
timestamp: May 25, 2022, 11:53:38 pm
content: That is comforting. Thank you. However, I am compelled to know my shortcomings. I am quite open to hear it.

author: metrologist
author_rank: Super Contributor
timestamp: June 06, 2022, 07:54:51 pm
content: Quote from: zrq on May 25, 2022, 07:05:30 pmWhat is more sensitive? The Uni-T sensitivity is less 60mK and HTI is 0.07C. Uni-T also has much wider temp range.HT-19 arrived and so far I have a few points:It clicks every few to dozen or so seconds. I think it is related to ranging the color temp. Sounds like a sensitive mechanical device prone to spontaneous dislocation of some moving part. It's annoying because every time I hear it I think that is one click cycle off the device's limited cycle life.There is no color temp range setting and it is always auto ranging. That means the image always displays the maximum hot and cold colors through the temperature range of what it's looking at. Without the numerical display turned on, there is no way you could tell the temperature of anything or even how different the temperature of an object really is. Here is a circuit board where the first image looks like several hot components, the second image has a soldering iron in the frame and it look much different. I wish the range could be set or at least locked.You can set the overlay of IR and visual images, in increments on the camera and with a slider in the software. The visual image can be manually offset (both x and y) from the IR image. This is sold as a feature, and it could be, but the visual image will offset itself (only vertically) if you move closer to, or farther from, whatever you are looking at. For example, I am looking at an Arduino from 1 meter away and the PCB fills the center 1/3 of the screen. I adjust the "registration" so that the visual and IR images are in alignment. If I then move the camera to 1/2 meter, the visual image of the PCB is now offset and fills the upper 1/3 of the screen. If I move the camera to 2 meters, the visual image of the PCB is offset in the opposite direction and fills the lower 1/3 of the display. In all cases, the IR image is centered. This offset is less affected at longer focal distances, so moving from 10 meters to 8 meters does not offset the images much, but if you are focusing on things that are close and far in your frame, something is not going to line up. I'm not sure if that is just normal for these devices (and why Uni-T has a "macro" lens), or if the two sensors are not properly aligned (defect)? I've asked the seller, who I think is an official HTI ebay distributor, if that is even a thing...I went out on a pitch black night to see how the world looked. I got freaked out because every tree had an animal crouching down at its trunk, but I later realized it was just that autorange made the very slightly warmer tree trunks look like glowing red animal balls.Of course, the first thing you do is check the temp of something the device cannot measure, but look, it records video:https://i.imgur.com/Nf2kufs.mp4

author: Bill W
author_rank: Super Contributor
timestamp: June 07, 2022, 09:12:25 pm
content: Quote from: metrologist on June 06, 2022, 07:54:51 pmQuite normal, it is 'flat field calibration', hopefully slows down once the camera warms up and stabilises in temperature.Quote from: metrologist on June 06, 2022, 07:54:51 pmThis effect is inevitable, due to parallax with the IR and visual lenses / sensors separated by a small amount. It is not a faultConsider the extreme of viewing at a few mm, your object would cover the thermal lens and the visual would miss it completely.Bill

author: metrologist
author_rank: Super Contributor
timestamp: June 07, 2022, 09:25:50 pm
content: Thanks. I figured it was a parallax issue but it seems more extreme than would be expected.The lack of color range control is more bothersome though. Maybe it's just something I need to learn to use/interpret better.

author: Bill W
author_rank: Super Contributor
timestamp: June 08, 2022, 10:02:10 am
content: Quote from: metrologist on June 07, 2022, 09:25:50 pmDo some geometry on the images and see what you get. Probably easiest to work out by aligning at distance and then seeing the mismatch close up.As for colour=temperature, that is a more complex task to do as it is all pixels every frame. As such it is the sort of thing that cannot be done on a cheaper camera with a lower powered processor.

author: metrologist
author_rank: Super Contributor
timestamp: June 17, 2022, 01:41:38 pm
content: I still need to learn how to use this thing. I was trying to see the temps of this shiny metal piece on my surface grinder. -400F and I could still touch the 1200F part

author: Bill W
author_rank: Super Contributor
timestamp: June 17, 2022, 08:58:31 pm
content: e=0.01 will always give stupid answers.This is saying that 1% of the energy from this thing is due to its;s temperature and the rest is reflected energy off of things the camera does not know the temperature of.So the smallest inaccuracy in guessing the 99% is translated into imaginary energy from the 1% - and multiplied by 100 !Bill

author: metrologist
author_rank: Super Contributor
timestamp: June 17, 2022, 09:36:32 pm
content: Yes, last time I used it I was just playing around with the settings and emissivity did not come to mind at all when I checked this part.Thank you for the explanation - it's much better than thinking of it as glossy or matte.I'm still sore over the autorange thing, but then I saw the PC210 got a FW upgrade so maybe HTI can add the feature if they want.