—

OIML R 51-1 Edition 2006 (E) Essential requirements in MID (Annex I and MI-006 common part and Chapter II) OIML R 51-1 Edition 2006 (E) Comment Annex I 1.1 T.4.3.1, T.4.3.7, 2.5, 2.6, 4.1.1 Covered 1.2 4.1.3, 4.1.6, 4.2.2 Covered 1.3 5.2.1 Covered 1.3.1 2.9.1.1, 2.9.1.2 Covered except that: Condensing/non-condensing and open/closed is not covered 1.3.2 Not relevant 1.3.3 (a) E1 (a) E2 A.6.3.1, A.6.3.2, A.6.3.4, A.6.3.5.1, A.6.3.5.2, A.6.3.3 Covered (a) E3 A.6.3.1, A.6.3.2, A.6.3.4, A.6.3.5.1, A.6.3.5.2, A.6.3.3, A.6.3.6 Covered on the provision that test pulse 5, test level IV according to ISO 7637-2:2004(E) is used (b) A.6.3.1, A.6.3.2, A.6.3.4, A.6.3.5.1, A.6.3.5.2, A.6.3.3 Covered on the provision that the severity levels specified in OIML D11 (2004) are used in some instances 1.3.4 2.9.2, A.6.2.4, A.6.2.5, A.6.2.6, A.6.2.7, 4.2.3, 6.4.3, A.5.2, 2.9.3, A.6.2.8, 2.9.1.3, A.6.2.2, 6.1.4 & A.5.8, 2.8.1, 6.4.4 & A.5.7 Covered except the mains frequency variation and power frequency magnetic field 1.4 1.4.1 A.6.1.1, Annex A Covered 1.4.2 A.6.2.3, 4.1.2 Covered except for the damp heat condensing test 2 A.7, 6.5.3, 2.10 Covered 3 T.3.6 Covered 4 T.3.5, 3.3.1, 2.5.1, 5.2.3 Covered 5 4.1.4, A.7, 2.10, 6.5.3, T.3.7 Covered 6 3.2.2, 4.1.3, 4.2.2 Covered 7 7.1 3.2.1, 3.2.2, 3.2.4 Covered 7.2 3.1 Covered 7.3 Not relevant 7.4 Not relevant 7.5 3.1 Covered 7.6 3.4.5. 2.11, 6.1.5 Covered except for the description of test procedure in the operation manual 8 8.1 4.2.4 Covered 8.2 3.2.6 Covered 8.3 3.4.5, 3.2.3 Covered 8.4 3.2.6, 3.2.3, 3.4.4 Covered 8.5 Not relevant 9 9.1 3.11.1, 3.11.2 Covered except for the indication of presence of additional device 9.2 Not relevant 9.3 3.11.1, 3.11.2, 3.11.3 Covered except for information (on its operation, installation, maintenance, repairs, permissible adjustments, correct operation), condensation, open or closed location 9.4 Not relevant 9.5 3.3.2 Covered 9.6 Not relevant 9.7 2.7 Covered 9.8 3.11.4, 3.12 Covered 10 10.1 3.3, 3.4, Covered 10.2 3.3.1, 2.4, 3.4.2, 3.3.2 Covered 10.3 3.4.3, 3.4.4 Covered 10.4 3.4.3 Not covered directly, although some requirements are relevant 10.5 Not relevant 11 11.1 3.4.4, 3.4.3 Covered when the measuring instrument is fitted with a data storage device or a printer 11.2 3.4.4, 3.4.3 Covered when the measuring instrument is fitted with a data storage device or a printer 12 2.11, 6.1.5, 6.1.8 Covered Annex MI-006 CHAPTER I — Requirements common to all types of automatic weighing instruments 1 5.2.1 Covered 1.1 T.3.1.1, T.3.1.2, T.3.1.3, 2.2.2, 2.2.1, 2.3.1, 2.3.3 Covered 1.2 2.9.2 Covered 1.3 2.9.1.1, 2.9.1.2, 2.9.3, 3.2.5 Covered 1.4 6.1.4, 2.9.3, 3.2.3 Covered 3 3.1 2.9.3, 3.2.3, 3.2.5 Covered 3.2 3.2.7 Covered 3.3 3.3.1 (3.3, 3.2.4, 3.6.4, 3.6.5, 3.7.3, 3.9.4) Covered 3.4 4.2.1 Covered 3.5 3.5, 3.6 Covered 3.6 3.3.3, 3.10.3 Covered CHAPTER II — Automatic catchweighers 1 1.1 2.1 Covered 1.2 2.1.1, 2.1.2 Covered except that IV is used in the Directive 2004/22/EC instead of IIII 2 2.1 2.1 Covered 2.2 2.1.1 Covered 3 2.1, 2.1.2 Covered 4 4.1 2.5.1.1, 2.5.1.2, A.3.9.2.2 a) Covered 4.2 2.5.1.1 Table 4 Covered 4.3 2.2.1 Table 1 Covered 4.4 2.2.1, 2.3 Covered except when the accuracy class is XIII or Y(a) with 0,1 g ≤ e ≤ 2 g 5 2.2.2 Covered 6 6.1 3.2.3 Covered 6.2 3.2.3 Covered 7.1 7.1.1 2.6.1 Covered 7.1.2 2.6.2 Covered 7.2 T.4.3.9, 4.1.3 Covered 7.3 2.9.1.2 Covered Notes: The column ‘Comment’ indicates the compliance between OIML R 51-1 and the relevant requirement in Directive 2004/22/EC. The indication ‘Covered’ means that: — the requirement of OIML R 51-1 is identical to the one of Directive 2004/22/EC, or — the requirement of OIML R 51-1 is more severe than the one of Directive 2004/22/EC, or — all of the requirement of OIML R 51-1 fulfils requirements in Directive 2004/22/EC (even when Directive 2004/22/EC allows other alternatives), — in case the requirement is not fully covered, a short statement explains what is covered. The indication ‘Not covered’ means that the requirement in Directive 2004/22/EC is either not compatible with the relevant OIML R 51-1 requirement or not included in OIML R 51-1. The indication ‘Not relevant’ means that the requirement in Annex I of Directive 2004/22/EC is not relevant for automatic catchweighing instruments.