Hantek HDM3055 - Page 1
Forum section: electronics / testgear

author: k0st1z
author_rank: Newbie
timestamp: October 28, 2024, 07:45:27 pm
content: Hello All,I am having hard time with the Hdm3000 software and the Hdm3055 multimeter. They show different messurments. The measurement at the multimeter is correct.I attach photos.Please for your help

author: coromonadalix
author_rank: Super Contributor
timestamp: October 29, 2024, 12:31:53 am
content: could be a glitch in the software, a matter of division values etc ... report to them ... unless the Meg Ohms would be Milli Ohms ...posting here could not be useful as reporting them any error you see, just saying ..and they may fix it ?

author: Kean
author_rank: Unknown
timestamp: October 29, 2024, 02:48:53 am
content: Looks like a software bug.Both values have significant digits starting with "63", and the screens both say "Auto 100Ω" indicating it is in the 100 ohm range, and yet the PC shows a value in mega ohms.Edit: It also looks like the remote display says "100 PLC" vs the DMM which is set to "10 PLC", and the trigger mode show above the reading is also different.

author: Aldo22
author_rank: Super Contributor
timestamp: October 29, 2024, 06:04:57 am
content: Quote from: coromonadalix on October 29, 2024, 12:31:53 amAsk zhangqiang@hantek.com , so you have a chance of getting an answer.Whether it will be fixed is a different question.Han ZhangqiangQingdao Hantek Electronic Co.,Ltd.Add:35# Building, No. 780 Baoyuan Road,Qingdao City, Shandong Province, China 266114Website:www.hantek.comEmail:zhangqiang@hantek.comTel:+86-0532-55678770-662

author: Microcheap
author_rank: Frequent Contributor
timestamp: October 29, 2024, 07:20:29 pm
content: The PC software is buggy and hasn't been updated, unfortunately.I added support for these Hantek multimeters for theTestControllersoftware that was developed by another forum member and is discussed here:https://www.eevblog.com/forum/testgear/program-that-can-log-from-many-multimeters/msg5685345/#msg5685345The TestController only supports LAN and RS-232, not USB. You will have to use the serial port of the DMM or, if you dare, add the missing parts to have the LAN working:https://www.eevblog.com/forum/testgear/finally!-hantek-hdm3065-6-5-digit-dmm/msg5530465/#msg5530465If you know how to program, you could try make your own software using Hantek'sSDK

author: k0st1z
author_rank: Newbie
timestamp: February 23, 2025, 05:24:30 pm
content: Hello microcheap, please help me.i tryed to connect the multimeter with usb to serial. I get error "time out"I add the missing parts from the lan port but i can't get connection.Any idea ?

author: Microcheap
author_rank: Frequent Contributor
timestamp: April 15, 2025, 10:58:29 pm
content: I know it's been some time, but just recently I found the cause of the issue with the DMM software, it's a silly bug but I don't use the software anyway.The problem is caused by the locale configurations in the Windows Region settings. If the format is set to use comma as decimal symbol the Hantek software will show the wrong readings. Just change the decimal symbol to use point instead (in this case I also recommend to change the digit grouping to comma, otherwise you can have problems with some other application, like FreeCAD).