elitech wjl-6000 hvac leak detector halogen gas tester hvac refrigerant ac sniffer r22 r410a r134a elitech <br>general descriptions: <br> <p>incorporated with advanced leak detect technology. elitech wjl-6000 is our new product with the stable function and small appearance; it’s an ideal and economic halogen leak detector. <p> <br>under normal working circumstance, the detector will beep when it is turned on. <br> <br>it is different with the rapid sound of warning alarm. <br> <br>applications: <br> <br>the wjl-6000 may also be used to: <br> <p>detect leaks in other systems and storage/recovery containers. it will respond to all halogenated (contains chlorine of fluorine) refrigerants. <p> <br> this includes, but is not limited to: <br> <br> cfcs e.g. r12, r11, r500, r503 etc.. <br>. <br> hcfcs e.g. r22, r123, r124, r502 etc... <br> <br> hfcs e.g. r134a, r404s, r125 etc... <br> <br> blends such as az-50. hp62. mp39 etc... <br> <br>detect ethylene oxide gas leaks in hospital sterilizing equipment (it will detect the halogenated carrier gas). <br> <br>detect sf-6 in high voltage circuit breakers. <br> <br>detect most gases that contain chlorine, fluorine and bromine (halogen gases). <br> <br>detect cleaning agents used in dry cleaning applications such as per chloroethylene. <br> <br>detect halogen gases in fire-extinguishing system. <br> <br> <br>technical parameters: <br> <br>operating temperature: 0~52°c (30°f to 125°f) <br> <br>maximum sensitivity: 6 gr/yr, for all halogenated refrigerants<br> <br>battery life: approximately 50 hours normal use<br> <br>response time: instantaneous<br> <br>work mode: continuous, no limitation<br> <br>fixed probe length: 20cm<br> <br>unit dimensions: 1.37 x 1.37 x 5.90 in<br> <br>warm-up time: approximately 6 seconds<br> <br>reset time: two to ten seconds<br> <br>power supply: 6v dc four cell 7# excellent batteries<br>