REPRO PS-305D power supply - bringing it back to life - I have questions - Page 1
Forum section: electronics / repair

author: bpjl
author_rank: Contributor
timestamp: May 24, 2024, 07:06:44 pm
content: Hi,My name is Przemek, I come from Silesia, and this is my first post on this forum. Of course, I come straight away with problems and questionsLet's get to the point:Hi,I plan to buy a new workshop power supply, but I also decided to revive the old REPRO PS-305D. After replacing the damaged transistors, the equipment came back to life. I'm also planning to replace the radiator to relieve them.I also replaced the old potentiometers (all four) because there were problems with setting low current and voltage values. When I tried incrementing the values ​​from 0 - first, nothing happened, then some value appeared, but it fluctuated and only within a further range could stable settings be obtained. After replacement, the problem went away.I also struggled with calibration -https://www.eevblog.com/forum/reviews/yh-305d-power-supply-calibration-procedure/.After all these treatments, everything seems to be OK, but a few things bother me:1. I think, but I could be wrong, that in the past it was possible to set the desired voltage by having the current potentiometers turned to 0. Currently, in order for any voltage to appear, I have to turn the current to at least the smallest minimum. Is it supposed to be like that or is it a glitch?2. By calibrating the power supply readings with potentiometers on the back of the display PCB, I set the mA and A readings without any problems. However, with voltage it's a different story. The differences between what the multimeter shows and the power supply's readings are as follows:When the readings converge at low voltages, the higher voltages start to diverge significantly. And similarly - if you synchronize the top, the bottom is terrible. As you can see above, I have set it as such accuracy up to 2V. I know that I cannot expect hyper-accuracy and "being razor sharp" from such a power supply, but maybe there is a way to improve the values readings ​​shown above?I don't mean that the power supply should reach the declared 30V (although it would be great). I care more about showing the voltage more accurately throughout its entire range. Below are the photos of the display PCB:Fun fact: Yes! I know that all these power supplies are one and the same inside. The only difference is the print on the front of the case.But to that extent? Outside it's REPRO and the display PCB is YIHUA?3. What is the "mA and A offset" potentiometer used for, because I don't really understand it?I am attaching a diagram for the Yihua YH-305D power supply. Maybe it will be helpful.

author: DanielDlouhy
author_rank: Contributor
timestamp: May 27, 2024, 05:11:08 pm
content: Hello,1. That's fine. There needs to be some small current to set the output voltage (to charge output caps and there's also propably a resistor on the output). Maybe the old pot didn't go all the way to 0 with resistance when it was set to 0, so it always set a very small current...2.That might be because of that cheap panel voltmeter... But I don't have the schematic.3.That is propably the "error" between mA and A range. For example: If you set the output current to 900mA in a mA range, then is should show 0,9A in a A range. But I suppose that ranges are selected automatically, so it is probably so it doesn't jump from 0,999mA to 1,1 A for example... But again, without a schematic, it's hard to guess.If you happen to find a schematic for your unit with the pots labeled, I might be able to help you more...-Daniel

author: bpjl
author_rank: Contributor
timestamp: May 27, 2024, 06:44:34 pm
content: Quote from: DanielDlouhy on May 27, 2024, 05:11:08 pmYou're probably right, and that could have been the case.Quote from: DanielDlouhy on May 27, 2024, 05:11:08 pmAs I have already mentioned, there is a YIHUA print on the PCB of the measuring panel. I have a diagram for the Yihua YH-305D power supply. I think they are the same panels. I am attaching the file to this entry.I also received this answer and advice on another forum. It may be helpful:Quote from: Vogelek23 @ [url=http://www.elvikom.plQuote from: DanielDlouhy on May 27, 2024, 05:11:08 pmThanks for the clarification. I will check it when I assemble the power supply. Currently, I am in the process of replacing the heat sink for the power transistors with a larger one.Yihua YH-305D.pdf(203.53 kB - downloaded 182 times.)

author: DanielDlouhy
author_rank: Contributor
timestamp: May 27, 2024, 07:15:35 pm
content: Looking at the schematic, you can't do much about the voltmeter. As "Vogelek23" from "elvikom.pl" said, the voltmeter is 100 % software. You can try to averige it (as he also said), by replacing the R17 and V11. But that's pretty much all you can do.Daniel

author: bpjl
author_rank: Contributor
timestamp: May 27, 2024, 08:39:36 pm
content: As soon as I assemble the power supply, I will try this solution. Of course, I will write what came out of it.For now, thank you for your help and time.

author: 807
author_rank: Frequent Contributor
timestamp: May 28, 2024, 01:42:54 pm
content: There are 3 components on that schematic marked V11. One on U6 pin 2, one on U6 pin 6 & one between R48 & Voltmeter+ Out+ (bottom left page 2).

author: bpjl
author_rank: Contributor
timestamp: May 28, 2024, 05:06:49 pm
content: Yes, I know that. And these are exactly:Quote from: Kaptein QK on April 24, 2013, 11:31:34 pmThe problem is that I am unable to set the voltage readings so that they are consistent with reality in their entire range. I described it in the first entry.

author: bpjl
author_rank: Contributor
timestamp: June 07, 2024, 02:17:57 pm
content: I finally found time to get down to work. I replaced V11 and R17 with the indicated values, set 15V on the multimeter and "calibrated" the built-in voltmeter to this value. The differences between the readings of the multimeter and the display in the power supply were as follows:Enough for me. I consider this problem solved.