## 5 Conclusions  
This study explains the importance of utilizing IoMT systems when monitoring patient information in a hospital setting, while collecting and comparing the data of different target variables using three different networks, which are the Sigfox, Hospital, and Non-Hospital networks, whilst monitoring the locations of infusion pumps, the temperatures of pharmaceutical refrigerators, and the body temperatures of nurses. Based on the findings drawn from an examination of the implementation of the IoMT system in the hospital, specific data, measurement, and user requirements should be taken into account when using the system for patient monitoring in the future. 1. Data requirements: The system must be able to accurately obtain temperature, acceleration, and gyro data. The gateway must comply with medical standards (HL7 and IEEE11073), and the reliability and security of the wireless network must be ensured when transmitting the measured patient's data. 2. Measurement requirements: When measuring body temperature, as the measurement cycle and set temperature value are different for each patient, it is necessary to ensure that the environment is appropriate for the patient and the ward condition and to make changes in cases of error. In addition, it is necessary to increase the reliability of the device by using a suitable battery in consideration of the average hospital stay (7 days) and the patient's biometric profile. 3. User requirements: Users of the IoMT system can improve the efficiency of service provision only when they understand the inconvenience of wearing a device, as well as the surgical and treatment schedule of each patient, the ward condition, and asset movement. A Design of Technology Element-based Evaluation Model and its Application on Checklist for the IoT Device Security Evaluation; [PERSON], [PERSON], published in 2018 was written in the position of developer and service proposer and covers inclusive and generalized security rather than that specific to each IoT, which makes it hard to \begin{table}
\begin{tabular}{c c c c} \hline \hline
**Network** & **Sigfox** & **Hospital Network** & \begin{tabular}{c} **Non-Hospital** \\ **Network** \\ \end{tabular} \\ \hline Accuracy & 69.2\% & 72.5\% & 83.3\% \\ \hline \multirow{3}{*}{Target (count)} & \multirow{3}{*}{Infusion pump (31)} & \begin{tabular}{c} Infusion pump (31), \\ pharmaceutical \\ refrigerator (2) \\ \end{tabular} & \
\begin{tabular}{c} Infusion pump (6), \\ Pharmaceutical refrigerator \\ (2), Nurse (12) \\ \end{tabular} \\ \hline \multirow{2}{*}{Sensor} & \multirow{2}{*}{Location} &
\begin{tabular}{c} Location, \\ Temperature \\ \end{tabular} & Location, Temperature \\ \hline Sensing cycle & Within 1 s & Within 1 s & Within 1 s \\ \hline Communication & & BLE to Wi-Fi \\ method & & & \\ \hline \hline \end{tabular}
\end{table} 
Table 10: Summary of Results According to the Network Method. identify the security of IoT for particular IoT, as a general user [14]. This study proposes a security evaluation model, based on the existing guidelines and related documents, for more specific IoT devices and proves that this approach is more convenient to ordinary users by creating checklists via the use of smart watch. IoMT amid the COVID-19 pandemic: Application, architecture, technology, and security; [PERSON] et al., published in 2021, suggests that the Internet of Medical Things (IoMT) has been deployed in tandem with other strategies to curb the spread of COVID-19, improve the safety of front-line personnel, increase efficacy by lessening the severity of the disease on human lives, and decrease mortality rates [15]. A number of on-going studies show that the adoption of secure IoMT applications is possible by incorporating security measures with the technology. The development of new IoMT technologies which merge with Artificial Intelligence, Big Data, and Blockchain offers more viable solutions.