I need to know the History of FLUKE 87  ( 1988 - 2010 )  - Page 1
Forum section: electronics / testgear

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 07, 2010, 05:50:29 pm
content: Yep I am thinking to buy one Fluke 87 ... ( lots of digits on the screen )And I found several versions of it ...The First one called " Fluke 87 " Made at 1996 - 1998 .Then , its another version called as " Fluke 87 III " (provably third generation )And another one called as " Fluke 87 V " ( The Latin " V " = fifth generation )Can you in light me .. about the deferences between the three versions ?Does it worth today , to buy the oldest first release ?Thanks ..

author: EEVblog
author_rank: Unknown
timestamp: April 07, 2010, 10:46:52 pm
content: I don't remember when the original Fluke 87 came in, but I thought it was a bit earlier than 96?The original took the market by storm, and is what cemented it as probably the #1 general use industrial meter.I don't ever remember seeing an 87-II anywhere, although I guess maybe there was one at one point, at least internally at Fluke?The 87-III was the next major update. It improved the accuracy from 0.1% to 0.05%, but it defaulted to AC instead of DC for current (annoying IMO). It also added a removable battery door, and the ability to switch back and forth from 4.5 digit mode without turning off the meter.The 87-IV does exist, but it was like the 180 series. I figure Fluke knew they made a mistake by calling it an 87, so they spun off into the 180 series.The 87-V added temperature measurement and low pass filter.That's from memory anyway!Dave.

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 07, 2010, 11:05:32 pm
content: Thanks Dave , about the "date of birth "about the " 87 V " ,found from the FLUKE book , that was made :May 2004 Rev. 1, 8/042004 Fluke Corporation, All rights reserved.All product names are trademarks of their respective companies.So its all ready 6 years old !!So from your reply , I can conclude , that I will have to stay away from the 87 III line .Does this Fluke 80 family , suffers from display issues , found some stories about display ghosting ,and repair kits about it ... are they so sensitive ?And as last , what worries me , or bothers me , are to get the one with the faster refreshing times ,display and calculations ..Does all act with the same speed ?Even the first 87 until the latest 87 V ..

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 08, 2010, 03:48:22 pm
content: Dave ... I got it ..http://cgi.ebay.co.uk/ws/eBayISAPI.dll?ViewItem&item=270555009910Not cheaply , but hell its an sparkling one .Now I have new , questionsIs the full manual of it, available on PDF ?

author: EEVblog
author_rank: Unknown
timestamp: April 08, 2010, 09:31:24 pm
content: I've never seen the display issue myself, but you can buy repair kits on ebay to solve it, so I guess at least some people have the problem.Fluke have most of their manuals online just google "fluke manuals".I believe all the models have the same update speed.Dave.

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 08, 2010, 10:42:08 pm
content: It looks that the called repair kit , its just the ( elastic carbon contacts strip ) .So the called damage, could possibly happen, only if the instrument gets exposed to extreme heat ,that could possibly cause disinformation of the soft strip, with side-effects ,loss of tension = poor contact , between LCD + PCB.The sellers recommend the fix , for the first 87 line ... (1996)About the documents , I found them ... but I had some hard times , I was finding old revisions of them.Finally found truly fresh material .. (2005 - 2008 ) on the Canadian sitehttp://ca.fluke.com/Fluke/caen/support/manuals/default.htmBest ever document, this one ..http://assets.fluke.com/manuals/8xv_____cmeng0104.pdf

author: switcher
author_rank: Regular Contributor
timestamp: April 09, 2010, 09:15:56 pm
content: The 80 series were introduced in the mid 80s IIRC.the original 87 manual is dated 1988 on the fluke website.Here is a good write up covering the different types:-http://reviews.ebay.com/Fluke-Meters-80-Type-Models-and-Series_W0QQugidZ10000000007491382

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 10, 2010, 12:31:54 am
content: Woa ... amazing linkThumps Up ...I do not know, if I have the right to make comments about it here.Any way , in all this pile of info, they are usable tips.But there is also and some " bloat" in it , or better expressed " misinformation " ,like about getting instruments on ebay , from the " right hands" , or " worthy " or " expert sellers".I feel 1000 times better , by knowing that I got my Fluke , from an young Deejay who lives in a small place in Honolulu , and his main interest are playing music .Than getting one covered with grease , or other hazardous deposits , found only in heavy industry .And who knows how many times , it had be dropped down ( even accidentally) .And about " expert sellers" ... they are the first ones that they can " cook " an heavily damaged multimeter , to look as shiny , and internally to be like hell .Any way , I will stop here , getting out of topic , its not my hobby .

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 10, 2010, 12:40:23 pm
content: Finally found some pictures of the Add-ons , when it comes boxed ..Dave .... stop laughing man ..Yes its just the " Getting started " ... tight assesLOLI watched your recent review about the wireless one , and your reactions , of screaming ... why ! why ! why ! ....So for any one who gets the instrument , with out the book, there is no real loss ..... PS: If you like the picture , save it on your computer too , imageshack its an unstable service,uploaded pictures , could be lost .. or some times it does ..Dave, we should have the option to upload directly on the forum server...Or valuable information and pictures , are in danger over the time .I know that an small cost involved for the service , but I like to believe ,that all of us , we could help if needed ..

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 13, 2010, 07:30:10 pm
content: Its my hobby , or personal tactic , for everything that I get – Buy , to collect in advance any information that circulates ,about it .So , I did the same about the Fluke 87-5 , and I constructed one F.A.Q. made by an technician , for the technicians.Now if the word“ F.A.Q.”sounds too much , call it as“unofficial Tips”list .1) Deference’s between the 87V (5 Generation )The87V has a switchable low pass ac filter for those trying to measure the output of an adjustable speed motor drive, the 87-III does not.The87V has a type K temperature mode and includes the 80BK beaded thermocouple probe, the 87-III does not.The87V can measure capacitors to 10,000uF, the 87-III can only measure to 5uF.87: 5uF maximum87 III: 5uF maximum87 V: 9,999 uF maximum or 9.99mFThedefault A/D converter on the 87V is 6000 counts, the 87-III is only 4000 counts. Both are switchable to 20,000 counts.The87V's has a CAT IV 600V safety rating, the 87-III does not.The87V has a smoothing mode, the 87-III does not.The87V's new holster (model H80M) has the necessary locking slots to work with Tool Pack and Locking Pack accessories, the 87-III does not but you can use the new holster with it to get that capability.Otherwise the 87V and 87-III have the same ranges and functions and physical size.2) The story about the “ 87-4 “In 1999, the 87-4 was introduced as new a model that combine the features of the 87-3 and the 8060A meters. The 89-4 is the same as the 87-4 with the addition of the logging capabilities. The 87-4 and 89-4 were never considered to be the same meter as the 87-3 witch is now the 87-5. They just happen to share a similar model number. About two years later the 87-4 and 89-4 were updated and re-branded as the 187 and 189.The 87-IV and the 89-IV had a Pulse width measure range that only when down to 50.000 msec.3) Function Tip .. enable - disable high impedance .To enable the high impedance mode for the 87-3 or 87-5 press and hold the Hz button during power-up. The input impedance is changed from 10 Megohms to >4,000 Megohms. This function is useful to measure mili-volts on very low energy circuits where a very high input impedance is needed.4) Repair TIP damaged multimeter by high power RFSign of damage , LCD on - meter does not respond to selector changes.Look to find an fusible resistor of 1 K Ohm , replace it with equal type .5) Display repair part 1Sign of damage : No back light ..Cause : Bad transistor that turns on the light, it is right before the light socket.Cure :Its replaceable with a standard transistor, you could try 2n3906 or a 2n3904 .One of the two will do the trick 100% .6) Display repair part 2Sign of damage : Faulty display – Ghosting – numbers fadesCause : Between the display and the PCB , there is elastomeric pads ,at the connection point between elastomeric pads and PCB , there is soft signs of oxidation.Cure :a*Remove the screws on the back.b*Disconnect the batteryc*Be careful and remove the outer LCD cover (clipped on)d*Gently pull off the LCD portion (it sits on the possibly oxidized connections that you'll soon clean with alcohol)e*Pull off the bottom white LCD backing still in the meter (connected by 2 pins on the left)f*Clean all the circular button connections and the pins below the LCD in the skinny slots top and bottom for the LCD about roughly 40 pins.g*Cleaning solutions .. denatured alcohol or glass cleaner ( as less drastic solution ) ,anything powerful could shorten the life of the elastomeric pads .k*Use tweezers to remove the old Pink elastomers if they stay down in the slot(s) to avoid any 'skin oil' deposits being left....they usually stick to the bottom of the glass in most cases, which makes it easier.The elastomeric kit is available from Fluke, and the part number is 627494. This is not the display upgrade kit (model DR80). Replacing the elastomeric connectors will correct the problem.Most problems was show up when your meter has an Pink colored elastomeric pad ,Fluke found that this pads was causing issues , upgraded the material with new type grey color this time.If your meter are an old one, and uses the Pink one , its best to upgrade it , with the latest Grey elastomeric.The ones with the Grey elastomeric, do not suffer so far from the above issue .7) Does my meter need to be calibrated?If you short the test leads together, does the meter read zero volts (or at least very close to zero)?If yes , you are ok .8 ) Do you repair your own tools ?If yes, you worth to have , the complete schematic of the Fluke 87 & PCB layout &,part list = full PDF file : 83_85_87smeng0500.pdf (11.850 KB)http://assets.fluke.com/manuals/83_85_87smeng0500.pdf9) Fluke special Fuses …a) Type 10,3X38FLU11(11A)-Volt Rating 1KV- Breaking point 17000A ( recently Fluke updated the specs to 20000A)b) Type 10,3X35 DMM 44/100= 440mA- Volt Rating 1KV- Breaking point 10000ATip : Its from the few times , that I will advice too " call Fluke ", so to get them , or the nearest representative !!At the begging of 2009 , there was a battle between the Fuse manufactures like Bussman & LittleFuse ,and they had triple the prices , of the same fuses , when they get sell in the open market !!Fluke , has its own stock , and keeps the pricing almost low , because they making money, from their products,and not mainly from the fuses !!EU Pricing 11A ( 8.50 – 12 EUR ) VAT includedEU Pricing 440mA ( 5.32 – 11 EUR ) VAT includedBe awareabout similar fuses fromSIBAbrand , looks identical in specs ,but they have Breaking point 30000A .Check the page called asURZ 23http://www.sibafuses.com/pdf/urz.pdfI am not aware if they can be used , but they are the closest in specs , from any other solution worldwide.If you have, any similar tips , just add them .. and help this Tips list to grow .Thanks ..

author: saturation
author_rank: Super Contributor
timestamp: April 14, 2010, 01:17:25 am
content: Here's my page for the service manual of the original 83,5, 7 series. It was for my 85 bought in late early 1990s. You'll see this series is as old as 1989.

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 18, 2010, 02:12:24 am
content: Now that I finally got my 87-5 , noticed few details , thats its good to be known.When I power it up , I get an fast boot screen as {F87}Well, I am a bit lost in here , is it a firmware version? .. or model version ?In the alphabet the " F " stands for the number " 6 ".I am not worry if the instrument are genuine , only older face plates , from 87 III ,could possibly masquerading one old 87, to look as 87III .. ( theoretically speaking)One quirk that I dislike:At High Resolution mode ... at 20V (end of the high resolution road )The meter correctly show as {19.999} above that drops to {20.10},that is logical.In order to show again the high Resolution , it does not need to lower the voltage to less than 20 again , but to drop it down to 16.80 ... then it shows again High resolution .( My DC PSU has an 30 turns potentiometer on it ... I have to do 3-4 complete rotations to drop it that low !! )..Anotherquirk that I found , are the clear plastic on the display ,its so shiny that causes reflections " glare " , even with the indoor lighting.( An not shinny LCD protector film , could be the answer )The holster , with the extended rear ( foot - leg - support point ) , does not look to be well designed ,it causes the feeling that the instrument will fall - flip , by an accidental sock , or even by pooling the test leads in your direction.( I am working on a solution about it , I had not made my mind yet )..Other than that looks solid ...I had test and the "high impedance" function ...I could measure the standard impedance at about 11Mohm ,but for the high one , I could not do it , as my other DMM stops at 200Mohm.

author: EEVblog
author_rank: Unknown
timestamp: April 18, 2010, 03:26:33 am
content: Quote from: Kiriakos-GR on April 18, 2010, 02:12:24 amOooo, I like the vernier dial on that supply!Dave.

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 18, 2010, 10:48:43 am
content: This specific dial its like 38 years old..found it last year , when we updated some machinery in my work, actually I save it, from becoming scrap.The PSU its model of 90s , It belongs to the over 1000$ price range.I had recently replace the volts pot with a Burns 10 turns (3540S-1-103).http://datasheet.octopart.com/3540S-1-103-Bourns-datasheet-9381.pdfClose up , for the younger ones , to see how it looks like.What it does as part , are to count the rotations , the smaller handle gives more resolution in movement,and the top right ( lever) .. works as lock , so to not be turned accidentally, or by vibrations.Lets see if the attachments tool works..

author: saturation
author_rank: Super Contributor
timestamp: April 19, 2010, 05:23:47 pm
content: I still use this type of PS, dual tracking. I brought them back from the Pacific when I lived there, most mid range tools are Japanese, mine is a Trio, I think that's a brand used by Kenwood in the 1980s.Quote from: Kiriakos-GR on April 18, 2010, 10:48:43 am

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 20, 2010, 01:01:55 pm
content: I am very happy today , my 87V passed the test of one voltage calibrator with success.I was just a bit worried , because I got it as used, and so there was a little doubt puzzling me.The installation that the test took place are "restricted", so I will not say anything more about it.I will just say what happened, and one picture special made for public view.The " calibrator " was one air-tight device , with 5 or 6 selectors , its selector had a scale of 0-9.And by setting the proper combination on the rotary knobs , you where getting the specific output.It needs to heat up , in one specific temperature ( probably has an internal heater) , takes about 3 minutes to get ready.The output of it, was only 11V to zero .. ( if it helps, I could also describe it, as one PLL synthesizer that produces voltage instead of RF )The Fluke 87 are above it ..in the picture.Under the "calibrator" Its a rack-able Fluke DVM with resolution as {12.00000} =12VAt the right side its another "old" hand-held Fluke DMM , its recognizable, next to it , its an Metex DMM , no auto range .The test was quick 5-7 minutes ( do not ask why ) ..The Fluke 87-5 got tested at 7V and 5V and 5mV & 1mV.At the volts scale , it proved to be off set at the last digit ( high resolution ) like "Minus 1"When the voltage had set as {5.00035} confirmed by the rack-able Fluke ,the 87-5 was indicating {4.999} ,The old Fluke and the Metex was measuring as {5.0} they are low resolution.Both remain accurate in all tests , by acting alike.At the mini-volts was totally accurate , no matter the setting, was fantastic there.We had test it, at 1mV and 5mV , and all the digits was an perfect {1.000}1mV and {5.000}5mVFrom today, I can trust it 100% or 99.99999%( I took the picture at the end of the task and after, so any readings pictured has no true value )

author: Excavatoree
author_rank: Frequent Contributor
timestamp: April 23, 2010, 05:16:19 pm
content: Quote from: Kiriakos-GR on April 10, 2010, 12:31:54 amHello! First, I apologize for the off-topic nature of this post.Thanks for your comments. I've heard almost no feedback about my E-bay reviews, and it's good to see someone has thought about them.I'm interested in a few of your comments:First, I never used the term "worthy" sellers at all. I never meant to imply that some sellers are "worthy" and some are not. I meant that sellers differ, and the type of seller will affect whether or not you want to bid on a meter, and what you should pay. Your DJ seller, for example, is a good one to buy from. Yes, some meters are beat to crap, and some change hands in pawn shops, etc, and are sold by folks who know nothing about them. Those are the meters you take a risk buying. Now, if you want parts, or a "fixer upper" you will be looking at sellers differently than a student who needs a fully-checked out, guaranteed meter.I may need to clarify what I call an "expert seller." There are some "so called expert sellers" that will do as you say - rip someone off. The ones I recommend (I don't give names for two reasons - one: It looks like I'm colluding with them and not being objective and two: There are probably expert sellers I don't know about. The sellers I call "expert" would never sell a pretty, but fried inside meter. The would have checked it with their calibrators (or a meter of suitably higher accuracy) and will guarantee that it's good. In my guide, I try to help folks identify the seller's knowledge, so buyers can be informed. I've seen an "expert" purposely mislead someone - but using common sense when reading their ad will identify these - no knowledge of electronics is required at all. (obviously, I can't mention this seller's name, either)I never meant to say that you should only buy from "expert seller." As you found, you can get a very good meter from someone who may not be a "meter expert."Again, I'm sorry if this is "off topic" but I was happy to see a link to my guide. I welcome any comments, suggestions, criticism, etc. etc.

author: saturation
author_rank: Super Contributor
timestamp: April 23, 2010, 07:25:32 pm
content: Wow, that old Fluke is still alive and working? That model isover 30y old!It looks like an 8060A, my 'dream' DMM, I could never afford back then.

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 23, 2010, 07:54:05 pm
content: QuoteHello @Excavatoree and welcome.Your first reply here are right on target , as confirmation , about the matters that I had "nag" about.The DJ seller, represents in my way of thinking, the "reliable" home user , who got something out of his waters (overkill product for his needs ) , and just needs to cash back.But to be honest, its a matter of how careful you are in a personal level, as shopper.1)I foundthat he was an DJ by looking the return policy and his profile.2)I foundthat he was an Real DJ , as he was buying lots of CD's ( Feedback history)3) He hadHigh resolutionpictures.4)Goodfeedback.5) I had even use Google maps , and his address , and the satellite landed on a Villain Honolulu.So as total , its not just one word or many, that could lead someone to have an successful "transaction".Its more complex than that.Even so, by my own opinion , you came close to the 70% of correct buying tips,and I like to congratulate you, even for that percentage.To All :Generally on Ebay , some people "sellers" , they use several tactics , to gain trust ,the most ridicules expression that I had spot, was like "comes from smoke free home" .You could live in a low-end neighborhood , with 50 factories next door ,and because you do not smoke a cigarette , the merchandise are smoke free,and free of any contamination .Any way , the used ones , always contain some sort of risk, its up to the buyers "talent or education or expertness ", to get the one who will be called as " safe bet ".By the knowledgethat I got until today , and have recorded it here,about the specific Fluke 87 models ....They are specific reasons , to not buy anything older than the 87-5 .1)The elastomer pad (soft contacts) on the display , are anofficial flaw.. ( will you ever buy something with an flaw ? )2)Any older than the 87-5 has missing specs , why someone to pay even the half price,for something like that ..3)Fluke has "silently" ( you have to search their white papers to find out)updatethe rating of the Fuses ... from 17KA breaking point to 20KA ... Thistheoretically, can cause the older Fuses to blow easier than theNewones....Why no one on eBay or elsewhere , points out such details ??For such expensive Fuses , as Buyer Idemandto know such details.Does anyone knows, that on the Genuine cables of the 87-5 are stamped with an production Date possibly of the DMM it self ?So possibly ... if you get an DMM with out cables" as Is ", its an deliberately made move, so to loose track of the production date ? or worst ... to not have an proof of the condition of the environment that the DMM was "living at" .For all those reasons , that I consider them as serious ones, I recommend nothing less than the 87-5 .Fluke did the smart move to not officially recall all this products , so any expense as repair , goes to the poor one who will buy from ebay , an completely out of warranty product, because Fluke like so.And I am also poor , but only in my pocket .. Thanks God.Edit:@saturation congrats for your DMM, We have many of those in the place that I work,recently one needed an new display,and we found it on ebay ...About accurate , it is one of the trusted ones.. Well done.Ebay could easily turn from bless to curse , this is all that I am trying to point out.

author: Unknown
author_rank: Unknown
timestamp: April 23, 2010, 09:36:48 pm
content: Quote from: Kiriakos-GR on April 23, 2010, 07:54:05 pmI don't understand. A quick search on the Fluke website showed me the user manual for the original 87, which seems to contain full specifications.Quote from: Kiriakos-GR on April 23, 2010, 07:54:05 pmI'm sure this is the maximum interrupt rating, not the current at which it will blow (can't imagine a 17kA fuse in a non-industrial setting). 20kA fuses are slightly safer if you're dealing with CAT IV stuff. I don't see why the 17kA fuses would blow easier.Quote from: Kiriakos-GR on April 23, 2010, 07:54:05 pmOr, more likely, they got damaged or lost. The 87V is offered in various kits with different sets of leads. The Fluke leads that I have on my bench (TL-71) don't have any visible date stamp.Quote from: Kiriakos-GR on April 23, 2010, 07:54:05 pmIn my opinion it depends on the price. A 87V is obviously newer and better, but I wouldn't mind an older 87/87-3 if it was cheap enough and working. The extra features of the 87V like higher CAT ratings and low-pass filter are fairly useless for non-industrial use.I'm sure only a part of the units does exhibit this flaw, and it's not a safety issue, so I don't see why they'd have to issue a recall. It might be nice to extend warranty coverage for this issue, but it's quite possible that they did.

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: April 23, 2010, 10:28:27 pm
content: QuoteIf you look today , for an Fluke 87 , the only official marketed are the 87-5 ..So you will read the today specs ... and withthosein mind you will decide getting one.If you do not do your homework properly , you will end up with an older one , with lower specs .Simple as that !!QuoteRight now we have as marketed Fuses, the two types as 11A and the 440mA ..The 11 Amperes one updated ... But externally remained the same !!I had read somewhere that one small "b" added , as extra description about the change,but its just words.QuoteThe retail basic version , is what most people gets ... lets keep it simple.QuoteIf thats the case , get an New Fluke 83 , why bother looking the 87 ?About recall , they did nothing , and even from respect to their customers, they should had one,and replace the known bad part , free of charge and to all customers , with or with out warranty.They would had less reasons to worry about pissed off customers , or half knowledgeable Greeks to bark for their products , not to mention about their locked forum, that they do not accepting your registration, so to have control against the pissed off customers .. aka " fluke community spamers " .If any Fluke , likes to become an friendly trusted brand, its up their hand to change all those irritating tactics.At list from the Taiwanese brands , I was aware of what to expect as warranty and after sales service.From Fluke , I was probably be more optimistic than usual , that why I am acting that way.I feel pain , and I act that way , because even if I got it as used , I was hopping, that I will get some respect with it , from the manufacturer.They do not grasp , that the people who are involved with electronics repairs , are equal to them.They probably believe that they are above all , and they acting like Gods or something !!At list the 87-5 its an successful product, and this fact works as an pill , That calms me down ,and I start smiling again.edit:local link, about the 87-5 accessories.https://www.eevblog.com/forum/index.php?topic=359.0

author: switcher
author_rank: Regular Contributor
timestamp: May 08, 2010, 06:44:56 pm
content: Horses mouth:-www.flukebiomedical.com/Fluke/usen/Community/Fluke+Plus/ArticleCategories/RD/25-Years+of+DMMs.htm

author: switcher
author_rank: Regular Contributor
timestamp: May 08, 2010, 06:47:39 pm
content: Quote from: saturation on April 23, 2010, 07:25:32 pmI remember when these were in the RS catalogue, they were nearly £500-

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: May 08, 2010, 10:43:41 pm
content: Quote from: switcher on May 08, 2010, 06:44:56 pmStraight from the horse's mouthMeaningFrom the highest authority.OriginIn horse racing circles tips on which horse is a likely winner circulate amongst punters. The most trusted authorities are considered to be those in closest touch with the recent form of the horse, i.e. stable lads, trainers etc. The notional 'from the horse's mouth' is supposed to indicate one step better than even that inner circle, i.e. the horse itself.It is a 20th century phrase. The earliest printed version I can find of it is from the USA and clearly indicates the horseracing context - in the Syracuse Herald, May 1913:"I got a tip yesterday, and if it wasn't straight from the horse's mouth it was jolly well the next thing to it."Kiriakos speaking :I am the rider ..And when the horse says : Fluke must balance the cost of holding an oversize parts inventory against the need to invest in developing new and better tools .It translates ... profits to them, loss for you ...The consumer always pays for the R&D , but you have to build first something so to get payed ... And not by getting money in advance by cutting as corners,the best interest of your all ready customers !! who had honor you with their trust !!

author: mikeselectricstuff
author_rank: Super Contributor
timestamp: May 09, 2010, 12:08:33 am
content: Another very practical and useful improvement between the original and the III is the backlight was changed from a green LED to white, and is way brighter.

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: May 09, 2010, 09:45:00 pm
content: Ok here is a gift for every wild spirit like me ..Here are high resolution pictures of my Fluke.I found few points of interest on the bottom housing...First , there is an small9thermally printed ( that stands for the manufacturing year )Secondly , at the fuses rating , it says min 17000A ( an indication that Fluke still uses old stock at the back cover ) , the specs have be updated to 20000A for many years now.Any way , nothing passes from the crazy inspector unwatched !!

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: June 08, 2010, 11:24:59 pm
content: Link addition about the Mini test : Chauvin Arnoux PAC12 AC/DC PRO Probe Clamphttps://www.eevblog.com/forum/index.php?topic=660.0

author: exodia
author_rank: Contributor
timestamp: August 18, 2010, 05:56:07 am
content: Hello kiriakos. I would like to know how much you paid +shipping to Greece for that Fluke 87V. I'm in the phase of purchasing my 1st serious DMM from Ebay but those V series are quite expensive. I like the greater capacitor range and temperature measurement of the latest generation but the 200+€ are abit high for me. I also like the lifetime warranty but i don't know if it's valid if i buy from Ebay. I think i'll try find a series III 87. Any suggestions?The 1st generation of 87's does it have full 0,01 resolution?

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: August 18, 2010, 01:11:23 pm
content: HiI had select the 87-5 as an fresh and True RMS meter ..About the pricing it matters of where you live ..If you do not care about the True RMS , get another meter and forget the Fluke 87 versions ...And get something that works for your needs ...The 87-5 or V , its not a God , yesterday I found that it can not measure pF as capacitors meter..Here is a picture of my invoice ... In comparison with the street price in Europe, my price can be called as bargain one..

author: ThunderSqueak
author_rank: Unknown
timestamp: August 18, 2010, 05:33:47 pm
content: Quote from: mikeselectricstuff on May 09, 2010, 12:08:33 amYep, I posted this earlier... My old 87 and my new 87-vQuote from: saturation on April 23, 2010, 07:25:32 pmI have techs working under me that have this meter... Old but they still work and do the job

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: August 19, 2010, 05:38:29 pm
content: Well I did one adjustment by software on the above picture ,the reason was to see, of how deferent are the quality, and the illumination of the new display.Well , its like night and day , even if the original picture , was taken with wrong settings ,and the display lighting looks more powerful in both , than the real thing.What I did was to drop down the illumination of the picture , at the point that the Green display starts to fade for good , the New display looks 200% more brighter and readable ... at list .

author: ThunderSqueak
author_rank: Unknown
timestamp: August 19, 2010, 09:26:06 pm
content: Quotehey, I took that picture with my phonebut yes, it is like night and day.

author: exodia
author_rank: Contributor
timestamp: August 21, 2010, 05:01:32 pm
content: Kiriakos i live in Greece like u :-)Now about Fluke 87V. I've come to a deal with a seller on ebay that can sell me a V for 255 calibrated tested so i think it's a good deal. Just today i bought a Rigol 50Mhz and a fluke 17B for a starter DMM and maybe my 2nd DMM will be a 87. Do you suggest anything better for a pricepoint at about 180€ max? To tell you the truth i don't think a true RMS is a must for me i'm more into dc stuff, microcontrollers and such.

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: August 21, 2010, 05:28:28 pm
content: Hello @exodia , and welcome ..I do understand , that you are a bit confused, the Fluke 87 line , had some wide recognition as model.But there is many other meters equally well made for bench use , that no one had put his hands on them, for an review ..My experience with brands and models are small , but by experience about to tell, if something works correctly or not, are huge.I will advice you to watch the reviews of Dave , so to get an picture of how the modern meters stand this days , about build quality.In my priority s , other than the TRUE RMS , was and the 20.000 counts ability .I think that its an good idea , to suggest to Dave , to make an review of meters at 20.000 counts.So my suggestion to you , are to start looking about meters with the 20.000 counts ability.From the other hand , UNI-T has some very fresh models loaded with nice futures,its up to you , if you will explore this new product range.Some models can be found here ..local link (GR) ..http://www.emimikos.gr/shop/

author: exodia
author_rank: Contributor
timestamp: August 22, 2010, 04:01:55 pm
content: Thanks kiriakos, i'll have a look. The 20.000 count is the resolution of the unit?

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: August 22, 2010, 04:14:45 pm
content: Yes it is ... so to be able to see max 19,999 volts .. above that it drops to 20,00 volts ...So the 20.000 count its five displayable digits ( at the range of 20 volts max )The 30.000 counts its five displayable digits ( at the range of 30 volts max )etc etc ...

author: batee
author_rank: Newbie
timestamp: August 23, 2010, 05:31:32 am
content: Bought my first Fluke 87 in approx 1991-2. Cost about $350.Suffered from LCD delamination. Before Fluke released the revamp kits, it was too expensive to fix, so I sold it in eBay.Bryan

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: August 23, 2010, 02:17:01 pm
content: Quote from: batee on August 23, 2010, 05:31:32 amHello Bryan , this issue are alive even today ...The third party repair Kits , even if they look available on ebay, its not something that you can count on it, about their existence and normal pricing , in long term .Thats why my advice to all the Europeans, are to get nothing less than the 87-5 (V) ,because it uses the improved one .

author: marianoapp
author_rank: Regular Contributor
timestamp: August 23, 2010, 03:32:05 pm
content: i may be able to get a used Fluke 87 (the first revision) locally for about $275, which is a good deal around here since a 87V is $600+so, reading the topic i found that the main differences between the original and V revision are:1- 0.1% accuracy2- cap measurement up to 5uF3- no temp measurement4- no low pass filteram i missing something?

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: August 23, 2010, 03:39:04 pm
content: Quote from: marianoapp on August 23, 2010, 03:32:05 pmSize and lighting of the display , plus .. I think that in order to replace the battery, it must completely opened.

author: ThunderSqueak
author_rank: Unknown
timestamp: August 23, 2010, 07:57:58 pm
content: Quote from: Kiriakos-GR on August 23, 2010, 03:39:04 pmyep, this is true. The old model also suffered from "ghosting" of the display. This could be fixed by a disassembling the display and cleaning the contacts but they eventually deteriorate again. Fluke put out a kit to replace the whole display with a newer part that does not have this issue. The cost of this kit and the price of an old 87 however would make purchasing an 87V more cost effective if you are starting from scratch. The bar graph also works differently on the meters. (see earlier picture I posted). To put the old meter into 20,000 count mode is different, you have to hold a button down while turning the meter on. On the new meter it is just a button press.I love my old 87, it has been with me for a long time, but I love my 87v more (I think the old one might be getting jealous).A family picture of my fluke meters. I replaced the holster on the old 87 as a while back a kid stole the meter, his father found the meter with my name on it and made him return it, the holster however did not make it back to me. Stupid thieves

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: August 23, 2010, 08:43:13 pm
content: Quote from: ThunderSqueak on August 23, 2010, 07:57:58 pmWow, you just pinch my interest , and I have to ask , how did you do an such marking,that personalizes the meter ..Just an sticker at behind ... or something more ?

author: ThunderSqueak
author_rank: Unknown
timestamp: August 23, 2010, 09:57:07 pm
content: I did a shallow engraving on the back and filled it with white paint so it stands out. I also have my name written on the inside battery. I am well known in this state (I have a good reputation), so people know where to find me.

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: August 23, 2010, 11:12:30 pm
content: Well from my side as freelancer , because I visit my customers , in their shops or homes..Some their targeting , my small hammer , or the wire cutters , but not an meter !!!I have also an sticker in the battery compartment of my mobile phone.Thanks for sharing the info ..

author: gcb
author_rank: Newbie
timestamp: May 24, 2012, 09:25:44 am
content: also found some ET87.regular series 1"ET-87 DDM"(images from Craigslist, will be gone soon. if you know how to upload images here, please do)

author: brandonb
author_rank: Newbie
timestamp: September 13, 2012, 02:20:22 am
content: Kiriakos-GR nice write up on the fluke, im a huge fan, have a fluke 87-3, but the ground bannana plug has a broken connection inside the component its self (not the solder joint), do you guys know where to buy a new bannana plug component so i could solder in a new one in the meter?

author: retiredcaps
author_rank: Super Contributor
timestamp: September 13, 2012, 04:03:02 am
content: Quote from: brandonb on September 13, 2012, 02:20:22 amDo you mean something like this?http://www.ebay.com/sch/i.html?_trksid=p5197.m570.l1313&_nkw=fluke+87+input&_sacat=0Personally, I find the prices super high for replacement parts since used fully working Fluke 87 and sometimes the 87 III sell for about $100 on ebay.This auction was more reasonable if you didn't care about the modificationhttp://www.ebay.com/itm/271053504078

author: brandonb
author_rank: Newbie
timestamp: September 13, 2012, 04:34:55 pm
content: yes thats what im looking for, thanks...they are proud of those, i was told that the meter is life time warranty, wonder if thats true?

author: jcpx
author_rank: Newbie
timestamp: October 05, 2012, 09:50:32 pm
content: Hello all,I purchased second hand fluke multimeter 87-III.I just received it and noticed the following thing:When I turn ON the fluke, on the AC Voltage measurement, the multimeter makes 2 small popping (or clicking) noises that come from the inside. it is not loud, similar to a small relay being activated.Sometimes, the display blinks at the same time than the noises.I noticed also that when turning the multimeter ON, on the AC Voltage measurement, the graph bar jumps like crazy in all the directions then settles down. it lasts 2 seconds before settling down.None of the fluke I used 87-5 and 187 had a similar behavior. I am worried with the popping noises happening at startup.Have you noticed that on the 87-III already?I also have to replace the 11A fuse (F2) because it is blown. Hopefully no damages happened to this fluke with a high current.Thanks for your help.

author: Aldobrandi
author_rank: Regular Contributor
timestamp: October 05, 2012, 10:38:54 pm
content: I recently got a 87III from eBay and it exhibits the same behavior you are describing (clickling sound at startup and bargraph excitability on AC range. I don't have the equipment to perform a full battery of tests so I can't tell if there's anything else unusual. I'm curious as well to hear what people have to say. Thanks!

author: jcpx
author_rank: Newbie
timestamp: October 05, 2012, 11:31:27 pm
content: Thanks for the feedback, yours having a similar behavior is already a good sign (hopefully). Let see if other people have some feedbacks to provide.On my side, I did a quick check on the basic functions and it seems to work correctly.AC Votlage OKDC voltage OKFrequency OKResistor and continuity OKDC current OK (I was only able to check low ranges since the F2 fuse is blown, so I can't measure large current for the moment).the different buttons (range, min max etc) seem to work also correctly

author: saturation
author_rank: Super Contributor
timestamp: October 06, 2012, 01:37:29 am
content: Chances are the rotary switch contacts are dirty. If disassembly and cleaning works then add deoxit cleaner and shield to protect it and that should hold for many years.

author: Aldobrandi
author_rank: Regular Contributor
timestamp: October 06, 2012, 03:56:23 am
content: So I took my meter apart to clean the contacts (range switch, screen and piezo contacts) and I realized that the clicky sound comes from the piezo itself whenever the screen comes on, i.e. when the meter is first turned on and any time you switch from normal to 4 1/2 mode and then back to normal mode. Has anybody else ever experienced something similar ?

author: retiredcaps
author_rank: Super Contributor
timestamp: October 06, 2012, 07:44:19 am
content: Quote from: jcpx on October 05, 2012, 09:50:32 pmI got a Fluke 85 III for fairly cheap off ebay because the seller could not spell Fluke properly and I was the only bidder over the 7 day auction.When I first received it, I put in a low voltage (it measured about 6.6V) 9V battery and the multimeter clicked, clicked and clicked. Then I put in a fresh 9V battery and it worked. If I listen very closely, I get the same 2 quick clicks as your 87 III.QuoteI hate sellers who write "everything works perfectly" and once you receive the item, there is a blown fuse. If Fluke fuses weren't so bloody expensive and ebay sellers didn't ripoff Canadians in shipping charges, then it would be a different story.

author: ModemHead
author_rank: Frequent Contributor
timestamp: October 06, 2012, 11:59:41 am
content: I have had a first-generation 87 since whenever it came out (20+ yrs?), and I never noticed that it clicks twice when you turn it on, but it surely does. Learn something new everyday.Quote from: jcpx on October 05, 2012, 09:50:32 pmI have replaced a number of blown 11A or 15A fuses in eBay Flukes and never found any further damages from over-current.

author: EEVblog
author_rank: Unknown
timestamp: October 06, 2012, 12:18:35 pm
content: Quote from: ModemHead on October 06, 2012, 11:59:41 amAnd it's got an "ultrasonic" data output mode too.Always wondered if it's actually useful, but never bothered to actually capture and decode it.Dave.

author: ModemHead
author_rank: Frequent Contributor
timestamp: October 06, 2012, 12:30:10 pm
content: Quote from: EEVblog on October 06, 2012, 12:18:35 pmYes it will make a lot of noise if you turn that on. As far as I know, Fluke never sold anything that would interface with it. I did once run into someone's project on the net that used an ultrasonic transducer to pick it up and decode it.Edit:Found it (PDF):http://www.circuitmaestro.com/Projects/DataHolster/DataHolster.pdfIt was just a Radio Shack microphone...

author: jcpx
author_rank: Newbie
timestamp: October 06, 2012, 01:09:31 pm
content: Thanks everybody for your feedbacks.It seems then that the clicking noises are common to the old generation fluke multimeters.Seems everything is ok with my acquisition!Waiting for the fuse to be delivered and I'm back on track!!See you.

author: 001
author_rank: Super Contributor
timestamp: November 20, 2017, 06:02:42 am
content: Sorry for necrostingBut what about fluke 87 benefits in 2017?So model is 29 years old but still high pricedMost of revolucionary features for 1988 now avaliable in $40 range gear

author: EEVblog
author_rank: Unknown
timestamp: November 20, 2017, 06:39:43 am
content: Quote from: 001 on November 20, 2017, 06:02:42 amYeah, but people trust the Fluke 87.

author: 001
author_rank: Super Contributor
timestamp: November 20, 2017, 07:24:44 am
content: Quote from: EEVblog on November 20, 2017, 06:39:43 amI.e. it is no more when a sort of fetish now, isn't it? (Sorry for my english, but I hope You can understand me, thanx)I pick one for me at occasional sale. Works great but it is bulky and not so speedy as described here

author: mikeselectricstuff
author_rank: Super Contributor
timestamp: November 20, 2017, 12:06:08 pm
content: Quote from: EEVblog on November 20, 2017, 06:39:43 amI've had an 87 in daily use for over 20 years - would today's $40 units last that long ?

author: ElektroQuark
author_rank: Unknown
timestamp: November 20, 2017, 02:37:13 pm
content: If it last 2 years you will have the same profit

author: Zhor
author_rank: Newbie
timestamp: February 05, 2020, 05:56:22 am
content: Quote from: Kiriakos-GR on April 07, 2010, 05:50:29 pmI still use my Fluke 87 first generation, I send it to calibrate march 1995 and I think I did this 2 years after I bought it, so I think first gen was on 1992-1993, I calle it the best gear of work on electronics I have ever buy, it cannot be betterSecond best gear for work is my PACE MBT-250 Rework station

author: bluey
author_rank: Regular Contributor
timestamp: April 24, 2020, 11:30:12 pm
content: I bought my Fluke 87 in 1991. The manual errata dates the original manual as August 1988.I got it to be a lifetime hobby meter. It was my second multimeter after an analog Radioshack kit meter. It had a dud test lead from the beginning, replaced under warranty. Battery leak corroded battery clip and has been replaced. LCD had fading out problems caused by deteriorating flexible zebra strip. Display works perfectly since that was replaced. Don’t think i’ve ever managed to blow a fuse. “Input alert” beeps if the leads are plugged in the wrong sockets.The most amazing thing to me is how little has improved in 30 years. Multimeters are much the same size and functionality is not vastly improved. I only miss not having built in thermometer - have a thermometer plug in module for that. Main construction weakness is the three self tapping screws into plastic posts to attach the back, which are still ok because i am careful to keep the screw running in the original grooves.

author: EEVblog
author_rank: Unknown
timestamp: April 25, 2020, 03:10:33 am
content: Quote from: bluey on April 24, 2020, 11:30:12 pmYes, I can remember the review in Electronics Australia magazine back in the 80's, I marveled at this wonder of a multimeter!

author: Electro Detective
author_rank: Super Contributor
timestamp: April 25, 2020, 04:49:21 am
content: Quote from: bluey on April 24, 2020, 11:30:12 pmIf the screw in posts are not cracked yet (like mine are) you'll be fineOtherwise if they are cracked or snapped, the meter has lost it's ability to contain any serious pops in your handand no better than a dropped cheap plastic toythe only quick fix afaik without stuffing about with glues, epoxies and tech prayer,is to tape a up the sides (doubled up for more strength) and put back in the holster, as in the 87V photo below which my 87-1 also has,or go ugly with cable tiesNot great or pretty, but quick n easy to do, and may help keep the meter intact on a bad drop toofwiw peel off tape carefully and reuse for battery swaps

author: bluey
author_rank: Regular Contributor
timestamp: April 25, 2020, 07:15:36 am
content: For the nostalgic, original Fluke 80 Series brochures: (the files were too big to post directly)https://photos.app.goo.gl/4F1GfDfYBDxCfN956The 2 page one is copyrighted 1990.Radio Electronics November 1988 has Fluke 80 series in “New Products”.https://archive.org/details/radio_electronics_1988-11/mode/2upElectronics Australia April 1989 has some interesting history.50 year anniversary issue has reprint of the April 1939 edition. Page 80 has an article on the next 50 years. Fluke 87 is in an ad on page 211.https://archive.org/stream/EA1989/EA%201989-04%20April#page/n81/mode/2upElectronics Australia June 1990 has an excellent article on choosing a DMM on page 160 - describes the theory and operation of DMM.https://archive.org/stream/EA1990/EA%201990-06%20June#page/n159/mode/2up

author: bluey
author_rank: Regular Contributor
timestamp: April 25, 2020, 09:38:31 pm
content: Electronics Australia July 1989 page 26 has review of Fluke 85, not 87.https://archive.org/stream/EA1989/EA%201989-07%20July#page/n23/mode/2up"We don't have access to a calibration lab facility, so we weren't really able to give the mode 85 a rigorous calibration check."

author: bluey
author_rank: Regular Contributor
timestamp: April 25, 2020, 10:01:17 pm
content: Fluke 87 plastic quick reference card.https://photos.app.goo.gl/CXhHMZboJKdydBYA8

author: Electro Detective
author_rank: Super Contributor
timestamp: April 26, 2020, 04:05:36 am
content: Thanks bluey87-1 still rocks, if I had to flog one of the two I'd sell the 87V first...and get better money for it

author: bluey
author_rank: Regular Contributor
timestamp: April 26, 2020, 09:31:02 pm
content: Quote from: Electro Detective on April 26, 2020, 04:05:36 amSo according to the 87-V manual, the 87-V doesn't have the self diagnostics that the 87-I has. Why would they delete a feature like that???I presume that a valid self test 1k ohm resistance value implies that the voltmeter and constant current generator are both working properly (or both completely wrong in sync - unlikely).

author: Electro Detective
author_rank: Super Contributor
timestamp: April 27, 2020, 01:46:09 am
content: Not sure on that, maybe the 87V has that tool less software calibration thing (?) and or operates differently ?Have to hang for someone who knows better as it's been a while since I looked into the innards and differences on meters,they've been working great and the batteries last ages, so no reasonI'll have to score an 87-111 too one day to complete the 87x set,and confuse myself more

author: bluey
author_rank: Regular Contributor
timestamp: April 28, 2020, 03:00:20 am
content: Quote from: EEVblog on April 07, 2010, 10:46:52 pmReading the service manuals, the first manual version is dated 1989-1996 (covers 1988-presume 1996 models). The second version is dated 1997, and applies to models marked front and back with "CAT III 1000V" on the top and bottom case. I presume this is the unlabelled 87-II CAT III compliant. (The first version is just marked "1000V", and has 600V fuses.)http://assets.fluke.com/manuals/83_85_87smeng0500.pdfThe 87-II service manual is named "838587_1000_smeng0000.pdf". I presume i got the link from here somewhere but can't find it now.(Found a sourcehttps://xdevs.com/doc/Fluke/83_1000_smeng0000.pdf)87-II [vs 87-I] specifications are very similar - difference listed below (the ones I could find):------------------------------------------------------------------------------------------------------------------------------------------VAC accuracy 400V 5kHz-20kHz unspecified [+/-(2% + 20)]VAC accuracy 1000V 1kHz-5kHz unspecified [+/-(2% + 4)]ohms accuracy 400ohms +/- (0.2% + 2) ** using REL [+/- (0.2% + 1)]Capacitance accuracy 5.0 microF +/- 1.9% +3 [+/- (1% +3)]mA AC and DC 40mA nad 400mA burden voltage typical 1.8mV/mA [1.6mV/mA]typical ohms short circuit current:400 ohms 200 microA [700 microA]4k ohms 80 microA [170 microA]40k ohms 12 microA [20 microA]400k ohms 1.4 microA [2 microA]ohms short circuit current <200 microA [<500 microA]diode test short circuit current typical 0.6mA [1.0mA]Fuses:mA or microA 44/100A 1000V FAST [1A 600V FAST]A 11A 1000V FAST [15A 600V FAST]Safety protection class: IEC 1010-1 1000V, CAT III; UL3111; TUV EN61010-1; CSA C22.216 1010.1 [Class II per IEC348, ISA-DS82, UL1244]------------------------------------------------------------------------------------------------------------------------------------------80 Series III service information is dated April 1998-2002. Specifications look the same as presumed series II which only lasted about a year.It looks like all models can verify against 1k ohm internal resistance (R43 metal film 1k 0.1% resistor- F1 fuse test result: 1000+/-5) by connecting V/ohm/diode input to microA/mA input (upper half). In the 87-I service manual, this is described under "Performance Tests". For the others it is described as a "F1 fuse test". Somewhere I read that the 1k ohm resistor is a high precision resistor, therefore if the number is correct it implies that the voltmeter and constant current generator are working to spec.

author: bluey
author_rank: Regular Contributor
timestamp: June 06, 2020, 10:48:19 am
content: Reading the theory of operation more carefully, the resistance function is a comparison function and doesn’t use a “constant current circuit” per se. Therefore the “sanity check” internal resistor test doesn’t achieve that much apart from being a fuse test. The voltmeter could be completely wrong and resistance of the internal resistor would still measure perfectly correct.Fortunately, the band gap voltage reference in the 87-I to 87-III is quite stable. The LM285BXZ-1.2 Micropower Voltage Reference Diode is described on page 7-58 of the National Semiconductor Linear Databook Vol 2 1988. Long term stability is 20ppm per 1000 hours. Temperature stability is 30ppm/degC.LMx85-1.2-1988NationalLinearDatabookVolume2.731107566.pdf(269.77 kB - downloaded 445 times.)https://usermanual.wiki/Document/1988NationalLinearDatabookVolume2.731107566.pdfMy 1991 87-I never adjusted reads the same as my new Hioki DT4222 - at least for volts.https://www.eevblog.com/forum/testgear/hioki-dt4222-compact-trms-multimeter-teardown/msg3089246/#msg308924686-V and presumably 87–IV (cant find a document listing the parts) use a higher spec 2.5V voltage reference.https://www.eevblog.com/forum/beginners/what-is-the-voltage-reference-part-in-the-fluke-87-v/msg3089331/#msg3089331LT1790ACS6-2.5 Micropower SOT-23 Low Dropout Reference Family has tolerance of 0.05% and thermal stability of 10ppm/degC. The voltage reference in 87-I to 87-III has tolerance of 1% and needs trimming to be accurate.