****

( 1 ) JO NO L 184 DU 15 . 7 . 1975 , P . 1 .

( 2 ) JO NO L 316 DU 6 . 12 . 1975 , P . 17 .

DECISION DE LA COMMISSION

DU 13 OCTOBRE 1978

EXCLUANT DU BENEFICE DE LA FRANCHISE DES DROITS DU TARIF DOUANIER COMMUN L ' APPAREIL SCIENTIFIQUE DENOMME " BECKMAN LIQUID SCINTILLATION SYSTEM , MODEL LS 8000 "

( 78/894/CEE )

LA COMMISSION DES COMMUNAUTES

EUROPEENNES ,

VU LE TRAITE INSTITUANT LA COMMUNAUTE ECONOMIQUE EUROPEENNE ,

VU LE REGLEMENT ( CEE ) NO 1798/75 DU CONSEIL , DU 10 JUILLET 1975 , RELATIF A L ' IMPORTATION EN FRANCHISE DES DROITS DU TARIF DOUANIER COMMUN DES OBJETS DE CARACTERE EDUCATIF , SCIENTIFIQUE OU CULTUREL ( 1 ),

VU LE REGLEMENT ( CEE ) NO 3195/75 DE LA COMMISSION , DU 2 DECEMBRE 1975 , FIXANT LES DISPOSITIONS D ' APPLICATION DU REGLEMENT ( CEE ) NO 1798/75 ( 2 ), ET NOTAMMENT SES ARTICLES 4 ET 5 ,

CONSIDERANT QUE , PAR LETTRE DU 6 AVRIL 1978 , LE GOUVERNEMENT ALLEMAND A DEMANDE A LA COMMISSION D ' ENGAGER LA PROCEDURE PREVUE AUX ARTICLES 4 ET 5 DU REGLEMENT ( CEE ) NO 3195/75 EN VUE DE DETERMINER SI L ' APPAREIL DENOMME " BECKMAN LIQUID SCINTILLATION SYSTEM , MODELE LS 8000 " DOIT ETRE CONSIDERE OU NON COMME UN APPAREIL SCIENTIFIQUE ET , EN CAS DE REPONSE AFFIRMATIVE , SI DES APPAREILS DE VALEUR SCIENTIFIQUE EQUIVALENTE SONT PRESENTEMENT FABRIQUES DANS LA COMMUNAUTE ;

CONSIDERANT QUE , CONFORMEMENT AUX DISPOSITIONS DE L ' ARTICLE 4 PARAGRAPHE 5 DU REGLEMENT ( CEE ) NO 3195/75 , UN GROUPE D ' EXPERTS COMPOSE DE REPRESENTANTS DE TOUS LES ETATS MEMBRES S ' EST REUNI LE 6 OCTOBRE 1978 DANS LE CADRE DU COMITE DES FRANCHISES DOUANIERES AFIN D ' EXAMINER CE CAS D ' ESPECE ;

CONSIDERANT QU ' IL RESSORT DE CET EXAMEN QUE L ' APPAREIL EN QUESTION EST UN COMPTEUR A SCINTILLATION LIQUIDE UTILISE DANS DES TRAVAUX DE RECHERCHE EN VUE DE MESURER L ' ACTIVITE SYNTHETIQUE DE SYSTEMES BIOLOGIQUES APRES OU EN RELATION AVEC DES RADIATIONS ; QUE SES CARACTERISTIQUES PERMETTENT NOTAMMENT DE DETERMINER QUANTITATIVEMENT LES RAYONNEMENTS BETA DE FAIBLE ENERGIE DANS DES ECHANTILLONS BIOLOGIQUES ;

CONSIDERANT QUE LES CARACTERISTIQUES ET L ' USAGE QUI EST FAIT DUDIT APPAREIL EN FONT UN APPAREIL SPECIALEMENT APTE A LA RECHERCHE SCIENTIFIQUE PURE ; QU ' IL DOIT DES LORS ETRE CONSIDERE COMME UN APPAREIL SCIENTIFIQUE ;

CONSIDERANT TOUTEFOIS QUE , SUR LA BASE DES INFORMATIONS RECUEILLIES AUPRES DES ETATS MEMBRES , DES APPAREILS DE VALEUR SCIENTIFIQUE EQUIVALENTE AUDIT APPAREIL , SUSCEPTIBLES D ' ETRE UTILISES AUX MEMES USAGES , SONT FABRIQUES DANS LA COMMUNAUTE , AU SENS DE L ' ARTICLE 3 PARAGRAPHE 3 DU REGLEMENT ( CEE ) NO 1798/75 , DEPUIS 1975 ,

A ARRETE LA PRESENTE DECISION :

ARTICLE PREMIER

1 . L ' APPAREIL DENOMME " BECKMAN LIQUID SCINTILLATION SYSTEM , MODELE LS 8000 " DOIT ETRE CONSIDERE COMME UN APPAREIL SCIENTIFIQUE .

2 . LES CONDITIONS VISEES A L ' ARTICLE 3 PARAGRAPHE 1 SOUS B ) DU REGLEMENT ( CEE ) NO 1798/75 DU CONSEIL , DU 10 JUILLET 1975 , POUR L ' ADMISSION EN FRANCHISE DES DROITS DU TARIF DOUANIER COMMUN DE L ' APPAREIL REPRIS AU PARAGRAPHE 1 NE SONT PAS REMPLIES .

ARTICLE 2

LES ETATS MEMBRES SONT DESTINATAIRES DE LA PRESENTE DECISION .

FAIT A BRUXELLES , LE 13 OCTOBRE 1978 .

PAR LA COMMISSION

ETIENNE DAVIGNON

MEMBRE DE LA COMMISSION