TITLE: Analysis of ethylene glycol and 1,4-butanediol by HPLC-RID
AUTHORS: Hannah M. Alt, Alexander F. Benson, Stefan J. Haugen, Morgan A. Ingraham, William E. Michener, Sean P. Woodworth, Kelsey J. Ramirez, Gregg T. Beckham

[DESCRIPTION]
An analytical method was developed using high performance liquid chromatography with refractive index detection (HPLC-RID) to quantify the concentration of ethylene glycol and 1,4-butanediol in aqueous samples. This list is non-exhaustive as this isocratic method with an acid modified mobile phase is compatible with analysis of many alcohols, small acids, and sugars. This method utilizes a Bio-Rad Aminex HPX-87H Ion Exclusion Column to provide chromatographic separation. 


[BEFORE_START]
All solvents and chemicals used are listed in the ‘Materials’ section. These are excluded from in-line references to maintain clarity and keep the steps concise.  


[GUIDELINES]
This protocol utilizes an high pressure liquid chromatography refractive index detector (HPLC-RID) system manufactured by Agilent Technologies as referenced in 'Materials'. A similar HPLC-RID system can be utilized; however, some parameter nomenclature may deviate depending on the manufacturer. 

[STEPS]
SECTION: Preparation of standards
3. Standards

Prepare a 10 mg/mL ethylene glycol stock standard by weighing 50 mg of the ethylene glycol raw material into a 20 mL amber vial.  Record the weight of the raw material to the nearest 0.1 mg.  Add 5 mL of UPW to amber vial and vortex thoroughly until raw material is dissolved.
Repeat step 1 in order to create a 10mg/mL 1,4-butanediol stock standard.
Make separate calibration curves for ethylene glycol and 1,4-butanediol by following the calibration table below.

 

 

SECTION: HPLC-RID analysis
5. Method Specifications

Analysis is performed using an Agilent 1200 Series High Performance Liquid Chromatography (HPLC) system. An isocratic concentration of 0.01N sulfuric acid through an Aminex HPX-87H column (300 x 8.7 mm, 9 μm particle size) is used to achieve separation at a flow rate of 0.6 mL/min. Quantitation is determined using refractive index detection (RID). Column and RID are both held constant at 55 °C and each sample and standard is injected at a volume of 6.0 μL.

 
Retention time of analytes is dependent on the configuration of the HPLC and will vary from instrument to instrument. Retention time markers for each analyte should be run individually to assess the total required run time of the analysis based on elution. Additional analytes not listed in this protocol may be compatible with these instrument parameters. This re-emphasizes the need to run single analyte retention time markers to prevent co-eluting peaks.  The method run time is 27 minutes.

 

SECTION: Preparation of mobile phase and instrument equilibration
1. Mobile phases

To make 0.01 N sulfuric acid (0.005 M), dilute 1.0 mL of 10 N sulfuric acid into 1.0 L of 18.2MΩ⋅cm ultrapure water (UPW). Volumetric preparation of this mobile phase will yield the most reproducible chromatography. See note below.

 
 
SECTION: Preparation of mobile phase and instrument equilibration
2. Instrument equilibration

Purge instrument with 0.01 N sulfuric acid solution made in the previous step. Be certain the instrument is purged through the entire flow path including the detector, before the analytical column is plumbed in. The Aminex 87H column is sensitive to solvents, for example, a higher amount of methanol will damage the column. The purge step is needed to remove all solvents/mobile phases from the analytical system. (See Bio-Rad's 'Instruction Manual' for Aminex resin-based columns for solvent compatibility and installation details).
Add the guard column and analytical column to the system and begin equilibrating the column at a low flow rate of 0.2 mL/min while the column reaches analysis temperature.
During the column equilibration, begin purging the reference cell of the refractive index detector (RID), this process will continue through the final equilibration of the column.
At intervals of at least 10 minutes, increase the flow by 0.2 mL/min until you reach the method flow of 0.6 mL/min.
Once the column is up to method flow and both the column compartment and the RID are at method temperature and stable, the RID reference cell can be closed. This typically takes around 30 minutes after method flow is reached. A longer purge of the reference cell is not detrimental.
After the reference cell is closed, wait for the RID signal to stabilize before starting analysis.
SECTION: Analytical Quality Control
6. Multiple strategies are utilized when performing this analysis to ensure instrument stability and reproducibility. 
SECTION: Analytical Quality Control
6.1. Calibration Curves

All compounds must have a correlation coefficient (r2) of 0.995 or greater using a linear calibration fit and ignore the origin.
SECTION: Analytical Quality Control
6.2. Calibration Verification Standards (CVS)

A calibration verification standard (CVS) is a level provided by the manufacturer that is re-analyzed every 20 or fewer samples to ensure instrument drift remains within the determined acceptance criteria. Acceptable CVS recoveries for this analysis are within 10% of the expected amount. Acceptance criteria may differ between instruments and should be determined experimentally.
SECTION: Example Chromatography
7.      


 












SECTION: Preparation of samples
4. Samples
Samples must be filtered through a 0.2 µm or smaller filter prior to injection on the HPLC ('Materials' section includes part numbers for filters to use depending on matrix composition)    
Samples expected to be over the linear range of the instrument should be diluted to ensure accurate analysis and avoid carryover.