—

OIML R 140 2007 Directive 2004/22/EC Essential requirements of Annex I and Annex MI-002 OIML R 140 (2007) Comment Annex I 1.1 T.2.6, T.3.4, 9.1.1 Covered 1.2 T.2.16, T.2.17.2, 9.1.2, A.1 Covered for disturbances 1.3 T.1.22, 7.1.3, A.2, A.4 Covered provided the manufacturer specify all needed information 1.3.1 7.1.3, A.4.1, A.4.2, T.1.22 Covered 1.3.2 (a) M1 M2 7.1.3, A.4.4 Covered for vibrations M3 Not relevant 1.3.2 (b) 7.1.3.1, A.4.4 Covered for vibrations 1.3.3 (a) E1 7.1.3, 10.2.8, A.4, A.4.5.a, A.4.5.b, A.4.6, A.4.7, A.4.8, A.4.11, A.4.12, A.4.13 Covered E2 Not relevant E3 Not relevant 1.3.3 (b) A.4 Covered, except short interruptions in the case of DC 1.3.4 6.4.9, A.4, A.4.9, A.4.10 Covered except for mains frequency variations and power frequency magnetic fields 1.4 1.4.1 A.1, A.4 Covered if modular approach is used 1.4.2 A.4.3 Covered 2 Not relevant 3 Not covered 4 8.2.1.3, 8.6.1.3 Covered 5 9.1.1, 9.1.2, 9.1.3 Covered 6 9.1.1, 9.1.2, 9.1.3, 9.1.5, 9.3 Covered 7 7.1 7.6 Covered 7.2 7.3.4, 8.5, 8.6, 8.7 Covered 7.3 9.3.5, 9.3.6 Covered 7.4 Not relevant 7.5 8.6.1.1, 8.6.1.2 Covered 7.6 9.3, 10.2.3 Covered except for test procedure 8 8.1 7.1.1.1, 7.1.1.2, 7.3.2, 8.6.1.4, 8.6.1.5 Covered 8.2 7.6 Covered 8.3 7.6.2, 10.2.3 Covered except providing the identification of software on the instrument 8.4 7.6.2 Covered 8.5 7.1.4.6, 8.7.2 Not Covered 9 9.1 7.5.1, 7.5.2, 7.1.1.2 Covered for information on data plate 9.2 Not relevant 9.3 7.5.1, 10.2.3, 10.2.8.2 Covered 9.4 Not relevant 9.5 8.2.1.2, 7.1.4.1 Covered 9.6 Not relevant 9.7 5.1 Covered 9.8 7.5.1 Covered 10 10.1 8.2.1.1 Covered 10.2 7.1.4.1, 7.1.4.3, 8.2.1.1 Covered 10.3 T.1.5, 8.3, 8.4 Covered 10.4 Not relevant 10.5 Covered provided that the volume conversion device is equipped with a local display 11 11.1 Not relevant 11.2 7.1.4.6, 8.7.2, 9.1.5 Covered 12 Covered Annex MI-002 Definitions 1.1, 2, 6.3, 7.3, 8.5, 8.6 Covered Gas meter Conversion device Minimum flowrate Maximum flowrate Transitional flowrate Overload flowrate Base conditions T.1.9, T.1.12, 7.3.2.1, T.1.14, 7.3.1 Covered 1 7.1.2.1, 8.6.1.1, 9.3.6 Not covered 1.1 Not relevant 1.2 Not covered 1.3 7.3, 7.5.1 Covered 1.4 7.1.3, A.4.1, A.4.2, T.1.22 Covered 1.5 7.1.3, A.4, A.4.9, A.4.10, T.1.22 Covered 2 2.1 Not relevant 2.2 Not relevant 3 3.1 3.1.1 T.2.17.2, T.2.17.4, 9.1.2, 9.1.3, 9.1.4 Covered 3.1.2 9.1.1, 9.1.2, 9.1.3, 9.1.5, 9.3.6 Covered 3.1.3 T.2.17.2 Covered 3.2 Not relevant 4 4.1 4.1.1 Not relevant 4.1.2 Not relevant 4.2 4.2.1 Not relevant 4.2.2 Not relevant 5 5.1 9.2.1 Not covered 5.2 9.2.2 Covered when the battery has a 5 years lifetime 5.3 Not relevant 5.4 Not relevant 5.5 Not relevant 5.6 Not relevant 6 5.1 Covered except when mass is expressed in tonnes 7 7.3.1, 10.2.2 Covered 8 6.3.3 Covered provided that ambient temperature is managed such that it remains within ± 3 °C at reference conditions and except for OIML R 140 class C 9 9.1 9.3.6 Covered 9.2 7.1.4, 8.2, 8.7.1 Covered only for parameters and provided that the conversion device is equipped with a local display Note: The column ‘Comment’ indicates the compliance between OIML R 140 and the relevant requirement in Directive 2004/22/EC. The indication ‘Covered’ means that: — the requirement of OIML R 140 is identical to the one of MID, or — the requirement of OIML R 140 is more severe than the one of MID, or — all of the requirement of OIML R 140 fulfils MID requirements (even when MID allows other alternatives), — in case the requirement is not fully covered, a short statement explains what is covered. The indication ‘Not covered’ means that the MID requirement is either not compatible with the relevant OIML R 140 requirement or not included in OIML R 140. The indication ‘Not relevant’ means that the MID Annex I requirement is not relevant for conversion devices.