Hameg HMC8012 Porn - Page 1
Forum section: electronics / testgear

author: dr.diesel
author_rank: Super Contributor
timestamp: October 31, 2013, 03:13:17 pm
content: Just arrived, quick video of the boot time, about 11 seconds.Busy day today, will try and get a few more up.

author: FrankBuss
author_rank: Unknown
timestamp: October 31, 2013, 03:20:02 pm
content: Nice. I have a HM8012, boot time is 3 seconds. But of course the rest is not as good as your shiny new HMC8012.

author: Jon86
author_rank: Frequent Contributor
timestamp: October 31, 2013, 03:32:08 pm
content: That's almost better than my chinky-chonk-chinese multimeter!Ah, one day I'll get my hands on something like that...

author: marshallh
author_rank: Unknown
timestamp: October 31, 2013, 03:55:23 pm
content: Love the euro-DIN font.I'd take it over a Rigol anytime

author: dr.diesel
author_rank: Super Contributor
timestamp: October 31, 2013, 04:18:15 pm
content: Busy playing instead of working!The Hameg Explorer software works great in Wine!

author: mcinque
author_rank: Unknown
timestamp: October 31, 2013, 04:26:49 pm
content: Really porn!! I love it!

author: psycho0815
author_rank: Regular Contributor
timestamp: October 31, 2013, 04:27:25 pm
content: Quote from: dr.diesel on October 31, 2013, 04:18:15 pmI'd prefer a native solution obviously, but that makes me want it more than i already did.

author: dr.diesel
author_rank: Super Contributor
timestamp: October 31, 2013, 04:57:24 pm
content: Dual Measure Functionality:

author: dr.diesel
author_rank: Super Contributor
timestamp: October 31, 2013, 06:46:39 pm
content: A few picts, I'll get some proper picts up with a macro lense soon.

author: Hydrawerk
author_rank: Super Contributor
timestamp: October 31, 2013, 11:12:37 pm
content: The internal design is clear and nice.

author: dr.diesel
author_rank: Super Contributor
timestamp: October 31, 2013, 11:22:12 pm
content: Quote from: Hydrawerk on October 31, 2013, 11:12:37 pmFirst impressions are quite good. The unit is shallow in depth, fits on the bench well, menus and navigation were well thought out. The display makes good sense, screen updates are quick and clear. You can adjust the brightness, contrast, screen back light and keyboard back light independently.More to come, I should have macro shots up tomorrow, something where you can actually read component numbers!

author: Vgkid
author_rank: Super Contributor
timestamp: November 01, 2013, 12:34:23 am
content: Interesting meter, looks to be set p for an ATE environment.

author: tszaboo
author_rank: Super Contributor
timestamp: November 01, 2013, 11:32:19 am
content: I will never understand these Hameg instruments. Every time I meet one, I always find that the specs are inferior to anything, and they are overpriced. I cannot understand these Germans.Who would buy this for 650 Euros, where for 730 there is the Agilent 34460A? They specify DC accuracy for 150 ppm/year, and 2.5% on the AC current dammit! What is that? Estimator?

author: dr.diesel
author_rank: Super Contributor
timestamp: November 01, 2013, 11:45:40 am
content: Quote from: NANDBlog on November 01, 2013, 11:32:19 amWell, I can't argue about the price, a bit higher than it should be.As for the 34460, it runs a m$ OS, so it has zero value to me.Yeah, I admit it, I have a problem..

author: Kjelt
author_rank: Super Contributor
timestamp: November 01, 2013, 11:56:17 am
content: Real thing of beauty.How much would a real Rohde & Swarz have set you back, I think that problem is very relative

author: Dark Prognosis
author_rank: Regular Contributor
timestamp: November 01, 2013, 11:56:22 am
content: Is that a cold boot or is it a phantom load device?

author: dr.diesel
author_rank: Super Contributor
timestamp: November 01, 2013, 12:01:20 pm
content: Quote from: Dark Prognosis on November 01, 2013, 11:56:22 amThat was the first boot after switching on via the hard power switch in the back. I suppose it could be "soft booting" in the background but I bet not. I'll switch on the hard power switch and immediately hit the soft button, see if there is any difference.

author: Dark Prognosis
author_rank: Regular Contributor
timestamp: November 01, 2013, 12:02:51 pm
content: Quote from: dr.diesel on November 01, 2013, 12:01:20 pmIf it really is cold booting with no activity behind the scenes that is pretty sweet.

author: con-f-use
author_rank: Unknown
timestamp: November 01, 2013, 12:59:38 pm
content: Why did you buy it over a Rigol DM3068 or the agilent? It just doesn't look that good on paper to me to justify the price. What am I missing?

author: dr.diesel
author_rank: Super Contributor
timestamp: November 01, 2013, 01:02:05 pm
content: Quote from: Dark Prognosis on November 01, 2013, 12:02:51 pmYup, from a complete cold boot, ~11 seconds.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 01, 2013, 01:15:22 pm
content: Quote from: con-f-use on November 01, 2013, 12:59:38 pmA couple reasons:- The Rigol is 291mm deep, the Hameg 280mm. Very small difference but a noticeable plus on my bench.- The display on the Hameg is far better to the Rigol.- Rigol support is utter garbage.- IMO Hameg engineering is better, even if the specs are mid-ranged. The specs meet my needs and I want an instrument that works out of the box, without months of firmware upgrades (or hardware revisions like the 832 PS), and will last 15 years.As for the Agilent, I've already explained that one!Besides, my wife was born in West Germany, she likes it when I buy German stuff!

author: con-f-use
author_rank: Unknown
timestamp: November 01, 2013, 01:37:31 pm
content: Okay. Thank you very much, for the insight. I have a friend who works at Rohde&Schwarz so that's basically why I'm interested.I myself was born in Germany and like the general build quality and engineering. Especially Festool, Wiha, Miele, Ersa, Wacker, Hazet and Zweibrüder have never disappointed me. However most reputable German companies could learn a thing or two when it comes to customer service, marketing and pricing.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 01, 2013, 01:46:02 pm
content: IMO Hameg equipment has always been rock solid, if perhaps a bit over priced and maybe a bit slower to innovate, kinda like Tek.If your considering this particular bench meter, feel free to ask any question, I'm happy to setup a video or test or whatever.Over the years each and every time I've emailed Hameg I've gotten a response within a day, often the same day. I'm confident on the rare chance I run into an issue they'd be quick to help, although I did write them asking for the service manual for the HMC8012 and they kindly declined! (I knew this going in, but asked anyhow)

author: RRobot
author_rank: Regular Contributor
timestamp: November 01, 2013, 01:54:00 pm
content: Quote from: dr.diesel on November 01, 2013, 01:15:22 pmAfter purchasing many pieces of equipment from Rigol, this is the same reason I've switched to Agilent, even if it is twice the price.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 01, 2013, 02:03:31 pm
content: Quote from: RRobot on November 01, 2013, 01:54:00 pmExactly, price really isn't an issue if your getting known good shit from day one.I'm giving Rigol one last chance however, I've got an MSO4024 on order. I needed the memory and wanted to pay with the hackability, I'm fully expecting to have issues off the bat, but I knew this going in and simply have the spare cash to play. If they dick this one up bad they will be blacklisted just like m$ products from my house.

author: con-f-use
author_rank: Unknown
timestamp: November 01, 2013, 03:07:44 pm
content: Quote from: dr.diesel on November 01, 2013, 02:03:31 pmI can see the Microsoft part, though Apple has been on the way to replace them in terms of frequent dick-moves. I'd just wish there was only one working (!) OS, one language (maybe three programming languages) and one mains system (incl. plugs, frequency and voltage). Those three things would make my life just so much easier.But on topic: Will you do a full review + teardown of the Hameg? It seams promising. If been looking for one and found nothing. Same for the HMP20XX and HMP40XX series of power supplies which I find very intriguing as well.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 01, 2013, 09:50:54 pm
content: Mmmmm, well, there are a few teardown picts in Post#8, plus I'll add macro shots (was hoping today but have been busy digging a trench through my driveway!!) very soon.I may do a quick 5 minute review, and will do just about anything upon request.

author: Attorney
author_rank: Regular Contributor
timestamp: November 01, 2013, 11:52:41 pm
content: Very nice DMM. I do see a tremendous amount of wasted, empty space inside. Reminds me of my Blue-ray player I recently serviced. It consists of two pieces: (1) the SMPS; and (2) the optical drive mechanism were the PCBs attached. 90% empty space. I guess marketing figured miniaturizing the device would hurt sales.In the case of the Hameg HMC8012, that much SMD condensing and minimum wiring is a sign of excellent engineering. Me, I would have mounted the PCBs vertically, then crunched the depth to about 1/4 to get some real benchtop space-saving utility. Finally, add some iron shot to weight it down and minimize sliding on the bench surface.Paul

author: dr.diesel
author_rank: Super Contributor
timestamp: November 02, 2013, 12:00:07 am
content: Quote from: Attorney on November 01, 2013, 11:52:41 pmRight on. There are only 3-wires, cause theywiselymounted the IEC connector to the pack panel and not on the PCB, otherwise there would have been none!Well, aside from the display ribbon cable.

author: Stonent
author_rank: Super Contributor
timestamp: November 02, 2013, 08:45:01 am
content: What was the LXI logo in the corner?

author: Circlotron
author_rank: Super Contributor
timestamp: November 02, 2013, 10:53:22 am
content: No overlay on the board so none of the SMD parts have any identification. That'll be fun come service time.

author: Dark Prognosis
author_rank: Regular Contributor
timestamp: November 02, 2013, 12:37:16 pm
content: Why is it that the Chinese companies have horrible CS? I know every single time I deal with China, or even worse Japan, they refuse to admit anything is wrong and how dare this round eyes fool even suggest that. As if they were going to lose face if they fixed anything because fixing something means they released something half assed. Software is no different from them (look at Japan's Konami to see how bad it can get when Silent Hill 3 was documented as not playable on 90% of personal computers and instead of admitting a failure and patching it they released SH4 instead).

author: dr.diesel
author_rank: Super Contributor
timestamp: November 02, 2013, 12:43:32 pm
content: Quote from: Stonent on November 02, 2013, 08:45:01 amQuote from: Stolen From Hameg WebsiteSounds like a feature in development.

author: FrankBuss
author_rank: Unknown
timestamp: November 02, 2013, 01:05:33 pm
content: Quote from: Dark Prognosis on November 02, 2013, 12:37:16 pmNot all Chinese companies. Rigol and Siglent have good customer support, as seenhereandhere.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 02, 2013, 02:42:29 pm
content: Quote from: FrankBuss on November 02, 2013, 01:05:33 pmMyself and many others have had horrible luck getting anything from Rigol. I've had better luck with Verizon than Rigol, which makes it 10 orders of magnitude more pathetic.

author: blackdog
author_rank: Frequent Contributor
timestamp: November 02, 2013, 04:08:15 pm
content: Hi con-f-use,Every RMS converter has a dynamic range withs is far smaler than de DC dynamic range.I think your meter is in spec...Wat you see is the noise level of the RMS converter.The changing of de residu value is probably the change in impedance the RMS converter sees.The specs are valid voor >5% with a sinus signal.Your noise level is far below that!Test is whit a calibrated signal generator and you wil see that probely your meter is within spec.Please read this info at Geller Labs about low level AC measurements!http://www.gellerlabs.com/34401A%20AC%20zero.htmKind regartsBlackdog

author: con-f-use
author_rank: Unknown
timestamp: November 02, 2013, 04:11:11 pm
content: Thanks, but not me and not my meter. I like to think my accent is not as thick.Just thought it might be relevant to the general Rigol-bashing in this thread. Offset is still a bit high on woopsserg and then there's a three month wait.

author: Dark Prognosis
author_rank: Regular Contributor
timestamp: November 02, 2013, 07:24:36 pm
content: Quote from: con-f-use on November 02, 2013, 02:55:32 pmWTF?!? 3.5 months he was without his equipment? Eff that crap.

author: con-f-use
author_rank: Unknown
timestamp: November 02, 2013, 07:30:03 pm
content: Proper companies give you a replacement, at least in the warranty time...Was the case for my HP/Agilent stuff. They let me keep the new unit once the old was repaired.

author: Dark Prognosis
author_rank: Regular Contributor
timestamp: November 02, 2013, 07:36:28 pm
content: Quote from: con-f-use on November 02, 2013, 07:30:03 pmYeah, I have never ever heard, or seen, a company just hold your equipment hostage like that. Most of the time they will either give you a loaner or do a cross shipment and the replacement (always is a refurbished unit) is sent to you while yours you will never see again if yours was under warranty.I have to say I was saving for the Rigol 1102E but with the stuff I am hearing I dunno what to do with my 400 Amazon.com dollars now.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 02, 2013, 08:08:39 pm
content: Quote from: alm on November 02, 2013, 03:16:09 pmProbably the RTC, hopefully that is all!

author: dr.diesel
author_rank: Super Contributor
timestamp: November 02, 2013, 09:22:14 pm
content: Ok, finally macro shots. These are big so I didn't embed them.Macro Shot 1Macro Shot 2Macro Shot 3Macro Shot 4Macro Shot 5Macro Shot 6Macro Shot 7Macro Shot 8Macro Shot 9The main ADC (ADS1259) doesn't appear to have anything connected to pin 16, the reference in, or perhaps a trace under the chip. I wonder if they are using the internal reference of the ADC? I don't recognize that AD part just to the left of the ADC, marked A OW maybe, external reference perhaps?

author: Gunb
author_rank: Regular Contributor
timestamp: November 03, 2013, 07:29:35 am
content: Quote from: dr.diesel on November 02, 2013, 08:08:39 pmYes, battery should be there for the RTC.Would be a bad idea to store time & date into EEPROM ...RgdsG?unb

author: tszaboo
author_rank: Super Contributor
timestamp: November 03, 2013, 10:45:43 am
content: Quote from: dr.diesel on November 02, 2013, 09:22:14 pmThat in fact looks like a reference. Just look at the layout. There is a capacitor probably on the bandgap (top), two signals differentially comes out, differentially filtered (RC) goes under the ADC. The ground of the IC comes from somewhere else, not from the groundplane.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 03, 2013, 10:53:52 am
content: Quote from: NANDBlog on November 03, 2013, 10:45:43 amYeah, good be, has the right number of pins, but seems unnecessarily bare in terms of markings for a precision reference?

author: Zbig
author_rank: Frequent Contributor
timestamp: November 03, 2013, 12:38:49 pm
content: These "marching ants" (first photo, under the third relay) series resistor (or are they?) strings are funny. What is the purpose of such a layout? I think at least one of these is a differential pair. Some transmission line impedance optimizing, perhaps?

author: dr.diesel
author_rank: Super Contributor
timestamp: November 03, 2013, 12:47:28 pm
content: Quote from: Zbig on November 03, 2013, 12:38:49 pmYes, precision resistor array.

author: Zbig
author_rank: Frequent Contributor
timestamp: November 03, 2013, 03:14:38 pm
content: Ok, but what's the point when you don't have any vias between them i.e. cannot tap to any intermediate value? Is it just that they needed 15 times the nearest available value?EDIT:Maximum per-resistor power dissipation. Stupid me

author: Hydrawerk
author_rank: Super Contributor
timestamp: November 03, 2013, 04:08:40 pm
content: This is Porn, right?

author: Dark Prognosis
author_rank: Regular Contributor
timestamp: November 03, 2013, 04:19:03 pm
content: Quote from: Hydrawerk on November 03, 2013, 04:08:40 pmYeah, just a different kind and that poor woman and her back. She is getting up there in age where she will soon, if she hasn't already, feel remorse for those big things.

author: Hydrawerk
author_rank: Super Contributor
timestamp: November 03, 2013, 05:15:43 pm
content: For those who prefer compact size I recommend Anna Ohura from Japan, LOL.

author: owiecc
author_rank: Frequent Contributor
timestamp: November 03, 2013, 05:16:34 pm
content: Quote from: Zbig on November 03, 2013, 03:14:38 pmNot only. It may be due to:– safety: increased effective distance between two ends– reliability: one fails short but the resistance remains fairly unchanged– tolerance: many 1% resistors in series will yield smaller total tolerance

author: SeanB
author_rank: Super Contributor
timestamp: November 03, 2013, 05:31:03 pm
content: Teasers here was advertising one "Dancer" as going in soon for much needed breast reduction surgery.........Then again I know a woman who had this as well. Here the regular stores stock bra sizes from A to ll, and sell a lot of both sizes as well. Some of the women at work have a hard time going through a door sideways. Some are 44GG, 40 ,55.........

author: Dark Prognosis
author_rank: Regular Contributor
timestamp: November 03, 2013, 08:48:04 pm
content: Quote from: SeanB on November 03, 2013, 05:31:03 pmLord help them because every single woman I have seen with breasts that large wanted to, or found a way to, have a much need breast reduction surgery when they got older. The really intelligent ones had it done while they were young.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 03, 2013, 09:17:13 pm
content: I like where this thread is headed, where is the NSFW forum category?

author: Hydrawerk
author_rank: Super Contributor
timestamp: November 03, 2013, 10:29:56 pm
content: That lady named Milena Velba has clothes, LOL. Why NSFW??

author: kxenos
author_rank: Frequent Contributor
timestamp: November 03, 2013, 10:44:18 pm
content: Now that's an MMB (MELF) package!

author: dr.diesel
author_rank: Super Contributor
timestamp: November 03, 2013, 11:12:03 pm
content: Quote from: Hydrawerk on November 03, 2013, 10:29:56 pmIt's not yet, but needs to be!

author: kilobyte
author_rank: Regular Contributor
timestamp: November 04, 2013, 08:42:35 am
content: Hi,http://streamlinemechatronics.com/picts/macro9.JPGThe part marked with A0W is a AD8672ARMZ. That looks like the driver for the ADC.The reference for the ADS1259 could be the AD780A.Regards,Kai

author: anotherlin
author_rank: Frequent Contributor
timestamp: November 04, 2013, 12:47:22 pm
content: Quote from: con-f-use on November 01, 2013, 01:37:31 pmNothing to do with electronics, but I just love my Festool TS55EBQ circular saw and Rotex 90 sander. Next buy would be a CTL MIDI vaccum from Festool !

author: Kjelt
author_rank: Super Contributor
timestamp: November 04, 2013, 01:15:09 pm
content: Quote from: anotherlin on November 04, 2013, 12:47:22 pmI use my Festool VAC SYS vacuumpump in combination with a vacuum reservoir and vacuumvalve for my Weller desoldering iron, works like a charm with -0.9 bar.

author: Gunb
author_rank: Regular Contributor
timestamp: November 05, 2013, 04:13:17 pm
content: Quote from: dr.diesel on November 01, 2013, 11:45:40 amHmm, yes, the Agilent is not much more expensive.But on the other hand the Hameg can measure up to 10A and frequency up to 700kHz opposed to 3A and 300kHz as the Agilent can do.The ethernet interface of the Agilent has to be paid extra, for the HMC it's included.Also the Agilent does not include measuring lines - OK, not the most important criteria, but it makes the Agilent again more expensive.If accuracy is the most important issue, I would prefer the Agilent, if the complete package is more important, I would prefer the HMC.

author: Zbig
author_rank: Frequent Contributor
timestamp: November 05, 2013, 06:24:34 pm
content: Quote from: Gunb on November 05, 2013, 04:13:17 pmAlso, capacitance measurement. Don't forget about capacitance.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 05, 2013, 07:05:25 pm
content: Quote from: Zbig on November 05, 2013, 06:24:34 pmThe display! This pict really doesn't do it justice, but the yellow/black contrast really works well.

author: Gunb
author_rank: Regular Contributor
timestamp: November 06, 2013, 09:12:31 am
content: Quote from: dr.diesel on November 05, 2013, 07:05:25 pmYes, that's true. It's an optical feature but it's the interface to the user!Quote from: Zbig on November 05, 2013, 06:24:34 pmOh yes, you're right! Had a quick view over the spec of the Agilent. It seems that it cannot measure capacitance, right?With respect to the datasheet the HMC can measure up to 500µF (!!!) Especially if space on the desk is limited it's more than a nice feature to have included in a single device.Hmm, slowly you move me to spend some money ....

author: con-f-use
author_rank: Unknown
timestamp: November 06, 2013, 09:47:57 am
content: Quote from: Gunb on November 06, 2013, 09:12:31 amConsider the Rigol dm3068, too. It can measure between 60pF - 100mF (!!!) accurately. It has 6 1/2 digits (>2 Mcounts) and a better temperature function. The display is much worse of course and Rigol service sucks. Also it's bigger but a tiny bit cheaper. I think it's still a tough call.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 06, 2013, 11:04:43 am
content: Quote from: con-f-use on November 06, 2013, 09:47:57 amOne important note, the Hameg can currently only read RTDs, not say type K thermocouples. The DC resolution is there, I see no reason they couldn't enable this with a firmware update, probably never happen though.

author: kayvee
author_rank: Regular Contributor
timestamp: November 06, 2013, 11:20:39 am
content: Haha dr.diesel, I love your top shelf adornmentThe DMM its pretty sexy tooDo you think Hameg will update the HM8118 with a new HMC series model, that would be sweet.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 06, 2013, 11:26:40 am
content: Quote from: kayvee on November 06, 2013, 11:20:39 amQuote from: kayveeI sure hope so, the new HMC format is perfect, I'd bet that have plans to bring more equipment to this size.

author: Gunb
author_rank: Regular Contributor
timestamp: November 06, 2013, 12:44:20 pm
content: Quote from: dr.diesel on November 06, 2013, 11:26:40 amYes, seems only the beginning.Visit their erotic blog:http://blog.hameg.com/?p=1133#more-1133

author: dr.diesel
author_rank: Super Contributor
timestamp: November 06, 2013, 12:49:04 pm
content: Quote from: Gunb on November 06, 2013, 12:44:20 pmYup, this pretty much sums is up:Quote from: Hameg

author: kayvee
author_rank: Regular Contributor
timestamp: November 06, 2013, 02:10:18 pm
content: Hopefully some new HMC series launch at productronica next week then.

author: Gunb
author_rank: Regular Contributor
timestamp: November 07, 2013, 08:36:42 am
content: Quote from: kayvee on November 06, 2013, 02:10:18 pmInteresting thought. Makes me curious to keep an eye on it. Also curious to see which device of the HMC series comes next.

author: anotherlin
author_rank: Frequent Contributor
timestamp: November 12, 2013, 04:53:32 pm
content: Any news for new HMC products from Hameg ?I'm looking for a signal generator.

author: dr.diesel
author_rank: Super Contributor
timestamp: November 12, 2013, 05:00:18 pm
content: Quote from: anotherlin on November 12, 2013, 04:53:32 pmI wrote them yesterday, specifically asking, haven't heard back yet.

author: Gunb
author_rank: Regular Contributor
timestamp: November 13, 2013, 09:45:37 am
content: Quote from: dr.diesel on November 12, 2013, 05:00:18 pmThink they're very busied at Productiva trade fair in Germany this weekKind regardsGunb

author: kayvee
author_rank: Regular Contributor
timestamp: November 13, 2013, 10:07:17 am
content: It does not look like there will be any new HMC announcements at the exhibition.There was a writeup about them in the Day 1 newspaper, to be found here:http://productronica.com/en/home/journalists/press-services/dailiy-newsThey did announce a new SA , it looks uber expensive.http://blog.hameg.com/?p=1384#more-1384So it looks like the other HMC stuff is still WIP.

author: Gunb
author_rank: Regular Contributor
timestamp: November 13, 2013, 10:57:33 am
content: Makes sense to replace the 5 analyzers by a single device and offer options to extend functionality later - prefer that.Think extending HMC series might take a bit longer. Usually Hameg releases new equipment not before firmware is working almost reliable.I'm also curious to see what comes next.

author: dr.diesel
author_rank: Super Contributor
timestamp: December 19, 2013, 08:38:55 pm
content: New firmware just posted, with some new handy commands:Code:2013-12-17V01.059-> Bugfix: time stamp in logging file fixed-> Bugfix: if logging is activated, the "Log" message on the display will be changed to yellow-> Bugfix: communication problem with screenshot module of the HMExplorer software-> Bugfix: file browser fixed-> New SCPI commands:CALCulate:FUNCtion {NULL | DB | DBM | AVERage | LIMit | POWer}CALCulate:FUNCtion?CALCulate[:STATe] {OFF | ON}CALCulate:STATe?CALCulate:POWer?[SENSe:]ADCRate{SLOW|MEDium|FAST}[SENSe:]ADCRate?DATA:LOG[:STATe] {0|1|OFF|ON}DATA:LOG[:STATe]?DATA:LOG:FNAMe {<?File_Name?>},[{INT | EXT | DEF}]DATA:LOG:FNAMe?DATA:LOG:FORMat {CSV|TXT}DATA:LOG:FORMat?DATA:LOG:MODE {UNLimited|COUNt|TIME}DATA:LOG:MODE?DATA:LOG:TIME <time in sec>DATA:LOG:TIME?DATA:LOG:COUNt <no of samples>DATA:LOG:COUNt?DATA:LOG:INTerval <interval in seconds>DATA:LOG:INTerval?