FS/For Sale in Germany/EU: new price Panasonic VP-7723A Audio Analyzer with GPIB - Page 1
Forum section: electronics / general

author: THDplusN_bad
author_rank: Regular Contributor
timestamp: February 07, 2020, 11:50:30 am
content: Good Day,for sale is a Panasonic VP-7723A Audio Analyzer, up to 110 kHz.The unit is in a good state, with a few stains from other equipment's rubber feet on the top cover and a few cosmetic issues/scratches on the front. Nothing is bent or broken.Specified at better than -80 dB THD+N up to 110 kHz, and -100 dB THD+N 20 Hz to 15 kHz with 80 kHz bandwidth.The unit is beating its own specifications by far, which I have verified with an Audio Precision AP555 Audio Analyzer. Very good linearity and a typ. performance of the oscillator of >110 dB THD+N.I am including some performance figures as measured below. I am happy to provide screenshots taken from the AP555 measurements and pictures of the unit per email upon request.This unit has a GPIB-interface, the wow and flutter option and option 20. When used with a GPIB-adapter and some programming, one can turn this into a "poor-man's Audio Precision". I am happy to provide some screenshots taken from my LabView program to illustrate this. If sold to a fellow enthusiast here, I will provide the LabView code for free.Specifications are available here, see last page:http://www.testequipmenthq.com/datasheets/PANASONIC-VP-7723A-Datasheet.pdfupdated price -I am asking750€650€ or best offer and would prefer local pick-up in Germany, Munich, Stuttgart or Nürnberg. I can ship the item within Europe, but would need to source a card box etc. for proper packaging.Thanks,THD+N_badPanasonic VP-7723AVP-7723A measuring its own distortion, oscillator output directly connected to analyzer input, Auto-tune notch filter enabled10 Vrms: THD+N: -103.7 dB (unweighted, no filtering or full measurement bandwidth), measured amplitude: 10.28 Vrms10 Vrms: THD+N: -107.3 dB (unweighted, 80 kHz measurement bandwidth filter enabled)10 Vrms: THD+N: -108.3 dB (unweighted, 20 kHz measurement bandwidth filter enabled)10 Vrms: THD+N: -108.3 dB (unweighted, 15 kHz measurement bandwidth filter enabled)10 Vrms: THD+N: -108.2 dB (unweighted, 100 Hz and 20 kHz measurement bandwidth filter enabled)10 Vrms: THD+N: -108.5 dB (unweighted, 200 Hz and 20 kHz measurement bandwidth filter enabled)2 Vrms: THD+N: -100.5 dB (unweighted, no filtering or full measurement bandwidth), measured amplitude: 2.033 Vrms2 Vrms: THD+N: -104.8 dB (unweighted, 80 kHz measurement bandwidth filter enabled)2 Vrms: THD+N: -105.7 dB (unweighted, 20 kHz measurement bandwidth filter enabled)2 Vrms: THD+N: -105.8 dB (unweighted, 15 kHz measurement bandwidth filter enabled)2 Vrms: THD+N: -105.7 dB (unweighted, 100 Hz and 20 kHz measurement bandwidth filter enabled)2 Vrms: THD+N: -105.9 dB (unweighted, 200 Hz and 20 kHz measurement bandwidth filter enabled)1 Vrms: THD+N: -98.8 dB (unweighted, no filtering or full measurement bandwidth), measured amplitude: 1.017 Vrms1 Vrms: THD+N: -107.3 dB (unweighted, 80 kHz measurement bandwidth filter enabled)1 Vrms: THD+N: -108.8 dB (unweighted, 20 kHz measurement bandwidth filter enabled)1 Vrms: THD+N: -109.1 dB (unweighted, 15 kHz measurement bandwidth filter enabled)1 Vrms: THD+N: -109 dB (unweighted, 100 Hz and 20 kHz measurement bandwidth filter enabled)1 Vrms: THD+N: -109.1 dB (unweighted, 200 Hz and 20 kHz measurement bandwidth filter enabled)200 mVrms: THD+N: -82.4 dB (unweighted, no filtering or full measurement bandwidth), measured amplitude: 204.2 mVrms200 mVrms: THD+N: -99.5 dB (unweighted, 80 kHz measurement bandwidth filter enabled)200 mVrms: THD+N: -103.6 dB (unweighted, 20 kHz measurement bandwidth filter enabled)200 mVrms: THD+N: -104.2 dB (unweighted, 15 kHz measurement bandwidth filter enabled)200 mVrms: THD+N: -103.8 dB (unweighted, 100 Hz and 20 kHz measurement bandwidth filter enabled)200 mVrms: THD+N: -104.2 dB (unweighted, 200 Hz and 20 kHz measurement bandwidth filter enabled)Performance testing with Audio Precision AP555 and APx 600 V4.6.0.255.130221 Software:1) Panasonic VP-7723A oscillator output tests:With 10 Hz to 22.4 Hz measurement bandwidth, using the AP555 high-performance sine analyzer:Measured 2.035 VrmsAt 0 dBV output settings into 600 Ohm, or 2 Vrms into high-impedance load; Measured THD+N -115.9 dB(unw.)At +14 dBV resulting in 10.25 Vrms: Measured THD+N -116.4 dB(unw.)With AES-17 filter used for 20 kHz measurement bandwidth, using the AP555 high-performance sine analyzer:At 0 dBV output settings into 600 Ohm, or 2 Vrms into high-impedance load: Measured THD+N -116.1 dB(unw.)At +14 dBV setting resulting in 10.25 Vrms: Measured THD+N -116.4 dB(unw.)High-bandwidth measurements:10 Hz to 90 kHz measurement bandwidth, using the AP555 high-performance sine analyzer:At 0 dBV output settings into 600 Ohm, or 2 Vrms into high-impedance load: Measured THD+N -113.8 dB(unw.)At +14 dBV setting resulting in 10.25 Vrms: Measured THD+N -114.9 dB(unw.)**********************2) Panasonic VP-7723A analyzer and input tests:When using theAPx555's high-performance sine generatorwith 2 Vrms output, 1 kHz sine wave, the VP-7723A measures as follows:-105.9 THD+Nand 1.996 Vrms, displays +6 dB (with 200 Hz HP-filter and 20 kHz LP-filter enabled)-105.9 THD+Nand 1.996 Vrms, displays +6 dB (with 15 kHz LP-filter enabled)-105.7 THD+Nand 1.996 Vrms, displays +6 dB (with 20 kHz LP-filter enabled)-104.8 THD+Nand 1.997 Vrms, displays +6 dB (with 80 kHz LP-filter enabled)-100 dB THD+Nand 1.997 Vrms, displays +6 dB (unlimited bandwidth, no filter)High-level measurements, all with a 1 kHz sine waveAPx 555 output:2 Vrms, VP-7723A measures: 1.996 mVrmsAPx 555 output:5 Vrms, VP-7723A measures: 5.01 VrmsAPx 555 output:10 Vrms, VP-7723A measures: 10.02 VrmsAPx 555 output:12 Vrms, VP-7723A measures: 11.86 VrmsLow-level measurements, all with a 1 kHz sine wave:APx 555 output:1 Vrms, VP-7723A measures: 1.000 mVrmsAPx 555 output:500 mVrms, VP-7723A measures: 0.500 VrmsAPx 555 output:200 mVrms, VP-7723A measures: 199.7 mVrmsAPx 555 output:100 mVrms, VP-7723A measures: 99.9 mVrmsAPx 555 output:50 mVrms, VP-7723A measures: 50.0 mVrmsAPx 555 output:20 mVrms, VP-7723A measures: 20.1 mVrmsAPx 555 output:10 mVrms, VP-7723A measures: 10.1 mVrmsAPx 555 output:5 mVrms, VP-7723A measures: 5.1 mVrmsAPx 555 output:2 mVrms, VP-7723A measures: 2.1 mVrmsAPx 555 output:1 mVrms, VP-7723A measures: 1.1 mVrmsAPx 555 output:500 µVrms, VP-7723A measures: 0.6 mVrmsAPx 555 output:200 µVrms, VP-7723A measures: 0.3 mVrmsAPx 555 output:100 µVrms, VP-7723A measures: 0.2 mVrmsAC level over frequency, 1 Vrms sine waveInput 5 Hz, VP-7723A measures: 0.970 Vrms (Slow response settling enabled)Input 10 Hz: VP-7723A measures: 0.986 Vrms (Slow response settling enabled)Input 20 Hz: VP-7723A measures: 0.993 Vrms (Slow response settling enabled)Input 50 Hz: VP-7723A measures: 0.995 VrmsInput 100 Hz: VP-7723A measures: 0.993 VrmsInput 200 Hz: VP-7723A measures: 0.994 VrmsInput 500 Hz: VP-7723A measures: 0.998 VrmsInput 1 kHz: VP-7723A measures: 1.000 VrmsInput 2 kHz: VP-7723A measures: 0.998 VrmsInput 5 kHz: VP-7723A measures: 0.998 VrmsInput 10 kHz: VP-7723A measures: 0.999 VrmsInput 20 kHz: VP-7723A measures: 0.999 VrmsInput 50 kHz: VP-7723A measures: 1.001 VrmsInput 100 kHz: VP-7723A measures: 1.012 VrmsInput 110 kHz: VP-7723A measures: 1.014 Vrms (specified upper limit for VP-7723A)Input 200 kHz: VP-7723A measures: 1.058 Vrms

author: Zucca
author_rank: Unknown
timestamp: February 07, 2020, 11:55:16 am
content: Service Manualhttps://elektrotanya.com/panasonic_vp-7723a_audio_analyzer_sm.pdf/download.html

author: THDplusN_bad
author_rank: Regular Contributor
timestamp: February 14, 2020, 05:18:42 pm
content: Good Day,Updated price to 650€ - c'mon, chaps, I need the bench space and so let's do this!There is a similar unit for sale from a dealer on ebay for 1,240€... seeeBay auction: #293134971230- just to put this into perspective.Cheers,THD+N-bad

author: Noy
author_rank: Frequent Contributor
timestamp: February 20, 2020, 03:43:55 pm
content: Hi,send a PM.Please check.Best regardsFelix

author: THDplusN_bad
author_rank: Regular Contributor
timestamp: April 29, 2020, 11:36:36 am
content: This item is reserved for now.Thanks,THDplusN bad