Tektronix DPO7104 repairs - Page 1
Forum section: electronics / repair

author: pasta2018
author_rank: Regular Contributor
timestamp: December 07, 2024, 09:41:19 am
content: When the oscilloscope is turned on, there is a fan that does not turn. It has been confirmed that there is no problem with the fan. Checking the acquisition board, the measurement found that the input terminal of the U1750 (LT1963) was shorted to ground, and the measurement found that the two ends of the C1750 were also short-circuited. May I ask who has DPO7104, please measure whether these two ends of the normal instrument are short-circuited.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 07, 2024, 09:47:33 am
content: I removed the acquisition board and tested it again, and found that there was no short circuit. It seems that the short circuit before the test may be caused by the power interface board.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 07, 2024, 10:07:51 am
content: It's really interesting, and then disconnect the power interface board from the mother board, then I find that the mother board 3.3V is short-circuited with GND

author: squadchannel
author_rank: Frequent Contributor
timestamp: December 07, 2024, 10:14:38 am
content: Check again with the acq board alone and nothing connected.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 07, 2024, 10:55:24 am
content: Thanks，now I can confirm it's the mother board failure. 3.3V out is shorted to GND.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 07, 2024, 01:31:46 pm
content: it's the power supply CVR460-96p01b01,3.3V out is shorted.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 12, 2024, 08:29:25 am
content: Still not fixed. At present, the measured power supply voltage is normal, when I remove the acquisition board, both fans on the side of the oscilloscope rotate, but after the acquisition board is installed, only one fan rotates. Looking at the maintenance manual, it says that the acquaintion board will detect the temperature and then adjust the speed of the fan, who knows where the circuit for detecting and adjusting the fan is?I saw that there is a place marked with sense on the acquisition board, it seems to be C100, and the two ends of the measurement are in a short-circuit state, I wonder if this is normal?

author: iAleks
author_rank: Regular Contributor
timestamp: December 12, 2024, 09:39:09 pm
content: Quote from: pasta2018 on December 12, 2024, 08:29:25 amI check resistance on work board this capacitor one side 0 Ohm to GND other side 8 Ohm to GND.Update:Little bit reverse engineering, FAN's connect to transistors Q632 and Q631 on main board (R632,R641,R631,R642=475 Ohm), next PWM signals go to acquisition board, on acquisition board PWM signals connect to U1305 its ADT7476 (bottom side), PWM1 connect to pin24 U1305, PWM2 connect to pin10 U1305.Ihighly recommendwhen you do some tests, connect GOOD fan's for cool acquisition board (when i repair my scope i cool this board with 3 FAN's, two for ADC's one for NI BGA chip) and PSU-Motherboard. And bevery carefulwith acquisition board try hard not bend this board (when you disconnect this), don't apply any force or litle hit to BGA heatsinks ADC, Preamp, DEMUX.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 13, 2024, 07:38:04 am
content: Really appropriate for your information, that's really useful. I will check the signals and be careful with acquisition board.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 14, 2024, 12:25:04 pm
content: I checked my scope, the power interface board is the same with yours. The acquisition board is different, pwm1 and pwm2 is shown in the picture,it is connect to pin 10 and pin24 of adt7476

author: pasta2018
author_rank: Regular Contributor
timestamp: December 14, 2024, 01:15:34 pm
content: I guess the Q631 or Q632 is fault.The two pin resistance of each one is not the same，while Q631 is about 2k，and Q632 is infinity.It,s marked with 2A X，it's MMBT3906,a PNP transistor.

author: coromonadalix
author_rank: Super Contributor
timestamp: December 14, 2024, 02:17:41 pm
content: i would say clean your pcbit is crusty and that could cause problems if it continues and humidity goes in that

author: pasta2018
author_rank: Regular Contributor
timestamp: December 14, 2024, 04:22:44 pm
content: I don't know what to clean well. I tried cleaning with water or alcohol and it didn't seem to work. It looks very clean when it gets wet, and it looks like this when it's dry.

author: iAleks
author_rank: Regular Contributor
timestamp: December 14, 2024, 09:45:58 pm
content: Quote from: pasta2018 on December 14, 2024, 01:15:34 pmNo, this is some N channel mosfet. Fan current 0,52A. You need some mosfet with similar parameter Vthreshold

author: pasta2018
author_rank: Regular Contributor
timestamp: December 14, 2024, 10:17:48 pm
content: Oh my god, thanks! Can you tell what is marked on Q631 of your scope? Mine is 2A XK8

author: iAleks
author_rank: Regular Contributor
timestamp: December 14, 2024, 10:22:53 pm
content: Quote from: pasta2018 on December 14, 2024, 10:17:48 pm2ax3A

author: pasta2018
author_rank: Regular Contributor
timestamp: December 15, 2024, 01:01:32 am
content: Thanks again.It may be SI2302ADS-T1-E3.By the way, I saw you upgrade the 7104 to 7354 and generate an option key, have you successed?

author: iAleks
author_rank: Regular Contributor
timestamp: December 15, 2024, 01:21:00 am
content: Quote from: pasta2018 on December 15, 2024, 01:01:32 amYes i think this mosfet fine for this.Upgrade DPO7104-DPO7354 possibly only if scope have acqusition board with 4 ADC, 8DEMUX and track and hold ic, and in this moment SPC fail, i search someone who have DPO7254-DPO7354 for copy some files from it.Your acquisition board have 2ADC 4DEMUX you can activate only some options and max memory, on this forum have links to keygen, or you can send me your sn and i generate key for you.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 15, 2024, 01:59:34 am
content: I get some information from the service manual,hope it's useful.Can you post the keygen link here,I can not find it.

author: calibrationfixture
author_rank: Regular Contributor
timestamp: December 15, 2024, 05:14:26 am
content: Hi pasta2018,You need the Program(s):- gen.py and- validate.pyThe last is used to split out the Option Mask for existing Keys.Both can be found at:https://0xacab.org/mitic/tek-ssc/tree/masterAnd a list of possible Options (page 2-9) for your DPO7104 attached.Success,Calibrationfixture

author: pasta2018
author_rank: Regular Contributor
timestamp: December 15, 2024, 07:14:24 am
content: Thanks. I will learn it, hope I can fix my dpo7104 first

author: pasta2018
author_rank: Regular Contributor
timestamp: December 19, 2024, 09:56:39 am
content: It turned out that R632 was desoldered

author: iAleks
author_rank: Regular Contributor
timestamp: December 19, 2024, 01:56:31 pm
content: Your scope work now? SPC pass?

author: pasta2018
author_rank: Regular Contributor
timestamp: December 19, 2024, 02:16:04 pm
content: Not yet.Now the two fans are running normally.I uninstalled the tekscope before and am now reinstalling it, not sure if the acquisition board is still faulty.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 20, 2024, 05:37:36 am
content: Quite difficult.When I install tekscope, it reports ddsigninstall32.exe error and then continue.After installation, the tekscope start and then it report x08e error, then shutdown and restart.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 20, 2024, 11:00:56 am
content: With another DPO7054 operating system, I can now access the oscilloscope test interface. However, there are some errors, registers 224 and acquisition error

author: iAleks
author_rank: Regular Contributor
timestamp: December 20, 2024, 03:19:34 pm
content: Download here your console.txt and diag.txt from C:\TekScope\calibration

author: pasta2018
author_rank: Regular Contributor
timestamp: December 20, 2024, 03:45:29 pm
content: it seems mainly point to U200,but I havn't find this chip on acquisition board.Code:Fri Dec 20 23:06:08 2024: DIAGNOSTIC EXECUTION starting ... Pu=2 T=-127CTest "111" - Processor-->Memory-->DRAMWalk1 startedTest execution time 2.7937e-006 secTest "112" - Processor-->Memory-->DRAMCell startedTest execution time 0.025139 secTest "113" - Processor-->Memory-->DRAMMarch startedTest execution time 0.001479 secTest "114" - Processor-->Memory-->DRAMCal startedTest execution time 1.9556e-006 secTest "211" - Registers-->PCI-->MIA startedTest execution time 0.0001422 secTest "221" - Registers-->Acquisition-->ACL startedTest execution time 0.00068221 secTest "222" - Registers-->Acquisition-->PLL startedTest execution time 0.02519 secTest "223" - Registers-->Acquisition-->Preamp startedTest execution time 0.31132 secTest "224" - Registers-->Acquisition-->TrkHld startedDataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x1/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0xfe/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x2/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0xfd/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x4/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0xfb/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x8/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0xf7/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x10/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0xef/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x20/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0xdf/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x40/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0xbf/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x80/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x7f/0x0DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x1fe/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x1fd/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x1fb/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x1f7/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x1ef/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x1df/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x1bf/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x17f/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x100/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x000 exp/act 0x1ff/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x1/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x1fe/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x2/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x1fd/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x4/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x1fb/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x8/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x1f7/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x10/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x1ef/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x20/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x1df/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x40/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x1bf/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x80/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x17f/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x100/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0xff/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x001 exp/act 0x1ff/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x1/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x1fe/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x2/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x1fd/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x4/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x1fb/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x8/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x1f7/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x10/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x1ef/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x20/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x1df/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x40/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x1bf/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x80/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x17f/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x100/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0xff/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x002 exp/act 0x1ff/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x1/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x1fe/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x2/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x1fd/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x4/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x1fb/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x8/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x1f7/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x10/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x1ef/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x20/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x1df/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x40/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x1bf/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x80/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x17f/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x100/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0xff/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ+0x003 exp/act 0x1ff/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x1fe/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x1fd/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x1fb/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x1f7/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x1ef/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x1df/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x1bf/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x17f/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x100/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x11) - U200(HFD201)/HFADJ_A+0x000 exp/act 0x1ff/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x1fe/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x1fd/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x1fb/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x1f7/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x1ef/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x1df/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x1bf/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x17f/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x100/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x12) - U200(HFD201)/HFADJ_B+0x000 exp/act 0x1ff/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x1fe/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x1fd/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x1fb/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x1f7/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x1ef/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x1df/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x1bf/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x17f/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x100/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x13) - U200(HFD201)/HFADJ_C+0x000 exp/act 0x1ff/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x1fe/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x1fd/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x1fb/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x1f7/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x1ef/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x1df/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x1bf/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x17f/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x100/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x14) - U200(HFD201)/HFADJ_D+0x000 exp/act 0x1ff/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_A+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_A+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_A+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_A+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_A+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_B+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_B+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_B+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_B+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_B+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_C+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_C+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_C+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_C+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_C+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_D+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_D+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_D+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_D+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/IN_MUX_SELECT_D+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/LP_SEL_A+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/LP_SEL_A+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/LP_SEL_B+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/LP_SEL_B+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/LP_SEL_C+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/LP_SEL_C+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/LP_SEL_D+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/LP_SEL_D+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x3fe/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x3fd/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x3fb/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x3f7/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x3ef/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x3df/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x3bf/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x37f/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x100/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x2ff/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x200/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x1ff/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL+0x000 exp/act 0x3ff/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DEBUG+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DEBUG+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DEC+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DEC+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DEC+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DEC+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DEC+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DIV+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DIV+0x000 exp/act 0xe/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DIV+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DIV+0x000 exp/act 0xd/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DIV+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DIV+0x000 exp/act 0xb/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DIV+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DIV+0x000 exp/act 0x7/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_DIV+0x000 exp/act 0xf/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_EN+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_EN+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_SEL+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_SEL+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_SEL+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_SEL+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x71) - U200(HFD201)/SD_ADC_CTRL_SEL+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xffe/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xffd/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xffb/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xff7/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xfef/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xfdf/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xfbf/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xf7f/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x100/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xeff/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x200/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xdff/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x400/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xbff/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x800/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0x7ff/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SIG_MUX_SELECT+0x000 exp/act 0xfff/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SMP_MUX_SELECT_A+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SMP_MUX_SELECT_A+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SMP_MUX_SELECT_B+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SMP_MUX_SELECT_B+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SMP_MUX_SELECT_C+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SMP_MUX_SELECT_C+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SMP_MUX_SELECT_D+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x00) - U200(HFD201)/SMP_MUX_SELECT_D+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x34) - U200(HFD201)/SPARE_1+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x35) - U200(HFD201)/SPARE_2+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x36) - U200(HFD201)/SPARE_3+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x37) - U200(HFD201)/SPARE_4+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x38) - U200(HFD201)/SPARE_5+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x32) - U200(HFD201)/SYNTH_FILT_DAC+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xffe/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xffd/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xffb/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xff7/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xfef/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xfdf/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xfbf/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xf7f/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x100/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xeff/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x200/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xdff/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x400/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xbff/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x800/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0x7ff/0x0Wlk1HWMismatch (0x31) - U200(HFD201)/SYNTH_TUNE_DAC+0x000 exp/act 0xfff/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x7fe/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x7fd/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x7fb/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x7f7/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x7ef/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x7df/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x7bf/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x77f/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x100/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x6ff/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x200/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x5ff/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x400/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x3ff/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL+0x000 exp/act 0x7ff/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL_BLOCK+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL_BLOCK+0x000 exp/act 0x6/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL_BLOCK+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL_BLOCK+0x000 exp/act 0x5/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL_BLOCK+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL_BLOCK+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x70) - U200(HFD201)/TEST_BUS_SEL_BLOCK+0x000 exp/act 0x7/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x21) - U200(HFD201)/TEST_SIG_DAC_A+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x22) - U200(HFD201)/TEST_SIG_DAC_B+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x23) - U200(HFD201)/TEST_SIG_DAC_C+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x24) - U200(HFD201)/TEST_SIG_DAC_D+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0xfe/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0xfd/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0xfb/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0xf7/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0xef/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0x20/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0xdf/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0x40/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0xbf/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0x80/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0x7f/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SELECT+0x000 exp/act 0xff/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_A+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_A+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_A+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_A+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_A+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_B+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_B+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_B+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_B+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_B+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_C+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_C+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_C+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_C+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_C+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_D+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_D+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_D+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_D+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x20) - U200(HFD201)/TEST_SIG_SEL_D+0x000 exp/act 0x3/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0x1e/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0x2/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0x1d/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0x4/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0x1b/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0x8/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0x17/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0x10/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0xf/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/TH_CONTROL+0x000 exp/act 0x1f/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/ZMUX_DISABLE+0x000 exp/act 0x1/0x0Wlk1HWMismatch (0x10) - U200(HFD201)/ZMUX_DISABLE+0x000 exp/act 0x1/0x0*** TEST FAILED ***Test execution time 0.21408 secdiagMgr:diagLogFailure: Fri Dec 20 23:06:08 2024 D Pu=2 T=-127 "DataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x1/0x0"Test "225" - Registers-->Acquisition-->ADC startedTest execution time 0.010165 secTest "226" - Registers-->Acquisition-->DAC startedTest execution time 0.42803 secTest "227" - Registers-->Acquisition-->Demux startedTest execution time 0.0046355 secTest "231" - Registers-->Trigger-->SBTL startedTest execution time 8.6603e-005 secTest "232" - Registers-->Trigger-->SGTL startedTest execution time 0.01268 secTest "233" - Registers-->Trigger-->TrigComp startedTest execution time 0.011714 secTest "234" - Registers-->Trigger-->CommTrig startedTest execution time 0.57148 secTest "241" - Registers-->Misc-->FanCtrl startedTest execution time 0.43045 secTest "311" - Acquisition-->Demux-->RunAB startedTest execution time 0.21265 secTest "312" - Acquisition-->Demux-->AcqDone startedTest execution time 0.00092665 secTest "313" - Acquisition-->Demux-->SysRdy startedTest execution time 0.00055677 secTest "314" - Acquisition-->Demux-->Interrupt startedTest execution time 0.0010096 secTest "315" - Acquisition-->Demux-->IdcLoSpeed startedTest execution time 0.0032996 secTest "316" - Acquisition-->Demux-->IdcHiSpeed startedTest execution time 0.11145 secTest "317" - Acquisition-->Demux-->IdcAcqDataXfr startedTest execution time 0.002733 secTest "318" - Acquisition-->Demux-->IdcDispDataXfr startedTest execution time 0.0027344 secTest "321" - Acquisition-->Dma-->PaDma startedTest execution time 0.0024484 secTest "322" - Acquisition-->Dma-->IdcAcqDma startedTest execution time 0.0031037 secTest "331" - Acquisition-->Memory-->MemData startedTest execution time 0.010492 secTest "332" - Acquisition-->Memory-->MemAddr startedTest execution time 0.0019681 secTest "333" - Acquisition-->Memory-->MemSpeed startedTest execution time 11.094 secTest "341" - Acquisition-->Vertical-->Preamp Inputs started*** TEST N/A FOR THIS HW ***Test execution time 1.9556e-006 secTest "342" - Acquisition-->Vertical-->TH Inputs startedU200(HFD201) input IN_A failed at high/low input level (ref:0000 hi:0000 lo:0000 mar:10000)U200(HFD201) input IN_B failed at high/low input level (ref:0000 hi:0000 lo:0000 mar:10000)U200(HFD201) input IN_C failed at high/low input level (ref:0000 hi:0000 lo:0000 mar:10000)U200(HFD201) input IN_D failed at high/low input level (ref:0000 hi:0000 lo:0000 mar:10000)*** TEST FAILED ***Test execution time 0.30627 secdiagMgr:diagLogFailure: Fri Dec 20 23:06:08 2024 D Pu=2 T=-127 "U200(HFD201) input IN_A failed at high/low input level (ref:0000 hi:0000 lo:0000 mar:10000)"Test "343" - Acquisition-->Vertical-->ADC Inputs startedU10_A(HFD204) input IN2 failed at low input level (ref:813b18 hi:813c4f lo:813ba4 mar:0100)*** TEST FAILED ***Test execution time 0.75759 secdiagMgr:diagLogFailure: Fri Dec 20 23:06:08 2024 D Pu=2 T=-127 "U10_A(HFD204) input IN2 failed at low input level (ref:813b18 hi:813c4f lo:813ba4 mar:0100)"Test "344" - Acquisition-->Vertical-->ADC Outputs startedTest execution time 0.23422 secTest "345" - Acquisition-->Vertical-->50ohmOvld startedTest execution time 1.0732 secTest "346" - Acquisition-->Vertical-->LFComp startedTest execution time 0.83658 secTest "351" - Acquisition-->PLL-->Clock Freq startedTest execution time 0.67067 secTest "361" - Acquisition-->HFSource-->SINE startedHFSource:SINE:INTLV_CAL_VCO_N CH1 failed, amplitude too low (Act delta:1.386e-001; hi:-1.980e-001 lo:-3.366e-001; Exp Delta min:1.720e-001)*** TEST FAILED ***Test execution time 0.049042 secdiagMgr:diagLogFailure: Fri Dec 20 23:06:08 2024 D Pu=2 T=-127 "HFSource:SINE:INTLV_CAL_VCO_N CH1 failed, amplitude too low (Act delta:1.386e-001; hi:-1.980e-001 lo:-3.366e-001; Exp Delta min:1.720e-001)"Test "362" - Acquisition-->HFSource-->SINE33 startedHFSource:SINE33:ACQ_PHASE_CAL_CLK CH1 failed, amplitude too low (Act delta:1.426e-001; hi:-2.099e-001 lo:-3.524e-001; Exp Delta min:2.680e-001)*** TEST FAILED ***Test execution time 0.049997 secdiagMgr:diagLogFailure: Fri Dec 20 23:06:08 2024 D Pu=2 T=-127 "HFSource:SINE33:ACQ_PHASE_CAL_CLK CH1 failed, amplitude too low (Act delta:1.426e-001; hi:-2.099e-001 lo:-3.524e-001; Exp Delta min:2.680e-001)"Test "363" - Acquisition-->HFSource-->SQUARE startedHFSource:SQUARE:HF_LF_CAL_STEP CH1 failed, amplitude too low (Act delta:3.980e-002; hi:-4.219e-001 lo:-4.617e-001; Exp Delta min:4.340e-001)*** TEST FAILED ***Test execution time 0.05094 secdiagMgr:diagLogFailure: Fri Dec 20 23:06:08 2024 D Pu=2 T=-127 "HFSource:SQUARE:HF_LF_CAL_STEP CH1 failed, amplitude too low (Act delta:3.980e-002; hi:-4.219e-001 lo:-4.617e-001; Exp Delta min:4.340e-001)"Test "371" - Acquisition-->AcqProcessor-->SPI startedTest execution time 0.0001651 secTest "372" - Acquisition-->AcqProcessor-->INT Line startedTest execution time 0.00062606 secTest "373" - Acquisition-->AcqProcessor-->Interrupt startedTest execution time 0.00048917 secTest "374" - Acquisition-->AcqProcessor-->RelayDrive started*** TEST N/A FOR THIS HW ***Test execution time 1.9556e-006 secTest "375" - Acquisition-->AcqProcessor-->TWI startedTest execution time 0.00014248 secTest "376" - Acquisition-->AcqProcessor-->PrbInterrupts startedTest execution time 0.00036709 secTest "377" - Acquisition-->AcqProcessor-->TCpowerCtrl started*** TEST N/A FOR THIS HW ***Test execution time 1.6762e-006 secTest "378" - Acquisition-->AcqProcessor-->MAX517 startedTest execution time 0.00096716 secTest "411" - Trigger-->Inputs-->Ch1 startedTest execution time 0.10322 secTest "412" - Trigger-->Inputs-->Ch2 startedTest execution time 0.11976 secTest "413" - Trigger-->Inputs-->Ch3 startedTest execution time 0.11968 secTest "414" - Trigger-->Inputs-->Ch4 startedTest execution time 0.11995 secTest "415" - Trigger-->Inputs-->Line startedTest execution time 0.016928 secTest "416" - Trigger-->Inputs-->Video startedTest execution time 0.18757 secTest "417" - Trigger-->Inputs-->Events startedTest execution time 0.20396 secTest "418" - Trigger-->Inputs-->Serial startedTest execution time 0.30997 secTest "421" - Trigger-->Outputs-->523TrgOut startedTest execution time 0.00072942 secTest "422" - Trigger-->Outputs-->523CpuInt startedTest execution time 0.00052046 secTest "423" - Trigger-->Outputs-->Fedge startedTest execution time 0.040204 secTest "424" - Trigger-->Outputs-->Ftrig startedTest execution time 0.03961 secTest "425" - Trigger-->Outputs-->TrigInfo startedTest execution time 0.47688 secTest "431" - Trigger-->Timers-->Delay startedTest execution time 0.096932 secTest "432" - Trigger-->Timers-->Delta startedTest execution time 0.047023 secTest "433" - Trigger-->Timers-->Holdoff startedTest execution time 0.10354 secTest "434" - Trigger-->Timers-->PostTrig startedTest execution time 0.15703 secTest "435" - Trigger-->Timers-->PreTrig startedTest execution time 0.057601 secTest "436" - Trigger-->Timers-->Timeout startedTest execution time 0.04074 secTest "511" - TekLink-->Topology -->Signals started-- The instrument does not have a physical connection to teklink network.*** TEST N/A FOR THIS HW ***Test execution time 0.00010169 secTest "521" - TekLink-->Trigger -->Path started-- The instrument does not have a physical connection to teklink network.*** TEST N/A FOR THIS HW ***Test execution time 0.00010029 secTest "531" - TekLink-->Reference -->Path started-- The instrument does not have a physical connection to teklink network.*** TEST N/A FOR THIS HW ***Test execution time 9.9733e-005 secFri Dec 20 23:06:28 2024: DIAG MODE DIAGNOSTICS RESULTS: 6 (of 70) tests SKIPPEDFri Dec 20 23:06:28 2024: DIAG MODE DIAGNOSTICS RESULTS: 6 (of 64) EXECUTED tests FAILEDTest 224 - F ==> Registers->Acquisition->TrkHld, 0.21408 secDataLineTstStepFail- U200(HFD201)/SPARE_0+0x000 exp/act 0x1/0x0Test 342 - F ==> Acquisition->Vertical->TH Inputs, 0.30627 secU200(HFD201) input IN_A failed at high/low input level (ref:0000 hi:0000 lo:0000 mar:10000)Test 343 - F ==> Acquisition->Vertical->ADC Inputs, 0.75759 secU10_A(HFD204) input IN2 failed at low input level (ref:813b18 hi:813c4f lo:813ba4 mar:0100)Test 361 - F ==> Acquisition->HFSource->SINE, 0.049042 secHFSource:SINE:INTLV_CAL_VCO_N CH1 failed, amplitude too low (Act delta:1.386e-001; hi:-1.980e-001 lo:-3.366e-001; Exp Delta min:1.720e-001)Test 362 - F ==> Acquisition->HFSource->SINE33, 0.049997 secHFSource:SINE33:ACQ_PHASE_CAL_CLK CH1 failed, amplitude too low (Act delta:1.426e-001; hi:-2.099e-001 lo:-3.524e-001; Exp Delta min:2.680e-001)Test 363 - F ==> Acquisition->HFSource->SQUARE, 0.05094 secHFSource:SQUARE:HF_LF_CAL_STEP CH1 failed, amplitude too low (Act delta:3.980e-002; hi:-4.219e-001 lo:-4.617e-001; Exp Delta min:4.340e-001)DIAGNOSTIC EXECUTION finished in 19.985 secDRAM is Mt_ddr2_512M_667Dram Calibration results:-------------------------DramCal has PASSED on all Demuxs-- The instrument does not have a physical connection to teklink network.pipe 0:4 buf:-128, buf:-1, cross 17983pipe 0:5 buf:-128, buf:-5, cross 25000pipe 0:6 buf:-128, buf:-11, cross 25000pipe 0:7 buf:-128, buf:-11, cross 25000pipe 0:4 buf:-128, buf:-1, cross 19840pipe 0:5 buf:-128, buf:-6, cross 25000pipe 0:6 buf:-128, buf:-11, cross 25000pipe 0:7 buf:-128, buf:-13, cross 25000pipe 0:5 buf:-128, buf:-5, cross 25000pipe 0:6 buf:-128, buf:-11, cross 25000pipe 0:7 buf:-128, buf:-14, cross 25000pipe 0:4 buf:-128, buf:-1, cross 14286pipe 0:5 buf:-128, buf:-6, cross 25000pipe 0:6 buf:-128, buf:-11, cross 25000pipe 0:7 buf:-128, buf:-14, cross 25000can't get a good crossingBad cross measurementPhase cal pass 2dmx 0 took 32 iterationsdmx 1 took 32 iterationsdmx 2 took 32 iterationsdmx 3 took 32 iterationspipe 0:4 buf:-128, buf:-2, cross 7413pipe 0:5 buf:-128, buf:-7, cross 25000pipe 0:6 buf:-128, buf:-12, cross 25000pipe 0:7 buf:-128, buf:-13, cross 25000pipe 0:5 buf:-128, buf:-6, cross 25000pipe 0:6 buf:-128, buf:-12, cross 25000pipe 0:7 buf:-128, buf:-12, cross 25000pipe 0:5 buf:-128, buf:-5, cross 25000pipe 0:6 buf:-128, buf:-11, cross 25000pipe 0:7 buf:-128, buf:-13, cross 25000can't get a good crossinginitial cross 111 chip:quad 0:0initial cross 107 chip:quad 0:1initial cross 113 chip:quad 0:2initial cross 108 chip:quad 0:3initial cross 118 chip:quad 1:0initial cross 23873 chip:quad 1:1Bad cross measurementPhase cal pass 3dmx 0 took 32 iterationsdmx 1 took 32 iterationsdmx 2 took 32 iterationsdmx 3 took 32 iterationscan't get a good crossinginitial cross 111 chip:quad 0:0initial cross 107 chip:quad 0:1initial cross 114 chip:quad 0:2initial cross 108 chip:quad 0:3initial cross 142 chip:quad 1:0initial cross 23873 chip:quad 1:1Bad cross measurementPhase cal pass 4dmx 0 took 32 iterationsdmx 1 took 32 iterationsdmx 2 took 32 iterationsdmx 3 took 32 iterationscan't get a good crossinginitial cross 111 chip:quad 0:0initial cross 107 chip:quad 0:1initial cross 113 chip:quad 0:2initial cross 108 chip:quad 0:3initial cross 118 chip:quad 1:0initial cross 23873 chip:quad 1:1Bad cross measurementERROR (0xffffffff): digPhaseCal failed

author: iAleks
author_rank: Regular Contributor
timestamp: December 20, 2024, 06:56:19 pm
content: Downlod here console.txt file.On previous photo i see your acqusition board without track and hold ic (its U200), look like tekscope program trying get access to u200.Try delete callibration file C:\TekScope\calibration >calSPCConst.dat< then restart scope

author: pasta2018
author_rank: Regular Contributor
timestamp: December 21, 2024, 01:06:56 am
content: Hi iAleks，thanks for your advise.I delete calibration file and restart,the errors didn's change.I upload the console.txt file here.

author: iAleks
author_rank: Regular Contributor
timestamp: December 21, 2024, 06:20:05 am
content: Your acqusition board looks like this? right?My board logCode:------- VERSION INFO -------"5.3.5 Build 22"RTL versionExpected = "5.3.5.22"Actual = "5.3.5.22"UI versionExpected = "5.3.5.22"Actual = "Unknown - UI not Started"----------------------------================================================================SYSTEM INFORMATION:StingrayModel: DPO7104Serial number: C010424Scope.exe Version: 5.3.5 Build 22; RELEASEDiagnostics:Powerup mode: USERPowerups Count: 74Timed Demo Info: Timed Demo, 0 days left.Options:ASM: Advanced Search & MarkLT: Limit TestApp 75 Availnot installed.Hardware Configuration:CPU Board Id: 1, 1Windows Driver version: 1021DMIA Image version: 10011156ACL4 Image version: AC140707Number of channels: 4Acq board RevADEMUXES PRESENT: 4 DEMUXES IN USE: 4ATODS PRESENT: 2 ATODS IN USE: 2Track and Holds: 0ACQ MEM: 268435456ACQ BW: 1000ACQ SAMPLE RATE: 5000DEMUX ASIC REV: 0FTL PRESENT: 0Max Recovd Clk: 1.25e+009CalFilter support: 0BWE Support: 1CF StandardBits: 0x00000000BWE Bandwidth: 1.000e+009BWE SampleRate: 5.000e+009IsVista: 0================================================================system configuration flagsPREAMP HFD144PREAMP M66850 ohm termination1 meg ohm terminationAttenuator 10xAdvanced SearchLimit TestTrigger VideoTrack and Holds: 0

author: pasta2018
author_rank: Regular Contributor
timestamp: December 21, 2024, 07:36:23 am
content: yes.My acquisition board is the same with this one

author: iAleks
author_rank: Regular Contributor
timestamp: December 21, 2024, 04:22:24 pm
content: Im not shure were problem, check this resistors. Then i prepare HDD image for recovery.And i dont like this string from your console fileFri Dec 20 23:06:08 2024: DIAGNOSTIC EXECUTION starting ... Pu=2T=-127CMaybe this couse board have somewhere bad configuration, this explains the negative temperature value and the attempt to U200.Can you take good photo your board? Two side of pcb.

author: coromonadalix
author_rank: Super Contributor
timestamp: December 21, 2024, 05:58:16 pm
content: on some acq of other models you have to rewrite some eeprom, i forgetting for what code ... but it cleared some errors, you have to go in the big tds7k threadhere's some others, for the given scripts i need to restart a computer to get them, the eeprom rewrite ..... not sure all related, but yes the other thread answer seems to point to something who get very hothttps://www.eevblog.com/forum/testgear/tektronix-tds6000b7000b-series-nvram-dump-script/https://www.eevblog.com/forum/testgear/tektronix-tds8000-backup-calibration-parameters/msg5755891/#msg5755891

author: pasta2018
author_rank: Regular Contributor
timestamp: December 22, 2024, 12:55:16 am
content: After I downloaded the latest firmware from the Tektronix official website and reinstalled it, the oscilloscope would have a blue screen when I booted up TekScope.The backup dpo7054 system can be started normally with a backup and will not be blue screen.Pay attention to my console log, there is also an H10-A error that causes acquisition 361 etc. When I was measuring, I found that the vertical position of channel 2 can be adjusted, while the vertical position of other channels is not adjustable, and it will closely follow the baseline of channel 2.

author: iAleks
author_rank: Regular Contributor
timestamp: December 22, 2024, 01:48:07 am
content: About software, try thislink, this hdd image created with Macrium Reflect.And check resistors from my previous message, this resistors for board configuration.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 22, 2024, 04:21:36 am
content: Thanks for the backup. Unlucky I can not download the file for some net reason in the country.I will take time to check the resistence first as yesterday I have packed the scope.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 22, 2024, 04:59:06 am
content: The four resistance value is the same with yours.Attached is my acquisition board.

author: iAleks
author_rank: Regular Contributor
timestamp: December 22, 2024, 11:51:37 pm
content: What file sharing services you have access? I have Wechat, maybe I can send you a file through it? Image size 6GB

author: pasta2018
author_rank: Regular Contributor
timestamp: December 23, 2024, 12:15:30 am
content: I will try VPN to access the link.I'm glad to add your WeChat.My account is threemor

author: pasta2018
author_rank: Regular Contributor
timestamp: December 23, 2024, 01:31:28 pm
content: I'm glad to report, with iAleks HDD image, my scope is fixed. There is no error now.Thanks again for all the help.

author: pasta2018
author_rank: Regular Contributor
timestamp: December 23, 2024, 01:49:43 pm
content: Since I used the DPO7054 oscilloscope system to recover my oscilloscope, and the DPO7054 oscilloscope acquisition board has U200, my DPO7104 does not have this chip, which leads to errors in self-test. The correct DPO7104 system is now used and the machine is normal.SPC is also pass.