Repair for Fluke 123, 124, 125 Scopemeter LCD Display .. what ?  - Page 1
Forum section: electronics / repair

author: Kiriakos-GR
author_rank: Super Contributor
timestamp: November 22, 2011, 03:50:29 pm
content: I found on Ebay the offer of a replacement screen about those Scopemeter for 135$, and the new question will be: do they get damaged ? and how easy is to happen ?http://www.ebay.com/itm/ws/eBayISAPI.dll?ViewItem&item=230416504476&viewitem=&sspagename=ADME%3AB%3AWNAFP%3AMOTORS%3A1123

author: kaz911
author_rank: Super Contributor
timestamp: November 22, 2011, 04:48:56 pm
content: seems like quite a few of the 12x and 19x series B&W displays have the problem. I think the repair guys resets (rework) the sockets/connectors. But only if it is horizontal lines. Vertical lines require new screen. Other "normal" problems with the 19x series seem to be "dead channel A" or "dead channel B"I have followed quite a few talks on Fluke.com and other places since I considered a Scopemeter for a while.

author: Monittosan
author_rank: Regular Contributor
timestamp: November 05, 2014, 07:06:33 am
content: you can rework the LCD ribbon cable with a soldering iron on its lowest temp setting.The challenge is getting to the vertical ribbon cable as its behind the glass

author: all_repair
author_rank: Frequent Contributor
timestamp: November 05, 2014, 09:09:09 am
content: Be careful. Do not use strong solution. The ribbon is glued to the display and WD40 will sure dissolve them. Reheating may help with some pressure, but I have not tried them myself.

author: Monittosan
author_rank: Regular Contributor
timestamp: November 11, 2014, 03:44:37 pm
content: Well I was able to fix it!The trick is creating a tip for your soldering iron to re bond the ribbon cable. I made a longer tip out of copper wire from coax cable to get under the glass and head up the ribbon on the board. Temperature of the copper tip was about 100 degrees centigrade and in the beginning you will make more lines appear before fixing any. If you still have faint lines appearing you need more heatbeforeDuring repairNotice the faint lines.After

author: miguelvp
author_rank: Super Contributor
timestamp: November 12, 2014, 02:03:17 am
content: Good job,Did you use the scope in sub zero temperatures btw? I've seen that LCD dead pixel pattern before

author: Monittosan
author_rank: Regular Contributor
timestamp: November 13, 2014, 12:13:28 pm
content: I've purchased it second hand from the US. Originally I thought that yellowing was a blemish or something that was rubbed onto the screen?So the LCD fails in extremely cold conditions? makes sense as to why the screen had so many lines as well! (about 18lines :S )Is there anyway to restore those pixels?

author: dexters_lab
author_rank: Unknown
timestamp: November 13, 2014, 03:11:47 pm
content: interesting, i have a kurzweil synth keyboard with a similar issue with the LCD, i might try that technique!

author: Monittosan
author_rank: Regular Contributor
timestamp: November 14, 2014, 01:29:17 am
content: Post your findings!

author: zoiasso
author_rank: Contributor
timestamp: March 05, 2019, 10:34:56 pm
content: I bought here in Brazil a fluke 123 without display (for repair)Searching for displays for fluke 123 i found thishttps://item.taobao.com/item.htm?spm=a1z10.5.w4002-2339678517.20.SZvF5r&id=38243727275&spm=a1z10.5.w4002-2339678517.20.SZvF5rits a chinese lcd, its smaller than the original.I bought it today 03/05/2019, when it arrive in the next month, i will try to make a video, maybe it helps some people.On Brazil the price of the display "new" is 3.300 BRL (1k dolar~). I paid 150 BRL (less than 50 dollar) for the fluke 123 for repair, just to clearify why i did that.

author: zoiasso
author_rank: Contributor
timestamp: March 16, 2019, 02:19:55 am
content: product out of stock =/

author: Rriis
author_rank: Newbie
timestamp: July 30, 2019, 02:05:03 pm
content: Hi ZoiassoWas the chinese LCD out of stock?Did you figure something out? I need a new LCD aswell.

author: rvalente
author_rank: Frequent Contributor
timestamp: August 03, 2019, 02:36:40 pm
content: Off topic: do you guys like the use and feel of these equipment? Used a few times for diagnosing dc motors convertes, what a terrible experience. The screen refresh rate is horrible, the waveform does not looks like a waveform, but an abstract work of art in chunky pixels....Sent from my LG-H870 using Tapatalk

author: zoiasso
author_rank: Contributor
timestamp: September 19, 2019, 09:48:57 am
content: Quote from: rvalente on August 03, 2019, 02:36:40 pmy it was out of stock, the scope still sitting here. Im a newb in electronics i wanted to know if there's any way to adapt any kind of display into this fluke 123, but i have no idea from where to start to study this xd.

author: zoiasso
author_rank: Contributor
timestamp: September 19, 2019, 09:50:48 am
content: Quote from: zoiasso on September 19, 2019, 09:48:57 ami never used it, i got it for R$ 100,00 without a display, i think it is working. i use a bench hantek DSO1502p, but i still want to find a display for the fluke123

author: mansaxel
author_rank: Super Contributor
timestamp: October 12, 2019, 06:59:07 am
content: Quote from: rvalente on August 03, 2019, 02:36:40 pmYes, I've got one literally since it came out. Got it as a gift on the release party (where dealers got to see it in the flesh for the first time) because I was working with sound, video and lights on the show.No, it is not a full-fledged scope, but it is very useful for LF work and as an added confidence on top of the DMM. And that was its stated purpose! They wanted to build a graphing multimeter for a 200lb gorilla poking about in power circuitry, and there it performs splendidly. For starters, it is isolated. No worries where you put your probe or ground clip, you won't create a circuit to anything strange like you would with a normal scope.

author: CyberKnight
author_rank: Newbie
timestamp: February 10, 2020, 09:31:24 am
content: Monittosan, "I made a longer tip out of copper wire from coax cable to get under the glass and head up the ribbon on the board." - Do you have any pics of the exact repair area for the ribbon connections you mention? (Or will it be obvious once the display is removed from the case?) Thanks!