Commission communication in the framework of the implementation of Directive 2004/22/EC of the European Parliament and of the Council on measuring instruments (Publication of the references to normative documents drawn up by OIML and the list of the parts thereof corresponding to the essential requirements (in conformity with Article 16.1 of the Directive)) (Text with EEA relevance)
Date: 2009-11-10
Institutions: European Commission
Subjects: harmonisation of standards, international standard, measuring equipment, metrology, technical standard








C_2009268EN.01000101.xml










10.11.2009   


EN


Official Journal of the European Union


C 268/1





Commission communication in the framework of the implementation of Directive 2004/22/EC of the European Parliament and of the Council on measuring instruments (1)


(Publication of the references to normative documents drawn up by OIML and the list of the parts thereof corresponding to the essential requirements (in conformity with Article 16.1 of the Directive))

(Text with EEA relevance)
2009/C 268/01
MI-002:   Gas meters

in relation to:






—


OIML R 137-1 2006







Essential requirements in MID (Annex I and MI-002)


OIML R 137-1 (2006)


Comment




Annex I


 


 




1.1


5.3.1


Covered




1.2


7.4.15 table 6, A.1


Covered




1.3


7.3.5


Covered




1.3.1


5.1, A.4.2.1, A4.2.2


Covered




1.3.2


 


 




(a) M1
(a) M2


7.4.13, A.5.1, A.5.2


Covered




(a) M3


 


Not relevant




(b)


7.4.13


Covered for meter up to 10 kg




1.3.3


 


 




(a) E1


7.4.15, A.6.1.1, A.6.1.2, A.6.2, A.6.3, A.6.4, A.7.3, A.7.4, A.7.5


Covered




(a) E2


 


Not relevant




(a) E3


 


Not relevant




(b)


7.4.15


Covered




1.3.4


7.4.15, A.7


Covered for voltage variation




1.4


 


 




1.4.1


5.3.1, 7.4, A.1


Covered for class 0.5 or if MPE are fulfilled during application of influence quantities




1.4.2


A.4.2.1, A.4.2.2


Covered




2


 


Not relevant




3


7.4.4


Covered




4


6.1.3


Covered




5


7.4.9


Covered




6


3.1.1, 6.5


Covered




7


 


 




7.1


3.1.4, 4.3.1


Covered




7.2


3


Covered




7.3


7.4.11


Covered for mechanical meters




7.4


 


Not relevant




7.5


3.1.1, 3.1.4


Covered




7.6


6.2.1, 7.3.5, 7.4.16, 4.2.4.x


Covered except for test procedure




8


 


 




8.1


6.3.1, 4.3.4.2


Covered




8.2


4.3.14.3.3


Covered




8.3


7.4.16, 4.2.4, 4.3.4.1


Covered




8.4


4.3.1, 7.4.1.16


Covered




8.5


6.1.1


Covered




9


 


 




9.1


4.2.1


Covered except for the information about additional devices




9.2


4.2.1


Covered




9.3


4.2


Covered for information on data plate




9.4


 


Not relevant




9.5


6.2.2, 6.1.1


Covered




9.6


 


Not relevant




9.7


4.1


Covered




9.8


4.2


Covered




10


 


 




10.1


6.1.1


Covered




10.2


6.1.1


Covered




10.3


2.1.10, 6.3


Covered




10.4


 


Not relevant




10.5


3.1.5


Covered




11


 


 




11.1


 


Not relevant




11.2


 


Not relevant




12


 


Covered




Annex MI-002


 


 




Definitions


2.1.1, 2.3.3, 2.3.2, 2.3.4, 2.4.4 and 7.4.11


Covered




1


7.3.5


covered




1.1


5.2


Covered for class 1.5 if Qmax/Qmin ≥ 150
Covered for class 1.0 if Qmax/Qmin ≥ 20




1.2


 


Not covered




1.3


5.1


Covered




1.4


5.1


Covered except for three temperature ranges of OIML R137-1




1.5


4.2.4


Covered




2


 


 




2.1


5.3.3


Covered except for the requirement on errors of same sign in 2.1, Annex MI-002




2.2


5.3.4


Covered if selected temperature lies between 15 °C and 25 °C




3


 


 




3.1


 


 




3.1.1


7.4.15


Covered




3.1.2


7.4.15


Covered




3.1.3


7.4.15, 2.2.10


Covered for the first clause of the requirement




3.2


7.4.8


Covered




4


 


 




4.1


 


 




4.1.1


7.4.9


Covered




4.1.2


7.4.9


Covered




4.2


 


 




4.2.1


7.4.9


Not covered




4.2.2


7.4.9


Not covered




5


 


 




5.1


6.4.2


Not covered




5.2


6.4.4


Covered by one of the alternatives




5.3


6.1.2


Not covered




5.4


3.4, 7.4.5


Covered




5.5


6.2.1


Covered




5.6


3.2, 7.4.5


Covered




6


4.1


Covered if kg or m3 is used





Note:

The column ‘Comment’ indicates the compliance between OIML R 137-1 and the relevant requirement in Directive 2004/22/EC.
The indication ဘCovered’ means that:






—


the requirement of OIML R 137-1 is identical to the one of MID, or










—


the requirement of OIML R 137-1 is more severe than the one of MID, or










—


all of the requirement of OIML R 137-1 fulfils MID requirements (even when MID allows other alternatives),










—


in case the requirement is not fully covered, a short statement explains what is covered.




The indication ‘Not covered’ means that the MID requirement is either not compatible with the relevant OIML R 137-1 requirement or not included in OIML R 137-1.
The indication ‘Not relevant’ means that the MID Annex I requirement is not relevant for gas meters.








MI-002:   Volume conversion devices

in relation to:






—


OIML R 140 2007







Directive 2004/22/EC
Essential requirements of Annex I and Annex MI-002


OIML R 140 (2007)


Comment




Annex I


 


 




1.1


T.2.6, T.3.4, 9.1.1


Covered




1.2


T.2.16, T.2.17.2, 9.1.2, A.1


Covered for disturbances




1.3


T.1.22, 7.1.3, A.2, A.4


Covered provided the manufacturer specify all needed information




1.3.1


7.1.3, A.4.1, A.4.2, T.1.22


Covered




1.3.2 (a)


 


 




M1
M2


7.1.3, A.4.4


Covered for vibrations




M3


 


Not relevant




1.3.2 (b)


7.1.3.1, A.4.4


Covered for vibrations




1.3.3 (a)


 


 




E1


7.1.3, 10.2.8, A.4, A.4.5.a, A.4.5.b, A.4.6, A.4.7, A.4.8, A.4.11, A.4.12, A.4.13


Covered




E2


 


Not relevant




E3


 


Not relevant




1.3.3 (b)


A.4


Covered, except short interruptions in the case of DC




1.3.4


6.4.9, A.4, A.4.9, A.4.10


Covered except for mains frequency variations and power frequency magnetic fields




1.4


 


 




1.4.1


A.1, A.4


Covered if modular approach is used




1.4.2


A.4.3


Covered




2


 


Not relevant




3


 


Not covered




4


8.2.1.3, 8.6.1.3


Covered




5


9.1.1, 9.1.2, 9.1.3


Covered




6


9.1.1, 9.1.2, 9.1.3, 9.1.5, 9.3


Covered




7


 


 




7.1


7.6


Covered




7.2


7.3.4, 8.5, 8.6, 8.7


Covered




7.3


9.3.5, 9.3.6


Covered




7.4


 


Not relevant




7.5


8.6.1.1, 8.6.1.2


Covered




7.6


9.3, 10.2.3


Covered except for test procedure




8


 


 




8.1


7.1.1.1, 7.1.1.2, 7.3.2, 8.6.1.4, 8.6.1.5


Covered




8.2


7.6


Covered




8.3


7.6.2, 10.2.3


Covered except providing the identification of software on the instrument




8.4


7.6.2


Covered




8.5


7.1.4.6, 8.7.2


Not Covered




9


 


 




9.1


7.5.1, 7.5.2, 7.1.1.2


Covered for information on data plate




9.2


 


Not relevant




9.3


7.5.1, 10.2.3, 10.2.8.2


Covered




9.4


 


Not relevant




9.5


8.2.1.2, 7.1.4.1


Covered




9.6


 


Not relevant




9.7


5.1


Covered




9.8


7.5.1


Covered




10


 


 




10.1


8.2.1.1


Covered




10.2


7.1.4.1, 7.1.4.3, 8.2.1.1


Covered




10.3


T.1.5, 8.3, 8.4


Covered




10.4


 


Not relevant




10.5


 


Covered provided that the volume conversion device is equipped with a local display




11


 


 




11.1


 


Not relevant




11.2


7.1.4.6, 8.7.2, 9.1.5


Covered




12


 


Covered




Annex MI-002


 


 




Definitions


1.1, 2, 6.3, 7.3, 8.5, 8.6


Covered




Gas meter
Conversion device
Minimum flowrate
Maximum flowrate
Transitional flowrate
Overload flowrate
Base conditions


T.1.9, T.1.12, 7.3.2.1, T.1.14, 7.3.1


Covered




1


7.1.2.1, 8.6.1.1, 9.3.6


Not covered




1.1


 


Not relevant




1.2


 


Not covered




1.3


7.3, 7.5.1


Covered




1.4


7.1.3, A.4.1, A.4.2, T.1.22


Covered




1.5


7.1.3, A.4, A.4.9, A.4.10, T.1.22


Covered




2


 


 




2.1


 


Not relevant




2.2


 


Not relevant




3


 


 




3.1


 


 




3.1.1


T.2.17.2, T.2.17.4, 9.1.2, 9.1.3, 9.1.4


Covered




3.1.2


9.1.1, 9.1.2, 9.1.3, 9.1.5, 9.3.6


Covered




3.1.3


T.2.17.2


Covered




3.2


 


Not relevant




4


 


 




4.1


 


 




4.1.1


 


Not relevant




4.1.2


 


Not relevant




4.2


 


 




4.2.1


 


Not relevant




4.2.2


 


Not relevant




5


 


 




5.1


9.2.1


Not covered




5.2


9.2.2


Covered when the battery has a 5 years lifetime




5.3


 


Not relevant




5.4


 


Not relevant




5.5


 


Not relevant




5.6


 


Not relevant




6


5.1


Covered except when mass is expressed in tonnes




7


7.3.1, 10.2.2


Covered




8


6.3.3


Covered provided that ambient temperature is managed such that it remains within ± 3 °C at reference conditions and except for OIML R 140 class C




9


 


 




9.1


9.3.6


Covered




9.2


7.1.4, 8.2, 8.7.1


Covered only for parameters and provided that the conversion device is equipped with a local display





Note:

The column ‘Comment’ indicates the compliance between OIML R 140 and the relevant requirement in Directive 2004/22/EC.
The indication ‘Covered’ means that:






—


the requirement of OIML R 140 is identical to the one of MID, or










—


the requirement of OIML R 140 is more severe than the one of MID, or










—


all of the requirement of OIML R 140 fulfils MID requirements (even when MID allows other alternatives),










—


in case the requirement is not fully covered, a short statement explains what is covered.




The indication ‘Not covered’ means that the MID requirement is either not compatible with the relevant OIML R 140 requirement or not included in OIML R 140.
The indication ‘Not relevant’ means that the MID Annex I requirement is not relevant for conversion devices.








MI-007:   Taximeters

in relation to:






—


OIML R 21 2007







Essential requirements of Annex I and Annex MI-007


OIML R 21 (2007)


Comment




Annex I


 


 




1.1


2.4.5.4, 2.5.5


Covered




1.2


A.5.4.5


Covered




1.3


3.5.2


Covered




1.3.1


3.5.1, 5.1.2


Covered




1.3.2(a)


 


 




M1


 


Not relevant




M2


 


Not relevant




M3


A.5.4.4


Covered




1.3.2(b)


 


Covered




1.3.3(a)


 


 




E1


 


Not relevant




E2


 


Not relevant




E3


A.5.4.3, A.5.4.5, A.5.4.6, A.5.4.7


Covered when the field strength is 24 V/m




1.3.3(b)


Annex A


Covered




1.3.4


Annex A


Covered




1.4


 


 




1.4.1


A.5.1


Covered




1.4.2


5.1.2


Covered




2


7.5


Covered




3


5.2.6


Covered




4


 


Covered




5


4.1


Covered




6


3.3


Covered




7


 


 




7.1


4.2.1


Covered




7.2


4.1, 4.9.1


Covered




7.3


 


Not relevant




7.4


 


Not relevant




7.5


4.1


Covered




7.6


4.2.3, 4.11


Covered when the test procedure is included in the operation manual




8


 


 




8.1


5.2.3.2


Covered




8.2


4.2.5


Covered




8.3


4.11.2.


Covered




8.4


4.10


Covered




8.5


 


Not relevant




9


 


 




9.1


4.12, 4.12.1


Covered




9.2


 


Not relevant




9.3


 


Not covered




9.4


 


Not relevant




9.5


 


Not relevant




9.6


 


Not relevant




9.7


3.4, 4.9.1


Covered




9.8


4.12.2


Covered




10


 


 




10.1


4.9.1


Covered




10.2


4.9.1


Covered




10.3


4.9.2


Covered




10.4


4.9.1


Covered




10.5


 


Not relevant




11


 


 




11.1


 


Not relevant




11.2


 


Not relevant




12


4.2.3


Covered




Annex MI-007


 


 




Definitions


 


 




Taximeter


2.1.1


Covered




Fare


2.3.1.1


Covered




Cross-over speed


2.3.1.2


Covered




Normal calculation mode S (single application of tariff)


2.3.1.3.1


Covered




Normal calculation mode D (double application of tariff)


2.3.1.3.2


Covered




Operating position


2.3.3, 2.3.3.1, 2.3.3.2, 2.3.3.3


Covered




1


3.1


Covered




2


2.3.3.2, 2.3.3.3, 3.1


Covered




3


4.3


Covered




4


5.2.3, 4.2.2


Covered




5


4.2.3, 4.2.5


Covered




6.1


A.5.4.4


Covered




6.2


3.5.1, 3.5.2


Covered




7


3.2.1.1


Covered




8


 


 




8.1


See Answer to Annex I § 1.3.3


Covered




8.2


5.1.3


Covered




9


5.2.5


Covered




10


3.1


Covered




11


4.5.3(b)


Covered




12


4.5.4


Covered




13


4.9.1


Covered




14.1


4.2.4


Covered




14.2


4.2.5


Covered when 4.2.5.i of OIML R 21:2007 is mandatory




14.3


See answer to Annex I § 8.3


Covered




15.1


4.7


Covered




15.2


4.7


Covered




15.3


4.9.1


Covered




16


4.8


Covered




17


4.1, 4.14.1


Covered




18


5.2.4, 3.2.1.1


Covered




19


4.14.1, 4.2.1


Covered




20


4.2.1


Covered




21


3.3


Covered




22


3.7


Covered




23


3.4


Covered





Note:

The column ‘Comment’ indicates the compliance between OIML R 21 and the relevant requirement in Directive 2004/22/EC.
The indication ‘Covered’ means that:






—


the requirement of OIML R 21 is identical to the one of MID, or










—


the requirement of OIML R 21 is more severe than the one of MID, or










—


all of the requirement of OIML R 21 fulfils MID requirements (even when MID allows other alternatives),










—


in case the requirement is not fully covered, a short statement explains what is covered.




The indication ‘Not covered’ means that the MID requirement is either not compatible with the relevant OIML R 21 requirement or not included in OIML R 21.
The indication ‘Not relevant’ means that the MID Annex I requirement is not relevant for taxi meters.








MI-008 CHAPTER II:   Capacity serving measures

in relation to:






—


OIML R 138 2007







Essential requirements in MID (Annex I and Annex MI-008 II)


OIML R 138 (2007)


Comment




Annex I


 


 




1.1


5.1


Covered




1.2


 


Not relevant




1.3


 


Not relevant




1.3.1


 


Not relevant




1.3.2(a)


 


Not relevant




1.3.2(b)


 


Not relevant




1.3.3(a)


 


Not relevant




1.3.3(b)


 


Not relevant




1.3.4


 


Not relevant




1.4


 


 




1.4.1


 


Not relevant




1.4.2


 


Not relevant




2


 


Not covered




3


 


Not covered




4


4.4


Covered




5


4.2


Covered




6


 


Not covered




7


 


 




7.1


 


Not covered




7.2


4.2


Covered




7.3


 


Not relevant




7.4


 


Not relevant




7.5


4.2


Covered




7.6


 


Not relevant




8


 


 




8.1


 


Not relevant




8.2


 


Not relevant




8.3


 


Not relevant




8.4


 


Not relevant




8.5


 


Not relevant




9


 


 




9.1


5.2.4, 5.2.1, 5.2.5


Covered when the reference temperature is 20 °C




9.2


 


Not relevant




9.3


 


Not relevant




9.4


 


Not relevant




9.5


 


Not relevant




9.6


5.2.1


Covered




9.7


3


Covered when units are SI units




9.8


5.2.1


Covered




10


 


 




10.1


 


Not relevant




10.2


5.2.1


Covered




10.3


 


Not relevant




10.4


 


Not relevant




10.5


 


Not relevant




11


 


 




11.1


 


Not relevant




11.2


 


Not relevant




12


 


Covered




Annex MI-008 Chapter II


 


 




Definitions


 


 




Capacity serving measure


2.2


Covered




Line measure


2.9


Covered when the gauge mark is a line




Brim measure


2.7


Covered




Transfer measure


2.2


Covered




Capacity


2.7, 2.5, 2.9, 4.6.1


Covered




1


 


 




1.1


5.2.5


Covered when the reference temperature is 20 °C




1.2


4.5


Covered




2


5.1.1, 4.1.1


Covered




3


4.2


Covered




4


 


 




4.1


4.4


Covered




4.2


 


Not covered




5


 


 




5.1


5.2.1


Covered




5.2


5.2.2


Covered when there are no more than 3 capacities marked




5.3


4.2, 5.2.2


Covered





Note:

The column ‘Comment’ indicates the compliance between OIML R 138 and the relevant requirement in Directive 2004/22/EC.
The indication ‘Covered’ means that:






—


the requirement of OIML R 138 is identical to the one of MID, or










—


the requirement of OIML R 138 is more severe than the one of MID, or










—


all of the requirement of OIML R 138 fulfils MID requirements (even when MID allows other alternatives),










—


in case the requirement is not fully covered, a short statement explains what is covered.




The indication ‘Not covered’ means that the MID requirement is either not compatible with the relevant OIML R 138 requirement or not included in OIML R 138.
The indication ‘Not relevant’ means that the MID Annex I requirement is not relevant for capacity serving measures.










(1)  OJ L 135, 30.4.2004.