Wenzel Low Frequency Circulator - Page 1
Forum section: electronics / general

author: uofmrob
author_rank: Regular Contributor
timestamp: August 10, 2020, 06:58:38 pm
content: Hi Everyone,I've been following the EEVBlog channel for ages. It has taught me a lot about what I know about electronics. I'm a PhD chemist by training, but my position is an instrumentation position where I need to be able to test and repair a lot of fancy expensive equipment. I'm currently trying to build my own electron paramagnetic resonance spectrometer at home. Part of the challenge is getting an inexpensive low-frequency circulator. I found this really cool design and I'm trying to learn about op-amps and build my own. I think that it could make a really cool YouTube video!http://www.qrp.gr/circulator/RFDesign3.pdfThoughts?Rob

author: SilverSolder
author_rank: Super Contributor
timestamp: August 10, 2020, 10:32:33 pm
content: Sounds ambitious! - what does an electron paramagnetic resonance spectrometer actually do?

author: KE5FX
author_rank: Super Contributor
timestamp: August 10, 2020, 10:58:08 pm
content: Quote from: uofmrob on August 10, 2020, 06:58:38 pmIt's a nifty circuit, all right. An LMH6733 or AD8003 is a good fit. The same technique gets you a single-ended directional bridge that works down to DC, which occasionally comes in handy around the house.Quote from: SilverSolder on August 10, 2020, 10:32:33 pmSee page 335 ("A simple magnetic-resonance spectrometer")here. Same idea works withproton precession.Edit: actually they both use proton precession. For some reason I was misremembering the Scientific American project article in connection with paramagnetic electron resonance (EPR).

author: uofmrob
author_rank: Regular Contributor
timestamp: August 11, 2020, 12:22:53 am
content: Yup! Same physics behind both. EPRs look at unpaired electron spin while NMR spectrometers look at nuclear spin. Electrons are a bigger magnet so they require higher excitation energy, but smaller magnetic fields to split the quantum spin states. The big research spectrometers that cost $1.6 million dollars can do this stuff among many other things (I wrote the introduction and sections on naturally occurring radicals and metals).https://scholar.google.com/scholar?q=epr+study+biological+systems+mccarrick&hl=en&as_sdt=0&as_vis=1&oi=scholart#d=gs_qabs&u=%23p%3DdaE_5BGMGPMJMine is just for the sake of seeing anything and learning in the process.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 11, 2020, 12:34:35 am
content: I made a mock up on a breadboard and it didn't seem to work. I think I may have had op amps that aren't what I should have used. I have this one now and I soldered it to a 14 pin IC breakout board from Adafruit. It's a quad amp as I'm going for a 4 port circulator. I'm very much not an electrical engineer. I'll probably be back for lots of help from all of you experts!https://www.mouser.com/ProductDetail/Texas-Instruments/LMH6722MA-NOPB?qs=7lkVKPoqpbZgnjs%252BcyvqkQ%3D%3D

author: KE5FX
author_rank: Super Contributor
timestamp: August 11, 2020, 01:30:20 am
content: Yeah, I don't think an LMH67xx opamp is going to work very well on a breadboard.With 1 GHz-class parts, your breadboards need to look more likeQuote from: uofmrob on August 11, 2020, 12:22:53 amVery cool. I always wanted to build either the SciAm project orJoe Geller's versionbut have never seemed to get around to it. If I had, I'd presumably remember what sort of particles it uses...

author: uofmrob
author_rank: Regular Contributor
timestamp: August 11, 2020, 01:38:18 am
content: This was my first prototype using these op amps (with most of the bare wire replaced with insulated wire before I powered it up once I got the connectivity right).https://www.mouser.com/ProductDetail/NJR/NJM2137D?qs=54bfbthyeuO6TvPIDHWJog%3D%3DInitially, I just powered it directly from my bench power supply, then I watched a bunch of videos and learned more about op amps and figured out that I needed to use my dual tracking supply in serial mode. I'm only shooting for a 25 MHz spectrometer. I've never built something like what you had in the photo. Is the copper plate the ground and then everything else is wired directly above the plate?

author: uofmrob
author_rank: Regular Contributor
timestamp: August 11, 2020, 01:49:57 am
content: I built one of the cheap 1-30 MHz antenna tuners off of eBay for the coil matching and it works well. The basic layout will be the source going into port one, then the matching network and saddle coil with the sample in the middle of the coil in port 2, then an rf detector on port 3 and a 50 Ohm load on port 4. When the coil is well matched, there should be hardly any reflection to the detector in port 3. Then, when the magnet field causes the spins to go into resonance, the Q changes and you should see the signal as an increase in the reflected power.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 11, 2020, 01:53:28 am
content: At least that's how our big fancy EPRs work. But ours use microwaves (either 9 GHz or 34 GHz) and magnetic fields of 340 mT to 1.2 T.

author: KE5FX
author_rank: Super Contributor
timestamp: August 11, 2020, 02:33:22 am
content: Quote from: uofmrob on August 11, 2020, 01:38:18 amYep. It's just a piece of unetched/undeveloped copper clad PC board material. Do a Google image search forugly constructionordead-bug constructionfor inspiration. Some people like to carve islands and 'traces' in the foil with a Dremel tool, or employ additional bits of PCB material as 3D islands or standoffs, but none of that stuff is generally useful or necessary.ICs built on modern semiconductor processes often contain transistors with frequency limits of 10 GHz or more, even when the IC itself is rated for use at much lower frequencies. So it doesn't matter what the intended frequency of operation for your project is, stray capacitance and inductance is almost always going to be a concern.

author: helius
author_rank: Super Contributor
timestamp: August 11, 2020, 04:48:52 am
content: The most common key word is "Manhattan construction", I think because the Trinity test electronics were done that way.

author: Brumby
author_rank: Unknown
timestamp: August 11, 2020, 07:48:56 am
content: Quote from: helius on August 11, 2020, 04:48:52 amI thought it was because such construction ended up looking like Manhattan in miniature.

author: David Hess
author_rank: Super Contributor
timestamp: August 11, 2020, 10:38:27 am
content: Quote from: uofmrob on August 11, 2020, 12:34:35 amStart out with a lower frequency version using lower frequency operational amplifiers. Fast operational amplifier will require ground plane construction and suitable decoupling.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 11, 2020, 11:17:37 am
content: Thanks for all the information everyone!

author: uofmrob
author_rank: Regular Contributor
timestamp: August 11, 2020, 11:38:28 am
content: This individual did it this way. I'm not exactly familiar with this, either. Did he just hand drill this material?http://www.qrp.gr/circulator/index.htm

author: uofmrob
author_rank: Regular Contributor
timestamp: August 11, 2020, 12:43:44 pm
content: Quote from: David Hess on August 11, 2020, 10:38:27 amDavid, would you have a particular op amp that your would recommend for this project?

author: David Hess
author_rank: Super Contributor
timestamp: August 11, 2020, 03:53:29 pm
content: Quote from: uofmrob on August 11, 2020, 12:43:44 pmI would need some idea of the maximum operating frequency. But even a few 10s of MHz should use ground plain construction as others mentioned; even if the application frequency is low, fast operational amplifiers are not. Forget using solderless breadboard for this. An expert might get away with point-to-point wiring on soldered breadboard at lower frequencies but Manhattan or dead-bug style construction will actually be easier to get working.Maybe someone else has a better source, but pads for Manhattan construction are available here:http://www.qrpme.com/?p=product&id=SAMAnother limitation is that the operational amplifiers will be required to drive low impedance loads which limits you to parts intended for 50 and 75 ohm applications. (1) In this case they have to drive their own medium impedance feedback network, the low impedance output network, and a second medium impedance output network all in parallel; so general purpose parts are not suitable.I would probably start with something low cost which meets the above requirements like the LT1395, which also happens to work at +/-5 volts where less expensive parts are limited to lower voltages.(1) "Fast" operational amplifiers generally meet this requirement because they need to be able to drive transmission lines.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 11, 2020, 04:27:13 pm
content: Thanks so much for all the advice!Rob

author: uofmrob
author_rank: Regular Contributor
timestamp: August 11, 2020, 05:55:27 pm
content: Another quick question....since I'm looking at building it as a four port circulator, is there a reason to avoid just going with an IC that has four amps in one IC as opposed to four separate ICs? Is there more of a chance of signal leakage with the amps all being in one chip?

author: KE5FX
author_rank: Super Contributor
timestamp: August 12, 2020, 01:24:48 am
content: Quote from: uofmrob on August 11, 2020, 05:55:27 pmWith the LMH6722 or LMH6733, crosstalk at 25 MHz will be in the -70 dB to -80 dB range (see figure 9 of the data sheet), so it will not dominate the isolation.I'd favor the LMH6733 over the LMH6722 for this application,unless you have a use for the fourth opamp.(edit: I see you want four ports, so, yeah.)I'm not familiar with the LT139x family David mentions, but it looks fine as well. Not a fan of whatever those 'Manhattan' pads are that he mentioned, though, unless you just want to add some stray capacitance in random places. Current-feedback amps can be twitchy in that regard.This bridge uses a very similar approach to Wenzel's,as described by Sam Wetterlin(link is a mirror, Sam's site seems to have vanished):If you use good opamps and good construction practices, the achievable performance from this class of circuit is pretty awesome. With the AD8003 that I used, the directivity is still 20 dB at 900 MHz(!). The LMH6733 performs similarly.

author: David Hess
author_rank: Super Contributor
timestamp: August 12, 2020, 01:33:20 am
content: Quote from: uofmrob on August 11, 2020, 05:55:27 pmEven if you ignore isolation between the operational amplifiers, the close confines around a dual or quad chip can make isolation between sections more difficult. Using single parts allows designing one "section" and then replicating it.Quote from: KE5FX on August 12, 2020, 01:24:48 amI have seen some better DIP and SO8 pads somewhere but the example I linked are what I was able to immediately find online.

author: KE5FX
author_rank: Super Contributor
timestamp: August 12, 2020, 01:44:34 am
content: Quote from: David Hess on August 12, 2020, 01:33:20 amThat'sfor darned sure.QuoteJust solder.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 12, 2020, 02:44:25 am
content: So without any pads, do you just glue the ICs upside down on ground plate and solder to the connectors?I can 3D print supports for the BNC connectors pretty easily and glue them down.

author: coppercone2
author_rank: Super Contributor
timestamp: August 12, 2020, 02:53:00 am
content: yes, I have bought the parts for this circuit a while back (I chose a very high speed op-amp but still in SOIC or MSOP, something with some kind of leads)I need to etch a PCB. It is a interesting one.if you are a chemist etch the PCB

author: uofmrob
author_rank: Regular Contributor
timestamp: August 12, 2020, 02:54:26 am
content: I think I'm going to make a big order for all the stuff tomorrow and give it a whirl. Thanks so much for all the help everyone. This forum is such an amazing resource with so much knowledge and generosity!

author: coppercone2
author_rank: Super Contributor
timestamp: August 12, 2020, 02:54:46 am
content: Quote from: uofmrob on August 12, 2020, 02:44:25 amSometimes I super glue parts upside down and bend the leads down and solder them to ground but it depends on what part you have. IMO this kind of construction gets old after a while.. and it helps if you have precise hand tools.The problem is if you have a thick ground plane, the chip and glue gets real hot when you are tacking the nearby lead to ground, so it could help if you pre-tin it. And you can glue copper to something, cut a hole out under a chip, then glue the chip not to the copper, so you don't have heat issues..I found that its kind of harder then it looks, and you end up with messy construction unless you do alot of thinking.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 12, 2020, 03:00:47 am
content: Oh, and another question I had. For this type of circuit, should the ground be an earth ground?

author: coppercone2
author_rank: Super Contributor
timestamp: August 12, 2020, 03:05:37 am
content: i would power it off batteries. if you have a real circulator or isolator, they don't have any connection on them other then the ports. They are usually a passive magnetic element. I would design it as a module you can use on a RF bench with batteries so you can use it the same as a regular circulator to keep things simple and reliable on some AAs.Once you setup some microwave assemblies you will be happy there is less wires.those RF jacks connect to the RF equipment, which is usually connected to ground, by the connector on the chassis.in a RF assembly typically you see a can with little solder feed through connections that the DC power goes into. I don't think its typically earthed?

author: RoGeorge
author_rank: Super Contributor
timestamp: August 12, 2020, 05:22:11 am
content: Quote from: uofmrob on August 11, 2020, 01:38:18 amYou may want to look at the following prototyping techniques:For high frequency, pick a method that uses a ground (copper) plane. Shielding might help a lot, too.Solderless breadboards are to be avoided for high frequency circuits. Those breadboards produces unpredictable results, mostly because their internal traces have a lot of stray inductance, capacitance and high contact resistance. In breadboards, internal traces are in fact parallel sheets of metal.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 12, 2020, 11:33:52 am
content: Thanks!

author: uofmrob
author_rank: Regular Contributor
timestamp: August 12, 2020, 12:55:22 pm
content: I had actually just subbed to this channel. There's some great content! I had just watched all of the NanoNVA videos. I have a NanoVNA 2 coming soon to help with the coil design and matching. I have a super expensive Network analyzer at work, but I wanted to have one at home. Part of this project is to see how cheaply I can make an EPR. It's going to be baby steps, but eventually, I want to drive the source and magnet with a raspberry PI. There are a lot of these out there as people have linked above, but those are all based on the original instruments that detected the effects. I'm shooting more for a magnetic field sweeping spectrometer based off of the modern instrument designs, just operating at lower fields with off the shelf components. If I can get it to work well with relatively simple and inexpensive parts and methods, I'm probably going to present it at the Rocky Mountain EPR conference. You know, if it ever meets again.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 12, 2020, 02:06:35 pm
content: OK, I ordered some individual op-amps and some copper clad boards. They should all be here Friday. It looks like I have a weekend project!

author: David Hess
author_rank: Super Contributor
timestamp: August 12, 2020, 08:03:01 pm
content: Quote from: uofmrob on August 12, 2020, 02:44:25 amWhat I generally do is anchor the package right-side-up above the ground plane using the decoupling capacitors between the ground plane and supply pins. Then I use either wire wrap wire or enameled copper to do point to point wiring.However in a real layout, the equivalent of not using pads is to cut away the ground plane to raise the impedance which the above method does not allow whether pads are used or not.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 12, 2020, 08:12:06 pm
content: I was thinking that for the places in the circuit where multiple connections come together, I would use my rotary tool and cut an island in the coper board

author: David Hess
author_rank: Super Contributor
timestamp: August 12, 2020, 09:56:02 pm
content: Quote from: uofmrob on August 12, 2020, 08:12:06 pmThat is a common method; I still prefer to glue or solder a little pad down if I am not wiring things in the air. Strips of copper clad board can be used as controlled impedance transmission lines.High value surface mount resistors can also be used as terminals and to anchor other parts.The photograph below shows an example where I used the leaded multilayer ceramic decoupling capacitors to anchor DIP sockets. The RF bridge detector using 1N270 germanium diodes worked to 100s of MHz.

author: KE5FX
author_rank: Super Contributor
timestamp: August 13, 2020, 01:53:44 am
content: If you try that with modern current-feedback amps, you'll end up with a competitive entry in the "Who can build the highest-frequency oscillator on a breadboard?" thread. They will complain because you didn't actually use a solderless breadboard, but the spirit of the rules will certainly have been satisfied.But hey, knock yourself out.

author: SilverSolder
author_rank: Super Contributor
timestamp: August 13, 2020, 02:32:10 am
content: Quote from: David Hess on August 12, 2020, 09:56:02 pmArtwork, love it - Soldering the battery in is very hardcore!

author: uofmrob
author_rank: Regular Contributor
timestamp: August 14, 2020, 01:18:35 pm
content: I have another question for all of you experts. I've been thinking about this circuit design and I think it might not actually work for what I need. At the coil, I need about a watt of power, but will I actually get no power carrying forward in this circuit? Sure, I'll get a voltage carrying forward from the op amps, but will I get no significant current because of how the op amps work?

author: uofmrob
author_rank: Regular Contributor
timestamp: August 14, 2020, 01:21:21 pm
content: With the ferrite-based circulators in our instruments, they carry forward the 200 mW microwaves in our continuous wave instruments and even the 1000 W microwaves in the pulse instruments.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 14, 2020, 04:22:32 pm
content: Thanks for the explanation!

author: KE5FX
author_rank: Super Contributor
timestamp: August 14, 2020, 04:22:59 pm
content: Quote from: uofmrob on August 14, 2020, 01:18:35 pmIn combination with the available supply voltage, the power output is determined by the sum of the output impedance of the opamps, typically a few ohms, plus whatever series resistance is added at their outputs. The output current capability of the opamps is closely related to their output impedance; often the data sheet will show both.The circuit as published uses opamps rated for output currents up to 70 mA, so that's about a quarter-watt into 50 ohms. Power in the 1-watt range is achievable, but you'll need to go with beefier opamps. TheTHS3491would be a great choice since it can run from +/- 12V rails. At lower voltages you'd need to scale down the series termination resistors at the outputs, and you're better off not doing that for various reasons (notably the ability to tolerate a reasonable amount of load capacitance and compatibility with 50-ohm cabling and components).

author: uofmrob
author_rank: Regular Contributor
timestamp: August 14, 2020, 04:27:50 pm
content: It looks like the ones that I ordered have an output current up to 100 mA. Then I think I should be OK. 1 W would be great, but even a quarter watt should be enough to see something if I can get a well-matched saddle coil with a high Q. Thanks again!

author: David Hess
author_rank: Super Contributor
timestamp: August 14, 2020, 04:31:59 pm
content: Quote from: uofmrob on August 14, 2020, 01:18:35 pmPower levels are limited by the output voltage and current capability of the operational amplifiers. Source termination further lowers the output power capability.I think practical designs with integrated parts are going to be limited to less than 100 milliwatts. 36 volt and 200 milliamp parts can do a little better than this.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 14, 2020, 04:43:52 pm
content: Thanks for the advice! I think that I'll try out the design and the ugly construction techniques that you all suggested this weekend with the ones that I ordered from Mouser and see if I can just get one that works at all and then look into more powerful op-amps for a second version if I can get this one to work.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 14, 2020, 05:03:11 pm
content: So something like this might be better for the final product:https://www.mouser.com/ProductDetail/Texas-Instruments/THS3091DDAR?qs=ZmJdcv7QZ9oZqiofJS38Ww%3D%3D

author: uofmrob
author_rank: Regular Contributor
timestamp: August 14, 2020, 05:16:35 pm
content: Really, I should just bump the operating frequency up and buy something like this. It would certainly be easier. But not as fun and educational.https://www.ebay.com/itm/400-420MHz-RF-coaxial-circulator-Isolator-SMA-Joint-frequency/254530755929?hash=item3b43371d59:g:K2oAAOSwsiBeX2En

author: uofmrob
author_rank: Regular Contributor
timestamp: August 15, 2020, 07:12:09 pm
content: Well, I learned another lesson. Buy more components than you need. 🤣. I snapped a pin off of one of the op amps, and clipped another one getting it out of the package. Those buggers are tiny! I've got another order in, but for now two of them are wired up with the filter capacitors holding them up and they are wired for power!

author: KE5FX
author_rank: Super Contributor
timestamp: August 15, 2020, 10:11:23 pm
content: A good start!And yes, never order just one or two of anything.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 21, 2020, 01:21:16 am
content: Hi everyone. I have it a whirl again tonight. The small surface mount op amps that I got are just too small. Well I can deal with the size, but the leads are just too fragile. Once you bend them once, any further flexing causes then to snap. I got them all wired up to power, but when I started soldering the resistors on, the leads kept snapping. This is what I have:https://www.mouser.com/ProductDetail/Analog-Devices-Linear-Technology/LT1395CS5TRMPBF?qs=ytflclh7QUWCZinjfMpnVQ%3D%3DCan anyone recommend a part number for a good op amp for this application that would come in a DIP IC that would be easier to work with?Thanks!

author: uofmrob
author_rank: Regular Contributor
timestamp: August 21, 2020, 01:22:50 am
content: And maybe one that could give me the half watt output that I would need.

author: lowimpedance
author_rank: Super Contributor
timestamp: August 21, 2020, 01:48:29 am
content: Just a thought, what about using an sot to dip adapter which would make it more robust as the LT1395 is soldered to the adapter then you have rigid pins on the other side to solder other parts to.for example;https://www.mouser.com/Embedded-Solutions/Engineering-Tools/Embedded-Tools-Accessories/Sockets-Adapters/_/N-cxdjj?Keyword=sot+to+dip+adapter&FS=Truesee digikey for more etc.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 21, 2020, 01:57:53 am
content: Thanks for the advice! I have some of these, but the consensus seemed to be that pass and adapters aren't the way to go.https://www.adafruit.com/product/1210

author: uofmrob
author_rank: Regular Contributor
timestamp: August 23, 2020, 12:19:49 am
content: Hey everyone, that's for all the advice! I have a working circulator! There's no leakage at all. It passes reflections on with the exact signal voltage and if you put a 50 Ohm terminator in, everything down wind sees nothing. Thanks so much for all the help! Now in to designing a coil!

author: KE5FX
author_rank: Super Contributor
timestamp: August 23, 2020, 01:45:25 am
content: Awesome, good going! That's a perfectly reasonable way to do it as long as the IC adapter doesn't add a lot of stray capacitance.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 23, 2020, 01:53:58 am
content: I was having so much trouble with the fragility of the surface mount leads, I just decided to give this a whirl to see if it worked. This quad op amp is pretty low power on the specs (it's one that I had bought initially as a spec replacement for the one in the original paper). I'll probably have to swap out the IC for something that can put out more power, but I'll give this a shot first. At least I know I can get one to work!

author: uofmrob
author_rank: Regular Contributor
timestamp: August 24, 2020, 01:18:36 am
content: One more question. Is there a reason that it wouldn't be safe to test this out with an oscilloscope in 1 MOhm mode? I was looking at the signal and my scope is acting funny now (it's a Tektronix 2430 scope). Did I bork my scope?

author: uofmrob
author_rank: Regular Contributor
timestamp: August 24, 2020, 02:16:56 am
content: I checked and there are no shorts to ground or anything. It wouldn't be too surprising that my scope is dying just from being 30 some years old. But I certainly don't want to be doing something I shouldn't. I had a 20 MHz 2 Vpp sine wave on my function generator on port 1 and had a BNC cable running from port 3 to the scope and I was testing the effect of a 50 Ohm load on port 2. It works perfectly. Without the load on port 2, the signal gets reflected to port 3, but when you put the load in, the signal goes away completely.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 24, 2020, 03:48:20 am
content: The only thing I can think of is that I was counting on the source providing the ground. I had forgotten that and there was a point that I didn't have the source plugged in, but the scope was still connected. Without the ground from the source, there is a couple hundred millivolts between the ground plane of the circulator and the ground of the scope. That might have been the classic blowing up there scope through a current through the ground. Ugh, I should have grounded the circuit.

author: KE5FX
author_rank: Super Contributor
timestamp: August 24, 2020, 03:49:58 am
content: Vanishingly unlikely that you damaged the scope with this circuit. What is the scope actually doing?If you take it apart for troubleshooting, remember to use a fan on the heat sinks. Failure to do thatwillhose a 2430.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 24, 2020, 03:55:49 am
content: It already had an issue with the relay on the 500mV/1V division on channel 2 not triggering on the high voltage side. Now it's not able to get a consistent trigger on any signal on either channel. You can see the signal, but it's completely unstable in the time domain.

author: uofmrob
author_rank: Regular Contributor
timestamp: August 24, 2020, 04:13:13 am
content: I thought it was maybe the function generator putting out an unstable signal (it's a FeelElec FY6900 60 MHz AWG), but my analog scope locks onto its signal up to its bandwidth limit just fine.

author: uofmrob
author_rank: Regular Contributor
timestamp: September 14, 2020, 08:43:19 pm
content: Hi everyone,I realized that I forgot to send a final thank you to everyone for all the help. I got it working perfectly!Rob

author: uofmrob
author_rank: Regular Contributor
timestamp: September 14, 2020, 08:45:24 pm
content: Also, the scope is OK. My 4-year-old had messed with the knobs while I wasn't looking and had changed the horizontal scaling so much that it was seeming to have a signal that it couldn't lock on, but it was interpolating and when I went back to a proper scaling, it triggered fine.

author: SilverSolder
author_rank: Super Contributor
timestamp: September 14, 2020, 09:34:25 pm
content: Quote from: uofmrob on September 14, 2020, 08:45:24 pmOne of the unexpected dangers of having children!