Lady Heather configuration help, please - Page 1
Forum section: electronics / general

author: tkamiya
author_rank: Super Contributor
timestamp: May 13, 2019, 01:13:36 am
content: I have the latest (6.14?) version of beta Lady Heather.I am trying to place -d=10000 and -ga into heather.cfg and it is not working. They both work from command line using / notation. What are the literal string I need to use?? Spent 8 hours so far trying all sorts. Still needing help.I want adev displayed in graph, and queue depth to be 10,000.Thanks!

author: texaspyro
author_rank: Super Contributor
timestamp: May 13, 2019, 02:36:34 am
content: The command to set the queue depth is /q= (not /d=)With some screen/device combos the /ga command from the command line might be overridden. Try putting GA in the config file (no leading /). This will execute like it came from the keyboard.

author: tkamiya
author_rank: Super Contributor
timestamp: May 13, 2019, 03:03:12 am
content: /q= worked perfectly/ga did not workGA did not workbut @GA worked!Thank you.I tend to mix up p, q, d, and b A LOT....

author: tkamiya
author_rank: Super Contributor
timestamp: May 13, 2019, 08:46:21 pm
content: Um.... temperature in Paris Hilton?

author: texaspyro
author_rank: Super Contributor
timestamp: May 14, 2019, 03:14:10 am
content: Quote from: tkamiya on May 13, 2019, 08:46:21 pmhttps://www.theregister.co.uk/2007/10/28/additional_reg_standards/Heather also does altitude in linguini...

author: tkamiya
author_rank: Super Contributor
timestamp: May 14, 2019, 04:35:50 am
content: Now, I'm trying to show adevs in tabular format. I can't get rid of circular plot entirely. Are there special zuccini's I have to use?

author: texaspyro
author_rank: Super Contributor
timestamp: May 14, 2019, 06:01:34 pm
content: Quote from: tkamiya on May 14, 2019, 04:35:50 amThe "A" menu controls "all adev" mode which dedicates the screen to showing xDEV info. What device are you using?

author: tkamiya
author_rank: Super Contributor
timestamp: May 14, 2019, 06:03:07 pm
content: Right now, Nortel GPSTM and SRS PRS-10. But I have others.

author: texaspyro
author_rank: Super Contributor
timestamp: May 14, 2019, 06:28:30 pm
content: Quote from: tkamiya on May 14, 2019, 06:03:07 pmThe ADEV tables will only show on the main screen if there is enough room for them to fit. They are a lower precidence that all the other items. Maybe try something like screen mode "K" (/vk or $k). Also, unless you are getting actual PPS/FREQ measurements from an external counter, Heather calculates "bogo adevs" from the devices self-reported loop measurements and bogo-adevs may have little relation to reality.

author: tkamiya
author_rank: Super Contributor
timestamp: May 15, 2019, 02:00:14 am
content: Yes, I am aware of BOGO. I'm waiting for TICC, so no external measurements. I'm just trying to setup LH to exactly how I want it long term.

author: tkamiya
author_rank: Super Contributor
timestamp: May 20, 2019, 02:56:37 pm
content: I am now trying to "BOGO" PRS-10 and Resolution-T.The scale appears but the range is 1e-019 to 1e-23. Therefore, graph itself does not appear. It doesn't auto-adjust to more reasonable value like it did with GPSTM. I can see why it wouldn't with Precision-T as it does not have a 10MHz output though.Is there any solution to this?

author: texaspyro
author_rank: Super Contributor
timestamp: May 20, 2019, 07:26:28 pm
content: Quote from: tkamiya on May 20, 2019, 02:56:37 pmThose devices don't send any data that makes sense to calculate xDEVs over, so they don't do any xDEV calculations.

author: tkamiya
author_rank: Super Contributor
timestamp: May 20, 2019, 07:36:01 pm
content: Thanks again.