EEVblog 1378 - NEW Uni-T UT61E+ Multimeter - Page 1
Forum section: electronics / general

author: EEVblog
author_rank: Unknown
timestamp: March 01, 2021, 05:21:06 am
content: Is the new Uni-T UT61E+ Multimeter still the bang-per-buck king the existing UT61-E was? How does it compare? Is the increased cost worth it?https://s.click.aliexpress.com/e/_AaRaJShttps://s.click.aliexpress.com/e/_9foaZOAnd what about the ANENG AM870 at half the price?https://s.click.aliexpress.com/e/_AFiPjA

author: SeanB
author_rank: Super Contributor
timestamp: March 01, 2021, 06:15:13 am
content: Nice meter, though 20A DC measurement, with a 10A fuse...... Guess it is the same current shunt, just with the "30 seconds maximum" there to prevent you blowing the fuse too often.Waiting for Joe to blow it up, and if it will survive the grill starter lowest energy test that he has. My bets are it might just barely pass, but will fail on the higher energy tests. But at least they spent the 30c and added in the 3 MOV's to it.Looking at the high resolution photoset, soldering on the input side not the best, with the lovely splatter on the input side.https://www.flickr.com/photos/eevblog/albums/72157718441411852/with/50980777783/Nice photos there Dave, that cardboard box is doing great work!

author: Wytnucls
author_rank: Super Contributor
timestamp: March 01, 2021, 03:05:36 pm
content: Extra cost gives you some extra features:AC+DC VoltMin/MaxVFDhFEBacklightLatched continuityNCV

author: HKJ
author_rank: Super Contributor
timestamp: March 01, 2021, 03:26:19 pm
content: The safety of the old UT61E depended on model, there was a EU model with better safety.I wonder if the new model has same issue in the current ranges as the old model, the burden voltage is a bit high in the high uA and mA range.

author: MikeK
author_rank: Super Contributor
timestamp: March 01, 2021, 06:16:42 pm
content: Is it a problem that the new one won't accept a standard BNC female to dual banana male for mA?

author: madires
author_rank: Super Contributor
timestamp: March 01, 2021, 09:13:20 pm
content: 10 or 15 bucks more would be fine, but 30 is simply too much for the improvements. And I fully agree that the hFE measurement is a complete waste and should be replaced with a temperature measurement.

author: Brumby
author_rank: Unknown
timestamp: March 01, 2021, 11:34:32 pm
content: Get rid of that chunky transistor adapter and save a few bucks right off the bat!

author: CChin254
author_rank: Regular Contributor
timestamp: March 02, 2021, 12:36:20 am
content: Design Wins:IC Identification:The other version (UT61E+) contains a AD737 RMS/DC Converter instead of the Cyrustek Co. solution used in the UT61E+).

author: jivaei
author_rank: Newbie
timestamp: March 02, 2021, 10:21:49 am
content: One other thing to note is that the old UT61E is hardwired to not auto shutoff, presumably for RS232. The UT61E+ should support auto shutoff as it can detect if the RS232 receiver is inserted.

author: Fungus
author_rank: Super Contributor
timestamp: March 02, 2021, 06:14:08 pm
content: Where's the 10MOhm input resistor chain? I can't see it.Definitely give the Aneng 870 a spin but get some of these super needle-sharp gold probes to go with it:https://es.aliexpress.com/item/33006553337.html:-)The only thing I see that this has going for it over the Aneng is USB interface and faster autorange. And a better stand.Wasting a whole switch position on hFE and an NCV function that barely works is unforgivable. My Aneng 870 can detect a live wire from about 10cm away (4 inches).Edit: That's notquitetrue. It can detect a pug board with 4 plugs in it from about 10cm away. A single unloaded mains cable makes it beep at 3-4cm depending on the angle of the cable.

author: HKJ
author_rank: Super Contributor
timestamp: March 02, 2021, 06:15:45 pm
content: Quote from: Fungus on March 02, 2021, 06:14:08 pmIn the black box, together with the other input divider resistors.

author: Fungus
author_rank: Super Contributor
timestamp: March 02, 2021, 06:20:39 pm
content: Quote from: HKJ on March 02, 2021, 06:15:45 pmThe old version has MELFsanda black box.

author: floobydust
author_rank: Super Contributor
timestamp: March 02, 2021, 06:21:29 pm
content: Why aren't we calling out fake regulatory claims, especially when they are obvious?The DMM is the usual fail - 250VAC fuses (with lots of room for bigger ones) - Intertek would not give 61010 approvals beyond that, unless they were sleeping or bribed. How did this achieve claimed 600V or 1,000V certification? I remember only one or two of the Uni-T UT61 models had a legit certificate and the others did not. One reason to stretch the truth and cling to an old model number.You can't put two PTC thermistors in series and expect equal sharing. One always heats up first and then takes the brunt of the HV whilst the other one sleeps. It's like two polyfuses in series, all the voltage can appear across one when tripped. So I would check a single PTC is rated for the overload voltage expected.%Hz is plastered at seven places on the front labels, enough already.

author: mariush
author_rank: Super Contributor
timestamp: March 02, 2021, 06:40:47 pm
content: My guess is that they didn't want to make a custom dongle bit just for this series, so they added the hFe feature to support the transistor bits on the dongle.I didn't see it mentioned, but the another plus for the new meter is the 3 updates/s versus 2 updates/s for the old model.The 9mA power consumption is a bit weird, considering the old regular ut61e consumed much less, and worked with very low voltages... was showing low voltage at 5.6v but worked with around 3vBut hey, plenty of room to add a lithium "pouch" style battery and a basic charging circuit in that square.Also have to wonder, considering the multimeter chip advertises 30000 count, maybe it's possible to modify some parameters or code to make the meter stay within a range up to 29999 or something like that.The IC also advertises running at 3v DC so maybe you don't even need 4 batteries, or using a single 3.7-4.2v lithium battery would be very easy to use.The datasheet is available on Cyrustek website:http://www.cyrustek.com.tw/spec/ES232.pdfIt has info about the serial comunication between IC and mpu, along with the commands and how to decode everything, so you can

author: Fungus
author_rank: Super Contributor
timestamp: March 02, 2021, 06:44:14 pm
content: Quote from: floobydust on March 02, 2021, 06:21:29 pmIs Intertek certifying this one?There's a UT161E whichiscertified but it costs a lot more:https://www.batronix.com/shop/multimeter/UNI-T-UT161E.htmlI wonder what's inside? Could the 61E be converted to a 161E? It looks like there's holes on the PCB that would would let you desolder the fuse holders and move them further apart to accept 1000V fuses.Also interesting is that the MOVs are labelled "SG1", "SG2"... SG is usually "Spark Gap".

author: thm_w
author_rank: Super Contributor
timestamp: March 02, 2021, 10:05:21 pm
content: Quote from: Fungus on March 02, 2021, 06:14:08 pmIts dangerous, you create a false sense of security that the wire is not live, start working on it and get zapped.You want a relatively sensitive NCV if you are doing any sort of actual electrical work (eg drilling in walls), the ones in these meters is just trash.They wouldn't do it but: get rid of NCV, get rid of HFE plugin, get rid of the USB adapter (sell it as an optional extra). This would bring the price down a bit. But I suspect a lot of sales are made from these useless "features".

author: floobydust
author_rank: Super Contributor
timestamp: March 03, 2021, 02:13:39 am
content: Quote from: Fungus on March 02, 2021, 06:44:14 pmThe prior UT61E is a two-faced dog. I see Uni-T has obtained approvals for the UT61E under TÜV Rheinland GS Mark, then sells versions with stripped out protection components (MOV's). They likely carried over that 61010 certificate to other agencies such as Intertek.Intertek ETL product certification database:Company: Uni-Trend Technology (China) Co., Ltd. - Dongguan, Guangdong CHINATrade Name(s): BESANTEK, AMPROBE, EXTECH INSTRUMENTS, Minipa, UNI-T, actron, BOSCH.UT61A, UT61B, UT61C, UT61D, UT61Eare listed but not the UT61E+.edit: I expect these larger Asian test equipment manufacturers to have some integrity and not make fake 61010 claims about their multimeters.Why else would a BBQ lighter kill, why have 250V fuses with 1,000V claims, why depopulate the MOV's for everywhere but Germany?Although, Uni-T's 61010 test reports are quite a bit how ya doin' and the certifier could be glossing over the details.Irregardless, it's just shady and aren't we tired of this?

author: Fungus
author_rank: Super Contributor
timestamp: March 03, 2021, 09:33:06 am
content: Quote from: floobydust on March 03, 2021, 02:13:39 amYep. Cheap meters have their place but fake CAT ratings could put people's lives at risk. Not everybody reads EEVBLOG forums.

author: SilverSolder
author_rank: Super Contributor
timestamp: March 03, 2021, 03:01:04 pm
content: Quote from: floobydust on March 03, 2021, 02:13:39 amIsn't this down to how much "teeth" the national regulatory agencies have? Write to your local representative?

author: floobydust
author_rank: Super Contributor
timestamp: March 03, 2021, 06:51:32 pm
content: It's possible Germany is strict and bans imports of products with no safety approvals/certificate, unlike all the too nice nations of the world.I think it's more affectionately known as the "Golden Sample". The one that looks great, lures customers and passes regulatory. Then you swap in the cheaper version.The old UT61E MOV's are USD $0.08/1k for a set of three. So in the bait and switch they saved pennies. I don't know of any other company playing games like that, they should grow up.For the UT61E+ my beef is no approvals, the word 'Electrician' on the faceplate, they kept the allocated room for larger fuses (like the UT61E) but decided 250V is fine. Why bother with a plastic guard box around the fuse if you're only expecting 250V there? It makes no sense. I feel better when the meters have actually been tested to their rated 1,000V to prove the pcb clearances are correctly done.

author: Fungus
author_rank: Super Contributor
timestamp: March 03, 2021, 09:22:46 pm
content: Quote from: floobydust on March 03, 2021, 06:51:32 pm...or make nice meters but put real ratings on them.Even so, joe has killed many UNi-Ts con a grill starter. I still wantsomerobustness even on a meter thatisn'tindustrial-strength CAT rated.

author: Kleinstein
author_rank: Super Contributor
timestamp: March 03, 2021, 11:30:36 pm
content: The regulations / customs should be similar in the whole EU - so it would not only be Germany to get a different version. German customs may be a little more strict in inforcing the regulations than some other EU countries. This sometimes can be a problem ordering parts from China that miss the CE stamp. There were different versions in the EU. I don't remember much better protection, but a lower, more realistic CAT rating. The the EU version may end up with a 250 v CAT 2 or similar. At least they note the limited fuse Voltage rating and don't keep it secret.There is nothing really wrong with a CAT 2 meter if labled correctly. There is a market for these at the lower cost end.The transistor tester is probably not worth the weight - it has some interresting lookign contacts to put SMD transistors inside. Proper SOT23 test sockets are kind of expensive.Still one can get a much better transistor tester for some $20 or so.For the battery compartment, it may be interesting to see if it is water tight.Instead of HFE function they should add a leaking battery warning.

author: EEVblog
author_rank: Unknown
timestamp: March 03, 2021, 11:37:17 pm
content: Quote from: floobydust on March 03, 2021, 06:51:32 pmThe plastic guieard is there for the fuse to switch contact voltage clearance. That's a different failure mode to the fuse break voltage rating.

author: floobydust
author_rank: Super Contributor
timestamp: March 04, 2021, 02:31:49 am
content: After the fuse has cleared, it has hazardous live on one end and between both ends. The plastic guard is on the (blown) fuse's hot side. With a 250V fuse, that node would go up to its breakage voltage rating of 250V or 1kV, which seems low to warrant building a guard there. But the 3D of the rotary switch I guess you'd have to see or test.Let's say the guard might be for the other direction, no arc from rotary switch to the fuse. But again the guard is at only one end of the fuse, ignores the other end, so that's not the design intent. The MOV's are supposed to limit the voltage at the rotary switch though. The PTC's are considered highest voltage, so the trace running off from PTC4 to the rotary switch is likely a disaster.The point of a formal regulatory assessment is they measure creepage and clearances, and test with high voltage to verify a safe PCB/product design. Every PCB designer has the best intentions yet they do make mistakes.

author: Kleinstein
author_rank: Super Contributor
timestamp: March 04, 2021, 09:59:55 am
content: The other / non shielded end of the fuse has a link to the main part of the circuit / rotary switch anyway. So there is no need to also have a shield or guard at that end of the fuse.The line running from PTC4 to the rotary switch may be clamped with some device - it could be some clamping diode or similar. We don't know the resistance of the PTCs - if they are relative high resistance to start with, they may be oK without an extra series resistor.

author: free_electron
author_rank: Super Contributor
timestamp: March 08, 2021, 02:27:21 am
content: Two disqualifiers- numeric aspect ratio. sorry. only Fluke and HP get this right. these elongated characters are hard to read. disqualified- non standard plug arrangement. prevents me from using things like 0.75 inch pitch bnc to banana adapters, thermocouple interfaces and so many other banana plugs on 0.75 pitcheevblog multimeter has same issue . get those banana jacks in line. not all weirdo.

author: Monkeh
author_rank: Super Contributor
timestamp: March 08, 2021, 03:14:28 am
content: Quote from: free_electron on March 08, 2021, 02:27:21 amI'm pretty certain they are 0.75". All my 'weirdo' ones are, despite not being in a neat line.

author: free_electron
author_rank: Super Contributor
timestamp: March 08, 2021, 03:49:09 am
content: Quote from: Monkeh on March 08, 2021, 03:14:28 ambut i want them in line... many pomona boxes and other 'adapters' bank on the 4 terminals to be either in line or in 'quad' (calibrators for example have ready wired blocks you simply plug in) . The key ones are the pitch between ground and Volt-ohm and ground-200mA range. for low current measurements you may want to use coax or triax cables and then you need a bnc or to banana adapter. in the old version at least the mA/uA range is next to COM , in the new version it is too far away. the 10A range sits next to COM .. FAIL !But the big stinker is the aspect ratio of the numbers. They are too hard to read. Fluke has the ideal ratio becasue they copied it from HP's 7 segment displays. ( fluke is actually a smidgen wider than a HP 7 segment. Also the way the segments flow into one another ( the chamfered edges ) makes a big difference in readability. The numbers need to be designed just right when it comes to width , height , pitch and segment shape / angle. Fluke nailed it. So did HP. This elongated crap ? unreadable.

author: WackyGerman
author_rank: Regular Contributor
timestamp: March 09, 2021, 09:54:03 am
content: They did some improvements what is really good , but it is still a bad meter . They sell it as an electricians meter , but it s definetively not . The CAT rating is still a lie , the fuses are only 250 V and do not have the rupture capacity they need for a proper CAT IV 600 V meter , the input protection is still lousy and the clearance is not good enough . Still many work to do for Uni-T

author: Kleinstein
author_rank: Super Contributor
timestamp: March 09, 2021, 10:55:07 am
content: Quote from: WackyGerman on March 09, 2021, 09:54:03 amNot much work needed: just replace the CAT rating with a realisting 250 V CAT 2 .

author: Fungus
author_rank: Super Contributor
timestamp: March 09, 2021, 12:42:21 pm
content: Quote from: Kleinstein on March 09, 2021, 10:55:07 amOr... if you're an "electrician" you can buy a Fluke 101 for half the price and stop worrying about fuses.

author: madires
author_rank: Super Contributor
timestamp: March 09, 2021, 12:52:25 pm
content: A sparky uses a Duspol or similar tool for checking voltages and a clamp meter for measuring current.

author: WackyGerman
author_rank: Regular Contributor
timestamp: March 09, 2021, 02:23:30 pm
content: Yes for 0 V check I use a voltage checker ( Fluke T150 )like demanded by the safety regulations and for current measurement I use a clamp meter , also for safety reasons . But for voltage , resistance and diode measurements I need a multimeter . But that s not the point .The point is that UNI-T sells equipment with wrong CAT ratings , poor input protection under a wrong description , in this case as an electrician multimeter . For electricians this multimeter is absolutely not suitable .If you label a multimeter with CAT IV rating the user expects CAT IV rating , nothing else . If you order and pay a car with 450 hp you expect a car with 450 hp , not 45 hp . If they would label the multimeter with CAT 2 and sell it as an electronics multimeter all would be fine . The problem is that UNI-T is kidding and lying so they habe a bad reputation and that s right

author: madires
author_rank: Super Contributor
timestamp: March 09, 2021, 03:04:54 pm
content: I fully agree that labeling the UT61E+ as "Digital Multimeter - Electrician" is misleading nonsense which can harm people. And the common use of underrated fuses in El Cheapo DMMs is worrying. I've seen too many 250V glass fuses in DMMs.

author: Lango1818
author_rank: Contributor
timestamp: March 10, 2021, 02:31:47 am
content: Re UT61E+ software.Hi all, I had a hard time finding the link for the software for the UT61E+ . The UT61E software does not work on the UT61E+. Nor would UltraDMM work. After spending some time roaming around on the net I found this link to the softwarehttps://www.uni-trend.com/meters/html/product/NewProducts/UT61%20161%20Series/UT61E+.htmlWorked first time, although as Dave said on his vid, I would rather have a Temp probe than the transistor HFE doodad. Especially when you can buy a component tester from around $10. I hope some finds this of some use.

author: hve
author_rank: Regular Contributor
timestamp: March 15, 2021, 05:10:44 pm
content: What about this meter?Code:RuoShui 86E digital MultimeterMax 22000 countAuto rangeTemperature frequency capacitor with USB interface Transfer:https://www.aliexpress.com/item/4000178957414.htmlAll for under 50 European central bank inflatable coins ...Looks pretty similar to the UT61 and has temperature as opposed to HFe nice big screen, is this the UT61E+ killer ??Have ordered one and will share some pictures of the inner workings if you like..

author: jdutky
author_rank: Regular Contributor
timestamp: March 18, 2021, 05:53:56 am
content: Does it have a proper auto-hold function? You know, where the meter will hold the current value after some short stable period (e.g. 1/10 sec) without you having to push a button (because, however useful it might be, I don't have three hands).I've gotten really used to the auto-hold feature on my Tek DM916 and I'm not willing to pay for another digital meter unless it has gotten this right.

author: hve
author_rank: Regular Contributor
timestamp: March 18, 2021, 09:57:39 am
content: It does have HOLD, single press to activate, single press to release...

author: Fungus
author_rank: Super Contributor
timestamp: March 18, 2021, 10:05:34 am
content: That's not auto-hold though.

author: Trader
author_rank: Frequent Contributor
timestamp: March 19, 2021, 04:52:41 pm
content: Quote from: Fungus on March 02, 2021, 06:14:08 pmI have 2 pairs from 2 sellers, they are very sharp (good), but they oxide easily and the resistance is ~100 mOhm.A good (chinese) probe has resistance around 30 mOhm.ProbeMaster around 25 mOhm

author: HKJ
author_rank: Super Contributor
timestamp: March 19, 2021, 05:15:31 pm
content: Quote from: Trader on March 19, 2021, 04:52:41 pmI would expect about 10 times lower than that for good probes, i.e. around 30mOhm.

author: Trader
author_rank: Frequent Contributor
timestamp: March 19, 2021, 09:17:10 pm
content: Quote from: HKJ on March 19, 2021, 05:15:31 pmYes, my mistake, the correct tested values:Quote

author: rsjsouza
author_rank: Super Contributor
timestamp: March 20, 2021, 08:10:53 pm
content: Quote from: free_electron on March 08, 2021, 03:49:09 amAs Monkeh had said, all the meters I have/had that feature the triangular arrangement are also appropriately spaced to use the banana adapters in both current ranges - something impossible with the inline arrangement. But I also like the inline arrangement in larger meters. YMMV.Quote from: free_electron on March 08, 2021, 03:49:09 amfree_electron, I suspect you would be surprised how similar the displays of various meters are these days. Check my page at:https://sites.google.com/view/vbeletronico/display-de-mult%C3%ADmetros

author: joeqsmith
author_rank: Super Contributor
timestamp: May 13, 2021, 02:53:08 am
content: Quote from: SeanB on March 01, 2021, 06:15:13 amI was recently asked about this meter and was told they had copied many of my changes to the 61E. Doing a quick search, I didn't come across a schematic for it.The 61E would fail the first round of tests, pretty much like most UNI-T products I've looked at. I gave up some of the frequency response to improve it's ability to handle an ESD event. The PTCs were replaced and MOVs were added to handle the slower waveforms.The 61E had some of the worse burden voltages I have seen. I had added a gain stage to improve one of the ranges.It had the worse temperature drift, second to my CEM meter. I used a diode and resistor to compensate the reference. It's now one of my more stable meters.They had placeholders for the larger fuses which I took advantage of. I then designed and constructed a new shunt for it. Some of the PCB had to be beefed up.This meter would take some abuse now as I have shown in my testing.

author: jspencerg
author_rank: Regular Contributor
timestamp: June 11, 2021, 04:57:41 pm
content: This upgrade is obviously not perfect, but meter has some good qualities. Mr. Smith is the acknowledged expert about the 61e considering his extensive testing and improvements. Joe, I realize your interests are elsewhere these days, but could you give the 61e+ a look and assess the functional changes? I feel I've learned something by watching you modify the 61e.

author: Ricks
author_rank: Newbie
timestamp: September 17, 2021, 03:15:06 pm
content: The main difference between ut-61e+ and ut-161 is the fuses. ut-161 have larger 1000V fuses.USB communication:1. When the supplied usb cabel (D-09) is insertedn in the meter the 'S' for serial communication lights up in the display (automatically)2. if a ut-d07a blutooth dongle is inserted it is not detected, the 'S' do not light up (a newer ut-d07b dongle is needed ?)3. the apps, iDMM 1.0 (ut-71/ut-171/ut-181), iDMM 2.0 (ut-61/ut-161/ut171/ut-181). It did not install on an older android 5.5 phone.

author: Per Hansson
author_rank: Unknown
timestamp: September 21, 2021, 07:16:36 pm
content: Hi Dave, it would be cool if you could test this new meter with mains on ohms mode.It was an easy way to kill the old one, I'm interested if the new one is any better.I posted in Joe's thread here with more details:https://www.eevblog.com/forum/testgear/hear-kitty-kitty-kitty-nope-not-that-kind-of-cat/msg3698545/#msg3698545

author: floobydust
author_rank: Super Contributor
timestamp: September 28, 2021, 07:47:56 pm
content: FYI UT61E+Dave tried it 245VAC 50Hz on Ohms and it survived:https://www.eevblog.com/forum/testgear/hear-kitty-kitty-kitty-nope-not-that-kind-of-cat/msg3713581/#msg3713581joeqsmith tried it 300VAC 60Hz on Ohms and it survived:https://www.eevblog.com/forum/testgear/hear-kitty-kitty-kitty-nope-not-that-kind-of-cat/msg3713920/#msg3713920The PTC seems to cover it ok.

author: Scherms
author_rank: Regular Contributor
timestamp: March 30, 2022, 11:46:01 pm
content: Sorry, for the regular UT61E, not the UT61E+

author: Per Hansson
author_rank: Unknown
timestamp: April 03, 2022, 02:38:21 pm
content: That schematic posted above is for the regular UT61E, not the UT61E+

author: Eheran
author_rank: Regular Contributor
timestamp: March 16, 2023, 11:44:35 am
content: I just got a UT61E+ here in Australia (hi to all my Aussie content producers!),this one here, it came with the following issues:1. After turning the dial (and also randomly) it sometimes only displayed a "low battery" icon in the top left corner and had to be turned off and on again.Here is a videoof what the issue looks like.2. God that annoying, super loud beeping all the time! I am used to a nice and quiet 87 V. Put a bit of tape over the piezo, its ok-ish now.Anyway, I looked inside and found that 2 pins of one chip (I assume it is the µC, the numbers are ground off, so unknown pinout) had a solder bridge, see picture 1. I fixed that, see picture 2, and the issue did not appear again so far (not a lot of testing, just found the bridge and made the post about it here).What about the CAT II rating with the 250 V fuse? Would it help to just replace the fuse with a higher rated one (like what?) or were there other issues with the PCB, missing/wrong components or what not? It has all its MOFs etc. as far as I can tell, see picture 3. I do want to be able measure 400 V AC/DC somewhat safely, so any recommendations are welcome. And no, I cant use the 87 V, it is still in Germany :'(

author: sanour_islam
author_rank: Newbie
timestamp: September 25, 2023, 01:09:35 pm
content: I have a uni-t ut61E+. I used it a lot. I have a question.The uni-t ut61E+ need 4.5-6V. I want to use a 3.7V battery to run the multimeter with a booster module and make it 6V. Every Booster module have higher frequency and ripple. Is there any problem?