hybridfree/eevblog-posts huggingface dataset:
HP 3406A Millivoltmeter [SOLVED] - Page 1
Forum section: electronics / general

author: RF-Engineer
author_rank: Contributor
timestamp: April 29, 2024, 06:33:47 pm
content: HP 3406A RF Millivoltmeter wanted that actually passes the 1.0V calibration test with the probe inserted into the CAL test jack. Most don't. Reply with actual photo of the unit in the CAL position with the probe inserted and the 3406A on the 1 V scale. In return, I will make a generous offer if it really passes.

author: Zucca
author_rank: Unknown
timestamp: April 29, 2024, 06:55:39 pm
content: https://www.keysight.com/us/en/assets/9018-02640/user-manuals/9018-02640.pdfMind-blowing how much details were in a manual back in those days...Operating and Service Manual in one document... I miss those days so much....

author: RF-Engineer
author_rank: Contributor
timestamp: May 10, 2024, 05:16:06 pm
content: I can call off the cry for help with a purchase. Today, I buckled down and began looking at the sampling probe of an HP3406A that had been shelf-bound.Starting at the probe tip, I discovered that one of the Schottky bridge diodes had failed shorted. A 1N5711 is a temporary replacement. For a permanent fix, I ordered a sleeve of Avago/Broadcom HSMS-2828 quad bridge diodes, the benefit being a matched set of diodes on a single substrate.As these RF millivoltmeters get passed around with time, invariably someone applies way too much input voltage to the probe -- and there it sits until it's sold to someone like me.Paul
FS (EU/US) Dediprog SF100 Flash Programmer - Page 1
Forum section: electronics / general

author: OneGeekGuy
author_rank: Regular Contributor
timestamp: May 06, 2024, 02:41:32 pm
content: Hi,For selling Dediprog SF100 Flash Programmer, used but working.Datasheet can be found here:https://www.dediprog.com/product/SF100Ideally shipping within Europe, but also possible abroad.Price100Euro80EuroShipping EU around 10 EuroShipping abroad around 15 EuroThanks!

author: OneGeekGuy
author_rank: Regular Contributor
timestamp: May 10, 2024, 08:15:21 am
content: Price Drop to 80EuroThanks!
SOLD [worldwide] Fluke 54-II Dual Input Digital Thermometer - Page 1
Forum section: electronics / general

author: Kirkhaan
author_rank: Frequent Contributor
timestamp: May 05, 2024, 10:04:48 am
content: SOLDSelling a Fluke 54-2 thermometer. It is in excellent condition and works well.Scope of delivery as shown in the photos. Please note that the K-type thermocouple is not included.Asking price is€245€225, excluding shipping. Payment via Paypal. Buyer protection is fine (fees are for the buyer).

author: Kirkhaan
author_rank: Frequent Contributor
timestamp: May 09, 2024, 07:17:24 am
content: I am reducing asking price to €225.
SOLD: HP 3478A, two to choose from - Page 1
Forum section: electronics / general

author: iroc86
author_rank: Regular Contributor
timestamp: May 09, 2024, 06:27:36 pm
content: I have two HP 3478A 5.5 digit multimeters for sale. I bought these for a datalogging project and never used them.The first meter has a calibration "certificate" from the seller that I bought it from in 2021. It was a measurement and test company, so I have no reason to doubt the numbers. It checks out and seems to work just fine. $100 + shipping.The second meter is uncalibrated--probably due to a dead backup battery--and is in pretty rough cosmetic shape. Might be useful if you have a way to calibrate it yourself. $40 + shipping.Willing to offer a discount and combine shipping if you want both!

author: iroc86
author_rank: Regular Contributor
timestamp: May 13, 2024, 07:57:49 pm
content: Doesn't look like a whole lot of interest... will be posting these to eBay.

author: indeterminate
author_rank: Regular Contributor
timestamp: May 14, 2024, 01:08:49 am
content: Any idea on what shipping to AU would beill take both of them if the shipping is not to much

author: iroc86
author_rank: Regular Contributor
timestamp: May 14, 2024, 01:20:21 pm
content: Quote from: indeterminate on May 14, 2024, 01:08:49 amI'll check and PM you...
SOLD (EU / Worldwide) JBC DD-2A 2-channel soldering station control unit (230V) - Page 1
Forum section: electronics / general

author: KhronX
author_rank: Frequent Contributor
timestamp: April 23, 2024, 09:08:45 am
content: Selling aJBC DD-2A2-channel soldering station control unit - a good few years old but fully functional, under 1200h actual work hours in total (see stat screens).Bought used, and was in my personal use for a few months, until i got a great deal on a 4-channel unit.No stands or handpieces are included, but i can provide an EU or UK mains lead.Would prefer to ship within EU (25eu); open to shipping elsewhere as well, pending checking costs - customs and any other fees are the buyer's responsibility.Looking to get~400eu350eu, but i'm open to offersPayment via Paypal, Mobilepay or bank transfer.

author: KhronX
author_rank: Frequent Contributor
timestamp: May 05, 2024, 04:59:13 pm
content: Asking price updated

author: KhronX
author_rank: Frequent Contributor
timestamp: May 08, 2024, 05:15:54 am
content: Provisionally reserved, pending payment.SOLD!
WTB: HP Logo Pin - Page 1
Forum section: electronics / general

author: Ghastly
author_rank: Contributor
timestamp: May 07, 2024, 08:13:26 pm
content: Hi there, I need one or several HP Logo Pins as I get new equipment several of them are without the pin. I tried finding them online but I can't find any. I'm located in the US, see picture for details.
Looking for old document from MOS Technology - Page 1
Forum section: electronics / general

author: Dunkelwind
author_rank: Newbie
timestamp: May 06, 2024, 06:00:13 pm
content: Looking for document from MOS Technology Inc.:"Specification for Scientific Calculator Arrays MPS 2525-001, MPS 2526-001,' the second revision of which is dated Jan. 7, 1974.tnx
(SOLD) [EU] Siglent SDS1104X-E oscilloscope - Page 1
Forum section: electronics / general

author: bateau020
author_rank: Frequent Contributor
timestamp: May 02, 2024, 11:31:58 am
content: And another one up for sale. Victim of an upgrade to 12 bits.SDS1104X-E, "liberated" to 200MHz, and all options enabled. So you get a SDS1204X-E.Clean, no (easily visible) scratches or discolorations, latest firmware, no intense use (see picture).Included: new cables and probes (all from my new scope, PP510 100MHz probes), original box, original paperwork.Bought new Dec 2021 from one of the official German distributors.Price: 300 EUR, excl. shipping (at cost).PM me if you are interested.
GPIB Breakout Board - Page 1
Forum section: electronics / general

author: dazz1
author_rank: Frequent Contributor
timestamp: April 25, 2024, 03:52:58 am
content: I have made my own piece of test equipment. It is a GPIB breakout board. It plugs directly into the HP Logic Analyser cable so no messing around with individual probes. Just plug and play. It would be easy to make up a cable to adapt the GPIB breakout connector to another make/model of LA.In addition, the signals are also exposed on a separate header.I have made the first GPIB breakout board and updated the pcbway share project, available for ordering your own board ([url=https://www.pcbway.com/project/shareproject/GPIB_Break_Out_Board_f7812b3e.html] ).I now have 4x spare boards and parts available for sale. Cheaper than buying parts and boards for one, like I have done.I am selling the parts as a kit for assembly. Just requires a soldering iron and some time.To calculate the shipping cost, go to this website:https://www.nzpost.co.nz/tools/rate-finder/sending-internationally/parcels] https://www.nzpost.co.nz/tools/rate-finder/sending-internationally/parcels for a Size 1 box from Wellington.

author: oPossum
author_rank: Super Contributor
timestamp: April 25, 2024, 04:56:53 am
content: Quote from: dazz1 on April 25, 2024, 03:52:58 amThis may damage the logic analyzer (4V max typical). There must be a proper 10:1 termination network on your PCB or use a HP/Agilent 01650-63203Termination Adaptor.

author: dazz1
author_rank: Frequent Contributor
timestamp: May 11, 2024, 12:56:47 am
content: Quote from: oPossum on April 25, 2024, 04:56:53 amI have designed a new version with the termination network included in the breakout board. Just waiting for the pcbs to be delivered.
[SOLD] Tektronix FG501 - Page 1
Forum section: electronics / general

author: BlownUpCapacitor
author_rank: Frequent Contributor
timestamp: May 06, 2024, 04:04:17 am
content: I am selling my Tektronix FG501. It works with no issues. Front panel pots and switches were cleaned mostly.Multiplier cam shaft shield is a bit loose, but tape fixes that.EEVblog members get $10 off.https://www.ebay.com/itm/166748178906?mkcid=16&mkevt=1&mkrid=711-127632-2357-0&ssspo=CTI57OxBQDq&sssrc=2047675&ssuid=CTI57OxBQDq&widget_ver=artemis&media=COPY
FS [US to XXX] Heltec 12-16 Active Balancer 5A - $30 - Page 1
Forum section: electronics / general

author: Zucca
author_rank: Unknown
timestamp: May 06, 2024, 03:28:09 am
content: 30 USD shipped in the USA...others please ask.This is how it works:Note: it is the "always run" version.Reason for selling: I upgraded to a smart balancer.
[WTB]: HP 8340B Synthesized Sweeper - Page 1
Forum section: electronics / general

author: Marco_Mar
author_rank: Contributor
timestamp: May 03, 2024, 10:34:10 am
content: Hi,I'm looking for an HP 8340B Synthesized Sweeper, up and running, and in a decent state of calibration.Is anyone out there willing to part with it?Only offers from UE countries, because I live in Italy.Thank you for reading
For Sale Keysight 34420A 7 1/2 digit Nano-volt, Micro-ohm Meter,  $2300.00  OBO - Page 1
Forum section: electronics / general

author: kkayser
author_rank: Contributor
timestamp: April 05, 2024, 01:36:49 pm
content: For Sale Keysight 34420A 7 1/2 digit Nano-volt, Micro-ohm Meter. $2300.00 OBO Free Shipping.Four wire connecting cable, shunt, and power cord included. Handle is not original to the meter, but works perfectly. All reasonable offers acknowledged. Excellent cosmetically and functionally.

author: Zucca
author_rank: Unknown
timestamp: April 05, 2024, 01:56:11 pm
content: The DMM microscope... what you can see with it is very interesting.Unfortunately I do not have the budget for it.Oh WOW just realized Keysight did not released any Nanovoltmeter with LXI/LAN

author: HighVoltage
author_rank: Super Contributor
timestamp: April 05, 2024, 03:50:47 pm
content: Quote from: Zucca on April 05, 2024, 01:56:11 pmI am sure they are working on a new version, black edition, with LXI/LANThese 34420A are just really good DMM.Just the prices for used instruments have skyrocketed.The new list price is €8,485.89 in Germany (Including VAT)Not long ago, the list price was half of this.@kkayserIf you cover the serial number, you also need to cover the qr-code,Nice instrument and nice cables!

author: Hydron
author_rank: Super Contributor
timestamp: April 05, 2024, 08:55:52 pm
content: RE a new version, it seems that they've redesigned it relatively recently (mid-2021) for ROHS compliance - I wonder if they have changed anything significant other than component lead plating?It used most of the custom 34401A parts by the looks of it (basically everything after the front-end seems very similar), so they must still have a decent stock of them. Unless they've done a big change of the internal architecture but decided not to update it for LAN/LXI/Black edition etc?Sadly this is not a justifiable purchase as much as I'd like one, I guess I'll just need to keep an eye out for a broken one to repair.

author: Zucca
author_rank: Unknown
timestamp: April 07, 2024, 02:13:05 am
content: Quote from: HighVoltage on April 05, 2024, 03:50:47 pmWondering if it is because KS is not selling anymore to private...

author: WattsThat
author_rank: Frequent Contributor
timestamp: May 01, 2024, 11:59:34 pm
content: Bump from a satisfied buyer. Purchased the 34461A from kkayser and unit was delivered as promised, packed extremely well and of course it functions as expected. Always a pleasure to purchase privately as we all win rather than paying ridiculous fees to eBay.

author: kkayser
author_rank: Contributor
timestamp: May 02, 2024, 01:44:05 pm
content: Thank you!
FS: Fluke 910R - Page 1
Forum section: electronics / general

author: AMS09
author_rank: Newbie
timestamp: May 02, 2024, 03:41:20 am
content: Selling a Fluke 910R GPS Controlled Frequency Standard was a R&D surplus, attached photos are presented to warm up / clock synchronization / locked statement. It has been preforming an excellent stability result.Spec: 10MHz Frequency standardAsking for $6500.
Wandboards for sale   1x Quad kit, 1x Solo kit, 1x Solo  EDM 1 module - Page 1
Forum section: electronics / general

author: coromonadalix
author_rank: Super Contributor
timestamp: May 01, 2024, 10:18:37 pm
content: well if some people want to play with thesesThe Quad is bootable Android 7 can't recall, i think you can push android 9 ?? played 1 hr with it, wi-fi and bluetoothI'm no linux / droid guyThe solo's where never started up, no sd-card in them, i later found an Quad EDM 1 module to upgrade a carrierThe quad costed me more than the sale of this bunch lollInfos, datasheets and os'es are availablebest offer ? + shipping, worldwide possible
FS (EBAY): Keysight N9917A Fieldfox Analyzer 18GHz - Page 1
Forum section: electronics / general

author: Retrain5111
author_rank: Newbie
timestamp: May 01, 2024, 10:05:09 pm
content: I posted this to Ebay today but I'm open to lower offers from EEVblog, as I'll save on the EB fee, etc. If you're local to SF Bay Area you can check it out before buying [in front of the local Police station because of its value and portability].https://www.ebay.com/itm/166741587185
SOLD.FS:WW  Textronix P5205 100MHz 50/500x Differentail Probe - Page 1
Forum section: electronics / general

author: watchmaker
author_rank: Unknown
timestamp: May 01, 2024, 06:16:18 pm
content: I had purchased this to wait until I saw a PSU under $100; they do show up. But, I bought the MicSig on sale at Amazon and someone here can likely use this.No PSU is needed for use w/ the correct Tek scopes. All others do need the PSU (Dozens on eBay for $250, but they show up for under $125).The last calibration was in Sept 2011. All lead insulation is intact. Cover is not damaged.I paid $100 for it. If that was too much tell me. Will include US shipping; outside the US I will credit $15 toward shipping.Would like to get the $100 back, but....This is a no risk transaction. As always, the deal is not done until you say you are happy.Regards,Dewey
WTB: Leader LPS-2801 power supply - Page 1
Forum section: electronics / general

author: glitchcatcher
author_rank: Contributor
timestamp: May 15, 2024, 12:59:44 pm
content: Hello,I'm looking for a Leader LPS-2801 power supply. Those seem to be really rare, not much information on them either. I attached a picture.If you have one or can source one, or know someone who could have one or source one, let me know and I will highly appreciate it with my collectors budgetThanks.

author: coromonadalix
author_rank: Super Contributor
timestamp: May 15, 2024, 01:46:02 pm
content: only thing who pops out is this Doultech link ?? and Korean?http://www.doultech.co.kr/views/product/model_detail.asp?model_num=5669

author: glitchcatcher
author_rank: Contributor
timestamp: May 15, 2024, 03:18:33 pm
content: Yeah I've seen this, and I'm guessing it will prove difficult to source it from there, but I'll give it a try.
WTB: N4010A wireless test set - Page 1
Forum section: electronics / general

author: Xtremexp
author_rank: Regular Contributor
timestamp: May 01, 2024, 06:08:01 pm
content: Hello,I am looking for wireless test set Agilent N4010A wireless test set with WLAN 2.4 and 5Ghz and Bluetooth options. Let me know if anyone got it for sale.Thanks!
FS: Entire Medical Robotics Device Manufacturer  - Page 1
Forum section: electronics / general

author: jrc22ski
author_rank: Regular Contributor
timestamp: May 01, 2024, 03:57:53 pm
content: Advanced Notice:Online auction sale of Medrobotics:Medrobotics – Turnkey/Online AuctionOVER 40,000 SQ FT FACILITY: ROBOTIC SURGICAL DEVICE MANUFACTURERTue, Jun 11 at 10am (ET)Raynham, MA U.S.A.Flex® Robotic Systems, Flex 360™ Systems, Medical & Lab Equipment, Manufacturing, Test & Measurement, Plant Support, Material Handling, FF&E, and Inventory.Facility Available Turnkey - All Assets Including Intellectual Property Available for Immediate Sale (Includes Purchase or Lease of Building w/ Clean Room)Images and details are available at the Medrobotics sale page which can be found here:https://soldtiger.com/sales/robotic-surgical-device-manufacturer/Any other comments or questions can be directed to me through this forum or at my coordinates below.Thank you,John CoelhoSr. VP - Remarketing ServicesTiger Group | Asset Intelligent84 State St. 4th FloorBoston, MA 02109P: (617) 523-5866 | C: (617) 797-0430 | F: (617) 523-3007JCoelho@TigerGroup.comwww.TigerGroup.com
Giving away for free (Sydney AU): BROKEN HL-2070N printer - Page 1
Forum section: electronics / general

author: rs20
author_rank: Super Contributor
timestamp: April 30, 2024, 08:23:52 pm
content: Listedhere in Facebook marketplace, but no Facebook account required to view the listing and I'll give it away for free to anyone who messages me within the forum system here. Pick-up only though, from Balmain 2041.HL-2070N printer, presenting error E52: "SCANNER MOTOR MALFUNCTION". Apart from the error message and it totally not working, it is in good condition. I've put virtually zero effort/attempts into repairing it, I haven't even taken it apart!
(SOLD) FS: Siglent SDS1104X-E - Page 1
Forum section: electronics / general

author: mmora
author_rank: Newbie
timestamp: April 16, 2024, 09:43:13 pm
content: Hi all,I've been reading EEVblog for a long time now, but I never found a reason to post. I guess now it's the time.I've been into electronics for a while now and I mostly tinker with small projects related to retro video game consoles. At one point, I was really active with modding and repairing, so I needed an oscilloscope. After reading (a lot) here, I ended buying a Siglent 1104X-E from Welectron in April, 2022. So it's been a couple years now. I won't bore you with the details, but with two little kids now things got... hectic. So, I still believe I need and oscilloscope for the occasional check or repair, but I guess I won't need a four channel tool for that and it has been sitting in my desk without use for the last year. Since 2022, I only used it very occasionally, and even though it's been really helpful, I only ever needed to use a single channel. It is 200MHz enabled, AWG, MSO and WIFI enabled (just the keys entered, no hardware included - or used). I probably knew this was going to happen at some point, as I kept the original packaging and documents. So, it will come boxed, with all four probes, and the original invoice, if needed (I think there is still a third year of guarantee since the original purchase). I can send pictures, of course.As I don't have any feedback here, I can provide my eBay account (mrdangerousbeans) as an example of my historical feedback.Shipped from Spain via courier, the asking price is 350€ shipped to basically anywhere in the EU.Thank you,

author: mmora
author_rank: Newbie
timestamp: April 23, 2024, 09:05:16 pm
content: I took a few pictures today. I'll also list it in a local second hand app tomorrow. Also open to (reasonable) offers.

author: mmora
author_rank: Newbie
timestamp: April 27, 2024, 09:25:05 pm
content: It is SOLD now (not in this forum, however, but via the local app).

author: mcjeff93
author_rank: Newbie
timestamp: April 30, 2024, 06:11:09 pm
content: goddamn, i was late.
WTB Prova AVM-07 Anemometer wind turbine, working or not - Page 1
Forum section: electronics / general

author: Paceguy
author_rank: Regular Contributor
timestamp: April 29, 2024, 04:41:21 pm
content: I have a Prova AVM-07 anemometer that is missing the external wind turbine and cable assembly. Does anyone have a parted out unit or non-working one that they would be willing to sell?
[SOLD] - eBay (US): Vremya-CH VCH-1006 Digital Passive Hydrogen Maser - [SOLD] - Page 1
Forum section: electronics / general

author: christopher.robot
author_rank: Newbie
timestamp: February 14, 2024, 12:10:29 am
content: A little too rich for my blood, alas - but maybe one of you other Time-Nuts are willing to pony up $5k for a hydrogen maser (albeit passive).https://www.ebay.com/itm/364591313317https://www.vremya-ch.com/index.php/en/products-en/product-arkhiv-en/vch-1006-en/index.html

author: 5065AGuru
author_rank: Frequent Contributor
timestamp: February 14, 2024, 05:42:48 pm
content: Thanks for the heads up!I have a CH1-76 passive maser and I'd be interested in this more modern one but after pulling a lot of teeth to get decent documentation on that one I'd hesitate to get the newer one.Another problem is the Hydrogen supply. If it is empty it will be almost impossible to get it refilled.That said I'd still be tempted if I had the $ to burn.Cheers,Corby

author: KE5FX
author_rank: Super Contributor
timestamp: February 14, 2024, 06:05:27 pm
content: I guess I'll find out. Not to worry, I'm sure the Russians will be happy to help me with any documentation I need to repair and maintain it.Thanks for the heads-up... I think...

author: christopher.robot
author_rank: Newbie
timestamp: February 14, 2024, 06:27:42 pm
content: Haha, I'm certainly envious, but I'm glad it found a good home. What a cool bit of kit!

author: KE5FX
author_rank: Super Contributor
timestamp: May 01, 2024, 04:36:27 am
content: Some notes and photos, now thatBunnie'sName That Warecontest is over.I still have a lot to learn about this particular gadget, but so far it has been working very well.

author: Zucca
author_rank: Unknown
timestamp: May 01, 2024, 01:14:25 pm
content: in case you don't know what they are talking about...https://en.wikipedia.org/wiki/Hydrogen_maser
WTB (UK/EU)- Toroidal coil winder & bobbin coil winder - Page 1
Forum section: electronics / general

author: NOTsinceAnipper
author_rank: Newbie
timestamp: April 29, 2024, 01:29:12 pm
content: Hi - Looking for a toroidal coil winder and a bobbin winder - not worried about how old it is or hand / motor driven either fine.regardsTim

author: coromonadalix
author_rank: Super Contributor
timestamp: April 29, 2024, 02:57:06 pm
content: For professional use ?You surely know about diy ones ?
WTB (US/NA) Battery and Charger for Flir Ext (compatible with E4, E5, E6, E8) - Page 1
Forum section: electronics / general

author: ZGoode
author_rank: Frequent Contributor
timestamp: April 29, 2024, 03:54:29 pm
content: I'm looking to buy one battery and one charger for the Flir Ext family to replace charging over micro-USB. If anyone has any extras, especially of the charger, I would be happy to buy one off you.
FS: Motorola MVME 202, 310, 211 - Page 1
Forum section: electronics / general

author: Cloud
author_rank: Regular Contributor
timestamp: March 31, 2015, 03:47:34 pm
content: I just got those modules. They look like they are brand new and the date code on the board is 1983. I have two pcs of each: MVME202 512KB Dynamic Memory Module, MVME310 Universal Intelligent Peripheral Controller and MVME211 RAM/ROM/EPROM memory module. Does anyone has idea how much are worth it? Because on ebay, the prices are 250$+, so I was thinking 100€ per board.I also got user manual for Tektronix TDS2MEM and Programmer Manual for Tek TDS200, TDS1000, TDS2000 and TPS200. They are still originally sealed. I was thinking 10€ per manual?Shipping is cost of the buyer. I am from Slovenia, Ljubljana.Please don't hesitate to contact me.

author: miguelvp
author_rank: Super Contributor
timestamp: April 01, 2015, 02:47:17 am
content: VME bus, brings back memories

author: Cloud
author_rank: Regular Contributor
timestamp: April 04, 2015, 01:44:49 pm
content: I found 4 pcs of HP 01650-63203 100kOhm Termination Adapter for logic analyzer. Date code is 39/93. And it doesn't seem like they have been used much. They come with plastic bag and manuals.Anyone interested?

author: LittleFrog
author_rank: Contributor
timestamp: April 26, 2024, 06:26:46 pm
content: Hi,I realise this thread is old, but please could I ask if you sold these boards?Thanks, Dave

author: ZGoode
author_rank: Frequent Contributor
timestamp: April 28, 2024, 04:09:02 am
content: Quote from: Cloud on April 04, 2015, 01:44:49 pmI'd probably be interested in those if the shipping and price is reasonable. How much would it cost to ship to Gilbert, AZ 85295 in the USA?
FS: Band 1 Stereo FM Link Broadcast Warehouse TX25 Transmitter + Receiver - Page 1
Forum section: electronics / general

author: kloss
author_rank: Contributor
timestamp: April 26, 2024, 04:25:40 pm
content: Band 1 Stereo FM Link Broadcast Warehouse TX25 Transmitter + Professionally adapted Stereo ReceiverTX25 is the full bandwidth version capable of any Band 1 broadcast channelReceiver is XTAL controlled currently on 52.75 MHz some alternatives available from stock if requiredTested working, 1 owner from new these always gave great resultsThe transmitter and receiver both offer MPX Composite supportA new SBS Stereo Audio Limiter may be included free of charge on requestAny questions - let me know£1000 Collection or shipping available
FS: ENI 403LA wideband RF amplifier 3 Watts 37dB 150 kHz - 300 MHz - Page 1
Forum section: electronics / general

author: kloss
author_rank: Contributor
timestamp: April 26, 2024, 04:23:43 pm
content: ENI 403LA Class A wideband RF amplifier 3 Watts 37dB 150 kHz - 300 MHzTested in good working order (photo 3 shows working)Power cable fitted with IEC type connector to support variable length cables£200 Collection or shipping availableSOLD
FS: ENI 403L wideband class A linear amplifier 150 kHz ~ 250 MHz low O/P fault - Page 1
Forum section: electronics / general

author: kloss
author_rank: Contributor
timestamp: April 26, 2024, 04:21:14 pm
content: ENI 403L wideband class A linear amplifier 150 kHz ~ 250 MHz, designed to give up to 3W output.Showing low RF output. All discrete component design so should be an easy repair.£125 Collection or shipping availableSOLD
FS: Cranage Plipman EMC Emissions tester 30 kHz ~ 30 MHz - Page 1
Forum section: electronics / general

author: kloss
author_rank: Contributor
timestamp: April 26, 2024, 04:16:53 pm
content: Cranage Plipman EMC Emissions tester30 kHz ~ 30 MHz professional EMC emissions tester, portable in carry case with mains connection cable and charger.Connect to either mains or RF input to monitor emissions either from the mains or from RF signals.Photo available showing working, receiving BBC R4 LW from small indoor aerial (not included).Calibration carried out last year so spot on frequency.£200 Collection or shipping availableNO LONGER AVAILABLE
FS: HP Agilent 33120A 15 MHz Function Arb. Waveform Generator, OPT 001, NEW CAL - Page 1
Forum section: electronics / general

author: kloss
author_rank: Contributor
timestamp: April 26, 2024, 04:14:43 pm
content: HP / Agilent 33120A 15 MHz Function Arbitrary Waveform GeneratorFitted with option 001 (phaselock and external reference)The internal reference has also been freshly calibrated against an off-air calibration standard (Droitwich) after a suitable run time.In excellent cosmetic condition and excellent working order, never missed a beat during the years i've owned this nice item.Comes complete with original user manual and power cable.Tested working (photo 3). £349 Collection or shipping available.SOLD
sold: Sayrosa Automatic Modulation Meter FM/AM 1.5 MHz ~ 2 GHz (4 GHz) - Page 1
Forum section: electronics / general

author: kloss
author_rank: Contributor
timestamp: April 26, 2024, 04:13:00 pm
content: SOLDFarnell Automatic Modulation MeterFM/AM 1.5 MHz to 2 GHz (useful to at least 4 GHz)Made by Sayrosa (now RF Logic)Tested working. £150 Collection or shipping available.
FS: Leader 3282 Stereo RF Sig Gen - Page 1
Forum section: electronics / general

author: kloss
author_rank: Contributor
timestamp: April 26, 2024, 03:56:37 pm
content: Leader LG3282 FM Stereo and AM Stereo RF Signal Generator 10 kHz to 280 MHz top of the line model with many great features. Previously used for Car Audio System development work. Included is a copy of the user manual and an Aeroflex carry case, custom designed for similar equipment. Top quality item from Japan. Tested working, calibration recently carried out so spot on frequency. £599.00 Collection or shipping available.SOLD
Sold Signal Hound SA124B - Page 1
Forum section: electronics / general

author: gsmaudis
author_rank: Newbie
timestamp: April 26, 2024, 02:56:22 pm
content: SoldI'm selling my Signal Hound SA124B 100kHz to12.4 GHz spectrum analyzer.In full working order, it was bought directly from the manufacturer.I'm asking $980 including shipping within the wordwide.Included are the SA124B and USB cables.Software is available off Signal Hound's website:https://signalhound.com/spike/
FS (AUS): Weller WEP70 Soldering Station 70W - $180 - element14 - Page 1
Forum section: electronics / general

author: austfox
author_rank: Regular Contributor
timestamp: April 29, 2024, 08:59:33 am
content: After purchasing a Multicomp Power Supply detailed in an earlier post from the 'Fire Sale' at element14 in Sydney, I note they also have Weller WEP70 station kits for $180.47 with free delivery. Their standard price is $391.https://au.element14.com/weller/t0053298390/soldering-kit-450deg-c-85w-230v/dp/291552002?st=FireSaleAllProductsThe kit includes side cutters, solder, and a spare tip with the station. I have seen just the station from other suppliers for around $250 delivered, so a nice little saving. I haven't used one so can't vouch for them, but I have a couple of the earlier WESD51 50W versions that uses the same 'ET' tips and they work well for me.
. - Page 1
Forum section: electronics / general

author: Ookpik
author_rank: Contributor
timestamp: April 24, 2024, 07:20:26 pm
content: .

author: Crambone
author_rank: Frequent Contributor
timestamp: April 26, 2024, 02:51:24 pm
content: I will take if shipping available Mountainside, NJ 07092Andrew
LFS I want to buy a Fluke 9100 - Page 1
Forum section: electronics / general

author: qudianzhuang
author_rank: Newbie
timestamp: April 25, 2024, 07:30:30 am
content: I am looking for a Fluke 9100, fault machine and normal function are acceptable, the price is not too expensive.

author: coromonadalix
author_rank: Super Contributor
timestamp: April 26, 2024, 10:14:00 am
content: better specify the letter who goes after 9100calibrator / oven etc ... on another thread just writing 9100 bring confusion
SOLD 2x Yokogawa 701943 500MHz 1:10 passive voltage probe - Page 1
Forum section: electronics / general

author: Kirkhaan
author_rank: Frequent Contributor
timestamp: April 25, 2024, 06:18:38 am
content: SOLDAnyone interested in these 2 Yokogawa probes? Asking price is€100both together.They are in excellent condition and working properly.What you see on the pictures is what you get.I am based in Germany and willing to ship worldwide. PM me in case you are intersted.Payment via IBAN or Paypal. Buyer protection is possible if buyer pays the ~4% uplift.
SOLD — Learning Art of Electronics course kit - Page 1
Forum section: electronics / general

author: petrinch
author_rank: Contributor
timestamp: April 19, 2024, 04:53:09 pm
content: Nearly complete** kit including 2016 LAOE course book, components, and custom circuit boards (“Big Green”, keypad, lcd and programming module)—all in unused, mostly unopened, like-new condition. If it’s listed on the LAOE sitehttps://learningtheartofelectronics.com/then I probably have it.US$200 (plus shipping within continental US) gets the lot (and no, I won’t break up the collection).**See attached digital/analog parts list spreadsheet. By “nearly complete” I mean I originally ordered EVERYTHING highlighted on the spreadsheet but some substitutions may have been made and a few low-value items may in fact be missing. My asking price factors that bit of uncertainty into the deal.

author: watchmaker
author_rank: Unknown
timestamp: April 19, 2024, 08:21:35 pm
content: I will take it. Sent you a pm.Thanks.Dewey

author: YetAnotherTechie
author_rank: Regular Contributor
timestamp: April 19, 2024, 10:31:37 pm
content: Just a few days someone was looking for PCB design files, program code, Microcontroller Code, etc. because he couldn't find them anywhere anymore. Do you have them?https://www.eevblog.com/forum/buysellwanted/tom-hayes-the-art-of-electronics-hands-on-lab-course-program-and-code-listings/msg5442092/#msg5442092

author: petrinch
author_rank: Contributor
timestamp: April 19, 2024, 11:46:07 pm
content: Thanks. I just posted a reply in that topic you referenced.

author: watchmaker
author_rank: Unknown
timestamp: April 25, 2024, 04:16:51 pm
content: Just arrived and extremely well packed and organized. THANK YOU.Looking thru the Lab Manual and the instructions for the Dallas bread board, I think this follows the philosophy of the text. To me, it assumes a lot about the leaner and is not particularly kind.But, as an adjunct to real Analog videos and the text based course I am using, it fill a gap. Should be very helpful once I get past capacitors and inductors.
FS [CAN] : HP 8503A S-parameter test set - Page 1
Forum section: electronics / general

author: Xtremexp
author_rank: Regular Contributor
timestamp: March 16, 2023, 09:03:12 pm
content: I am selling HP 8503A S-parameter test set in good working condition. I no longer need it. I am open to offers. Its is located in Vancouver BC, Canada.I am asking CAD $100 OBO

author: Xtremexp
author_rank: Regular Contributor
timestamp: April 02, 2023, 06:13:56 pm
content: adding price in the listing

author: entegreci
author_rank: Newbie
timestamp: April 24, 2024, 10:35:01 am
content: Hello, this is Necati UNAL from Turkey. I am interested in this device. Is it still for sale? Please Information. Thank you
FREE: Electronics Australia, ETI, AEM Magazines - Page 1
Forum section: electronics / general

author: EEVblog
author_rank: Unknown
timestamp: February 13, 2024, 11:06:58 pm
content: FREE: Electronics Australia, ETI, AEM Magazines from 1960 to 1988Someone from Adelaide is looking for someone to take them off their hands for free, or to cover postage.Contact me if interested and I'll pass you on.

author: JoeRoy
author_rank: Regular Contributor
timestamp: April 23, 2024, 05:16:47 pm
content: https://datassette.org/revistas/electronics-australiahttps://datassette.org/revistas/electronics-today-international-au

author: Brumby
author_rank: Unknown
timestamp: April 24, 2024, 02:25:43 am
content: In discussions with them.
WTB/LFS Gossen Metraline DM 62 GREEN RUBBER CASE - Page 1
Forum section: electronics / general

author: GigaHurts
author_rank: Contributor
timestamp: April 23, 2024, 09:26:45 pm
content: I have recently bought a used but working DM 62 but it came without the lovely bright green rubber case/protector. (Photo for illustrative purposes)It seems they don't have this for sale anywhere (contacted the german Gossen branch for support but so far no response). They only sell spare cases for their Metrawatt product line, not the Metraline sadly. However, i read somewhere that this is a rebranded meter by some indian company. Does anyone know what model that is? Maybe the case can be obtained for the original model name.Maybe someone here has a broken one sitting on the shelf and wouldn't mind parting with the case? DM42, 61 and 62 should all be compatible.Suggestions on how to DIY a case, or maybe some universal rubber cases that might fit this model are welcome!
FS - Fluke 9100 Meter Calibrator - Page 1
Forum section: electronics / general

author: carlos100
author_rank: Newbie
timestamp: December 13, 2023, 12:04:43 am
content: Fluke 9100 Meter Calibrator for sale, in operational conditions, just out of calibration and with small number of errors, selling as is including all manuals. asking price $2500.Pickup from our office in Clontarf QLD, or we can ship anywhere in Australia for extra cost.

author: coromonadalix
author_rank: Super Contributor
timestamp: December 13, 2023, 02:18:49 am
content: aka Fluke Wavetek Datron 9100op manualsservice manualschematics ? seems there ...http://www.ko4bb.com/getsimple/index.php?id=manuals&dir=06_Misc_Test_Equipment/Datron/Datron_9100

author: deepfryed
author_rank: Unknown
timestamp: December 13, 2023, 04:38:29 am
content: Nice unit if one can fix the failures. Still not in my budget though, blown it all already

author: wolfy007
author_rank: Frequent Contributor
timestamp: December 13, 2023, 04:27:13 pm
content: Tempting as it is, my budget took a hit this past month, probably not a good time to be selling around Xmas.

author: vvolt
author_rank: Newbie
timestamp: March 01, 2024, 06:50:31 am
content: I'm interested.Please PM me to discuss...

author: qudianzhuang
author_rank: Newbie
timestamp: April 23, 2024, 01:05:42 am
content: How can I contact you if I want this
[SOLD] Two Siglent 5.5 digit DMMs for 6.5 digit Siglent or Keithley - Page 1
Forum section: electronics / general

author: KungFuJosh
author_rank: Super Contributor
timestamp: April 07, 2024, 11:32:46 pm
content: I made the mistake of buying an SDM3065X, and now I want a second one.I have an SDM3045X upgraded to SDM3055X-E (5.5 digits), and I have an SDM3055 (5.5 digits).Both look new and both work great & are in spec / cal. I'm interested in either selling these (separately or together) or trading for an SDM3065X. I would also be interested in a Keithley DMM6500, but probably prefer the Siglent since that's what I'm used to.Please DM me with any questions or offers.Thanks,Josh

author: coromonadalix
author_rank: Super Contributor
timestamp: April 08, 2024, 01:06:35 am
content: but you have some good offers here for 34461a ?us based ?https://www.eevblog.com/forum/buysellwanted/keysight-34461a-6-12-digit-multimeter-$800-free-shipping/

author: KungFuJosh
author_rank: Super Contributor
timestamp: April 08, 2024, 01:24:01 am
content: I emailed him before I bought the 3065X, and he already sold it. I think I'd prefer to stick with the DMMs I mentioned. I'm very happy with the 3065X so far, and I've heard good things about the DMM6500.

author: coromonadalix
author_rank: Super Contributor
timestamp: April 08, 2024, 02:15:07 am
content: i myself, would tend to theses models toogood luck

author: KungFuJosh
author_rank: Super Contributor
timestamp: April 10, 2024, 02:07:40 pm
content: I put the first one up on ebay:https://www.ebay.com/itm/166702727442

author: KungFuJosh
author_rank: Super Contributor
timestamp: April 12, 2024, 09:50:47 pm
content: I listed the 2nd one:https://www.ebay.com/itm/166711149452

author: KungFuJosh
author_rank: Super Contributor
timestamp: April 20, 2024, 10:43:52 pm
content: $420 shipped in the US for the SDM3055 or best reasonable offer.I would trade both meters for the SDM3065X, or both plus cash for a DMM6500.Send a private message if you're interested. I'll ship internationally, but you have to pay actual shipping.

author: KungFuJosh
author_rank: Super Contributor
timestamp: April 22, 2024, 09:56:22 pm
content: The SDM3055 has sold. I'm considering keeping the SDM3055X-E as a 3rd (but silent) DMM, but might still trade + cash for a 3065X if somebody is interested.If anybody is considering selling their DMM6500 or SDM3065X, please DM me.
SOLD: Yokogawa 700937 current probe 15A DC-50MHz - Page 1
Forum section: electronics / general

author: Kirkhaan
author_rank: Frequent Contributor
timestamp: April 22, 2024, 06:27:08 pm
content: SOLDDear hobbyists,For sale is a pre-owned Yokogawa 700937. There are some signs of use, but overall its in good condition.It has been used until recently and is working perfectly fine. I've attached some pictures here, but can make more per need.Asking price is €375.Shipping within EU will be around €20, depending on the country.Shipping to other continents is also possible. For USA will be around €40 (standard shipping).If you are interested, just contact me via PM, so we can sort out the shipping options.Payment can go via PayPal. Its also OK if you want to use buyer protection, but the fees are yours.I am selling as private person. So cannot make formal invoices or the like.First come, first served!
Selling Universal Genesis GX11S (1 unit) and GC30S (3 total) P&P machines - Page 1
Forum section: electronics / general

author: EJOOLEM
author_rank: Newbie
timestamp: April 22, 2024, 08:11:40 am
content: Hello!We are currently selling 2011 Universal Genesis GX11S (1 unit) and GC30S (3 total) P&P machines in working condition. With this one can also buy Rehm Nitro VXP 4200 reflow oven and DEK Horizon 03iX printer. We do daily deal with preowned SMT equipment and feel free to request more information on these units or any other machines that You might be looking for.We ship worldwide, however majority of our machines are located in Europe.
(Sold) Wamco KW-104S minitron displays (filament 7 segment) X3 - Page 1
Forum section: electronics / general

author: adriansmith31
author_rank: Contributor
timestamp: April 21, 2024, 12:21:12 pm
content: I have three Wamco KW-104S aviation grade 7 segment displays which are typically used in avionics such as radios. They are used but have been tested and work great with no burned segments and dead spots on the filaments. The manufacturer calls them "minitron" displays which are in a nutshell filament 7 segment displays in a plug in module package similar to 0.3" LED displays however the pin spacing is non standard.Data sheet herehttps://wamcoinc.com/assets/pdf/Display%20Catalog.pdfMake me an offer for them or I can sell via eBay if you prefer as this makes international shipping and customs much easier. They are currently listed on eBay so if you would like to purchase them that way I can give you the link. Otherwise we can sell directly.Note in order to take a good photo of them working I turned down the voltage as they were too bright overexposing the camera.

author: adriansmith31
author_rank: Contributor
timestamp: April 21, 2024, 09:47:00 pm
content: These have now been sold
FS Agilent 33220A Function / Arbitrary Waveform Generator, 20 MHz - Page 1
Forum section: electronics / general

author: Teichhermelin
author_rank: Regular Contributor
timestamp: March 11, 2024, 08:23:18 am
content: I rarely use the device anymore and would like to sell it here.I imagine the price to be 550 euros.We would have to clarify shipping individually.Device is fully functional and in good condition.would be happy to send more pictures upon request.–Fully compliant to LXI Class C specification–20 MHz Sine and Square waveforms–Pulse, Ramp, Triangle, Noise, and DC waveforms–14-bit, 50 MSa/s, 64 k-point arbitrary waveforms–AM, FM, PM, FSK, and PWM modulation types–Linear & logarithmic sweeps and burst operation–10 mVpp to 10 Vpp amplitude range–Graph mode for visual verification of signal settings–Connect via USB, GPIB and LAN

author: Retirednerd2020
author_rank: Regular Contributor
timestamp: March 11, 2024, 01:06:43 pm
content: I own one of these models. It is a very nice generator and easy to use. Good luck with your sale.

author: Teichhermelin
author_rank: Regular Contributor
timestamp: March 16, 2024, 12:36:29 pm
content: Does my asking price seem to be too high? Or no one is interested in the good piece!

author: tszaboo
author_rank: Super Contributor
timestamp: March 16, 2024, 12:55:18 pm
content: Quote from: Teichhermelin on March 16, 2024, 12:36:29 pmI don't think it's high. It's a good AWG for this price. Be a bit patient, and I think someone will get it.I would, but I bought an AWG last week.

author: Retirednerd2020
author_rank: Regular Contributor
timestamp: March 16, 2024, 01:29:38 pm
content: Price seems right to me (but I have one already).

author: Teichhermelin
author_rank: Regular Contributor
timestamp: April 21, 2024, 02:54:37 pm
content: Would still be available.
WTB - signal gererator for FM stereo work - Page 1
Forum section: electronics / general

author: ttx450
author_rank: Regular Contributor
timestamp: March 04, 2024, 11:48:42 pm
content: Looking for a unit to work on FM stereo. Something like a sencore sg-80, leader 3217, Philips, Kenwood, Panasonic, etc. Open to ideas/offers.. thks..

author: Retirednerd2020
author_rank: Regular Contributor
timestamp: March 06, 2024, 02:08:43 am
content: HiI have an RF Generator, Synthesized Wavetek 2520A 2-2200 MHz. It has come in handy for some of my projects and has worked well.It is not an analyzer like the Sencor you listed but is a nice, general purpose RF generator that will do AM, FM, sweeps etc..Have a look at the specs. online to see if it will help you. It is fairly heavy so shipping will be cost to consider too.What sort of FM work are you trying to do? What is your budget for this equipment?

author: ttx450
author_rank: Regular Contributor
timestamp: April 11, 2024, 12:55:26 am
content: These look like fraud... These were PM to me, not posted to the thread.M3Knight ---> harveygrout75@gmail.com --------> Mark Evans markevans56@yahoo.com Is under investigation by paypal for a fraudulent sell of a Leader 3217 to me. When paypal is finished, will update here and send to Admins and other sites.M3knightZeroPosterPosts: 0Hello Mate ,Any luck yet ?Contact Harvey he's has one he's willing to part with at a fair price. Pm him harveygrout75@gmail.com----------------------------------------------------------------------------------------------------------------------------------------------------------------Offline andrewkurks8ZeroPosterPosts: 0I think i saw a guy called Muncho posted a signal gererator for sale. Here is his email juniormuncho@gmail.com

author: Paceguy
author_rank: Regular Contributor
timestamp: April 21, 2024, 11:07:45 am
content: I have used a Sound Technology ST1000A for decades. They may be old but they are built like tanks and work very well. Many of the big audio companies used these.
SOLD - KNIGHT ML2010 MINI LAB - AN EE LAB IN A SUITCASE! - Ebay Link - Page 1
Forum section: electronics / general

author: t1d
author_rank: Super Contributor
timestamp: April 20, 2024, 05:27:17 pm
content: Here's the Ebay link:https://www.ebay.com/itm/276433140836This unit is in great shape. The description pictures are of the actual unit and show it working. The perf board has never even been used!Or, send me a private message with your email address and I will send all of the details directly to you.The price is just $170USD, plus shipping at cost - no markup, or handling fee. New price is $685.
[LFS UK] "The Art Of Electronics Hands On Lab Course" PCB/Program/Code Files  - Page 1
Forum section: electronics / general

author: Chris56000
author_rank: Frequent Contributor
timestamp: April 10, 2024, 11:39:58 am
content: Hi!. . .Does anyone know if any of the PCB design files, program code, Microcontroller Code, etc., etc., are available to make the "Big Green Breadboard" and the purpose made "Hex Keypad" used in Tom Hayes's book "Learning The Art Of Electronics" have been archived or saved anywhere please?. . .I couldn't find them using the Wayback Machine on the "Learning The Art Of Electronics" web site!. . .I could possibly make suitable "Big Green Breadboard" and "Hex Keypad" PCBs if I know exactly what model of breadboard was used, but obviously without the programming code there would be no point attempting to design one!. . .Tom Hayes's email appears to be dead, is he still living?. . .How much of this Lab Course is it still feasible to do today?Chris Williams

author: petrinch
author_rank: Contributor
timestamp: April 19, 2024, 11:43:43 pm
content: Program code and functional descriptions of the custom boards are available from the LAOE site via the “materials” and “parts, equipment, lab notes” tabs.As for the the pcb design files, AFAIK those are the exclusive property of GearLoose Development (Tom Hayes). I bought a set of board kits from him about a year ago and he included schematics in the package—you would have to communicate directly with Tom himself if you want your own set of boards or a copy of the design files. PM me and I can share the last email address I have for him from about a year ago.Without a doubt, the Lab Course is still perfectly feasible to do today once you’ve gathered all the bits and pieces.