## 3 Methods
### Data Acquisition and Analysis  
A custom multidimensional GC was used (Microanalytics, a part of Volatile Analysis Corporation, Round Rock, TX, USA), built on a standard Agilent 6890 platform (Agilent Technologies, Santa Clara, CA, USA). System automation and data acquisition software were MultiTrax v. 6.00.1 (Microanalytics, Round Rock, TX, USA) and ChemStation E.01.01.335 (Agilent Technologies, Santa Clara, CA, USA). Chromatography was performed on two capillary columns connected in series. The use of conventional retention indexes (e.g., Kovats retention index) is not appropriate for identification in this type of column configuration. The first column was 5% phenyl polysilphenylene-siloxane (30 m \(\times\) 0.53 mm inner diameter \(\times\) 0.5 \(\upmu\)m thickness, Trajan Scientific, Austin, TX, USA) with a fixed restrictor pre-column. The second polar column was bonded polyethylene glycol in a Sol-Gel matrix (30 m \(\times\) 0.53 mm inner diameter \(\times\) 0.5 \(\upmu\)m thickness, Trajan Scientific, Austin, TX, USA). The midpoint between the two columns was maintained at a constant pressure of 0.39 atm by a pneumatic switch. In this research, all effluent from the first column was directed into the second analytical column, that is, no heartcutting was performed. True multidimensional analyses were not performed (i.e., the system was used in full heartcut mode), meaning separation was performed on both columns in series. Effluent from the second polar column was simultaneously directed to a single quadrupole mass spectrometer (MS) (Model 5973N, Agilent Technologies, Santa Clara, CA, USA) and an olfactory (sniff) port via an open spit interface at atmospheric pressure. The sniff port was equipped with a purge flow controller and supplied with humidified air at 0.54 atm. Flow to the MS and sniff port was determined by fixed restrictor columns, one part to the MS and three parts to the sniff port. Olfactory was not utilized in the research. The GC inlet was operated in splitless mode at 250 \({}^{\circ}\)C. GC oven parameters started with an initial temperature of 40 \({}^{\circ}\)C, held for 3.0 min, followed by a 7 \({}^{\circ}\)C per min ramp to 240 \({}^{\circ}\)C, held for 8.43 min. Total run time was 40 min. Carrier gas was ultra-high purity (UHP) helium (99.999%, Airgas, Des Moines, IA, USA). Temperatures of the sniff port and MS transfer lines were 240 \({}^{\circ}\)C and 280 \({}^{\circ}\)C, respectively. MS full scan range was set from 34 m/z to 350 m/z. Scans were collected in electron ionization (EI) mode with an ionization energy of 70 eV. MS heated zones for quadrupole and source were 150 \({}^{\circ}\)C and 230 \({}^{\circ}\)C, respectively. Daily tuning of the MS was performed with perfluorotributylamine (PFTBA) before each analysis.  
Identification of compounds was performed using the Automated Mass Spectral Deconvolution & Identification System (AMDIS) target library search with at least 80% mass spectral match [8]. Target libraries included (a) the 6 libraries that are included with the AMDIS program, (b) an onsite library created from analysis of pure standards (200+ compounds), and (c) NIST11 mass spectral library. Retention times were also verified with pure standards.