EEVblog 1413 - Micsig CP2100B Current Probe TEARDOWN and TEST - Page 1
Forum section: electronics / general

author: EEVblog
author_rank: Unknown
timestamp: August 19, 2021, 11:48:20 am
content: Teardown and measurements with the Micsig CP2100B 2.5MHz oscilloscope current probe.Also a demonstration of mains switch mode power supply current and noise measurement, and also some performance tests.Use the coupon code FLUX for 15%And paying with crypto gets you a further discount on top of that!https://www.eevblog.com/product/cp2100b/Or get the cheaper lower bandwidth 800KHz CP2100A model:https://s.click.aliexpress.com/e/_A1gq4ihttps://s.click.aliexpress.com/e/_AKl2Y200:00 - Intro05:39 - Teardown of base unit14:25 - Clamp unit teardown23:40 - Mains switchmode current measurement34:00 - Measuring the Earth's magnetic field!34:42 - Noise measurement35:26 - Low current measurement

author: Kleinstein
author_rank: Super Contributor
timestamp: August 19, 2021, 01:00:04 pm
content: Whatching the scope in the last minute of so shows a funny effect: the amplitude is randomly changing quite a lot. I don't think this is a fault of the proble. It is more like a fault of the scope triggering right (a good scope should warn about that - a fluctuating frequency is kind of a warning though hidden). Averaging over shifted traces reduces the signal. It is a kind of trap for the young players when using the averaging mode.The shields are nice for a tear down, but maybe not too good a contact for a really sensitive signal.I would be a bit worried about the magnetic core not being laminated very fine. So higher frequencies may not pass through the core as much. They somehow seem to have a good compensation for that, even without the trimmer caps.Chances are a rectanglular waveform would not look as nice.For a little more signal one could wrap some 2 or 5 turns through the sensor if needed.It looks like they just switch the gain of the instrumentation amplifier - so maybe a trimmer for each range.Averaging 2 back to back sensors could reduce the offset drift - at least the part that correlates between the 2.

author: Rdx
author_rank: Contributor
timestamp: August 19, 2021, 04:45:29 pm
content: Wondering about component U4 on the head unit.It probably is an alternative footprint for U5 I would assume?What is a usual workflow for such stuff? Just place two components and wire them in parallel (like apparently they did here) and then manually delete one version each out of the BOM?Well I guess it depends on your CAD package whether it does have a workflow for these things.

author: thm_w
author_rank: Super Contributor
timestamp: August 23, 2021, 09:29:23 pm
content: Quote from: Kleinstein on August 19, 2021, 01:00:04 pmYoutube comment from Yuxian:QuoteQuote from: Rdx on August 19, 2021, 04:45:29 pmYes, place two.Another option is place one with a dual footprint in PCBLIB, but they clearly did not do that as you noticed the U4/U5.In the schematic you can select it as "no BOM" for altium, or manually delete it as you say.

author: tchicago
author_rank: Regular Contributor
timestamp: August 26, 2021, 04:05:50 pm
content: Curious why these HV probes are not made galvanically isolated. It would feel much safer knowing that there is no direct connection to the HV circuit whatsoever. And that the info about the signal passes via the highly rated isolation optocouples or transformers. Also should be easier to fight the common mode rejection issues.

author: Kleinstein
author_rank: Super Contributor
timestamp: August 26, 2021, 04:30:40 pm
content: The scope is usually grounded. For safty reasones the high voltage probe also needs to grounded one side. Some of the high voltage probes come with an extra ground lead, just in case the BNC connector at the scope is not enough.

author: 2N3055
author_rank: Super Contributor
timestamp: August 26, 2021, 04:34:34 pm
content: Quote from: tchicago on August 26, 2021, 04:05:50 pmErrr... It doesn't work that way.It is current that kills you and equipment.And current flows into ground, all microamps of it...Making DC-100MHz galvanically isolated probes is harder than you think.Tektronix has some fibre optic isolated probing solutions... How much your house cost?

author: thm_w
author_rank: Super Contributor
timestamp: August 27, 2021, 11:46:21 pm
content: Quote from: tchicago on August 26, 2021, 04:05:50 pmAs mentioned, its hard and expensive.Look up optocouplers and what kind of bandwidth you can get out of those. Optocouplers cannot easily transmit analog signal linearly anyway. Special linearized optos exist eg HCNR201 is only 1MHz, for $6.Dedicated analog optos exist (isolation amplifiers), only 100-200kHz bandwidth, price is over $5.https://www.digikey.ca/en/products/detail/broadcom-limited/HCPL-7510-000E/825318Its probably easier to digitize the signal and send it over isolated USB or something instead.

author: trophosphere
author_rank: Frequent Contributor
timestamp: August 28, 2021, 03:07:03 am
content: Quote from: thm_w on August 27, 2021, 11:46:21 pmAs an example:

author: Wolfgang
author_rank: Super Contributor
timestamp: October 21, 2021, 11:52:01 pm
content: Hi,I bought one and I wanted to see how reliable the specs are, The surprise was that even short runs of open wiring will result in major deviations from the bandwidth and rise/falltime specs.Details here:https://electronicprojectsforfun.wordpress.com/testing-a-micsig-cp2100b-current-probe/

author: thm_w
author_rank: Super Contributor
timestamp: October 22, 2021, 12:09:40 am
content: Quote from: Wolfgang on October 21, 2021, 11:52:01 pmcorrected link:https://electronicprojectsforfun.wordpress.com/testing-a-micsig-cp2100b-current-probe/