UTD420C .What do you need to know on the scope?(I have got one to play with ) - Page 1
Forum section: electronics / testgear

author: eevblogfan
author_rank: Frequent Contributor
timestamp: May 01, 2013, 05:04:36 am
content: heyI know some Uni-T distributor and he told me to "learn" all the hidden options ect of that scope , and I thought , don't you need some info about Uni-T's scopes ?well I can tell you that my DSOX2002 is better , we compared them , the Uni-t has 16pF while my agilent has only 11pF ( input capacitance ) hence the fall / raise time was more accurate in my agilent , also he has low count on that particular measurement , while mine was 17.5nS , the Uni-T was filckering 18~20nS and has only 1nS resolution while mine has 0.1nS ,the scope feel reasonably , the buttons feel okay but the knobs feel crappy , now way near the agilent , the scope has some Dual time base option ( I didn't discovered all but I can tell you it has 2 screens for say CH1 with two different time bases , wired )the scope has lame DMM option which can measure voltage current resistance and diode etc , but the actual inputs measures only voltage , you get two adaptors ( Shunts) for measuring 400/40/4 mA range ,that really annoying since you have to manually change the range on the adaptor and then change the range n the scope himself , very bad user interface - to my opinion )Unlike the agilent - you have only run/stop mode and not single on the panel , in order to get single shot(s) you need to go to the trigger menu and choose "single"the horrible small display is no way near the HD (?) display the agilent got , not talking about the horrible red(ish) Uni-T's color on the back , yuckthat Uni-T cost is aprox 2500$ for 4 CH 200Mhz device , I would definitely go for the 70Mhz 4CH agilent offer ( in my particular case , with 2500$ I could have bought 350Mhz 4CH , *BUT* that only me and the local Agilent's distributor ) hence I would only go for either old HP/Agilent/lecroy etc , and NEVER new or used Uni-T , But assuming you do , well , it's your ChoiceIf you need/want me to run some checks for that or any other Uni-T model ( including meters etc ) please let me know , I DO have access to Cal lab with fluke 5700 within cal so no worries , and some other 500Mhz 4CH scopes to compare , so please give me any taskCheers !

author: Wytnucls
author_rank: Super Contributor
timestamp: May 01, 2013, 06:45:06 am
content: I think you mean the 4 channel 200MHz 2GS/s 24kptsUTD4204C, which is an old design, with the low res screen.The one to test, if you can get hold of it, is the newer 16Mpts 100MHzUTD4102CM.

author: eevblogfan
author_rank: Frequent Contributor
timestamp: May 01, 2013, 07:20:44 am
content: no problem . I'll contact him today

author: Hydrawerk
author_rank: Super Contributor
timestamp: May 01, 2013, 04:08:18 pm
content: UTD4204C seems to be not bad, but it has the bloody small screen. Anyway, try to compare it with Rigol DS1000 somehow. Check, if it has cursors in XY mode.Their webpage is really crap. There is no support like firmware or user manual for download.http://www.uni-trend.com/UT4042C.htmlIs it possible to get the new firmware somehow?? The upgrade procedure is mentioned in the user manual.Anyway, I managed to get the user manual.http://www.gme.cz/dokumentace/720/720-110/orn.720-110.1.pdfThis scope is not cheap at all. It costs about 27000 CZK, but my Agilent DSOX2002A costs about 31000 CZK.http://www.gme.cz/osciloskopy-multifunkcni/digitalni-stolni-osciloskop-utd4104c-p720-110/#dokumentaceI don't know, who will ever buy this scope. If I wanted a cheap 200MHz scope, I would go for Owon SDS8202 for 16 630 CZK.

author: eevblogfan
author_rank: Frequent Contributor
timestamp: May 01, 2013, 09:29:51 pm
content: heyI returned the scope today since you told me the main interest is for the new one ( 100Mhz 4 CH 16Mpst ) so I told the man and he will give me that scope once he'll have it

author: Hydrawerk
author_rank: Super Contributor
timestamp: May 01, 2013, 09:44:42 pm
content: Great. I am looking forward to info about this scope... Well, but if it was a bargain, they would sell it in Czech Republic... It is still just 2×500Ms/s or 1Gs/s scope.

author: Hydrawerk
author_rank: Super Contributor
timestamp: May 01, 2013, 09:52:35 pm
content: Quote from: Wytnucls on May 01, 2013, 06:45:06 amWell, I am not sure if there is any UTD4102CM. I could only find UTD2102CM.http://www.pinsonne-elektronik.de/pi1/pd125.htmlhttps://www.eevblog.com/forum/reviews/has-anyone-seen-the-utd2102cm/

author: Hydrawerk
author_rank: Super Contributor
timestamp: May 01, 2013, 10:01:31 pm
content: Unitrend hasno info in englishabout this scope UTD2102CM. You must go to their chinese web.http://www.uni-trend.com.cn/cp-show.asp?yy=%D6%D0%CE%C4&ProductNO=719Why no info in english?? You can even download the user manual in chinese.http://www.uni-trend.com.cn/download/UTD2000M%20%E4%BD%BF%E7%94%A8%E6%89%8B%E5%86%8C%20V1.0.pdfAnd have you heard about scope UTD8202C? But apart from the large screen it's not interesting at all. Memory 24kpoints per channel and 15000 waveforms per second.http://www.uni-trend.com.cn/cp-show.asp?yy=%D6%D0%CE%C4&ProductNO=723Okay, they have an english page, too, but there is no info on these new scopes...http://www.uni-trend.com.hk/

author: Wytnucls
author_rank: Super Contributor
timestamp: May 01, 2013, 10:10:57 pm
content: The UTD4102CM is the same 100MHz scope as the UTD2102CM, except that the sample rate has been pushed up to 2GS/s and it weights nearly 1Kg more, at 3.8Kg. Both of them are 2-channel scopes only.

author: Wytnucls
author_rank: Super Contributor
timestamp: May 01, 2013, 10:22:09 pm
content: The top of the range Uni-T scope, at the moment, is theUTD5102C:2 channels, 1GHz, 5GS/s, 128Mpts, 20,000wfms/s and 8.4" screen.

author: Hydrawerk
author_rank: Super Contributor
timestamp: May 01, 2013, 11:14:35 pm
content: Well, most of these scopes will probably never be sold in Europe or english speaking countries. Seehttp://www.uni-trend.com.cn/cp.aspIf they wanted to sell them in Europe, they would make better english websites as Rigol does. I am not a Rigol fan, I will hardly ever buy their products but they are much more trustworthy.UNI-T is good at cheap multimeters, but I would never spend much money on a UNI-T product.

author: Wytnucls
author_rank: Super Contributor
timestamp: May 02, 2013, 02:34:39 am
content: The scopes are offered for sale all over the world.The operating software is in English and so is the manual. That's all what importers are interested in, apart from competitive pricing and technology, of course.The English language site hasn't been updated in years, so it is always better to consult the Chinese one and translate the important bits.I don't discount any product out of hand. The proof is in the pudding.Somebody with some established unbiased knowledge will eventually review these new scopes and let us know if the price/quality ratio is acceptable.A competitive entrant at the lower end of the market, can only be beneficial for all hobbyists out there.https://www.emtesco.com/index.php/oscilloscopes/benchtop/utd2102cm.htmlhttp://www.ebay.com/itm/UNI-T-UTD2102CM-2-Channel-100MHz-1GSa-s-Benchtop-Oszilloskop-/160785832832?pt=Mess_Pr%C3%BCftechnik&hash=item256f951b80http://www.reichelt.nl/Oscilloscopes-Spectrum-Analyser/UTD-2102-CM-R/3//index.html?ACTION=3&GROUPID=5898&ARTICLE=123970&SHOW=1&START=0&OFFSET=16&http://gsmserver.com/shop/equipment/measuring_equipment/digital_oscilloscope_uni_t_utd2102cm.php

author: Hydrawerk
author_rank: Super Contributor
timestamp: May 02, 2013, 01:46:19 pm
content: Thanks for info and pictures. These scopes are maybe an interesting alternative to Rigol DS1000 series. Well, Rigol has much better marketing, I must say. But my UNI-T distrubitorhttp://www.tipa.eu/en/oscilloscopes-generators/c-1164/doesn't sell these new scopes and they even don't answer emails.On the other hand, there are some distributors in Czech Republic who sell full assortment of Rigol, Owon, Hantek, Siglent and GW Instek...

author: Wytnucls
author_rank: Super Contributor
timestamp: May 04, 2013, 01:57:19 am
content: Things to check for on the UTD2102/4102CM:1) 150,000 wfm/s waveform acquisition rate...obviously2) Full use of the 7" 800x480 TFT screen with 'Menu On/Off' button (It seems like the working area doesn't widen when the menu is removed, which is pretty lame)3) Waveform intensity grading, if any4) Firmware stability and features5) Fan noise level

author: hgg
author_rank: Frequent Contributor
timestamp: May 04, 2013, 04:05:53 am
content: I can't wait to see the results!