TITLE: The design and manufacture of massively scalable inertial focusing prototype microfluidic devices
AUTHORS: Thomas Carvell

[DESCRIPTION]
Microfluidics is a rapidly expanding field and microfluidic devices have been used in a variety of biomedical applications such as cell sorting, disease diagnostics and various lab-on-a-chip systems. There is great demand for manufacturing techniques capable of fabricating ever more intricate microstructures for increasingly complex applications whilst remaining cost-effective for use in biomedical research. Conventional manufacturing techniques can be used to fabricate many complex microchannel architectures but are often expensive, low throughput, have poor microfeature resolution or are unsuitable to be used at scale. To address this issue, we describe a manufacturing technique that employs stereolithography 3D printing to produce a base substrate which can be sealed with a laser patterned adhesive layer cover and stacked in a compact configuration. The technique provides for the design and manufacturing of a device that can be
massively parallelized at relatively low cost and with a small laboratory footprint. This paper aims to fully explain the design and manufacturing process undertaken to allow the use of this technique in future research.

[BEFORE_START]
Read the entire protocol before starting. Note: section 1 describes the design process for a specific, but exemplar, microfluidic device. Steps 1.6 to 1.33 can be modified as appropriate for other applications. 

The characterisation of the device was performed to provide representative results only and does not need to be followed. This will highly depend on the final application of the device that can be manufactured using this protocol. 

[GUIDELINES]
Computer-aided design (CAD) software should be used to design both the 3D-printed and laser patterned parts. 
When using the stereolithography 3D-printer, follow the manufacturer’s instructions and any local safety regulations.
When using a CO2 laser cutter, follow the manufacturer’s instructions and any local safety regulations.


[STEPS]
SECTION: Design of exemplar parts
1. Open computer-aided design software and ensure 3D-modelling format is
selected and units are set to millimetres.

SECTION: Design of exemplar parts
1.1.       Select the rectangle tool and enter x-y coordinates for origin (0, 0). The width is 35 and the height is 70. This generates the ‘device rectangle’.
SECTION: Design of exemplar parts
1.2.       Save the device rectangle as ‘adhesive tape cover’ using the .dxf file type. Note: no further modifications to this file are needed. 
SECTION: Design of exemplar parts
1.3.       Save the device rectangle as ‘3D printed base’. 
SECTION: Design of exemplar parts
1.4.       Open the ‘3D printed base’ CAD file. 
SECTION: Design of exemplar parts
1.5.       Select the line tool and draw a line (line A) from the midpoint of the top side of the rectangle at 90° and 15 in length. 
SECTION: Design of exemplar parts
1.6.       Repeat step 1.3 for the bottom side at 90° and 15 in length to generate line B. 
SECTION: Design of exemplar parts
1.7.       Use the line tool and draw a line (line C) of 0.25 in length and at an angle of 180° relative to the central endpoint of line A. 
SECTION: Design of exemplar parts
1.8.       Use the line tool and draw a line (line D) of 0.25 in length and at an angle of 0° relative to the central endpoint of line B.
SECTION: Design of exemplar parts
1.9. Use the rectangular tool to draw a rectangle from the non-centred endpoints of line C and line D. Note: the rectangle (microchannel rectangle) should be 40 in length and 0.5 in width.
SECTION: Design of exemplar parts
1.10. Delete lines A, B, C and D. 
SECTION: Design of exemplar parts
1.11. Using the line tool, draw a line (135° angle, 10 in length) from the midpoint of the top side of the microchannel rectangle. 
SECTION: Design of exemplar parts
1.12. Draw another line (45° angle, 10 length) from the same midpoint as in step 1.12.
SECTION: Design of exemplar parts
1.13. Draw two lines from the bottom of the microchannel rectangle and extend these to the device rectangle side (both lines, 90°, 15 in length). 
SECTION: Design of exemplar parts
1.14. Use the offset tool to generate parallel lines (0.25 distance away) to the lines drawn in steps 1.12 and 1.13. There will be some lines that overlap with the microchannel rectangle, but a Y-shaped channel will be displayed. 
SECTION: Design of exemplar parts
1.15. Use the trim tool to remove any lines inside the Y-shaped microchannel and remove all parts of the microchannel rectangle drawing that protrude past the offset lines. This should generate a central channel with an inlet channel and 2 smaller outlet channels at a Y-shaped angle.
SECTION: Design of exemplar parts
1.16. Use the line tool and draw a line (90° angle, 10 in length) from each of the outlet channel lines. These will extend past the edge of the device rectangle.
SECTION: Design of exemplar parts
1.17. Use the trim tool to remove all extended lines drawn in step 1.16. 
SECTION: Design of exemplar parts
1.18. Connect the end of all channels. These should be 0.1768 in length for the smaller outlet channels and 0.5 in length for the inlet channel.
SECTION: Design of exemplar parts
1.19. Select all lines except for the device rectangle and use the join command. This should join 14 lines into one polyline.
SECTION: Design of exemplar parts
1.20. Select the device rectangle and extrude by 3, below the plane of the Y-shaped channel. This forms the device cuboid. 
SECTION: Design of exemplar parts
1.21. Select the Y-shaped polyline and extrude by 0.15 below the plane of the polyline.
SECTION: Design of exemplar parts
1.22. Use the Face tool and select the cross-section of the microchannel.
SECTION: Design of exemplar parts
1.23. Use the rectangle tool to draw a rectangle (with the same dimensions as the relevant channel cross-section) on each face of the end of the inlet and outlet channels.
SECTION: Design of exemplar parts
1.24. Draw a line below the channel plane (0.9 length, angle 90°) from each of the center-points of the 3 rectangles drawn in step 1.24.
SECTION: Design of exemplar parts
1.25. Delete each of the rectangles drawn in step 1.24.
SECTION: Design of exemplar parts
1.26. At the bottom endpoint of each of the three lines (drawn in step 1.25) draw a circle with 0.9 radius. 
SECTION: Design of exemplar parts
1.27. Delete the lines drawn in step 1.25.
SECTION: Design of exemplar parts
1.28. Select each of the circles from step 1.24 and use the extrude tool to extrude the circle by 5 towards the centre of the device to form cylinders. 
SECTION: Design of exemplar parts
1.29. Select the subtract tool and left click the device cuboid.
SECTION: Design of exemplar parts
1.30. Right click anywhere outside of the design.
SECTION: Design of exemplar parts
1.31. Left click the Y-shaped channel and the cylinders. 
SECTION: Design of exemplar parts
1.32. Right click anywhere outside of the design. 
SECTION: Design of exemplar parts
1.33. To confirm this has successfully removed the cylinders and channel from the device cuboid, select the X-ray visual style and rotate the design to visually inspect that all flow channels are connected.
SECTION: Design of exemplar parts
1.34. Click Ctrl+A to highlight the design and export as an .stl file for use with the 3D-printer.
SECTION: Manufacturing of base through stereolithography 3D-printing
2. Ensure the 3D-printer is clean and ready to use as per the manufacturer's instructions.
SECTION: Manufacturing of base through stereolithography 3D-printing
2.1. Upload a stereolithography file (.stl) to the associated 3D-printing software.
SECTION: Manufacturing of base through stereolithography 3D-printing
2.2. Orientate the print vertically with the ports running parallel to the Z axis. 
SECTION: Manufacturing of base through stereolithography 3D-printing
2.3. Where required, attach support structures to  the design. Note: it is important for no supports to be attached to the cover side of the design and supports must not obscure the ports. 
SECTION: Manufacturing of base through stereolithography 3D-printing
2.4. Attach the build platform to the 3D printer. 
SECTION: Manufacturing of base through stereolithography 3D-printing
2.5. Insert the resin tank and wiper. 
SECTION: Manufacturing of base through stereolithography 3D-printing
2.6. Insert the resin cartridge.
SECTION: Manufacturing of base through stereolithography 3D-printing
2.7. The display should confirm the tank and cartridge have been inserted correctly. 
SECTION: Manufacturing of base through stereolithography 3D-printing
2.8. The 3D printer should dispense resin into the tank and heat it to the required temperature.
SECTION: Manufacturing of base through stereolithography 3D-printing
2.9. Once the machine is ready, click print.
SECTION: Manufacturing of base through stereolithography 3D-printing
2.10. Fill a sample bag with 70% isopropanol (IMS).
SECTION: Manufacturing of base through stereolithography 3D-printing
2.11. When the machine has finished printing, carefully remove the 3D-printed part, and immediately place the part in the sample bag from step 2.10, ensuring that IMS flows over the part. Note: work quickly and avoid direct light exposure during steps 2.11-2.13.
SECTION: Manufacturing of base through stereolithography 3D-printing
2.12. Place the sample bag on a rocking platform for 10 minutes and then discard the IMS/resin mixture.
SECTION: Manufacturing of base through stereolithography 3D-printing
2.13. Use a pipette and eject IMS directly onto the open-faced microchannel to help clear residual resin.
SECTION: Manufacturing of base through stereolithography 3D-printing
2.14. Refill the sample bag with IMS and repeat step 2.11-2.13 before continuing with step 2.15.
SECTION: Manufacturing of base through stereolithography 3D-printing
2.15. Whilst under a fume hood, allow the IMS to evaporate fully and then expose the part in sunlight for a 24-hour period to allow for complete polymerization of the resin. 
SECTION: Manufacturing of laser patterned adhesive cover
3. Place the adhesive tape into the laser system bed.
SECTION: Manufacturing of laser patterned adhesive cover
3.1. Ensure the adhesive tape is lying flat and level. 
SECTION: Manufacturing of laser patterned adhesive cover
3.2. The laser system should be focused to a height that is appropriate for the thickness of the material. Note: this is likely to be highly system-dependent and the user should consult the laser system manual for the appropriate recommended laser focusing parameter.
SECTION: Manufacturing of laser patterned adhesive cover
3.3. The CAD design should be uploaded to the relevant software.
SECTION: Manufacturing of laser patterned adhesive cover
3.4. The laser parameters used are frequency: 5000Hz, power: 5%, speed: 15% and 10 passes selected.  Note: If an incomplete cut is observed, adjust the number of passes. 
SECTION: Manufacturing of laser patterned adhesive cover
3.5. Turn on the valve to venting of any generated gases. 
SECTION: Manufacturing of laser patterned adhesive cover
3.6. Click print on the software.
SECTION: Manufacturing of laser patterned adhesive cover
3.7. Click the GO button on the laser cutter.
SECTION: Manufacturing of laser patterned adhesive cover
3.8. The adhesive cover should be rectangular in shape and the edges of the cut should be smooth.
SECTION: Assembly of the device
4. Polytetrafluorethylene (PFTE) tubing should be inserted into the ports of the 3D-printed base layer. Note: ensure that the tubing is ~1 mm away from the start of the flow channels.
SECTION: Assembly of the device
4.1. Superglue is used to liberally cover the PFTE tubing 2 mm away from the end of the tubing, sealing it into the ports. Note: ensure the tubing is not sealed shut with glue. 
SECTION: Assembly of the device
4.3. The backing of one side of the adhesive cover is removed and the adhesive layer is placed over the base layer to seal the tubing into the ports and close the open face of the microchannel.
SECTION: Assembly of the device
4.4. Run a piece of rigid material along the face of the cover to seal the device closed and remove any air bubbles. 
SECTION: Assembly of the device
4.2. Allow the superglue to fully polymerize for 24 hours.
SECTION: Assembly of the device
4.7. Attach the tubing to the appropriate syringe pumping systems.
SECTION: Assembly of the device
4.8. With a successful assembly, the injection of dH20 at the inlet does not cause a leakage and all fluid should flow through the outlet(s).  
SECTION: Assembly of the device
4.9. OPTIONAL: At step 4.5, it is possible to remove the secondary backing of the adhesive cover to expose the upper adhesive side. Another IF device base layer can be stacked on top and the assembly repeated, or a substrate support such as polymethyl methacrylate can be used to increase the robustness of the device.
SECTION: Assembly of the device
4.5. Place the sealed device in a clamp and apply 1 tonne pressure for 3 minutes. 
SECTION: Assembly of the device
4.6. Allow 24 hours for the pressure sensitive tape to adhere. 
SECTION: Characterization of the device
5. The following steps 5.1 - 5.8 are optional are provided as exemplar characterisation steps. 
SECTION: Characterization of the device
5.1. Vortex the neat particle suspension for 30 seconds. 
SECTION: Characterization of the device
5.2. Add 100 µL of neat particle suspension to 49.9 mL of phosphate-buffered saline.
SECTION: Characterization of the device
5.3. Use a cell strainer with 40 µm pores and pipette the particle suspension through and count to determine the concentration.
SECTION: Characterization of the device
5.4. Prime the microchannel by injecting filtered PBS at the inlet. 
SECTION: Characterization of the device
5.5. Inject the filtered particle suspensions at the inlet of the IF device and wait for fluid to exit the outlet tubing (this is to allow the particles to reach equilibrium focusing positions).
SECTION: Characterization of the device
5.6. Collect the processed samples in microcentrifuge tubes at the outlets.
SECTION: Characterization of the device
5.7. The inlet and outlets samples are quantified for particle concentration. This can be performed using a haemocytometer, flow cytometer or other cell/particle counting system (not described in this protocol).  
SECTION: Characterization of the device
5.8. The separation efficiency, defined as the capacity of the device to separate particles to a specific outlet and can be calculated using the following equation: [outlet sample X] / [total outlet sample] = separation efficiency (%).