SIGLENT SDS1072CML (v5.01.02.22) next firmware feature requests  - Page 1
Forum section: electronics / testgear

author: alexan_e
author_rank: Newbie
timestamp: July 25, 2014, 10:40:22 pm
content: I purchased the SIGLENT SDS1072CML a couple of weeks ago. The device I have received has hardware version 11-62-3.5 and software version 5.01.02.22.What I'm disappointed with is that no matter what memory mode is selected (long or normal), the single button captures in low memory mode, this renders the long memory (2M) useless since you can't use it for captured signals unless you use run/stop button, but doing that doesn't have the same effect as the single trigger.Hoping that someone from Siglent is reading this, here is a wish list for the next firmware release:Single trigger should be able to capture in the user selected memory mode, normal or long instead of forcing normalWhen a channel is off, the auto button should not enable it and try to detect a signal from it but ignore it.For settings where there is an A and B setting (e.g. the two cursors cur A and cur B) the click of the intensity adjust button should change between them (like in Rigol) rather than having to use the screen side buttons. There should also be a mode that moves both cursors simultaneously (like in Rigol)The user should have the option to selectively add single measurements to the screen rather than having to add all the time and all the voltage related measurements. For example being able to add only Vpp, period and +DUT on the screen like shown when auto is pressedThere should NOT be a language selection popup every time the oscilloscope is turned on, this is annoying

author: tautech
author_rank: Super Contributor
timestamp: July 26, 2014, 07:53:04 am
content: Quote from: alexan_e on July 25, 2014, 10:40:22 pmA few good suggestions here, I will point Siglent to this post.But I ask, as your refer to Rigol, what model are you comparing the SDS1072CML to?If the Rigol model you refer to is at a similar price point, your comments are worth noting.You can select a range of single measurements to be displayed in the menu area.Language popup on boot, refer to this post:https://www.eevblog.com/forum/testgear/siglent-technical-support-join-in-eevblog/msg468451/#msg468451All DSO's have varied UI's and they all take time for user familiarity and the "second nature" of operation to develop.I am sure in time you will appreciate the robustness and functionality for your Siglent.

author: alexan_e
author_rank: Newbie
timestamp: July 26, 2014, 03:24:59 pm
content: Quote from: alexan_e on July 25, 2014, 10:40:22 pmI refer to the Rigol DS1052E which is a much older model and currently cheaper.Here is the relevant part of a video that shows what I describe regarding the cursorshttp://youtu.be/B0ajIBb4EU4?t=6m15sYou can select between curA, curB and cur A & B just by clicking the intensity adjust button. (Currently SDS1072CML has no curA&B at all)Quote from: tautech on July 26, 2014, 07:53:04 amYes but the right side measure menu takes space that may be valuable, plus it can only show up to five measurements and the only alternative is to add in the screen all the voltage or time related measurements (shown in the bottom side of the screen). Why not ne able to add only a few of them?Quote from: tautech on July 26, 2014, 07:53:04 amI will check it, thank you.I'm quite happy from the oscilloscope performance with the exception of the single trigger problem I have described, the rest of the feature I request will just make the usage of the oscilloscope more enjoyable and more effective.

author: Siglent
author_rank: Regular Contributor
timestamp: July 31, 2014, 02:14:42 am
content: Quote from: alexan_e on July 25, 2014, 10:40:22 pmAlexan, thanks for your advice, we will improve it.

author: havefun
author_rank: Newbie
timestamp: July 31, 2014, 03:05:24 am
content: Hi, I've got a SDS1152CML. I checked out the measure function. I found that the default measure parameters is VPP, Vmax,Vmin,Period,Vrms , but you can choose the parameters you are interested in with the type menu in the measure.

author: alexan_e
author_rank: Newbie
timestamp: July 31, 2014, 12:30:51 pm
content: Quote from: havefun on July 31, 2014, 03:05:24 amYes the five measurements in the right side can be selected independently,but I meant the measurements shown in the left bottom area of the screen, like when auto is pressed:There is currently the option to add ALL the time related measurements and/or ALL the voltage related measurements (like in the image below) but it would be more practical if the readings could be added selelctevelly (for example only Vpp, Vmean and Vrms or any user selected combination).

author: alexan_e
author_rank: Newbie
timestamp: July 31, 2014, 01:24:34 pm
content: Quote from: Siglent on July 31, 2014, 02:14:42 amThank you for your reply.Regarding the forced normal memory mode for single trigger, I'm not sure if this was done intentionally or if it is a bug.Since there is a model with a low memory (CNL) and a model with the high memory (CML) , I don't understand what is supposed to be the benefit of the 2Mpts memory when the user can not utilize it when capturing a signal. It can only be used while viewing a signal in real-time or randomly stopped (STOP button) and in both cases the long memory makes little sense.Here is what I describe with images. All images show the 1KHz test signal @250usHere is an image when the oscilloscope is running in real time (red square) and the memory is set to long mode (green square), Notice the Waveform Preview Display (blue square) that shown the captured length vs what is visible.Here is an image when the oscilloscope is stopped using the RUN/STOP button (red square), notice that the memory mode (green square) still shows long and the Waveform Preview Display (blue square) still shown the same length as in real time.Now this is the same signal captured with the single button (red square). Notice that the memory mode (green square) has become grayed and has forced the normal (low memory) mode and the Waveform Preview Display (blue square) is now showing that the captured wave is much smaller that what it was with the long memory mode.

author: havefun
author_rank: Newbie
timestamp: August 01, 2014, 12:46:54 am
content: Quote from: alexan_e on July 31, 2014, 12:30:51 pmYes , I know what you mean, but for different measures the parameters you want may be different, so I think it is ok to display all the measurements. But I think your idea is well worth thinking about for the SIGLENT.

author: azdem
author_rank: Newbie
timestamp: December 03, 2019, 05:05:33 pm
content: @SiglentI want to update the firmware of my SDS1072CML (HW Version: 11-62-3.7, bought in ~2015, Current FW Version: 5.01.02.29, never updated). None of the ADS files listed onhttps://www.siglent.eu/Downloadswork: I get always this message: "Product type does not match".

author: tautech
author_rank: Super Contributor
timestamp: December 03, 2019, 06:31:17 pm
content: Quote from: azdem on December 03, 2019, 05:05:33 pmWelcome to the forum.It is because the CML range was updated to the CML+ and now none of the official Siglent websites have the older CML model firmware. DO NOT attempt to install CML+ FW versions into older CML models.In my records I have one later CML firmware version, P32 from Nov 15.Summary of Changes:1.added bandwidth update option2.fixed some bugs (including trigger jitter problem. If use LongMemory and if timebase is 250us/div or less there is a jitter. If change to Normal memory no problem, if switch to Long Memory it appears.)The .ads file is not large, only 1.5 MB so I can post it here or send it to you privately in an email.PM your email and I can send it to you.

author: sameerdhiman
author_rank: Contributor
timestamp: April 14, 2020, 11:09:43 am
content: Quote from: tautech on December 03, 2019, 06:31:17 pmCan you please share that file on google drive or any location from where I can download the same.Regards,Sameer

author: tautech
author_rank: Super Contributor
timestamp: April 14, 2020, 12:19:19 pm
content: SDS1000CML (NOT for later CML+ models !)July 2015 Version: 5.01.02.32Files attached.Note, REMOVE .txt extension from P32.ADS firmware file.

author: sameerdhiman
author_rank: Contributor
timestamp: April 14, 2020, 01:34:01 pm
content: Quote from: tautech on April 14, 2020, 12:19:19 pmIt says "Download does not exist"Dear I want to confirm things before updating the firmware whether the product supports it or not. I purchased the re-branded DSO in Aug-2013 (INDIA) Metravi DSO-5070A 70MHz 2-Channel it came with Siglent User Manual and it's details are here underSoftware Version: 5.01.02.13Hardware Version: 11-62-3.3Serial No. NEUxxxxxxxxxxxProduct Type: DSO-5070A1. Can we identify from above details whether product is CML/CFL/CNL ?2. Do we need to install all firmware updates released so far in incremental order or the last released update includes all the patches from previous firmwares ?I very much require your help here. If you require me to open the case to identify correctly I can do that also if above details are not sufficient.Regards,Sameer Dhiman

author: tautech
author_rank: Super Contributor
timestamp: April 14, 2020, 02:24:23 pm
content: Edit made to previous post.

author: sameerdhiman
author_rank: Contributor
timestamp: April 14, 2020, 02:50:00 pm
content: Thank you so much.Please, how can I identify that the DSO I own is SDS1072CML ? from Hardware Version etc. ?

author: tautech
author_rank: Super Contributor
timestamp: April 14, 2020, 03:13:09 pm
content: Quote from: sameerdhiman on April 14, 2020, 02:50:00 pmYou will probably need to install a cfg file first to allow for the Siglent firmware to then be recognised and installed.You can find that cfg file here:https://www.eevblog.com/forum/testgear/siglent-ads-firmware-file-format/msg2998250/#msg2998250

author: sameerdhiman
author_rank: Contributor
timestamp: April 15, 2020, 10:15:07 am
content: Quote from: tautech on April 14, 2020, 03:13:09 pmIs there any risk involved in installing CFG file ? I am asking this because I do not have any other DSO this is the only oscilloscope I have.Regards,Sameer Dhiman

author: tautech
author_rank: Super Contributor
timestamp: April 15, 2020, 08:19:48 pm
content: Quote from: sameerdhiman on April 15, 2020, 10:15:07 amNo.It is necessary to install the firmware so DSO is changed from Metravi to Siglent and will accept the FW upgrade.Looking at your model specs here:https://www.metravi.com/single-product.php?prod=DSO-5070E&cat=Digital+Storage+OscilloscopeIt aligns to SDS1072CML:https://www.batronix.com/shop/oscilloscopes/Siglent-SDS1072CML.htmlExcept the USB socket placement is different.There is always a risk flashing another firmware into a DSO and only you can decide to take that risk.IMO, with the USB socket in a different place it seems the HW is different so I would not attempt to make the firmware change until more information comes available.YMMV

author: sameerdhiman
author_rank: Contributor
timestamp: April 15, 2020, 08:47:05 pm
content: Quote from: tautech on April 15, 2020, 08:19:48 pmThe model you looked at (DSO-5070E) is newer model and it is different from mine (DSO-5070A). They have removed DSO-5070A related material from their website but I dug out the web and I found the relevant document on another website. Please go through the attached document, I am 99% sure that 5070A is SDS1072CML now only 1% is left on your confirmation.BTW, had Siglent released any update between P13 and P22 for CML ?Regards,Sameer Dhiman

author: tv84
author_rank: Super Contributor
timestamp: April 15, 2020, 08:53:30 pm
content: Quote from: sameerdhiman on April 15, 2020, 08:47:05 pmFor that model, I think not.But in the end, despite your "similarity" feeling it should be a decision of your own. You can't expect tautech to certify a crossflash of that kind because no one has done it before.That doesn't mean it wont work. There is always a 1st attempt.

author: tautech
author_rank: Super Contributor
timestamp: April 15, 2020, 09:01:31 pm
content: Quote from: sameerdhiman on April 15, 2020, 08:47:05 pmAh, OKQuoteAt a quick look through old files, apparently not but I maybe wrong.Later 29 and 32 versions were released.Quote from: tv84 on April 15, 2020, 08:53:30 pm

author: tv84
author_rank: Super Contributor
timestamp: April 15, 2020, 09:04:52 pm
content: My suggestion is: open the case and look at the PCB references. That should provide the necessary infos.

author: sameerdhiman
author_rank: Contributor
timestamp: April 15, 2020, 09:10:16 pm
content: OK I'll do it and will update you.Thank you so much Tautech and tv84Regards,Sameer Dhiman

author: sameerdhiman
author_rank: Contributor
timestamp: April 16, 2020, 11:50:40 am
content: This is to let you know that update process is successfully completed on Metravi DSO-5070A.Special thanks goes to tautech and thanks tv84 for encouraging me.150700 and 150734 => Before update151151 and 151238 => After Config update (Boot picture and Device Type changed to OEM Siglent)152926, 153603 and 153826 => After P22, P29 and P32 updates.Regards,Sameer Dhiman

author: tv84
author_rank: Super Contributor
timestamp: April 16, 2020, 01:16:16 pm
content: Quote from: sameerdhiman on April 16, 2020, 11:50:40 amWell done!Fortune favors the bold.Now get me this SCPI command response:MD5_PR?

author: sameerdhiman
author_rank: Contributor
timestamp: April 16, 2020, 02:56:37 pm
content: Quote from: tv84 on April 16, 2020, 01:16:16 pmIt is not connected to PC. I use it as standalone unit.

author: tv84
author_rank: Super Contributor
timestamp: April 16, 2020, 03:01:52 pm
content: Quote from: sameerdhiman on April 16, 2020, 02:56:37 pmCan you connect it? Maybe an upgrade is possible...

author: tautech
author_rank: Super Contributor
timestamp: April 16, 2020, 05:43:51 pm
content: Quote from: sameerdhiman on April 16, 2020, 02:56:37 pmFollow tv84's lead as it may well benefit you.You should have a USB-B connection on the back of the scope for connection to a PC.You will need to download and install EasyScopeX from a Siglent website. You also need NIVISA Runtime for the USB connectivity SW. Install the Runtime package before EasyScopeX.Within EasyScopeX there is a SCPI command panel where you can enter and execute commands.To check it's working properly *IDN? returns the instruments information.Have fun.

author: sameerdhiman
author_rank: Contributor
timestamp: April 16, 2020, 06:03:28 pm
content: Of course, Let me setup NI-VISA, IVI and EasyScope.Fortunately, I have my own detailed post on this forum regarding setting up NI-VISA.https://www.eevblog.com/forum/testgear/siglent-sds1102cml-scope-and-easyscopex/msg285583/#msg285583

author: sameerdhiman
author_rank: Contributor
timestamp: April 16, 2020, 08:09:27 pm
content: Quote from: tv84 on April 16, 2020, 01:16:16 pmThese are the response*IDN? => *IDN SIGLENT,SDS1072CML,NEUxxxxxxxxxxx,5.01.02.32MD5_PR? => SDS1000CML

author: tv84
author_rank: Super Contributor
timestamp: April 16, 2020, 10:22:03 pm
content: Quote from: sameerdhiman on April 16, 2020, 08:09:27 pmSee pm and report.

author: sameerdhiman
author_rank: Contributor
timestamp: April 17, 2020, 08:31:11 am
content: Done, please check PM.

author: tv84
author_rank: Super Contributor
timestamp: April 17, 2020, 01:45:22 pm
content: As suspected. I think we would be able to upgrade to 150Mhz but we would have to patch the FW in order to bypass a disabling flag. Maybe one day...

author: sameerdhiman
author_rank: Contributor
timestamp: April 17, 2020, 02:04:58 pm
content: I'll wait for that day

author: tautech
author_rank: Super Contributor
timestamp: April 17, 2020, 07:57:52 pm
content: Quote from: tv84 on April 17, 2020, 01:45:22 pmI well remember one day installing the latest FW into a 100 MHz CML+, as I do with all DSO's we despatch, and it got dumped to a 150 MHz model !Dunno why but the customer was well pleased.

author: tv84
author_rank: Super Contributor
timestamp: April 17, 2020, 09:38:11 pm
content: Quote from: tautech on April 17, 2020, 07:57:52 pmCML+ are easy. CML are harder...

author: sameerdhiman
author_rank: Contributor
timestamp: April 18, 2020, 08:50:54 pm
content: Can you do me one favor if possible.I need SDS1072CML's frequency response up to 200 MHz. I do not have any frequency generator.Actually, amid lock down I got time to build a Variable frequency RF generator (Analog). It just touched 180 MHz with 20mV pk-pk as measured with above DSO. To know what is the actual amplitude is there I need oscilloscopes frequency response curve or data series in excel format. Can you please provide me that.Regards,Sameer Dhiman

author: tautech
author_rank: Super Contributor
timestamp: April 18, 2020, 09:00:14 pm
content: Quote from: sameerdhiman on April 18, 2020, 08:50:54 pmTypically the -3dB point for most DSO's is 10-20% beyond rated BW. At 200 MHz my guess would be 50+% down.But of course if the probe is in circuit it could be more if they are only 70 MHz rated probes.Standard, it should give good indicative results to approaching 100 MHz.You need make these kind of measurements with care as even the signal source amplitude accuracy impacts on results.

author: sameerdhiman
author_rank: Contributor
timestamp: April 19, 2020, 06:02:53 am
content: Tautek, you are Siglent's distributor and have access to their products. If you have time can you please make measurements using 70 MHz CML+ (I think CML model would not be available) with factory shipped 70 MHz probe in 1x and 10x modes not very precise data is required, only generator's reference amplitude (in pk or pk-pk assuming generator output is rock solid) and measured figures are required at 30, 40, 50 .... 200 MHz in 10MHz increments. As you know DSOs are based on Gaussian response it is cumbersome to make guess whether beyond -3dB point signal is hitting brickwall filter or are they rolling of at 6dB/octave or 12 or 18.It would be great help from your side.Regards,Sameer Dhiman

author: tautech
author_rank: Super Contributor
timestamp: April 19, 2020, 06:54:29 am
content: Quote from: sameerdhiman on April 19, 2020, 06:02:53 amCorrect.....but only their products I care or chose to stock.I only have SDS1102CML+ here and when sold will not get any more as their technology is outdated for today's use and SDS1202X-E is far better value for the $.

author: sameerdhiman
author_rank: Contributor
timestamp: April 19, 2020, 06:59:34 am
content: Oh ok...so no chance

author: tv84
author_rank: Super Contributor
timestamp: June 22, 2020, 04:37:12 pm
content: Quote from: tv84 on April 17, 2020, 01:45:22 pmThat day has come.Those gents that have an old SDS CML/CNL and want to upgrade them to 150/200 Mhz pm me. I may be able to help you.When I have the 1st photo, I'll post it.

author: sameerdhiman
author_rank: Contributor
timestamp: November 16, 2020, 07:58:36 am
content: Tautech, do you ship to India ? Now, I need 200MHz or above probes as a side effect of TV84 patch.

author: tv84
author_rank: Super Contributor
timestamp: November 16, 2020, 09:25:41 am
content: @sameerdhiman made the jump.1. Before Patch*IDN? => *IDN SIGLENT,SDS1072CML,S/N,5.01.02.322. Right after patch and reboot*IDN? => *IDN SIGLENT,SDS104X,S/N,5.01.02.323. Sent the SCPI commandMCBD xxxxxxxxxxxxxx4. After command and reboot*IDN? => *IDN SIGLENT,SDS1152CML,S/N,5.01.02.32Attached is theFW that allows Siglent SDS1000CML to go to 150MHz (and most probably the SDS1000CNL to go to 200MHz).You just have to search the forum for the generator that allows you to fill the xxxxxxxx.

author: sameerdhiman
author_rank: Contributor
timestamp: November 16, 2020, 10:55:44 am
content: Thank you so much TV84Here are the screenshots.Owners of SDS1000CML/CNL (without +) with lesser bandwidth oscilloscopes e.g. 1072, 1102 etc. and wants to upgrade to higher bandwidth please PM TV84.Regards,Sameer Dhiman

author: Sobakin76
author_rank: Contributor
timestamp: March 06, 2021, 10:34:36 pm
content: Thanks to tv84, my old SDS1072CML turns to 1152.Because original 70Mhz probes are both died now, I have to buy a bit more expensive 150Mhz pair

author: Sobakin76
author_rank: Contributor
timestamp: March 06, 2021, 10:56:05 pm
content: But I don't see any real difference in bandwith with cheap P6100 "100MHz" probe, only time/div limit now 2.5ns instead of 5ns.On picture Ref waveform is saved when it was original 1072. Square (I believe it is) 27MHz generated by SN74HC14N at 5V and with probe used small spring ground lead wire in both cases, probe of course in x10 mode, memdepth "Normal", 1GSa/s (second channel disabled).

author: Sobakin76
author_rank: Contributor
timestamp: March 06, 2021, 11:40:25 pm
content: Just make probe tests and for sure it's because of cheap probe.Tested with my new SDS1104X-E (updated to 1204), compared this P6100 with Siglent original 100MHz PP510 which came with 1104X-E. Probes was recalibrated with each reconnection.Pics in order: 1152CML & PP510, 1204X-E & P6100, 1204X-E & PP510.And pic in previous post is 1152CML & P6100. Unfortunately I didn't this test with 1072CML in it's original 70MHz state for more adequate compare

author: tautech
author_rank: Super Contributor
timestamp: March 06, 2021, 11:47:27 pm
content: P510 shipped with the SDS1102CML and later + versions since forever and your screenshot above shows how well they match the X-E vs P6100.IIRC at some 200 MHz they are only ~1dB down so they'll be fine to use with your newly modded SDS1152CML.And welcome aboard.

author: Sobakin76
author_rank: Contributor
timestamp: March 06, 2021, 11:55:54 pm
content: Quote from: tautech on March 06, 2021, 11:47:27 pmYes I saw some probes chart with 200MHz scope here, and "100MHz" PP510 appears not worse than Siglent 1202's 200MHz probes. So now I know that I need two more PP510'sQuoteThank YouBy the way, the 1204X-E shows a bit more detailed waveform than 1152CML with same probe and at same 1GSa/s

author: tautech
author_rank: Super Contributor
timestamp: March 07, 2021, 12:09:28 am
content: Quote from: Sobakin76 on March 06, 2021, 11:55:54 pmNormal.CML HW is limited by its 150 MHz HW whereas SDS1kX-E HW is 200 MHz capable and as there are far greater HF components in a square wave the CML's HW will not reproduce it as accurately.Dig out the thread: Show us your square wave and have a look at how different scopes handle the HF component.

author: Sobakin76
author_rank: Contributor
timestamp: March 07, 2021, 12:41:47 am
content: Quote from: tautech on March 07, 2021, 12:09:28 amOh I haven't more good scopes. I have 3rd scope for outdoor usage, but it's a $100 40Mhz Hantek 2C42 with 250MSa/s sampling (in single channel mode), so it can't do anything good with this waveform (used PP80B probe shipped with scope):Maybe we need another thread for this topic?

author: Sobakin76
author_rank: Contributor
timestamp: March 08, 2021, 07:49:48 pm
content: I founde that probe calibrated at 1204X-E has some overshoot on calibration square wave then connected to older 1152CML, looks like the old one has a bit smaller input capacitance, despite it marked 16pF vs 15pF at 1204X-E, if I understand the concept correctly (I may not).

author: la-xara
author_rank: Newbie
timestamp: August 27, 2021, 02:26:12 am
content: Hello friends, how to do generating code.Sent the SCPI commandMCBD xxxxxxxxxxxxxx1. Before Patch*IDN? => *IDN SIGLENT,SDS1072CML,S/N,5.01.02.322. Right after patch and reboot*IDN? => *IDN SIGLENT,SDS104X,S/N,5.01.02.321and 2. this is i do success!Help me

author: la-xara
author_rank: Newbie
timestamp: August 29, 2021, 02:36:13 am
content: Hello friends!Thank you very much for you help tv84!

author: robertstar
author_rank: Newbie
timestamp: November 14, 2021, 10:39:56 am
content: Hello friends, how to do generating code.Sent the SCPI commandMCBD xxxxxxxxxxxxxx1. Before Patch*IDN? => *IDN SIGLENT,SDS1072CML,S/N,5.01.02.322. Right after patch and reboot*IDN? => *IDN SIGLENT,SDS104X,S/N,5.01.02.321and 2. this is i do success!Help me

author: robertstar
author_rank: Newbie
timestamp: November 14, 2021, 10:49:21 am
content: Screen

author: Krohovich
author_rank: Newbie
timestamp: June 07, 2022, 01:46:51 pm
content: Hi !Hello friends, how to generate code ?

author: croma641
author_rank: Regular Contributor
timestamp: October 03, 2023, 04:53:48 am
content: From a (broken: screen + firmware) Atten ADS1062CAL to..... Siglent SDS1152CNL ..Thanks, TV84 !

author: croma641
author_rank: Regular Contributor
timestamp: October 03, 2023, 04:54:32 am
content: and... yes, trigger works at 200 MHz..

author: tv84
author_rank: Super Contributor
timestamp: October 03, 2023, 09:44:56 am
content: Quote from: croma641 on October 03, 2023, 04:53:48 amGreat recovery!PS: CML

author: tcoman
author_rank: Newbie
timestamp: March 21, 2025, 12:45:46 am
content: Quote from: robertstar on November 14, 2021, 10:39:56 amI can get the same steps 1-2 to work. The MBCD xxxxxxxxxxxxxx I am not sure about.I assume that I input this command first, before running the 32-Patch.Still not sure what the x's mean-- maybe the serial number of my sds1072cml?Please help!

author: tv84
author_rank: Super Contributor
timestamp: March 21, 2025, 11:18:29 am
content: Quote from: tcoman on March 21, 2025, 12:45:46 amThe x is the license generated with the keygen.1st you do the patch upgrade. Then you insert the license.

author: Spartan1
author_rank: Newbie
timestamp: May 30, 2025, 12:45:27 pm
content: Hey tv48,I think you helped me update my scope years back.Are there anymore updates for it?Thanks.

author: tautech
author_rank: Super Contributor
timestamp: May 31, 2025, 07:47:01 pm
content: Quote from: Spartan1 on May 30, 2025, 12:45:27 pmThe only update worth you doing is for the firmware version:Pull the latest and study the revision and install notes.https://int.siglent.com/download/firmwares/?ProId=40

author: ryger
author_rank: Newbie
timestamp: September 30, 2025, 03:23:01 pm
content: Step By Step For SDS1072CML1. Download the update, save it to a USB drive, connect the USB drive to the front USB port, upload it to SIGLENT, and then restart (SDS1072CML)https://www.eevblog.com/forum/testgear/siglent-sds1072cml-(v5-01-02-22)-next-firmware-feature-requests/?action=dlattach;attach=11104722. Script from the post:https://www.eevblog.com/forum/testgear/unified-keygen-script-for-siglent/msg5554039/#msg5554039Paste it on the website:https://www.programiz.com/python-programming/online-compiler3. Download EasyScopeX from the Siglent website:https://siglentna.com/download/30910/?tmstv=1759224406and install it. You also need NIVISA Runtime for the USB connectivity SW;4. You should have a USB-B connection on the back of the scope for connection to a PC;5. Within EasyScopeX there is a SCPI command panel where you can enter and execute commands:6. Use the SCPI command panel: MD5_SRLN? result enter them in the script in SN and ID;7. Uncomment Model = 'SDS1000X-E' in script;8. Generate CODE;9. Use generate code 150M - paste in SCPI command panel: MCBD YOUR_CODE.DONE - ENJOY SDS1152CML