Scoping Signals in the 10s of Kilovolts - Page 1
Forum section: electronics / projects

author: madshaman
author_rank: Frequent Contributor
timestamp: October 29, 2013, 01:47:33 pm
content: Hello,I'm looking for pointers/advice on how to go about the following:I'm going to be building a circuit to dump very fast high voltage pulses into a load composed of nitrogen gas (N2 laser).I would like to be able visualise the pulse in the time domain using an oscilloscope.I'm not rich so I'm planning to use a Tek7104 I'm restoring (got the mainframe working, yay, and I'm on to the 7A29 vertical amps once I'm back home) using an old Tek scope camera mount retrofitted with a CCD camera.I don't have enough knowledge or experience to acquire the signal properly but I do know that if I don't do it right, in the very best scenario I'll get a single shot trace that looks absolutely nothing like the actual signal (and I'll probably totally change the system behaviour with my test setup).In the worst case I'll simply destroy my scope.I know how to attenuate the signal using a simple voltage divider (and how to do that safely) but I'm positive this will, at best, get me garbage and I suspect this will alter the system's behaviour too much to be of real use.Even if I do my best to apply what little I know about signal propagation, I'm certain I'll fail and won't even know I'm failing, so a related question would be: is there any way I can verify my test setup?Any help would be greatly appreciated.Thank-you in advance.

author: ConKbot
author_rank: Super Contributor
timestamp: October 29, 2013, 01:56:12 pm
content: Do you need voltage or current measurements? just a waveform? A pulse current transformer would give current waveforms easily. If you have an exposed HV node, you could use capacitive pickup to get the voltage waveform, but neither of those would be accurate for measurements. The current transformer could be if you had a way to calibrate it.

author: Jon86
author_rank: Frequent Contributor
timestamp: October 29, 2013, 02:04:36 pm
content: Just get one of those high voltage automotive ignition probes, they're cheap as chips on ebay.

author: madshaman
author_rank: Frequent Contributor
timestamp: October 29, 2013, 02:09:42 pm
content: Do those cheap as chips probes accurately acquire signals up to 1Ghz?

author: madshaman
author_rank: Frequent Contributor
timestamp: October 29, 2013, 02:14:06 pm
content: Quote from: ConKbot on October 29, 2013, 01:56:12 pmYes, I should be more specific, i want to acquire the voltage waveform, without distortion, phase or otherwise.An N2 laser self terminates in nanoseconds so the rising edge needs to be very fast and the pulse width needs to be as short as possible. Any energy delivered to the system after the termination of lasing action is wasted.I need to visualise the pulses accurately to debug my circuit.

author: Fraser
author_rank: Super Contributor
timestamp: October 29, 2013, 02:18:32 pm
content: 10's of kV at 1GHz bandwidth....... you will be looking at a very expensive setup I am affraid.The PICOScope automotive ignition pickup is just a capacitive coupler around the insulated HT lead. It is not an accuarate oscilloscope probe and definitely not good to 1GHz !You could use a 1GHz bandwidth active Oscilloscope probe with an E field antenna to spatially pick up the radiated signal from your set up but no calibration will be possible. Only waveforms. A 1GHz active probe is also a very expensive piece of kit unless you already own one.At such high voltages I believe a non contact method may be best to avoid arching and influence on the signal you are trying to capture.

author: madshaman
author_rank: Frequent Contributor
timestamp: October 29, 2013, 02:37:20 pm
content: Quote from: Aurora on October 29, 2013, 02:18:32 pmThank-you, I *do* own both an active 3.5Ghz probe (untested as yet), and two 1.5Ghz passive probes (tested and verified with a 20Ghz sampling scope). Obviously I nabbed these used.Yeah, this isn't exactly an easy problem to solve, even with halfway decent equipment to start.Was kinda hoping someone would say: "Oh yeah, I had to do this at such and such and here's how you do it, and this is why..."I might browse IEEE to see if any papers sufficiently describe a similar test setup.

author: Unknown
author_rank: Unknown
timestamp: October 29, 2013, 08:39:00 pm
content: If the rise times are in the nanoseconds, then forget any divider you build yourself. You'd be lucky to get 1 MHz bandwidths, or rise times in the microseconds. Assuming you're skilled enough in HV construction, since making a probe for 10 kVs is non-trivial even at DC.Normally my answer would be a Tek P6015(A), but take a careful look at the frequency derating curve. You're likely exceeding the capabilities of this probe, which I believe is the only one of its kind. The P6015 is often available used for reasonable prices, but note that this needs to be filled with a freon compound to achieve its rated voltage specs, and availability of this stuff is extremely poor because of environmental regulations. That's why they designed the P6015A, which doesn't use freon and is therefore much more expensive used. This probe only has 75 MHz bandwidth, so it's also a little slow for your application. Good luck finding something with more bandwidth and a higher voltage rating, I don't think such a thing exists.I would probably indeed look at capacitive or inductive pickup. If you want good frequency response, maybe you can fit a well-insulated wire through a Tek CT1? These have a decent bandwidth and are fairly affordable, but you would obviously be measuring the current signal, not the voltage signal. I'm also very doubtful that you'll find wire rated for tens of kV small enough to fit through the hole. Maybe a slower probe like the P6022 would work? Forget about accurate voltage measurement with any of those setups.

author: pinkysbrein
author_rank: Contributor
timestamp: October 30, 2013, 03:34:55 pm
content: He can build one capacitor which can discharge in the ns time range ... why not build two to form a capacitive divider?Couldn't you just clear small part of the ground plane then put another piece of dielectric over the ground plane and then put some aluminium foil over that so it covers the cleared part and a larger section of the ground plane? It should have small capacitance to the high voltage plane and large capacitance to the ground plane ... thus forming a capacitive divider?Of course you'd need an active probe or an in place buffer to measure the voltage over the divider.

author: ElectronicTonic
author_rank: Contributor
timestamp: October 30, 2013, 09:28:03 pm
content: I was going to suggest the Tek P6015 as well, but certainly not for 1 GHz BW. Although, if you can find one, it would come in handy for calibrating your probe at HV, but at lower frequencies.What I can suggest is that you watch a series of videos I made about designing and building a 2 kV, 100 MHz scope probe from other scavenged probe parts. It was a long process. I started with just building it, then ran into problems. PSpice simulation really helped, and the finished product worked just as well as a professionally made probe.From my experience, I have two pieces of advice when designing a HV, high-BW scope probe.(1) Simulate your design to iron out the kinks. When every pF counts, simulation is key. And don't forget to take into account the inductance of your return ground path, especially at 1 GHz.(2) If your scope can be situated right next to the HV source you want to measure, then design the probe to be a simple box that plugs into scope input on one end and has HV input directly on the other end. If you don't have a long coax cable between input and output, then your design can be simplified because you wont need to compensate for the coax's capacitance, inductance, or internal reflections.Here is the first of four videos about this 2 kV probe I made. You might have a look at some of the other videos in my "High Voltage Scope Probes" playlist.

author: Rufus
author_rank: Super Contributor
timestamp: October 31, 2013, 12:04:41 am
content: I don't see it being necessarily that difficult providing what you are measuring has plenty of drive.A resistive probe which is just a resistor feeding 50 ohm coax has inherently high bandwidth. A 5M resistor will give you a X100k probe. The problem is getting 5M of resistors that can stand 10s of kV and the stray capacitance across them.You can compensate for stray capacitance with a capacitor to ground forming a capacitive divider with the same ratio as the resistance. A possible trouble is just 20fF of capacitance is about 25k at 1GHz and for 10s of kV that is a significant load, however, does the capacitance/stray of what you are driving make 20fF look significant?

author: madshaman
author_rank: Frequent Contributor
timestamp: October 31, 2013, 05:46:07 am
content: Quote from: ElectronicTonic on October 30, 2013, 09:28:03 pmThank-you very very much. I was also thinking that I'd have to become a *lot* more familiar with probe design and probing techniques. If the circuit was within the voltage range acceptable to the 7A29 I think I could use a short bnc cable soldered right on to the board.It does look like I'll need an approach with the scope right next to the circuit, but it's a scary prospect as the circuit itself is design to deliver a very large amount of energy into the load in a very short amount of time; I can imagine all that energy dumped into the scope. I'm fairly certain there would be fireworks.Thanks again for sharing. I won't be able to watch your video until I'm somewhere with a decent bandwidth connection.

author: madshaman
author_rank: Frequent Contributor
timestamp: October 31, 2013, 06:12:56 am
content: Quote from: Rufus on October 31, 2013, 12:04:41 amThe circuit will have enormous driving ability. Picture a Marx generator, but instead of spark gaps, imagine high voltage bjts in avalanche mode. For the capacitors, I have a large number of 1200V capacitors with the lowest internal resistance and inductance I could find.I plan to wire the whole circuit point to point, and I anticipate it's going to be a nightmare laying it out to balance avoiding inductance while preventing arcing. One of my ideas for prototyping it is to embed most of the circuit in paraffin wax.The load itself is a tube of N2 gas at 30 torr at the lowest. I'm going to start the tube length at 20cm and the gas will between two copper electrodes, one a flat strip, the other will be a 45° wedge "pointing" at the plate. This arrangement comes from an IEEE paper which evaluates the effectiveness of different electrode shapes for an N2 laser.The electrode spacing (wedge edge to strip) will be about a centimetre. However, I will be experimenting with different spacings as well as different gas pressures.The "tube" will be composed of polycarbonate.The gas itself will present some inductance and the electrode arrangement will of course form a series capacitor. I don't anticipate having much in the way of parasitics.To answer your question, I think I'll need to complete construction to measure the load's capacitance but I'm fairly certain it will exceed 20fF.None of my ideas are new, in fact I'm basically just ripping off design ideas from various papers.

author: TerraHertz
author_rank: Super Contributor
timestamp: October 31, 2013, 07:55:27 am
content: You have a very similar design problem as with something I'm working towards. And no, I haven't really come up with a waveform capture solution yet either.You might find some useful information here:http://w140.com/tekwiki/wiki/P6015It occurs to me that the P6015 with its Freon liquid insulator was probably designed before the availability of silicone oils. Also I don't yet understand why other high voltage insulator oils aren't suitable. Why Freon? Perhaps it's necessary to use a non-polar molecule, otherwise energy from the electrostatic field goes into rotational movement of the molecules - which won't be a constant drain over time - thus causing signal distortion.The central problem with dividing a high voltage fast waveform down, is that whatever resistive structure you use will have unevenly distributed capacitive and inductive parasitics, thus forming a horrible RC/RL 'distortion box'.And yet for high voltages you _have_ to use large geometry, high value resistances to avoid arc-over. Or in general exceeding the dielectric breakdown strength of the resistive material, or even its substrate.For very fast pulses perhaps a purely capacitive divider could work. But that has to avoid field variations, propagation delays in the electrode structures, and so on. I've heard somewhere there is a HV probe physicists use, that is mostly a pair of concentric metal spheres, insulated with Freon. But don't have any more info on it, not even the manufacturer.QuoteWhat's wrong with oil? One BIG advantage: it's 'self healing'. Secondly you can drain it off to work on the circuit.I'm nowhere near beginning construction on the HV part of my project, but I'm expecting to be using coaxial transmission lines operated in an oil bath. Suitable oil yet to be determined.I do have a Tek P6013A probe, though it's utterly inadequate for the project needs. Amusingly, it came with the large arc burn marks you see in the pic. Something vaporized some of the diecast metal of the probe compensation box, grounding to the scope frame. It doesn't seem to have done the probe or compensation box circuitry any harm, but I bet it gave the operator a surprise. They hadn't made any attempt to clean it up either. I filed the worst of the burnt metal jaggies off, and it works fine - within it's specs.Hopefully the other electrode for those arc burns wasn't the operator.I also have a 7104 and 7A29 modules, but a) the mainframe is currently dead, and b) really it's useless for single shot waveform capture. The microchannel plate screen is for beam intensity enhancement, not a storage tube. Coupling it to a CCD sensor sounds like a good idea! Did you try any fast single shot captures yet?The HP 54120T 20 GHZ digitizing scope I have - same problem, in that it can't capture single shot waveforms. Plus the very amusing +/- 2V MAXIMUM voltage limit on the 50 ohm sampler inputs. Or you destroy the GaAs input circuitry. Oh right, I'm sure I can get that working safely nearby something generating multi-kilovolt impulses - not. Well theoretically it's possible with good shielding/grounding, but as the sampler modules for the 45121T are no longer made, there's not much margin for trial and error.What might be usable: the Tek 7912AD, which is a really weird and cool thing. Lots on the net about it. But I don't like my chances of getting one at all, let alone getting it operational. I did get a couple of 7A29P plugins so far though, just in case.The 1985 Tek catalog lists 'destructive testing' as one application for the 7912AD. Which um... as I understood it, it was developed for use in nuclear weapons testing. So yeah...Hey, if anyone has a 7912AD they don't want, I'd be delighted to give you some freed-up space in return for it.

author: pinkysbrein
author_rank: Contributor
timestamp: October 31, 2013, 10:37:13 am
content: What high voltage transistor are you using BTW? I wasn't aware those could non destructively avalance.

author: TerraHertz
author_rank: Super Contributor
timestamp: October 31, 2013, 02:05:19 pm
content: Speaking of Tek P6015: ebay 370929628416Don't think I've ever seen a seller provide such crappy pixelated photos before.Whether that means he's hiding some details, or is just an idiot, who knows?If the latter it might be a good deal, as the stupidly bad pics will scare off some bidders.Shame I'm completely out of cash atm, or I'd try to buy it.

author: Unknown
author_rank: Unknown
timestamp: October 31, 2013, 02:17:31 pm
content: Quote from: TerraHertz on October 31, 2013, 07:55:27 amThey redesigned it with silicone isolation, look at the P6015A. Replacement of the freon in the P6015 has been discussed extensively on the TekScopes list, but is non-trivial. It requires a boiling point not far above room temperature, so just a few drops of liquid ensure a sufficiently high vapor pressure within the probe. The probe will slowly leak, and you don't want this to compromise the isolation. Weight might also have been a consideration, filling the space with oil might make the probe very heavy. Obviously the breakdown voltage will need to be high enough. And the dielectric constant will need to be fairly close to the freon they used, since it forms a capacitor that's taken into account in the compensation network. I believe people have considered and tried various coolants, I'm not sure if any of them was really successful.About buying a P6015 on eBay: note that without freon the maximum voltage is severely derated, I believe the specs will be fairly close to a P6013 then. There's a reason why P6015s without freon are much cheaper than P6015A probes. And note that even with freon the P6015 won't even come close to handling tens of kV pulses with ns rise times.

author: pinkysbrein
author_rank: Contributor
timestamp: October 31, 2013, 04:29:39 pm
content: Quote from: Rufus on October 31, 2013, 12:04:41 amThe shunt capacitance is generally not the problem, as you say it's easily compensated for, the problem is the distributed parasitic capacitance across the length of the resistor to ground ... without a ground shield it will be variable and can't be compensated for, you can only drown it out with enough capacitance. With a ground shield you can compensate, but now the ground shield adds a lot of capacitance ... no free lunch.Thinking in femtofarads is hopelessly optimistic.

author: Rufus
author_rank: Super Contributor
timestamp: October 31, 2013, 05:04:49 pm
content: Quote from: pinkysbrein on October 31, 2013, 04:29:39 pmI didn't think the divider resistors needed to be anywhere near a ground for the OPs application - he only really has one point to probe, however, you are right the self capacitance of the divider resistors would be very significant.

author: pinkysbrein
author_rank: Contributor
timestamp: October 31, 2013, 06:11:39 pm
content: madshaman, do you have ieeexplore access?"DC to 1 gigahertz multikilovolt voltage probe"describes a self-built probe, self built by master student with access to a metal shop and a network analyzer though.

author: Unknown
author_rank: Unknown
timestamp: October 31, 2013, 06:25:21 pm
content: I wonder how they tested that. Many high-voltage components will have a voltage coefficient, so frequency response at 1.5 V may not be equal to response at 15 kV.

author: madshaman
author_rank: Frequent Contributor
timestamp: November 01, 2013, 07:30:18 am
content: Quote from: pinkysbrein on October 31, 2013, 06:11:39 pmI do, I'll add it to my file cabinet, thanks!wrt high voltage avalanche transistors, afaik, any BJT will avalanche but exceeding a given peak current will destroy the device (by a method which I fully don't understand yet) as will normal thermal destruction (too much power dissipated in too short a time).I'm away from my home machine and out of the country so I don't have access to all the papers I collected for this project. From memory, one paper (not from my ieee file cabinet :-( ) described a general guideline for choosing candidates for avalanche mode transistors based on Vcbo being close to Vceo). The paper was evaluating regular (non hyper-expensive Zetex avalanche -specific) BJTs for avalanche operation. The authors chose a set of BJTs to test noting that some worked well, others did not.My plan is to evaluate a small number of high voltage BJTs driven to avalanche breakdown via different capacitor arrangements (modulating the peak current and total energy delivered) and graph my results.Digging back into my digikey orders, I ordered the following parts for this purpose when I was deep into the topic. I made these choices to the best of my knowledge mixed with a bit of intuition, and I honestly don't completely remember everything I considered (ever spent half a day pouring over data sheets?):http://www.onsemi.com/pub_link/Collateral/BUH50-D.PDFhttp://www.onsemi.com/pub_link/Collateral/2N6515-D.PDF(PNP)http://www.fairchildsemi.com/ds/KS/KSC5502D.pdfhttp://www.st.com/web/en/resource/technical/document/datasheet/CD00219617.pdfI ordered large numbers as I figure I'll end end killing many of them. I also plan to experiment with operating the [simple one transistor] circuit at temperatures down to ~= -70º C to see what effect this has. I also have a large number of lower voltage BJTs for prototyping.I might face total failure, but I figure that as long as I document all my observations, I'll learn something and will gain experience with this kind of circuit. I know I have an enormous amount to learn here. I also know this will be a huge endeavour as there are so many aspects to the circuit I want to build.Here's one paper that might be of interest to you although not directly linked to my particular application:http://ieeexplore.ieee.org/xpl/articleDetails.jsp?tp=&arnumber=5386197Also, if you put "avalanche bjt" or "avalanche transistor" into the IEEE XPlore search box you'll see many articles on the topic, describing many topologies and also some papers on the development of high voltage BJT specifically designed for high voltage operation.Although my electronics is half-way decent, academically I come from a computing and information science background so my in-depth knowledge of transistor operation at the physical level (beyond the basic theory) is sorely lacking; I'm hoping to rectify this through this project and I would greatly appreciate any help or pointers to facilitate increasing my knowledge.P.S.My apologies for not being able to provide as much information as I could from my home machine, honestly, connectivity sucks here (with my SO who's doing international development work in East Africa) and I had to hold my laptop out the window to access IEEE and my digikey account properly, which is very annoying. I'm glad to see so much interest in the topic and I'll certainly add more when I return home in December.

author: madshaman
author_rank: Frequent Contributor
timestamp: November 01, 2013, 07:54:37 am
content: Quote from: TerraHertz on October 31, 2013, 07:55:27 amThanks for the link! What are you working on?QuoteYeah, ideally I'd run into someone who's actually had to build such a thing :-(QuoteNot opposed to HV transformer oil of some sort. The reason I immediately thought of paraffin is that:a) it's an excellent insulatorb) it's a solid a lower temperatures and I can avoid accidentally spilling it all over my lab ^^' and I figure it's much less messy to work withc) I can go buy it in large quantities pretty much anywhere and I've used it before(it also makes a decent shield for high speed neutrons for when I'm older and crazier and start work on that Farnsworth reactor ;-))QuotellolQuoteNo, not yet. Ironically, I've got my mainframe working but repairing the 7A29 modules I have is still on my todo list. I've gotten as fas as buying the Tek camera mount I'm going to retrofit but that's all; I don't anticipate too much trouble, the only thing requiring real work will be rigging up the circuit to trigger it at the right time.QuoteA tinkerer after my own heart, guess what 20Ghz sampling scope I have too ;-). I've been drooling over the 50Ghz test set, but I cant justify paying $2000 for one (which is the best price I've seen so far), I don't really need it and imagine my joy if the thing turned out to be fried.QuoteI'll check it out, I love weird s**t like this.

author: ElectroIrradiator
author_rank: Frequent Contributor
timestamp: November 01, 2013, 08:00:00 am
content: Is this a commercial development project, or more of a hobby experiment?If the latter, then there may be ... other methods ... than an avalanche transistor to consider for your pulse generator, depending on how critical the generated pulse waveform is.

author: johansen
author_rank: Super Contributor
timestamp: November 01, 2013, 08:37:39 am
content: if I'm not mistaken, the peak KVAC rating for the P6015 of which i have one:is derated for continuous signals only, and the issue is the probe arcing over internally, not some kind of voodoo.so as long as you've got less than 40KVdc applied to the probe for less than 100ms and less than 10% duty cycle and you're below 8000 feet, you will do fine.(i had to build my own compensation box. it has 9 feet of mystery oscope lead on the order of 30 ohms per foot.)

author: Marco
author_rank: Super Contributor
timestamp: November 01, 2013, 11:57:39 am
content: Quote from: madshaman on November 01, 2013, 07:30:18 amAFAIK the term avalanche is a bit overloaded ... they can avalanche to the open base breakdown voltage instead of fully turning on, the former is slow and will cause the transistor to burn a lot of power. There might be some very high voltage BJTs which avalanche to turn on, but I guess it will take experimentation to see which they are.

author: Unknown
author_rank: Unknown
timestamp: November 01, 2013, 02:16:10 pm
content: Quote from: johansen on November 01, 2013, 08:37:39 amBut what if it's a 100 ms pulse of a 40 kVrms 100 MHz signal (assuming it's filled with freon)? Based on Fig. 1-2, I would say that the frequency would be derated significantly. The manual does not appear to make any statement on how frequency and duty cycle interact, so it's not obvious to me whether the maximum voltage would reach the same 40 kV(rms) maximum at short pulse durations, or a much lower maximum. A very short pulse with ns-level rise times would even have even more high-frequency content, but a much shorter pulse duration and lower duty cycle.

author: Eviltech
author_rank: Contributor
timestamp: November 01, 2013, 07:16:26 pm
content: What i want to ask is why do you need such crazy frequencies with that lazer and how exacly do you get >100Mhz signals trough a transformer. I'm asuming you are not using aircore.

author: TerraHertz
author_rank: Super Contributor
timestamp: November 02, 2013, 09:47:12 am
content: Quote from: pinkysbrein on October 31, 2013, 06:11:39 pmBah. $31 for non-members.Guess I'll just have to do without it, since I am currently broke. Bad case of Ebay splurge syndrome.Quote from: madshaman on November 01, 2013, 07:54:37 amIn general, an interest in the behavior of arc plasma paths as HF transmission lines.Specifics - don't wish to say, due to potential commercial applications.QuoteHmm... actually, you may have converted me. For small setups anyway. I've only used it as sculptural material for lost wax process, in which you burn it out as fast and hot as possible. But I suppose slow heating would be much less 'sooty'. Would still be very hard/impossible to do with large systems though.QuoteNow there's a well worn garden path, with some amusing twists.Here's a quote from Richard Hull, owner ofhttp://www.fusor.net"There is absolutely no posibility of ever making a fusion generating reactor here using any scientifc methodology and amateur funding." (sic)Heh. Saved for future value.QuoteAssuming the image capture time of the CCD can be faster than the decay time of the 7104 tube phosphor. Which I have no idea what that is.QuoteHave you looked at HP 83480 lightwave modular scopes? I haven't tried buying a system yet, but it looks like with lucky deals you could get a very cool set for well under $2000.For my use, the bonus would be getting optical sampling in the 10s of GHz too.ebay search: (83480a,83481a,83485b,83486a,11898a,83487a) -cdOh yeah, and F*CK ebay for their latest insane revision of searching. Goodbye 'saved searches', welcome our new 'followed searches' overlords, in which there is no way at all of avoiding your 'followed searches' cluttering your ebay landing page with hundreds of unwanted 'Feed' photos. Good job ebay management, you pack of retards. Why don't you all go get a job at Facebook?

author: Marco
author_rank: Super Contributor
timestamp: November 02, 2013, 10:36:55 pm
content: Quote from: TerraHertz on November 02, 2013, 09:47:12 amYar, matey do you really want it?You're not missing much though, he basically skips the math behind every important design decision and the exact measurements aren't in there either. The only thing I really learned from the paper (other than that it's possible to make such a probe relatively cheaply) is the idea of putting an antenna wire parallel to the high resistance leg of the divider, connecting to the centre of the divider through a resistor with an unexplained value, inside the ground shield to compensate for the high order fall off caused by the distributed capacitance to ground across the resistor.Whether he just experimented with the wire to get a reasonable transfer function or whether he used EM field simulation before fabricating ... no idea.

author: ElectroIrradiator
author_rank: Frequent Contributor
timestamp: November 03, 2013, 09:14:32 am
content: Quote from: TerraHertz on November 02, 2013, 09:47:12 amThe writing was on the wall, once they removed wildcard searches. I'm not even sure what all of this is supposed to accomplish. Do they expect their customers will buy *anything* with a subject line looking remotely similar to what they are looking for?Sorry for OT derail.

author: TerraHertz
author_rank: Super Contributor
timestamp: November 03, 2013, 11:29:50 am
content: Quote from: Marco on November 02, 2013, 10:36:55 pmAye, I do be wanting it. Even if it's as scurvey as you say, still interesting.QuoteA wire parallel to the divider resistor, in a HV probe? Er, in an insulating medium?QuoteI do. He didn't simulate. If he had he'd have been proud to show it.The problem with gear like this, is the lack of calibration signals. Where do you find a nice clean 100KV square wave?

author: Marco
author_rank: Super Contributor
timestamp: November 03, 2013, 04:38:21 pm
content: Quote from: TerraHertz on November 03, 2013, 11:29:50 amWell air, it's only designed for 15 kV ... and you can of course use an insulated wire, just pot the end.

author: madshaman
author_rank: Frequent Contributor
timestamp: November 04, 2013, 06:30:06 am
content: Quote from: Eviltech on November 01, 2013, 07:16:26 pmI might be less confusing if I talked in terms of risetime as opposed to bandwidth. The 1Ghz bandwidth is roughly what I need to see the pulse shape.The only transformer that *might* be involved in my project would be used to trigger the avalanche in the marx-generator-like circuit.This type of laser cannot operate in CW mode, increasing the pulse repetition rate is much much further down the line.(you might confusing my posts with other general discussion in the thread around the same topics)

author: madshaman
author_rank: Frequent Contributor
timestamp: November 04, 2013, 06:52:16 am
content: @TerrahertzYou're not the first person to consider using plasma to carry signals or as an antenna ;-).Best of luck though, I'll be doing some experiments with plasma soon, but more for artistic expression. I'll be happy to share notes.wrt ieee: I had access to both ieee and and acm at my former employer (named after a fruit ;-p), but I was a lowly senior SE.I pay for personal memberships in both out of pocket. Honestly, having access to so many academic papers (given one's math is halfway decent) is invaluable.In the loosely grouped hacker community I've participated in, there was an unspoken philosophy that holding back your techniques can lead to fear that you'll never ever come up with a better idea in the future.Now that I've switched fields and become independent (very recently) I'm more of the mind to publish *after* I've released a product to the market. If someone wants to use my ideas (which I assume aren't globally unique) they could always just RE my product anyway.I'm still not entirely sure where I fall on this, because it *is* painful to spend countless hours with pen, paper and mathematica, prototyping and refining and then have someone just copy you; here I'm referring more to software I've written.That being said, I generally assume there are at least one million people at least as smart and knowledgeable as myself and for who the exact same ideas will occur. It's depressing, *but*, only a small number will actually put in the work to bring their idea to life.If we didn't live in such a commercial global society it would be clear how inefficient non-sharing can be, but we don't live in an ideal world.Also, I do believe that if someone puts in long hours, works very hard and makes the sacrifices that this requires, they *do* deserve a better lifestyle. There are always those more than willing to leach off another's labours in one way or another.I'm trying to find a balance myself with regards to this in my whole life, I'll let you know if I find a solution ;-)

author: TerraHertz
author_rank: Super Contributor
timestamp: November 04, 2013, 02:01:07 pm
content: Quote from: madshaman on November 04, 2013, 06:52:16 amSure, I know, nothing secret about that. It's a particular application idea that I'd like to experimentally check out before opening my mouth. Partially to avoid looking like a fool if it's a stupid idea, partially for ... other reasons.QuoteCool. Strangely enough 'plasma art' is another thing I want to pursue. Since I'll have pretty much all the required gear set up for the other project.I hear that 'plasma spheres' are illegal to sell now. Not sure of the details re what kinds, but I can understand the worry about X-Rays. It's something I'll have to be very careful of with my main project. Hence spending $1K on a gamma ray spectrometer. Vacuum, ions and HV are not a good mix, on your own side of a big lead shield.QuoteCurrently I'm very poor. Also I take the position that 'knowledge' isn't valid unless it's freely available to all. Academic paper paywalls make me very angry. Even when I could afford to pay, I refuse to.QuoteHa ha... lack of ideas isnota problem I've ever had. Quite the opposite.And don't worry, if it works you'll hear the details.QuoteWe'd have an interesting conversation f2f, I'm certain.Too bad we've only the net (hi there NSA), since there are many things I won't talk about online.Which, considering the kinds of things I do sometimes talk about here (to the annoyance of the mods), should make you wonder about that.

author: ElectronicTonic
author_rank: Contributor
timestamp: November 05, 2013, 12:09:05 am
content: Glad to help. I just now remembered another helpful video for when you have some better BW. It's Mike Harrison showing how to probe a 100 MHz clock signal which is not easily done with a typical scope probe.

author: madshaman
author_rank: Frequent Contributor
timestamp: November 05, 2013, 07:00:28 am
content: @TerraHertz:wrt: cash strapped: I hear ya, I've been very poor at certain points in my life, here's a couple hints as to how I'd get by:1) shamelessly pirate softwareWhen you have real money, you really should dole out the cash for a tool you like and use.2) ask a good friend still in uni for their digital library credentials; in my day (after leaving uni) this usually took the form of a web proxy and login information for it; then you can (usually) freely access *all* the electronic journals through the university's library site (well, whatever set the school pays for). Not sure if it still works this way for most schools.Again, when you *can* afford, pay for the appropriate memberships and journal access.wrt: f2f, well, Australia's on our list for places to visit eventually, I'll let you know if we're ever in the area!wrt: secrecy, ABSOLUTELY no worries, I still wrestle with this. I would love a world where there weren't any patents and everyone would freely share all their research, but imho, it isn't remotely practical at this point in history. Best we can do is find a nice middle-ground. Most bleeding edge technology starts as peer-reviewed academic research anyway.wrt: plasma globes, I'm fairly certain that no off-the-shelf globe uses anywhere near the acceleration voltage needed to produce high energy photons, and even if they did, there's waaaay too much matter in the way for any particle to attain enough energy before colliding with another for it to be a risk. Spontaneous decay in the world around us is probably a more dangerous source of x-rays (as in no danger at all). I think you can still buy them in Canada, but I'm sure the paranoia will reach here eventually.Cheers!

author: madshaman
author_rank: Frequent Contributor
timestamp: November 05, 2013, 07:05:16 am
content: Quote from: ElectronicTonic on November 05, 2013, 12:09:05 amThanks again! Man, I'm grinding my teeth over not being able to see Dave and w2aew's newest videos, not to mention the latest eps of walking dead :-(

author: johansen
author_rank: Super Contributor
timestamp: November 05, 2013, 07:09:57 am
content: Quote from: madshaman on November 05, 2013, 07:00:28 ampsychopaths always rise to the top, its just how the world works.I would say that about 50% of bleeding edge technology starts as academic research, the rest as accidental inventions..however, most accidental inventions are rediscovered years later as academic research.