GPSDO brands? - Page 1
Forum section: electronics / testgear

author: Housedad
author_rank: Frequent Contributor
timestamp: October 22, 2018, 05:15:16 am
content: Which GPSDO units are the better or worse units? Feature sets? Symmetricon, Trimble, Oscilloquartz or any others?

author: LapTop006
author_rank: Unknown
timestamp: October 22, 2018, 08:00:08 am
content: Most of the major ones are fine, although one thing to look out for is newer models have newer GPS cores, which can be needed (week rollover issues just for a start).

author: awallin
author_rank: Frequent Contributor
timestamp: October 22, 2018, 05:37:56 pm
content: check this one out:http://www.jackson-labs.com/index.php/products/furydatasheet has both a phase-noise plot and a long-term ADEV plot - look for those in other products also

author: edpalmer42
author_rank: Super Contributor
timestamp: October 22, 2018, 06:12:10 pm
content: If you google for <gpsdo "allan deviation"> you can find info on some of the usual suspects.Here's some tests I've run on various units. Normally, Standard Deviation doesn't work well on oscillators, but since these are locked to GPS, I think it's valid.Code:GPS Devices -- Measure & analyze the period of the 1 PPS OutputDevice ............... Std Dev (ns).... Range (max-min)(ns) ... Device ...... NotesNavsync CW12 ......... 4 - 5 .......... 20 - 25 ............... GPS Rcvr .... 1,7Motorola UT+ ......... 40 - 55 ........ 95 - 110 .............. GPS Rcvr .... 2,7Rockwell Jupiter ..... 10 ............ 50 ................... GPS Rcvr .... 3,7Motorola M12M ........ 10 - 15 ........ 40 - 60 ............... GPS Rcvr .... 7Trimble Thunderbolt .. 0.4 - 0.5 ...... 2 - 4 ................. GPSDO ....... 6,8HP Z3801A ............ 0.1 - 0.2 ...... < 1 ................... GPSDO ....... 6HP Z3817A / CW12 ..... < 0.1 .......... < 1 ................... GPSDO ....... 4,6Jackson Labs GPSTCXO . 0.3 - 0.4 ...... 2 - 3 ................. GPSDO ....... 6NEC NWM-034241-201 ... 0.1 - 0.2 ...... < 2 ................... GPSDO ....... 5Trimble UCCM ......... 0.09 - 0.11 .... < 1 ................... GPSDO ....... 5Results are based on multiple runs of ~ 1000 measurements each.Sawtooth correction has not been used for any of the GPS receivers. Where supported, it would reduce the numbers substantially.All units were connected to the same antenna system.Notes1. Sawtooth correction not supported.2. Most 'range' results were in this group, but there were a few at 20 - 30.3. Only one test.4. Requires external 1 PPS input. Equipped with E1938 oscillator.5. Measurement made with Fluke PM6681.6. Measurement made with HP 5370B.7. Measurement made with HP 5372A.8. Standard parameters.9. Optimized parameters. (Not yet used in any tests)

author: tautech
author_rank: Super Contributor
timestamp: October 22, 2018, 07:56:41 pm
content: So you didn't get enough good info from the other thread you started:https://www.eevblog.com/forum/testgear/what-frequency-standard-is-sufficient/

author: edpalmer42
author_rank: Super Contributor
timestamp: October 24, 2018, 07:29:14 pm
content: Quote from: tautech on October 22, 2018, 07:56:41 pmI know that this question is intended for Housedad, but I'll answer it anyway. That thread is more about which type of standard is sufficient although there was discussion about a couple of specific GPSDOs. I think that a separate thread is appropriate. I've posted my data before, but I'd like to see some similar measurements for some of the newer models. I have no intention of trying to buy one of everything!

author: texaspyro
author_rank: Super Contributor
timestamp: October 24, 2018, 07:53:42 pm
content: Quote from: edpalmer42 on October 24, 2018, 07:29:14 pmWell, uhhh, I pretty much have at least one of each. I'm in the process of testing them all in the same config. Using a TAPR TICC clocked by a HP-5071A cesium beam oscillator. PPS goes to channel A and the 10 MHz divided down to 1PPS goes to channel B. GPSDOs are allowed to warm up for at least 24 hours.

author: edpalmer42
author_rank: Super Contributor
timestamp: October 24, 2018, 08:07:48 pm
content: Quote from: texaspyro on October 24, 2018, 07:53:42 pmI thought of you when I wrote that!But for anyone else who's reading this, you don't need a cesium to make the measurements. As long as your reference has negligible aging during a 1000 sec. run, that's good enough. My counter was referenced to my Efratom FRT Rb house standard for my measurements.

author: texaspyro
author_rank: Super Contributor
timestamp: October 24, 2018, 08:21:23 pm
content: My runs last over 18 hours and go to tau=20,000 seconds...

author: edpalmer42
author_rank: Super Contributor
timestamp: October 24, 2018, 09:54:51 pm
content: You don't need to go to 20K sec. to characterize a GPSDO. You know that if you average over a long enough time period, the frequency will be dead on. You can measure Allan Deviation, but they all resemble each other pretty closely - particularly at large tau values where they all converge due to the impact of GPS disciplining. The only place you'll see differences is in the central part of the graph where the OCXO and the disciplining parameters affect the graph. For a good example, look athttp://www.leapsecond.com/pages/gpsdo/.What's easier to measure and for most users more useful, is short term jitter. All you have to do is measure the period of the 1 PPS and calculate the standard deviation and min-max range. Do it a few times to catch outliers and you're done. If you're lucky, your counter does all the calculations for you and you just write down the results. Normally, you can't measure Standard Deviation of an oscillator because drift messes up the math. That's why they came up with Allan Deviation in the first place. However, if the oscillator is locked to GPS, there's no drift so the math should work.Texas, I know that you know all this, I'm preaching to the other readers.

author: texaspyro
author_rank: Super Contributor
timestamp: October 25, 2018, 12:33:16 am
content: Quote from: edpalmer42 on October 24, 2018, 09:54:51 pmActually you should... a lot of GPSDOs seem to want to occasionally go a bit bonkers. If you are only looking at 1000 seconds of data, you can easily miss the shenanigans. And to be really safe make sure you include the time around 00:00 UTC (and particularly around the time-of-week rollover)... evil spirits like to come out an play around the witching hour.

author: edpalmer42
author_rank: Super Contributor
timestamp: October 25, 2018, 03:31:45 am
content: Quote from: texaspyro on October 25, 2018, 12:33:16 amI agree with you, but that's a never-ending chase. There will always be weird 'corner cases' where strange things happen. Are you going to include entry to and exit from holdover? What about the next 1024 week rollover? Yeah, I know that's getting silly, but you get my point. Regarding 'evil spirits' and 'the witching hour', do we have to make special measurements on Halloween?IMHO, the 1000 second measurement, particularly if you do a few, will give you good confidence in the performance capabilities of a GPSDO under normal operating conditions.After all, look at my list. There aren't enough types or measurements to be definitive, but if you rank them by those numbers, do you disagree with the order? Actually the Trimble UCCM is very interesting. Look at how tight the results are! I did 7 tests of 1000 measurements each. The Std. Dev. readings are from 98 - 110 ps. and the min-max range values are from 0.58 - 0.86 ns. Very impressive!

author: MrW0lf
author_rank: Frequent Contributor
timestamp: October 25, 2018, 06:57:24 am
content: Quote from: edpalmer42 on October 25, 2018, 03:31:45 amQ&A red alertrandom glitch fanboy identified. You know what will happen if such attitudes will sneak up on growing generation and logic/code ends up in vehicle software, banking, etc? I know you know but just preaching the other readers...

author: edpalmer42
author_rank: Super Contributor
timestamp: October 25, 2018, 06:22:55 pm
content: Quote from: MrW0lf on October 25, 2018, 06:57:24 amI hear you and yes, people designing professional equipment must worry about 'corner cases' and other odd situations, but seriously MrW0lf, we're not designing life-critical equipment here. We're a bunch of hobbyists checking out some used equipment that we bought off fleabay for use in our home labs!Can we possibly maintain a reasonable sense of proportion and common sense?

author: Scratch.HTF
author_rank: Regular Contributor
timestamp: November 03, 2018, 07:52:31 am
content: Oscilloquartz is part of Swatch who makes world class timing equipment (under the Omega brand) for the Olympic Games; so therefore, Swatch applied their area of expertise to GPSDO units.

author: awallin
author_rank: Frequent Contributor
timestamp: November 03, 2018, 08:08:48 am
content: Quote from: Scratch.HTF on November 03, 2018, 07:52:31 amOSA is, since 2014, part of ADVAhttps://en.wikipedia.org/wiki/Oscilloquartz

author: tomato
author_rank: Regular Contributor
timestamp: November 03, 2018, 05:33:27 pm
content: Quote from: edpalmer42 on October 24, 2018, 09:54:51 pmYou're making the assumption that the GPSDO is operating correctly. The main reason for testing a device is to verify that itisoperating correctly.

author: edpalmer42
author_rank: Super Contributor
timestamp: November 03, 2018, 07:03:33 pm
content: Quote from: tomato on November 03, 2018, 05:33:27 pmThe assumption that probably should have been stated is that you've got lock, good satellite visibility and signal strength, completed survey, etc. You can't measure ADev or Std. Dev. until all the basics are covered. These steps alone will weed out most faults.Texas's argument is that even when the basics are covered, the 18 hour test is needed to test for glitches and since you're doing that, you might as well measure ADev while you're at it. I don't disagree with that, but an 18 hour test is just as arbitrary as a 1K (~17 min.) test. I'd rather do a few 1K runs on different days and times than one 20K test. I'm sure that you could find a unit that might pass a 1K sec. test but fail an 18 hour test. I'm also sure that you could find a unit that would pass an 18 hour test and ultimately fail.Murphy rules!

author: tomato
author_rank: Regular Contributor
timestamp: November 03, 2018, 07:15:11 pm
content: Quote from: edpalmer42 on November 03, 2018, 07:03:33 pmMost faults doesn't mean all faults. That's why you do the tests. (Especially since people on this forum seem to favor used equipment of unknown condition bought at bargain basement prices.)QuoteIt's not an either-or choice. It's just as easy to make a few 18 hr. runs on different days, and cover all your bases.

author: edpalmer42
author_rank: Super Contributor
timestamp: November 04, 2018, 04:18:44 pm
content: Quote from: tomato on November 03, 2018, 07:15:11 pmNow you're just trolling. You know as well as I do that this is a statistical rabbit hole that has no end. 'All faults'?? Good luck finding 'all faults'. Every user has to decide what level of performance is appropriate for their situation. But if someone is even thinking that they might need to do 'a few 18 hr. runs' to 'cover all your bases', they won't be buying their hardware on fleabay because their corporate lawyers won't let them.It's also worth keeping in mind that regardless of how long you test, there will still be data glitches caused by any number of things that have nothing to do with the experiment. In fact, it might be impossible to tell whether a glitch is due to internal or external factors. That's why all of the ADev programs have an extensive suite of data editing tools and warn that you have to edit out the glitches before you can analyze the data.This isn't a professional time-nut forum. It's a bunch of hobbyists who think that GPSDOs and other time-nutty things might be interesting to play with. Telling them that they have to spend many hours making a single set of measurements is neither helpful nor correct. You might as well tell someone that's making a 555 led blinky that his test is a failure unless the frequency is correct to withing a milliHertz.Ed

author: texaspyro
author_rank: Super Contributor
timestamp: November 04, 2018, 07:25:19 pm
content: I originally posted this image in the wrong thread...Here is an example of why you want to look at GPSDO data over extended periods of time. A 1000 second run would miss that big honkin' burst of crapitude.

author: edpalmer42
author_rank: Super Contributor
timestamp: November 04, 2018, 08:58:25 pm
content: Quote from: texaspyro on November 04, 2018, 07:25:19 pmI wondered what was going on there....QuoteActually, it looks like it was about 45 minutes long. That kind of stretches the definition of 'burst'! But now what? Was the cause internal or external? Weird microwave oven or someone playing with a GPS blocker? Bad solder joint or logic error? Was it a one time event or will it repeat? How long are you going to continue the test? What have you proven?Sure, a 1000 sec. test would have missed that noise, but that's not what it's trying to measure. The 1000 sec. test assumes that the unit is operating properly during the test and attempts to document the performance during that 1000 sec. interval. Even an ADEV test makes the assumption that everything is stable during the test. If all hell breaks out you either edit it out of the data or delete the data and try again. If it happens too often, you start looking for faults.I'm measuring jitter rather than ADEV because jitter is a lot easier for a newbie to measure and understand. ADEV isn't nearly as intuitive. As I said earlier, I'd really like to see some more measurements so we can decide if measuring jitter actually makes sense as a way to characterize GPSDO performance at a hobbyist level. Based on my measurements, it looks promising.Ed

author: rbm
author_rank: Regular Contributor
timestamp: November 04, 2018, 09:11:31 pm
content: Quote from: texaspyro on November 04, 2018, 07:25:19 pmTexaspyro, In real world terms, what is the significance of this "burst of crapitude"? Is this something that only affects the operation of a machine depending on the short term stability of the GPSDO signal? Or is this something that adds error to the long term accuracy of the GPSDO? I'm not very familiar with the concepts to work this out for myself.

author: texaspyro
author_rank: Super Contributor
timestamp: November 04, 2018, 10:04:19 pm
content: Quote from: edpalmer42 on November 04, 2018, 08:58:25 pmNo idea what caused it. I had Heather monitoring the GPSDO and nothing seemed to be amiss (like satellite dropouts) at the time. A lot of GPSDOs seem to occasionally have similar behaviors... something upsets their tracking loop and they go a bit wonky for a while.What does it prove? Well, it proves that GPSDO can go a bit wonky.Disturbances like that will degrade the overall ADEVs of the GPSDO. A 1000 second run would probably not catch such things. A longer run will. 24 hours or so seems to be a reasonable length to catch the majority of such things.BTW, to get the best performance from most GPSDOs try to set the satellite elevation and signal level masks to values that minimizes the changes in the tracked satellite constellation. Most GPSDOs seem to add noise to the outputs whenever the constellation changes.

author: tomato
author_rank: Regular Contributor
timestamp: November 04, 2018, 10:12:15 pm
content: Quote from: edpalmer42 on November 04, 2018, 04:18:44 pmI happen to agree with Texaspyro's opinion on this subject, and have given reasons why. That hardly makes me a troll.Quote from: edpalmer42An 18 hr. Allan Variance run of a GPSDO still does not guarantee one will catch 100% of faults, but it helps. The best thing is that it costs nothing to extend an Allan Variance from 1,000s to 20,000s; you start your computer program and let it run. Your computer doesn't care how long the data acquisition run lasts.Quote from: edpalmer42Corporate lawyers? You're going to have to translate this, because I have no idea what it means.Quote from: edpalmer42Nobody is telling anyone what they have to do. I simply agree with Texaspyro's opinion that one should measure the Allan Variance of a GPSDO out to longer (20,000s) times. I think this is especially important for hobbyists since many are buying GPSDOs of unknown operating condition. I disagree that one only has to measure the Allan Variance out to 1,000s because "at large tau values ... they all converge due to the impact of GPS disciplining" because that assumes a priori that the GPSDO is operating correctly.

author: edpalmer42
author_rank: Super Contributor
timestamp: November 06, 2018, 02:09:07 am
content: I reached out to another Time-Nut to get another opinion on this topic. After reviewing the thread, he agreed with Texas and Tomato that a short test isn't enough to draw any conclusions regarding the performance of a GPSDO because there are too many things that can go wrong.As I thought about that, I realized that my experience with GPSDOs isn't that great. Because of that, I just haven't seen the problems that you guys have. To me, a GPSDO is just as stable and reliable as a quartz oscillator. They just work! So I couldn't understand why you were being so negative.And since they just worked, and basically worked the same, I got bored with them and stopped buying them. That left me with older, perhaps higher quality units that were more likely to be stable and reliable.So the end result was that my idea of a possible quick way to rank GPSDOs was based on a limited data set that didn't reflect reality. Sorry if I caused any confusion or bad feelings.Homedad, after totally hijacking your thread, it looks like there's no easy way to rank GPSDOs. Personally, I've always thought that my Z3801A was the best one I've got and slightly better than my Trimble Thunderbolt. But they've both gotten rather expensive lately. The only two 'newer' GPSDOs I've got are the Trimble UCCM, and the NEC unit, both of which are still available on ebay. Both units tracked well, but I find that the outputs have more frequency/phase noise than the Z3801A although they both look good on my jitter test - which I still think has value even though it's not a definitive test.I find that other features aren't very important on a GPSDO because, for me, they're just install and forget. YMMVOne other thing I'll say is that you can cripple a good GPSDO by having a poor antenna setup. Work hard to get a location clear of obstructions and with good visibility towards the equator. That's the direction where you'll see more satellites. It becomes more important as you move away from the equator.Ed

author: cdev
author_rank: Super Contributor
timestamp: November 06, 2018, 02:49:16 am
content: How can one test a GPSDO 'device under test' without a better one to compare it to?

author: texaspyro
author_rank: Super Contributor
timestamp: November 06, 2018, 03:13:46 am
content: Quote from: cdev on November 06, 2018, 02:49:16 amAssuming you mean the output signals, you can't. You need to compare it to some higher-speced device. An HP-5071A with the high performance tube works fairly well. Or a maser.

author: David Hess
author_rank: Super Contributor
timestamp: November 06, 2018, 03:17:57 am
content: Quote from: cdev on November 06, 2018, 02:49:16 amA comparison to a local atomic standard including a rubidium will reveal problems in the GPSDO disciplining.

author: texaspyro
author_rank: Super Contributor
timestamp: November 06, 2018, 03:21:27 am
content: Quote from: David Hess on November 06, 2018, 03:17:57 amThat depends upon the quality of the atomic standard and of the GPSDO. A lot of rubidiums (particularly surplus telecom ones) are worse than a lot of GPSDOs. The HP-5065A rubidium is an exception...

author: cdev
author_rank: Super Contributor
timestamp: November 06, 2018, 03:47:27 am
content: OoooooooQuote from: texaspyro on November 06, 2018, 03:21:27 am

author: awallin
author_rank: Frequent Contributor
timestamp: November 06, 2018, 05:49:05 am
content: Quote from: texaspyro on November 06, 2018, 03:13:46 amyou can try with three similar-ish units, and three-cornered-hat analysis. with only one or two GPSDOs you can't do much - get three or morehttp://www.wriley.com/3-CornHat.htm

author: edpalmer42
author_rank: Super Contributor
timestamp: November 06, 2018, 07:07:21 am
content: You guys are so pessimistic!!You can confirm basic functionality by comparing two GPSDOs to each other. If they are both locking to GPS, the average drift between them will be zero. The signal will wobble around a bit due to noise and differences between the two devices. Since both devices are moving around, noise measurements won't be nice, but the average frequency will still zero out. You need a decent time interval counter and the ability to capture the data to confirm this. You might be able to use a counter with less functionality, but it would be painful.If your maser is in the shop and you loaned your 5065A to your brother-in-law (what were you thinking??), the Stanford PRS-10 is worth considering. Even the Efratom FRK or its military cousin the M-100 give surprisingly good results. All three will give you ADev results that dip well into the e-13 range.There are no reasonably priced FRK units on ebay right now, but the M-100 and PRS-10 are, or have recently been, available for a not too outrageous price.Ed

author: texaspyro
author_rank: Super Contributor
timestamp: November 07, 2018, 05:04:41 am
content: Quote from: edpalmer42 on November 06, 2018, 07:07:21 amNope, we're time nuts. We want our shineys to be all shiney down to the atto-hertz.

author: edpalmer42
author_rank: Super Contributor
timestamp: November 07, 2018, 06:06:20 am
content: Quote from: texaspyro on November 07, 2018, 05:04:41 amYeah, I've noticed....I can't handle that. I start to get lightheaded when I go below 1e-13.

author: texaspyro
author_rank: Super Contributor
timestamp: November 07, 2018, 04:22:40 pm
content: Quote from: edpalmer42 on November 07, 2018, 06:06:20 amPoor thing... time to climb back into your crib with your 555 timers and RC oscillators until you feel better...Take two cesium capsules with a glass of rubidium hyrdroxide and consult TVB in the morning.

author: Housedad
author_rank: Frequent Contributor
timestamp: November 16, 2018, 10:54:19 pm
content: Quote from: texaspyro on November 07, 2018, 05:04:41 amDoes that fall into the category of Obsessive-Compulsive Disorder?