Hantek DSO 1000S series - Page 1
Forum section: electronics / testgear

author: Paulv
author_rank: Newbie
timestamp: September 11, 2013, 01:38:06 pm
content: Hi,I want to buy a handheld scopemeter with isolated inputs. I doubt between two options:a Fluke scopemeter like Fluke199C and a Hantek DSO1062Sor DSO1122S. The Hantek scope has a 4(!) times larger display and is much cheaper. However, the Hantek has some missing features:- Is it right that the Hantek scope does NOT have the option to SAVE a FFT spectrum? I see in the manual that it can save channel 1 or 2, but not the MATH channel. Who can verify this?- The Fluke displays the frequency and amplitude of the peak in a FFT automatically. With the Hantek you must select the peak value with the cursors, which is much more cumbersome;- The Hantek scope has a smallest sensitivity of 5V/DIV. This means that when measuring the mains (640Vpp), a 1:100 probe has to be used. The Fluke has a max of 100V/DIV, so no special probe is needed;Does anyone own a DSO 1000Sseries and can confirm these findings?And is the bandwidth hack to 200MHz also possible for new 1000Shandheld hanteks?Thanks,Paul

author: tinhead
author_rank: Super Contributor
timestamp: September 11, 2013, 07:01:10 pm
content: Quote from: Paulv on September 11, 2013, 01:38:06 pmi don't have "S" handscope, but i know someting about the benchtop/other handheld models, which are as youmaybe already saw herehttps://www.eevblog.com/forum/testgear/hantek-tekway-dso-hack-get-200mhz-bw-for-free/very similar to "S" Handhelds, so i will try to answer at least some questions.Quote from: Paulv on September 11, 2013, 01:38:06 pmyes it isQuote from: Paulv on September 11, 2013, 01:38:06 pmdepends on the environment in which you suppose to use them, Fluke might be better idea. Not that Hantek is cheap plastic or so, but i trust Fluke much more (in case of what so ever).Quote from: Paulv on September 11, 2013, 01:38:06 pmthat's right, but Fluke have up to 2.5GSa/s per channel where Hantek is max. 1GSa/s in single and 500MSa/s in dual channel.Fluke have as well nice recorder (up to 8 days timespan), which i actually never ever tested on Hantek (i know it can be set to e.g. 1000 frames with frame each every 30s or so, but thats far from "8 days" on Fluke, not to mention the nice chart.)Quote from: Paulv on September 11, 2013, 01:38:06 pmnot sure if S model firmware is already on benchtop model stage (in principle it should as they using same platform),but on benchtop models itself you can save CH1/CH2 data as- Screenshot- REF- CSV- Recorder data (all waveframes, via recorder)- single Waveframe (via wave record)where the FFT can be only saved as REF.Quote from: Paulv on September 11, 2013, 01:38:06 pmon handhelds, with no rotary knobs, that sucks indeed. It takes "hours" to push all the buttons to right/left, and then switch from horizontal to vertical to read value, crazy. On benchtop you can use tracking cursor in FFT, it is not the best solutionbecasue you have still to scroll left/right with buttons, but at least it does detect peaks and shows the values. However, the last time i played with handheld there was no tracking cursor.Quote from: Paulv on September 11, 2013, 01:38:06 pmwith such needs i would not buy Hantek ^^ I don't trust myself, one day 1:10, then 1:100,ehm no, better directly handheld which is more robust. On the other side, when you can work with all the disadvantages, the price is for sure a big plus.I'm not the biggest fan of Siglent, but they have isolated handheld as well, you should check the specs as well.

author: Paulv
author_rank: Newbie
timestamp: September 11, 2013, 07:44:02 pm
content: Thanks Tinhead for your comment on the Hantek/Fluke.QuoteI am not sure what the difference is between REF and Screenshot and whether an unlimed number of REFs can be saved, but reading your comment, I prefer the Fluke. As you say, it is more robust and give more confidence.

author: Hydrawerk
author_rank: Super Contributor
timestamp: September 11, 2013, 07:45:24 pm
content: For 230V or 400V measurement, i would prefer Fluke. But it is expensive...

author: tinhead
author_rank: Super Contributor
timestamp: September 11, 2013, 09:54:04 pm
content: Quote from: Paulv on September 11, 2013, 07:44:02 pmi should have said "in FFT can resut can saved as screenshot and REF". The REF itself is nothing else than the information from the screen, and that can be loaded later to compare to other values, e.g. attached picture:- A (on bottom) with some two peaks at 125 and 250MHz- B (on top) with with no peaksThese DSOs/MSOs/Handhelds can currently save only 2 REF values, both located on internal NAND memory.Attached as well screenshot of how the tracking cursor look like on Hantek.

author: echen1024
author_rank: Super Contributor
timestamp: September 11, 2013, 11:46:55 pm
content: You might want to look at a used Tek THS720/730. I hear those are pretty good. You can get them for around $300/

author: Paulv
author_rank: Newbie
timestamp: September 12, 2013, 08:00:17 am
content: To bad, the Tek THS720/730 does not have a proper FFT nor the possibility to save waveforms or screen (only to RS232).If the Hantek has a possibility to SAVE screenshots, then this would make it possible to SAVE (an unlimited number of) FFT's as well? Or is the screenshot limited to CH1/CH2?

author: tinhead
author_rank: Super Contributor
timestamp: September 12, 2013, 08:46:04 am
content: Quote from: Paulv on September 12, 2013, 08:00:17 amscreenshot is simply picture of everything you see on screen, there is no restriction (maybe except file system max. amount of files, etc.) known.