Tektronix 11801 diagnostic and repair help [solved] - Page 1
Forum section: electronics / repair

author: rhb
author_rank: Super Contributor
timestamp: March 15, 2019, 08:10:02 pm
content: My 11801 arrived today. On power up it makes two beeps and puts up two lines of text on the screen:Diagnostics in progressComm Test in progressBut does not go past that. The seller sent a picture showing an E5622 error, "passed" below that and "????" for the timebase, main acq, mcu a-d acq entries and a picture of the screen with "Self test in progress".I have reseated all the boards and cables behind the CRT and the cables on the display board. The display board was missing the 4 corner screws. I have replaced those.So does anyone recognize the issue? While I was pulling the boards I checked the 4 lithium cells and all were over 3 V despite 1988 date codes. That did not match the service manual for the 11801C which is a little disturbing. Is this really just a CSA803 with 4 extra ADC channels? Does anyone know of a manual for this unit? I think all the ROMs are marked 1989, but was not meticulous about that detail as I expected reseating the connectors to have an effect.Thanks,Reg

author: pbarton
author_rank: Frequent Contributor
timestamp: March 15, 2019, 09:03:55 pm
content: "Successful completion of the Kernal diagnostics is indicated by the start of the Self Test diagnostics."single high/low beep and illuminated menu button indicators. = Executive Kernal failure.Manuals available herehttp://w140.com/tekwiki/wiki/11801

author: rhb
author_rank: Super Contributor
timestamp: March 15, 2019, 09:17:12 pm
content: Do you have any more information? I had worked out that the initial power on test was failing, but without any service data it's a little hard to proceed.All the manuals I've found on line, despite other labeling have been for the 11801C which is very different.

author: pbarton
author_rank: Frequent Contributor
timestamp: March 15, 2019, 09:42:36 pm
content: Manuals available herehttp://w140.com/tekwiki/wiki/11801Read the section on 'Diagnostic Troubleshooting'"If the error message does not appear on the screen, the error index code can be obtained from the status LEDs and pins on the A15 Memory Management Unit board (see Fig 6-34)."

author: rhb
author_rank: Super Contributor
timestamp: March 15, 2019, 11:08:37 pm
content: Unfortunately, the 11801 link is actually 11801C documentation. As is also the case at BAMA and a bunch of other sites. Having looked at the 11801C manual it's clear it's quite different from an 11801.However, after I pulled the boards a 2nd time to try to identify them it has come back to life! But only 3 of the boards are labeled with anything intelligible to a human. Just a number and two of them it's not clear it's even a Tek number! One has a label "Property of and copyright Silicon Software".I need replacements for the NVRAM which no one in the US has at the moment (and won't until June!!!), but a member of the tekscopes list in the UK has offered to get a couple and send them to me.For $185 delivered a good score. Now if the two SD-22s I got for $150 work....

author: rhb
author_rank: Super Contributor
timestamp: March 25, 2019, 01:24:21 am
content: After going around and around with it. I finally got it working:I set J710,711,712 & 713 to zero on the A14 I/O board. That still did not prevent it from getting stuck in the diagnostics at an E5622 error. With the "exit diagnostics" button disabled on the touchscreen.Then at the suggestion of a member of the TekScopes list I removed the battery backup sockets from U500 and U511. In this case it had NEC D43256 - 10L CMOS static ram in sockets with two small lithium cells and small board with a DS1213 with a 1997 date code. Removing the battery backup and directly seating the SRAM in the socket let it boot.After resetting J710-713 back to one, it halts in the extended with a T1331 timebase error. But pressing the "diagnostics exit" twice will allow the scope to function. With the calibrator connected to an SD-22 it reports a 30 ps rise time.I still don't really know how to operate it, but everything seems OK. It shows 44,000 hours of operating time and 2000 power ons.

author: arw
author_rank: Newbie
timestamp: June 15, 2019, 10:03:58 pm
content: Brilliant! Exact same fault, E5622, on my 11801, removed both battery backups on U500 and U511 and now works after exiting from diagnostic.

author: rhb
author_rank: Super Contributor
timestamp: June 24, 2019, 04:28:00 pm
content: It really is worth replacing the NVRAM chips. STM makes a replacement. Whether you install them now or later, don't wait to buy them. They were almost unobtainium when I needed them. I had to get a pair shipped from the UK and then I almost didn't get them as they sold some 36 pieces in about 10 days and RS claimed they were out of stock. Conductive anti-static foam will discharge the internal batteries, so watch out for that.While the UI is the worst ever, the performance with a fast head is quite a amazing.

author: GreyWoolfe
author_rank: Unknown
timestamp: June 24, 2019, 08:28:07 pm
content: Quote from: rhb on June 24, 2019, 04:28:00 pmI wonder of those engineers were having drunken conversations with HP engineers for the 54201D scope, truly a different UI.