YR2050 Milliohm Meter High-Precision Handheld DC - Page 1
Forum section: electronics / testgear

author: enut11
author_rank: Super Contributor
timestamp: October 20, 2023, 06:48:38 am
content: Good milliohm meters can be very expensive, and, the older they are the more they cost!Found this relatively cheap (~$60US) meter on eBay and was wondering if anyone has tested one.It certainly has a lot of digits and, from what I see on the back label, the test current appears to be up to 500mA? EDIT - see next reply.Brief specs from the listing below:- Resolution 120000- Temperature drift: ±100ppm (10~40 ℃)Gear Minimum position Accuracy indication ±(reading%+range%)1Ω 0.01mΩ ±(0.12%+0.03%)10Ω 0.1mΩ ±(0.08%+0.02%)100Ω 1mΩ ±(0.08%+0.02%)1KΩ 10mΩ ±(0.10%+0.03%)https://www.ebay.com.au/itm/314044912094

author: enut11
author_rank: Super Contributor
timestamp: October 20, 2023, 09:37:25 am
content: I found another listing which has a copy of the manual showing the test current for various ranges.https://www.ebay.com.au/itm/354995090853?_trkparms=amclksrc%3DITM%26aid%3D1110006%26algo%3DHOMESPLICE.SIM%26ao%3D1%26asc%3D20201210111314%26meid%3Db6256c080fd14ca09072b1538c48b51d%26pid%3D101195%26rk%3D1%26rkt%3D12%26sd%3D314044912094%26itm%3D354995090853%26pmt%3D1%26noa%3D0%26pg%3D4429486%26algv%3DSimplAMLv11WebTrimmedV3MskuWithLambda85KnnRecallV1V2V4ItemNrtInQueryAndCassiniVisualRankerAndBertRecallWithVMEV3CPCAuto%26brand%3DUnbranded&_trksid=p4429486.c101195.m1851&amdata=cksum%3A354995090853b6256c080fd14ca09072b1538c48b51d%7Cenc%3AAQAIAAABYObhgc4Nk8%252BdtAwOww4FKLaj%252FQ5qqgDlQCuqZA43WcPFUWDERCUugbbOk7XQv0JXlBfqCg2xKF3WcPghxGMFw2oSlXvfExEaMYr7I7LmrHcP6czY1wIMt0ORyKiCWt95xldincyyBx3g%252BNDW%252B%252FhWUgTaBhK6xAm%252BJIbCOMehu%252BdwAKIos3lSV2cWI%252FfMjOs%252Fi0yQMRy3Wk2yf0yEUJyGzN4hArsdKH1LckEzGK9TWa5JzJ4%252FcrjcMmUenOReljRwpnqh%252Bug3WW5L3OxFmrBPNJdurtk%252FVo3ow5l1c3tXXqHhRtfD%252BUFtDeWTDzJkHjkUw5nO%252FvVQGLxbesKo211fiHFl63rykbb4AGMfUfGij8UB3%252BnVv6ZpXSQ5q55i%252BaidGhPsUYWcRYXsDzpapOw6Pn3qrZlKCadIBFlj3FPNY8OABteN1Ssyg0KYHEW4talyRLNFVdi2CqhJp7kOARqDUQw%253D%7Campid%3APL_CLK%7Cclp%3A4429486

author: enut11
author_rank: Super Contributor
timestamp: October 20, 2023, 08:50:57 pm
content: YT video feedback says there is an english manual available - somewhere!The YR2050 Modes:1.general mode; 2.checking mode; 3.background setting; 4. energy saving setting; 5.up down limited alarm;6.port setting; 7.calibrate/adjust; 8. restore calibration/adjust; 9.contrast; 10.resistance ratio settingStill looking for an interior shot to see if it might be hackable...

author: UK
author_rank: Regular Contributor
timestamp: October 20, 2023, 09:37:10 pm
content: Hey! It's a nice milliohm meter, I use it frequently.I'm currently out of town, but on Tuesday I could take some pictures of its insides... of course if you're not in a rush to buy this unit today

author: thm_w
author_rank: Super Contributor
timestamp: October 20, 2023, 09:47:32 pm
content: You can find a few short reviews here:https://www.aliexpress.us/item/2255800323558045.htmlSounds like the menus are in chinese only.video:https://www.youtube.com/watch?v=h4MDwn6nUH4What are you planning to hack on it, the test current?

author: enut11
author_rank: Super Contributor
timestamp: October 21, 2023, 02:18:37 am
content: I would like to be assured that the test current is adequate and not excessive.Also interested in verifying the stated accuracy which seems to be better than some more expensive units.There must be some Vref in there and it would be nice to check that out.

author: alm
author_rank: Super Contributor
timestamp: October 21, 2023, 07:39:54 am
content: Quote from: enut11 on October 20, 2023, 09:37:25 amI don't see test current, just power consumption. There may well be a DC-DC convertor that's driving the current source.Test current is generally not so important. What's important is dissipation in the DUT (I^2*R) and the voltage drop. For contact resistance measurements, in which you don't want the test signal to break through any oxides, since that's what you want to measure, the voltage is usually kept under 20 mV. Otherwise whatever voltage doesn't damage your DUT will work. For in-circuit measurements it would be useful if the voltage was below a diode drop.Quote from: enut11 on October 21, 2023, 02:18:37 amProbably nothing special for 0.08%.

author: UK
author_rank: Regular Contributor
timestamp: October 21, 2023, 04:34:48 pm
content: Quote from: enut11 on October 21, 2023, 02:18:37 amRegarding the test current for its different modes, I've found it in my notes:1 Ω -97.88 mA10 Ω -9.874 mA100 Ω -0.989 mA1 KΩ -0.097 mAAll measurements was done with my trusty UT171B.Hope that helps you a bit while I'm traveling.

author: enut11
author_rank: Super Contributor
timestamp: October 21, 2023, 08:20:42 pm
content: Thanks@UK. That is certainly what I wanted to know. These are not nice round figures so I am guessing the 0.08% calibration is done through saved constants.When you get home, could you also check the compliance voltage?enut11

author: UK
author_rank: Regular Contributor
timestamp: October 21, 2023, 10:15:57 pm
content: I've already measured it after buying the unit... and if I remember correctly it was about 1-2 mV.But anyway I check it again for each mode.

author: alm
author_rank: Super Contributor
timestamp: October 21, 2023, 10:22:46 pm
content: Quote from: UK on October 21, 2023, 10:15:57 pmHow can ~100 mA through 1 Ohm be limited to 2 mV unless it's a constant voltage instead of a constant current source?

author: enut11
author_rank: Super Contributor
timestamp: October 21, 2023, 11:30:28 pm
content: Need to measure open circuit voltage of test leads. Should be several volts.

author: UK
author_rank: Regular Contributor
timestamp: October 24, 2023, 07:26:13 am
content: Of course, I've measured it with closed clamps. Since the seller said in capital letters that it is designed to measure only passive components and can be easily damaged by any powered circuit... since I haven't it principal circuit, so that's why I do not want to throw it into the bin just out of curiosity.Hope you'll be happy with the taken photos. Chinese manufacturing conspiracy makes me sad with all these unmarked ICs.Also, have to say its plastic shell is not the worst I've seen, but it definitely can be unscrewed only two or three times at max.

author: arturmariojr
author_rank: Contributor
timestamp: October 24, 2023, 08:36:02 am
content: I wrote a manual in portuguese for it.I hope it is better than the chinese one!

author: enut11
author_rank: Super Contributor
timestamp: October 24, 2023, 09:38:03 am
content: Quote from: UK on October 24, 2023, 07:26:13 amThanks@UK. Nice pix showing lots of detail. Single lion cell may limit the compliance voltage unless there is an up-scaler. In any case, measuring the open circuit test lead voltage with a DMM will not damage the device.enut11

author: enut11
author_rank: Super Contributor
timestamp: October 24, 2023, 09:43:22 am
content: Quote from: arturmariojr on October 24, 2023, 08:36:02 amThanks@arturmariojr. Yes indeed, it is easier to understand than Chinese.I have downloaded the file and now looking for an MSword translator from Portuguese to English.The front panel button labels can easily be overlaid in English.enut11

author: UK
author_rank: Regular Contributor
timestamp: October 24, 2023, 10:14:11 am
content: Quote from: enut11 on October 24, 2023, 09:43:22 amMaybe I'll help you a bit since I already have it but only in the picture format.

author: arturmariojr
author_rank: Contributor
timestamp: October 24, 2023, 11:50:31 am
content: Hey, thanks a lot!But has anyone ever studied a way to translate internal software? With paper manual or not, is very difficult to work with it with software in chinese...A great equipment, but I think twice before using it!

author: UK
author_rank: Regular Contributor
timestamp: October 24, 2023, 04:16:59 pm
content: There no translation is needed practically.In 95% of cases, you will use only the 1st menu item (resistance measure), and possibly 5% of the 2nd (wire measure) where both screens are without any Chinese characters. 4th (battery saving) and 9th (contrast) you will set only once after buying, and literally 9th is some kind of useless setting because the one acceptable value for contrast is 5 (from the 0-9 range). Then menu items 6th, 7th, and 8th are not even active... assume they are for dev purposes. And finally, the 10th just contains a ton of calibration values.

author: DaneLaw
author_rank: Frequent Contributor
timestamp: February 05, 2024, 11:43:50 pm
content: Quote from: UK on October 24, 2023, 10:14:11 amThx - will definitely be a helping hand, as it's still Chinese language variants being sold, as far as I can tell.. doesn't look to be any newer revision of the YR2050 milliohm meter.?What is the general feedback on these Yaorea YR2050 milliOhm units ??at around 32 euro incl. both delivery & 25%EUVAT (early 2024 prices) it's not like milliohm meters are hanging on the trees for a self-contained display unit in that lower pricebracket..

author: Kean
author_rank: Unknown
timestamp: February 06, 2024, 12:06:21 am
content: Quote from: DaneLaw on February 05, 2024, 11:43:50 pmI posted some tests of my YR2050 with some 0.1% and 0.01% resistors.Seehttps://www.eevblog.com/forum/testgear/absolute-best-multimeter-test-leads-very-low-resistance/msg5269221/#msg5269221*****Also, unlike the YR1030 which is available as YR1035 with a better connector for the test leads, the YR2050 only seems to be available with the test leads using a USB connector.

author: DaneLaw
author_rank: Frequent Contributor
timestamp: February 07, 2024, 07:04:12 pm
content: Quote from: Kean on February 06, 2024, 12:06:21 amYes, I have seen your post Kean, thx.My impression so far is that the YR2050 is a decent milliohm meter, as long as you have its price in mind & the lack of contending alternatives in this lower-end price bracket for a stand-alone unit, unless you wanna build one yourself..So far I have only seen the model with the USB to Kelvin clamp interface for the YR2050 series, and not the screw-terminal even though the molding from UK's pictures is clearly optimized for one in the bottom, but seems its a housing that are used on numerous models..

author: DaneLaw
author_rank: Frequent Contributor
timestamp: March 08, 2024, 08:33:59 pm
content: My YR2050 arrived a few days ago, and my first impressions are positive.I purchased my unit directly from what seems to be their official sales channel.Arrived in a double box with Yaorea branded packing tape.The build quality is better than what I had expected for a price under 35 US (incl. both free delivery and 25% EUVAT) the housing seems rigid and the 4 leads are silicone wires with gold-plated kelvin clips.The biggest con is obviously the Chinese language.My unit arrived with software vers. 1.2209It came with an 8-page English user manual that Yaorea now seems to incl. for western buyers, but it does leave a lot to be desired for when you down into the sub menu's.Instead turn to Google's Translate app.Will post it here the relevant parts of the menu where the Chinese text are in play and things ain't obvious, so I can look it up when in doubt about the settings.* Menu 10 (Res settings) from segment 13 to 20 are not valued with substances (13-20 customizable) .Adjusted my screen backlight setting (menu3) to 60 sec delay so it doesn't turn off so fast & screen brightness to 55%.Your null your value with "HOLD" button.. a short press equals HOLD and a long press equals nulling (it will show a U)

author: enut11
author_rank: Super Contributor
timestamp: March 09, 2024, 03:38:54 am
content: Thank you@DaneLaw. You have done many YR2050 owners a great serviceenut11

author: ass20
author_rank: Regular Contributor
timestamp: March 10, 2024, 01:39:46 pm
content: i analize photo of yr2050U14 U13 AD7705/7706 or TM7705/06 TM7707/08 it is need Quartz 2.4576 mHz Y2D15 tl431U1 STM8L151C6+U4 LM321U17 uln2003J5 swim connector for flash U1U19 U22 battery protection

author: DaneLaw
author_rank: Frequent Contributor
timestamp: March 10, 2024, 05:04:09 pm
content: Quote from: UK on October 24, 2023, 04:16:59 pmYou save your contrast settings with "HOLD"It works as intended on my unit (fw v1.2209) I got the contrast on setting 0.When entering menu 9.. you cycle the value forward with the button under HOLD and then hit HOLD, to save the value.Menu 5 seems to be lower & upper limit values and you can plug in these, and you active it at the bottom, where you can cycle the Chinese label for (Off/ON) the front measurement page will then display them (LO/UP) at the left, and the meter will give audio feedback-signal for the ohm reading, with a label on the screen (NO/YES) if its inside the tolerances.

author: DaneLaw
author_rank: Frequent Contributor
timestamp: April 01, 2024, 11:46:08 pm
content: Has anybody tried the 1030+ probes with the YR2050.?Been really impressed with the YR2050 I purchased a month or two back (my first & only milliohm meter) been running around and testing resistance left & right with the freedom that a handheld unit gives, but the incl. 4wire Kelvin clips are not ideal for certain scenarios, so decided to also pursue the 4-wire probes in the hope they will work with the YR2050, it looks like it uses pogo-pins,.The kit with a YR1030+ (battery resistance 4-wire tester) was being sold at prices, so it didn't make sense to purchase the 4-wire-probes alone.YR1030+ with incl. 4wire probes, were merely 21US incl.25% EUVAT & delivery (at the latest sale) but the YR1030+ and YR2050 are two quite different products, so I wonder if anybody has actually validated that the incl. 4wire probes with YR1030+ also works with the YR2050.?

author: csuhi17
author_rank: Frequent Contributor
timestamp: April 02, 2024, 09:27:26 am
content: I didn't manage to help you, but because I've taken it apart, I'll post a picture of the two.When I bought these, I made a mistake and accidentally bought two different connectors.The two are from the same manufacturer Yaorea, it would not make much sense for them to add a different type of measuring probe to the two devices.I was thinking about the connections in the connectors.

author: DaneLaw
author_rank: Frequent Contributor
timestamp: April 02, 2024, 07:43:22 pm
content: Yep, it's a shame that the YR2050 is not fitted with the screw-terminal from the other series' or even better 4mm banana jacks.As USB-A just seems so delicate, and asking for problems down the line.But the YR1030+ should arrive in a few days., so will check the pin out when it arrives.Look like there are two different 4-wire probes for these YR1030+/1035+ that I think also would work on YR2050, beside the standard Kelvin clipsThere is the side-by-side, what look to be springloaded pogo-pins (picture above #26) and then there is also these (picture below)- where the probe-pin(s) looks like its, one sliding barrel inside another for the 4 wire expect?.They cost from around 12½ USD and up, depending on sellers (incl. VAT & delivery) can also be had with the screwterminal interface. (YR1035) _https://www.aliexpress.us/item/1005003630321521.html?

author: csuhi17
author_rank: Frequent Contributor
timestamp: April 02, 2024, 08:52:33 pm
content: The enchanted version is similar to the meter unit of the battery holder. "Battery Stand".Inside that meter tip there is another isolated meter tip.It is suitable for four-wire measurements. but since its diameter is smaller, it is questionable how long it will last.The pogo pin ones couldn't even touch each other because they weren't in each other.And if you use this a lot, the inner insulating layer can wear out quickly.The way I look at it, you have to push it all the way in to be able to measure with it.

author: DaneLaw
author_rank: Frequent Contributor
timestamp: April 02, 2024, 09:19:42 pm
content: For the batterystand it makes better sense, as there, I recoon you can lock it in with a firm/tight connection, while as a crude handheld probe I could imagine your reading would be dancing all over, while trying to get a solid connection on both, & holding the inner spring down to align with the outer.Looks like there are different models..of these 1000 series resistance meters..also looks to be a TR1030?? varianthttps://www.aliexpress.us/item/1005001624187092.html?This one seems to be the old OG YR1030 model.

author: Phil1977
author_rank: Frequent Contributor
timestamp: April 03, 2024, 08:19:04 am
content: Do you also know this device:https://github.com/ludwich66/RC3563/wikiIt´s a dedicated ESR meter, using a 1kHz AC signal to measure the equivalent series resistance.The probes that are delivered with that device look very similar to the YR2050 packages. From my own experience this device works quite well as long as you have reliable contact of all 4 contact points. If one of the probe contacts is weak, the meter sometimes shows an unplausible *LOW* ESR value.For everydays work I built a set of probes where the 4W-cables are shortcut shortly before a standard 2W-probe-pin.

author: csuhi17
author_rank: Frequent Contributor
timestamp: April 03, 2024, 08:26:10 am
content: The TR series looks like a different brand to me, maybe "Terrific" and the YR is Yaorea.Navy blue and red are Terrific.And the gray and light blue are Yaorea.I guess "Terrific" is just a rebranded product.At least when comparing its properties, you can find the same in both brands.

author: csuhi17
author_rank: Frequent Contributor
timestamp: April 03, 2024, 09:29:50 am
content: Quote from: Phil1977 on April 03, 2024, 08:19:04 amyes, very similar, only the size is different.The red Aether BA1020 and the black Yaorea YT1035+.I think Aether is the same one you linked.

author: Phil1977
author_rank: Frequent Contributor
timestamp: April 03, 2024, 09:57:14 am
content: Exactly, the RC3563 uses a 4-pin GX16 aviation plug - I think that´s a better choice compared to off-label-USB-plug-usage.Has anyone ever directly compared the RC3563 to the YR2050? Are the measurement basics (1kHz, ranges, etc...) the same?

author: DaneLaw
author_rank: Frequent Contributor
timestamp: April 15, 2024, 11:58:04 pm
content: Quote from: csuhi17 on April 03, 2024, 09:29:50 amThe branded Yaorea uses a press-fitting and seems to rely on some ball joints for the lock, IMO it looks to be the preferred way.The BA1020 and the RC3565 and the likes of it VR1220 etc... seems to use a screw fitting and the same connection I also have on an older cheap Tesla/gauss-meter..Got the YR1030+ today, its a non-Yaorea branded unit.. it claims 1030B on the screen with FW2.2206 when starting upthere are quite a few differences from this unit I received to the Yaorea YR1035+ with the press-balljoint-fittingThe YR1035+ got a decimal more on the voltage and the menu options also vary.You got A B C binning functions that you can program, and it also looks like you can adjust the current in menu9 between 200mA and 400mA on this YR1030+, where on the branded YR1035+ its buzzer and contrast in menu9 while you have RS232 option on menu6 on the Yaorea YR1035+.Went for the YR1030+ for the 4wire probes to be used with YR2050 as the price difference was not a lot between the probes alone and the YR1030+ unit with a set.It works well with YR2050 so same USB pinout.

author: mawyatt
author_rank: Super Contributor
timestamp: January 20, 2025, 12:51:56 am
content: Is there a new UI in English?Best