Fluke 867B Backlight Problems - Page 1
Forum section: electronics / repair

author: kdl1950
author_rank: Contributor
timestamp: April 06, 2024, 09:48:05 pm
content: Hello everyone, I have a problem with my Fluke 876B.The display, or rather the backlight, flickers very quickly. In some cases the backlight has failed completely. After switching it off for a few minutes, it will work again.Now my questions are:-Is the problem known?-does anyone know any advice?-is it the electronics of the backlight or the lighting itself?At first I thought it was power supply related. Unfortunately it doesn't matter whether it's with a battery or a power supply.Greetings from bavaria/GermanyAlex

author: mahi
author_rank: Regular Contributor
timestamp: April 08, 2024, 09:00:52 am
content: Fluke 863/867B Service Manual(4.5 MB)The internal power supply of the 860 series multimeters is a DC-DC converter with multiple outputs and post regulators to provide the different voltages required by the various system components. The backlight is fed from the unregulated output of the DC-DC converter, controlled by a current sink circuit. I would expect that if there is a problem with the power supply of the backlight, this would also impact other power supply outputs. Since you do not mention any other issues with the meter, I'm tempted to assume the internal power supply is fine.My guess is the problem is with the LED backlight on the Hitachi LMG7135PNFL LCD display. Bad contact or solder joint, maybe a failing LED, getting worse when heating up. I haven't inspected the LCD up closely, so I don't know whether the backlight is accessible/serviceable.Hitachi stopped manufacturing these displays, but it seems there is still plenty of stock (at the moment of writing). For example:https://www.aliexpress.com/item/32847956772.html(no affiliation - I just picked a random result from Aliexpress).When in doubt, you can try to power the backlight without the multimeter connected. Sadly I could not find a datasheet for the LCD, but you can measure the voltage drop over the backlight (IBL+ and IBL- in the schematics) and according to the service manual the current through the backlight should be approximately 80 mA. If it flickers with an external power supply as well, you can be sure the problem is with the LCD module.

author: kdl1950
author_rank: Contributor
timestamp: April 12, 2024, 02:43:08 pm
content: Hello mahi,Thanks for the feedback and information.I have a replacement display. I'll install it one day and try it.I'll report back then.Greetings

author: kdl1950
author_rank: Contributor
timestamp: April 21, 2024, 04:41:55 pm
content: Today I installed the replacement display.At the moment the error seems to have disappeared.Only time will tell whether it was really the display.I will keep you up-to-date.Greetings from Germany

author: Laval
author_rank: Regular Contributor
timestamp: April 21, 2024, 06:24:07 pm
content: Have you considered it might just be a dodgy solder joint ? I guess it's always great to have a brand new display

author: kdl1950
author_rank: Contributor
timestamp: April 21, 2024, 07:08:23 pm
content: Hello laval,thanks for the tip. I will follow the advice.I can't break much anymore anyway. The display is no longer available anyway.Only used ones.I will report.Greetings from Bavaria