Micsig TO1104 Firmware Upgrade - Page 1
Forum section: electronics / testgear

author: thogevoll
author_rank: Newbie
timestamp: July 24, 2018, 12:38:25 am
content: Does anyone know if Micsig has or will come out with a firmware update to correct the time base sequence in the Micsig TO1104 scope?My scope has a 4mS setting which should be 5mS to be in keeping with the 1-2-5 sequence of the time base. It doesn't hinder me from using the scope but it is an annoyance.I've written to Micsig about this multiple times and have never gotten an answer.

author: SteveOregon
author_rank: Contributor
timestamp: July 24, 2018, 01:23:45 am
content: I purchased a TO1104 through Amazon about three weeks ago. Time base in that range is 10ms/5ms/2ms/1ms. No 4ms."Version Info" under settings is9.15.1.306

author: tsman
author_rank: Frequent Contributor
timestamp: July 24, 2018, 01:39:38 am
content: Quote from: SteveOregon on July 24, 2018, 01:23:45 amHmm. Interesting. They've made a newer version it seems of the hardware and firmware. My TO1104 from late last year is 8.14.1.287 and shows 4ms. The last number is the firmware version/build so yours is newer. The EU online update site only shows version 273 and the China online update site only shows version 287.

author: Micsig_support
author_rank: Regular Contributor
timestamp: July 24, 2018, 03:38:46 am
content: Hello, We are sorry that our sales did not reply you ASAP. next time if you do not get the answer in time, you can email me here and i will do my best to answer you.For your question, i tried to explain in English.Measurement duration = memory depth/sampling frequency.our tbook mini is 14 grids.so 4ms/div=28M/（500M S/s*14div）our scope uses 1 -2-5 steps, but when they are in special time base, it is not 1-2-5. some scopes uses same way.all firmware versions are same.However, thanks for your remind, I told our IT team to update the newest firmware soon.As I know, they are testing automotive functions right now, but it is not ready to be released. so they do not update the firmware in our server system. if someone want to test free automotive(PS, it is not perfect right now), you can contact me here to get the free license. but i do not promise it works to all customers.

author: maginnovision
author_rank: Super Contributor
timestamp: July 24, 2018, 04:20:22 am
content: Quote from: Micsig_support on July 24, 2018, 03:38:46 amI would be interested in the automotive functions but what is included with that? Protocols, automated measurements, diagnostic functions?

author: SteveOregon
author_rank: Contributor
timestamp: July 24, 2018, 04:40:07 am
content: Strange, but when I went to look at my TO1102 again, the 5ms range had disappeared and I was only getting the 4ms range. After checking all the settings, discovered that the "sample" mode is what causes the difference. If I chose "average", I get the 5ms range, if I chose "normal" I get the 4ms range. Had not made that connection before. I think that's what Micsic was trying to explain:Quote

author: exe
author_rank: Unknown
timestamp: July 24, 2018, 07:39:38 am
content: Quote from: Micsig_support on July 24, 2018, 03:38:46 amThere is a new firmware for TO1104?? This is awesome!

author: StillTrying
author_rank: Super Contributor
timestamp: July 24, 2018, 12:59:47 pm
content: Quote from: SteveOregon on July 24, 2018, 04:40:07 amYes strange, I still judge waveform changes against the grid lines so I think I'd count the ms/Div changing betweensampleandaverageas a bug!

author: exe
author_rank: Unknown
timestamp: July 24, 2018, 02:37:59 pm
content: Quote from: StillTrying on July 24, 2018, 12:59:47 pmSame here. I used to use average quite a lot and didn't notice any timebase changes. So, I guess, it's a display/rendering issue (don't have my to1104 at hand to check this).

author: I4E
author_rank: Regular Contributor
timestamp: October 17, 2018, 10:14:30 pm
content: We are Micsig Distributors in the UK and Ireland.Are you still have in the issues you've mentioned about and is there anything I can help with?SincerelyJoy TorresInstruments 4 Engineers LtdTel + 44 (0) 161 871 7450www.instruments4engineers.com

author: Dave Boechler
author_rank: Newbie
timestamp: September 24, 2019, 11:59:25 pm
content: Where did you fine a version 9.15.1.306? I am still on 7.13.

author: orion242
author_rank: Unknown
timestamp: September 25, 2019, 12:59:07 am
content: Thought maybe there was something new for it so I grabbed mine to peek what it was running.Under InformationVersion Info 10.15.1.349Firmware Date 1-10-2019 16:41When I first got mine I just connected to my wifi and ran update - network and it just fetched this and updated itself. Check of the network update today, reports its running current.

author: ForceFed
author_rank: Newbie
timestamp: September 25, 2019, 04:02:14 am
content: Checked mine today too, no updates...Version Info 10.15.1.349Firmware Date 2018-09-11 16:12

author: exe
author_rank: Unknown
timestamp: September 25, 2019, 03:49:11 pm
content: I think only the last number is the firmware version. Other numbers are hardware revision or something.

author: voltsandjolts
author_rank: Unknown
timestamp: September 25, 2019, 06:40:05 pm
content: Checked mine today too, no updates...Version Info 10.15.1.349Firmware Date 2018-12-19 13:53 <-- Maybe this is the date when the firmware last flashed. Dunno.

author: orion242
author_rank: Unknown
timestamp: September 26, 2019, 12:14:15 am
content: Quote from: voltsandjolts on September 25, 2019, 06:40:05 pmBazaar.Three of us all with the same version and all different dates? Looked up my order and the order was placed may '19, so unless the clock was off... I know it ran an update when I got it. Not sure how it would have the current date/time unless it run NTP or something. You don't seem to be able to enter it anywhere. Even then...yesterday was the second time I ever connected to the my network.

author: dekagon
author_rank: Contributor
timestamp: November 17, 2019, 04:57:14 pm
content: Hello all,my TO1102 scope had firmware 9.15.1.349 from 2018-08-21 16:09 installed.If I check for updates I get a message that the firmware is latest. Can that be confirmed?No newer ones as for the TO1104 model (10.15.1.349)?ThanksChris

author: myval
author_rank: Contributor
timestamp: March 15, 2020, 10:16:04 am
content: Hi,Just wondering does anyone know how to get the latest firmware? Micsig Download page is useless there is nothing new when I try to update through the scope it always say it is the latest (clearly not).my version is 7.14.1.349Thanks

author: beanflying
author_rank: Super Contributor
timestamp: March 15, 2020, 10:27:15 am
content: Just hooked mine up via WiFi to double check and as with dekagon above I have the same firmware 9.14.1.349 so are you sure about the 7 in front of yours?Reply back from Migsig was "Current version is latest!" btw. on a TO1104 with most of the options.

author: exe
author_rank: Unknown
timestamp: March 15, 2020, 01:22:23 pm
content: Afaik first number or two are hardware revision. Mine is, afaik, 7.xx I thought that was the latest. I wonder what they changed.

author: beanflying
author_rank: Super Contributor
timestamp: March 16, 2020, 01:25:33 am
content: Interesting so the latest firmware should read X.14.1.139 then.Just tried looking at the Micsig website and page 1 of the firmware downloads is Brokenhttp://www.micsig.com/html/list_60_3.htmlso the only way to get a current firmware update is via LAN or WiFi.

author: exe
author_rank: Unknown
timestamp: March 16, 2020, 12:02:15 pm
content: Yes, their website is horribly broken since a long time ago.I also found they have a Chinese forum on their website. I used google translate to read it. Not sure how I found it, may be by registering onhttp://micsig.com/User/Login/.So, it seems the community is fragmented.PS I think there is a way to download firmware update. I think the scope checks this link for downloading new updates:http://www.micsig.com.cn/upgrade/config.xml. There you can see firmware URLs too. See discussions here:https://www.eevblog.com/forum/testgear/micsig-tablet-oscilloscope-tbook-mini-to1000/50/

author: myval
author_rank: Contributor
timestamp: March 17, 2020, 07:37:00 am
content: if the first number is HW version it could be latest I just got email from micsig that it is the latest.well I will have to trust them.

author: langlv
author_rank: Contributor
timestamp: March 22, 2020, 01:09:41 am
content: In my opinion, the XML file may be not up to date. I recently received my STO1104 with firmware 1.0.102. I wanted to try the WIFI update process, so I click the 'Un-install update' of Oscilloscope App. Now it reverts to 1.0.99. In the XML, the latest version for STO1104 is 1.0.56.I don't know how to get back the 1.0.102.

author: Roperm01
author_rank: Newbie
timestamp: October 07, 2024, 06:54:15 pm
content: Is it possible to receive automotive functions upgrade for the to1104

author: csuhi17
author_rank: Frequent Contributor
timestamp: October 10, 2024, 07:30:29 am
content: Quote from: Roperm01 on October 07, 2024, 06:54:15 pmI asked them about it before, but I didn't get an answer from them. But they answered my other questions in that email.