Rigol DS1104Z Plus unlock? - Page 1
Forum section: electronics / testgear

author: cw_cdz
author_rank: Newbie
timestamp: July 08, 2017, 06:26:09 am
content: I've been scanning through the forumn and having a hard time locating if this is still a thing? I found this site:http://gotroot.ca/rigol/riglol/and I am running firmware 00.04.04.SP1. I have not yet tried entering a code (seems there coudl be a lockout) and I'm unsure if I really NEED to update the firmware on this model. Most of the models I seem to have found "hacks" for are for the 1054Z upt to the 1104Z. Any info would be much appreciated.

author: CharlieWorton
author_rank: Regular Contributor
timestamp: July 08, 2017, 08:35:17 am
content: Not sure which scope you have. The link you provided will - I believe - unlock the embedded features in a Rigol 1054Z scope and effectively turn it into an 1104Z. But firmware updates - as opposed to the above hack, which is not a firmware update - are obtained from the Rigol website. Rigol generally ships their scopes with the latest version of the firmware, so unless you've had your scope for a while you probably don't need to update it (although it never hurts to check).The link you provided will only interact with a Rigol 1054Z scope. If you have a different scope, then the link would not be applicable to you.>Charlie

author: ebastler
author_rank: Super Contributor
timestamp: July 08, 2017, 05:54:32 pm
content: CharlieWorton is close, but not quite correct. The key generator you linked to will work nicely with all DS1000Z series scopes (i.e. DS1054Z, DS1074Z, DS1104Z). For these, this simple hack certainly is "still a thing".However, the DS1000ZPlusseries is a close relative of the MSO1000Z series. And for those scopes, the simple key generator hack has never worked. There is a significantly more complex approach which achieves the same for these scopes. Some threads on this forum discuss this -- not sure whether these are the latest and simplest instructions:https://www.eevblog.com/forum/testgear/you-can%27t-unlock-a-mso1000z-series-scope-without-a-memory-dump-and-other-lessons/https://www.eevblog.com/forum/testgear/sniffing-the-rigol's-internal-i2c-bus/msg569236/#msg569236

author: Fungus
author_rank: Super Contributor
timestamp: July 09, 2017, 11:27:02 am
content: Quote from: CharlieWorton on July 08, 2017, 08:35:17 amThe clue is in the title: "DS1104Z Plus"

author: LuckyMan
author_rank: Newbie
timestamp: September 11, 2017, 11:30:48 am
content: Hello, I was reading a lot over here, lot of people is writing about 1054 unlock, but nobody is writing how to lock it back. It is possible?After unlocking, your warranty is gone, so there is someone, who succesfully locked back his 1054Z ?If not I think is nonsense unlocking it.

author: Fungus
author_rank: Super Contributor
timestamp: September 11, 2017, 11:59:17 am
content: Quote from: LuckyMan on September 11, 2017, 11:30:48 amIt's been mentioned many times (but not as often as unlocking, for obvious reasons).Quote from: LuckyMan on September 11, 2017, 11:30:48 amYes. Look for "uninst" in the programming manual.Quote from: LuckyMan on September 11, 2017, 11:30:48 amDon't worry, it's easy.

author: rstofer
author_rank: Super Contributor
timestamp: September 11, 2017, 05:52:00 pm
content: Quote from: LuckyMan on September 11, 2017, 11:30:48 amMy warranty was gone the moment I decided I wanted a quieter fan and detents in the selection rotary encoder.Given that, I just went ahead with the rest of the upgrades. The features I wanted, the specific reasons for buying the scope given that I have a 350 MHz Tek 485, would require a bit of a hack.I had read nothing re: infant mortality so I had no reason for concern about warranty repair. I don't remember the actual purchase date but I suspect my warranty is just about up. If I had one...

author: Fungus
author_rank: Super Contributor
timestamp: September 11, 2017, 07:04:06 pm
content: Quote from: rstofer on September 11, 2017, 05:52:00 pmThere's been very few complaints about reliability here, if any. There's been a few units with a bad channel when new but long term reliability doesn't seem to be a problem.They're solidly built and the tight integration on the main board should make the electronics very reliable. The weak pointmightbe the power supply but I don't remember reading about power supply problems even after 2.5 years.Whatever else you might have to say about the DS1054Z, the hardware gets a thumbs up!

author: Jacon
author_rank: Regular Contributor
timestamp: September 12, 2017, 02:38:02 pm
content: Quote from: rstofer on September 11, 2017, 05:52:00 pm+100

author: LuckyMan
author_rank: Newbie
timestamp: September 13, 2017, 06:43:48 pm
content: Thank You. Let me write here for other beginners how to lock back device, and get back to Trial options:- Connect with USB your device to your computer.- Start on your computer Ultra Sigma software, right click on your device, choose SCPI Panel control- In the field SCPI Command type exactly what is here: :SYSTem:OPTion:UNINSTall- Click Send Command- Done. Trial versions are backDo not hack devices. Someone was working hard to make it. Respect they job and others will respect yours

author: Fungus
author_rank: Super Contributor
timestamp: September 13, 2017, 06:50:26 pm
content: Quote from: LuckyMan on September 13, 2017, 06:43:48 pmUltra Sigma?? Ewwwwww!Save yourself 500Mb of disk space and a horrible experience: Connect an Ethernet cable then telnet to port 5555 on the Rigol.