SIGLENT SDS1202X-E troubles - Page 1
Forum section: electronics / repair

author: luistrano
author_rank: Newbie
timestamp: March 09, 2021, 12:57:20 pm
content: Hi All. It's my first post in this blog.I'm searching around looking for a solution for the problem arised with my 1202X-E: the instrument is quite new (end 2019), and it was running with no problems, until last week....Last week I've loaded on the machine a wrong licence key (in fact not dedicated to the 1000X series), and after the reboot the instrument had lost all the connection features, so is not more seen and reacheable from a PC via USB or LAN, the bandwidth is downgraded from 200 to 50MHz, and from System Info I can notice that the string of the Product ID is blank (only that string): the instruments don't know to be a 1202X-E anymore...All the functions and features seems to run correctly, boots normally, but no way to return to a factory reset and recover from that condition of totally deaf and heavy sight-impaired: over the limit of 50MHz the signal levels displayed decreases too much, as it did before but with signals at 400MHz...I've tried all the known remedies, from PowerON+MATH, last FW reloaded (the USB pen are seen normally), etc... No way to have the system back.The LAN interface is certainly alive and pingable at his last-known IP but the OS refuses an SSH connection.I've read in some post that someone was able to connect to the instrument via telnet, but at the moment it seems to me not a way: how? and the credentials are unknown to me.Having some skills in electronics and IT, I suppose can't avoid to open the box, (losing any possible official warranty right), to find some serial port on the PCB useful to reach the system.Anyway if there is a solid way to recover that situation by remote I'll be happier....So I hope that between the numerous skilled user of this Forum I'll find someone who can help me, driving me through a clean procedure to restore the pristine conditions of the scope.Many thanks for your kind attention. I hope you will excuse my poor english...best regards, ciao.Luigi

author: tautech
author_rank: Super Contributor
timestamp: March 09, 2021, 07:40:17 pm
content: Welcome to the forum.License key ?Do you mean you loaded an incorrect firmware version ?Can you give more information please.

author: luistrano
author_rank: Newbie
timestamp: March 10, 2021, 06:48:52 am
content: Hi Tautech.Yes: I've registered a wrong licence key, not uploaded a wrong firmware.Don't ask me why....I kow my sins.Hope ypu know a solution.L.

author: tautech
author_rank: Super Contributor
timestamp: March 10, 2021, 07:19:36 am
content: Quote from: luistrano on March 10, 2021, 06:48:52 amYour mistake is there are no improvements possible for SDS1202X-E. It already is a 200 MHz only platform and there are zero options available for it so it is best not tampered with in any way.I have a request to the factory for a recovery package for you however no reply as yet....maybe tonight.Send me a PM with your email and we will try to get you fixed in the next few days.

author: luistrano
author_rank: Newbie
timestamp: March 10, 2021, 09:17:52 am
content: Yes tautech, no doubt I've done a stupid thing. But I've realized it 10" too much later.....Your words are for me something like this verse from the "Divina Comoedia" (Inferno, canto V): ...come colombe dal disìo chiamate...I'm sure you will translate correctly without problems, but anyway I say to you a big thanks!!Hope you'll receive the package from the factory very soon.In the meantime I'll send you a PM immediately.Thanks again, ciao!!Luigi

author: tonyalbus
author_rank: Super Contributor
timestamp: February 04, 2023, 03:20:29 pm
content: How it was solved in my case:restore firmware to 1.3.17R1- now scope reports SDS1052X-ESCPI is working i can send MCBD with 200 MHz key and reboot- now scope reports SDS1202X-E again and responds fastnow update firmware to 1.3.26do self/ca .. all is okey againiguessin the 1.3.26 the support for lower bandwidth was removed, so sending a wrong bandwidth, 350 MHz in my case... sets the scope to lowest.. 50 MHz..but in the latest firmware it does not excist so it goes all crazy... downgrading it to 1.3.17R1 does have that support.. and give you the change to send the correct key again.this were exiting daysAlso thanks to @Tautech who was there to support me, amazing service sir

author: tautech
author_rank: Super Contributor
timestamp: February 04, 2023, 09:57:54 pm
content: Quote from: tonyalbus on February 04, 2023, 03:20:29 pmYep that's know bug in the keygen for all X-E models.

author: tonyalbus
author_rank: Super Contributor
timestamp: February 13, 2023, 10:25:57 am
content: @luistrano just confirmed this solution worked for him too