COMMISSION REGULATION (EEC) No 2676/90 of 17 September 1990 determining Community methods for the analysis of wines

THE COMMISSION OF THE EUROPEAN COMMUNITIES,

Having regard to the Treaty establishing the European Economic Community,

Having regard to Council Regulation (EEC) No 822/87 of 16 March 1987 on the common organization of the market in wine (1), as amended by Regulation (EEC) No 1325/90 (2), and in particular Article 74 thereof,

Whereas Article 74 (1) of Regulation (EEC) No 822/87 prescribes the adoption of methods of analysis for establishing the composition of the products indicated in Article 1 of that Regulation and of rules for checking whether these products have been subjected to treatments in violation of authorized oenological pratice;

Whereas, inasmuch as the Community has not yet laid down maximum levels for substances whose presence indicates that certain oenological practices have been used and has not yet adopted tables enabling analysis data to be compared, there is good reason to authorize Member States to determine such maximum levels;

Whereas Article 13 (1) of Regulation (EEC) No 822/87 provides for an analytical test, including at the least an assessment of the characteristics, as listed in the Annex to that Regulation, of the quality wine psr in question;

Whereas the verification of the particulars on documents concerning the products in question calls for the introduction of uniform methods of analysis to ensure that accurate and comparable information is obtained; whereas, consequently, these methods should be compulsory for all commercial transactions and all verification procedures; whereas, however, in view of the trade's limited facilities, a restricted number of usual procedures should be admitted enabling the requisite factors to be determined rapidly and with reasonable accuracy;

Whereas, as far as is possible, generally recognized methods such as those developed under the 1954 International Convention for the Unification of Methods of Analysis and Appraisal of Wines, which are published in the Recueil des méthodes internationales d'analyse des vins (Compendium of international methods for the analysis of wines) by the International Office of Vine and Wine, may usefully be retained;

Whereas the Community methods of analysis applicable to wine were laid down in Commission Regulation (EEC) No 1108/82 (3); whereas scientific progress has made it necessary to replace certain of the methods by some that are more suitable, to modify others, and to introduce new methods, particularly those approved since the aforesaid Regulation by the International Office of Vine and Wine; whereas, because of the profusion and complexity of these changes, all the analyses should be reassembled in a new Regulation, and Regulation (EEC) No 1108/82 should be repealed;

Whereas, in order to ensure the comparability of the results obtained by applying the analytical methods referred to in Article 74 of Regulation (EEC) No 822/87, steps should be taken to refer, in regard to the accuracy, repeatability and reproducibility of these results, to the definitions laid down by the International Office of Vine and Wine;

Whereas, in order to recognize the scientific advances on the one hand and the technical equipment of official laboratories on the other, and with the aim of increasing the efficiency and profitability of these laboratories, there is good reason to allow automated analytical methods to be applied under certain conditions; whereas it is important to specify that, where a dispute arises, the automated methods may not replace the reference methods and the usual methods;

Whereas the results of a density measurement using the automated method based on the principle of the frequency oscillator are, in respect of their accuracy, repeatability and reproducibility, at least as good as the results obtained by the methods listed in section 1 of the Annex to the present Regulation for measuring the density or specific gravity; whereas it is therefore indicated, by virtue of Article 74 (3) of Regulation (EEC) No 822/87, that this automated method shall be considered as equivalent to the said methods listed in the Annex to the present Regulation;

Whereas the procedure described in Chapter 25 under point 2.2.3.3.2 in the Annex hereto for analysing the total sulphur dioxide content of wines and grape musts of a presumed content of less than 50 mg/l results in better extraction of that substance compared to the methods described in Chapter 13 under point 13.4 of the Annex to Regulation (EEC) No 1108/82; whereas the result is higher total sulphur dioxide contents of the products analysed, which may exceed, in particular, in the case of certain grape juices, the maximum limit laid down; whereas, in order to avoid difficulties in the disposal of grape juice already prepared at the time of entry into force of this Regulation and until such time as the production processes are adapted to result in more complete de-sulphiting of grape musts with fermentation arrested by the addition of alcohol, the procedure described in the abovementioned Regulation should be allowed to be used during a transitional period;

Whereas the measures provided for in this Regulation are in accordance with the opinion of the Management Committee for Wine,

HAS ADOPTED THIS REGULATION:

Article 1

1. The Community methods for the analysis of wine making it possible, in the context of commercial transactions and all control operations, to:

- establish the composition of the products listed in Article 1 of Regulation (EEC) No 822/87,

- check whether these products have been subjected to treatments in violation of authorized oenological practice,

are those set out in the Annex to this Regulation.

2. For substances for which reference methods and usual methods are prescribed, the results obtained by the use of the reference methods shall prevail.

Article 2

For the purposes of applying this Regulation:

(a) the repeatability shall be the value below which the absolute difference between the two single test results obtained on identical test material, under the same conditions (same operator, same apparatus, same laboratory and a short interval of time), may be expected to lie with a specified probability;

(b) the reproducibility shall be the value below which the absolute difference between two single test results obtained identical test material, under different conditions (different operators, different apparatus and/or different laboratories and/or different time), may be expected to lie with a specified probability.

The term 'single test result' shall be the value obtained when the standardized test method is applied fully and once to a single sample.

Unless otherwise stated, the probability shall be 95 %.

Article 3

1. Automated analytical methods shall be acceptable, under the responsibility of the Director of a laboratory on condition that the accuracy, repeatability and reproducibility of the results are at least equivalent to those of the results obtained by the analytical methods listed in the Annex.

Where a dispute arises, the methods listed in the Annex may not be replaced by automated methods.

2. The automated method for measuring density based on the principle of the frequency oscilliator shall be considered as equivalent to the methods listed in section 1 of the Annex to the present Regulation.

Article 4

Wherever mention is made of water for solution, dilution, or washing purposes, this shall mean distilled water or demineralized water of equivalent purity. All chemicals shall be of analytical reagent quality except where otherwise specified.

Article 5

Regulation (EEC) No 1108/82 is repealed.

However Article 1 (4) of that Regulation shall apply until 31 December 1990.

Article 6

This Regulation shall enter into force on the day of its publication in the Official Journal of the European Communities.

It shall apply with effect from 1 October 1990.

This Regulation shall be binding in its entirety and directly applicable in all Member States.

Done at Brussels, 17 September 1990.

For the CommissionRay MAC SHARRYMember of the Commission

(1)OJ No L 84, 27. 3. 1987, p. 1. (2)OJ No L 132, 23. 5. 1990, p. 19. (3)OJ No L 133, 14. 5. 1982, p. 1.

ANNEX

1. DENSITY AND SPECIFIC GRAVITY AT 20 °C 1. DEFINITIONS

The density is the mass per unit volume of wine or must at 20 °C. It is expressed in grams per millilitre and denoted by the symbol r 20 °C.

The specific gravity at 20 °C (or the 20 °C/20 °C relative density) is the ratio, expressed as a decimal number, of the density of a certain volume of the wine or must at 20 °C to the density of same volume water at the same temperature. It is denoted by the symbol d20 °C

20 °C

. 2. PRINCIPLE OF METHODS

The density and specific gravity at 20 °C are measured on a test sample:

- either by pigmentary: reference method,

- or by hydrometry or densimetry using a hydrostatic balance: usual methods.

Note:For very accurate measurement, the density must be corrected for the effect of sulphur dioxide by using the formula:

r20 °C = r220 °C 0,0006 × Swhere r20 °C = corrected densityr220 °C = observed densityS = total quantity of sulphur dioxide in grams per litre3. PRELIMINARY TREATMENT OF SAMPLE

If the wine or must contains appreciable quantities of carbon dioxide, remove most of it by stirring 250 ml of the wine in a 1-litre flask or by filtration under reduced pressure through 2 g of cotton wool placed in an extension tube.4. REFERENCE METHOD

4.1. Apparatus:Normal laboratory equipment, and in particular:

4.1.1. A Pyrex pycnometer (1), of approximately 100 ml capacity, with a removable ground-glass jointed thermometer calibrated in tenths of a degree from 10 to 30 °C (Figure 1). The thermometer must be standardized.

Figure 1 The pycnometer and its tare

The pycnometer has a side tube 25 mm in length and 1 mm (maximum) in internal diameter, ending in a conical ground joint. This side tube may be capped by a 'reservoir stopper' consisting of a conical ground-glass joint tube and terminating in a drawn-out section. This stopper serves as an expansion chamber.

The two ground joints of the apparatus should be prepared with great care.

4.1.2. A tare bottle, consisting of a vessel with the same outside volume (to within at least 1 ml) as the pycnometer and with a mass equal to the mass of the pycnometer filled with a liquid of specific gravity 1,01 (sodium chloride solution 2,0 % m/v).

A thermally insulated container, exactly fitting the body of the pycnometer.

4.1.3. A twin-pan balance with a range of at least 300 g and a sensitivity of 0,1 mg,ora single-pan balance with a range of at least 200 g and a sensitivity of 0,1 mg.4.2. Calibration of pycnometer

Calibration of the pycnometer involves a determination of the following quantities:- tare of the empty pycnometer,- volume of the pycnometer at 20 °C,- mass of the water-filled pycnometer at 20 °C.4.2.1. Method using a twin-pan balance

Place the tare bottle on the left-hand pan of the balance and the clean and dry pycnometer, fitted with its 'reservoir stopper', on the right-hand pan. Add weights to the pan holding the pycnometer and record the weight required to establish equilibrium: let it be p grams.

Carefully fill the pycnometer with distilled water at the ambient temperature and fit the thermometer; carefully wipe the pycnometer dry and place it in the thermally insulated container. Mix it by inverting the container until the temperature reading on the thermometer is constant. Accurately adjust the level to the upper rim of the side tube. Wipe the side tube dry and put the reservoir stopper on; read the temperature t°C carefully, possibly correcting it for the inaccuracy in the temperature scale. Weigh the water-filled pycnometer against the tare and record the weight p2 in grams required to establish equilibrium.

Calculation (2):

Taring of empty pycnometer:tare of empty pycnometer =p + m,where m = mass of air contained in the pycnometer,m = 0,0012 (p p2).

Volume at 20 °C:V20 °C = (p + m p2) · Ft,where Ft = a factor taken from Table I for the temperature t °C.V20 °C should be known to within ± 0,001 ml.

Mass of water at 20 °C:M20 °C = 0,998203 V20 °C,where 0,998203 is the density of water at 20 °C.

4.2.2. Method using a single-pan balance

Determine:- the mass of the clean and dry pycnometer: let this be P,- the mass of the pycnometer filled with water at t °C, following the procedure described in 4.2.1 above: let this be P1,- the mass of the tare: T0.

Calculation (3):

Taring of empty pycnometer:tare of empty pycnometer =P m,where m = mass of air contained in the pycnometer,m = 0,0012 (P1 P).

Volume at 20 °C:V20 °C = [P1 (P m)] × Ftwhere Ft= a factor taken from Table I for the temperature t °C.The volume at 20 °C should be known to within ± 0,001 ml.

Mass of water at 20 °C:M20 °C = 0,998203 V20 °C,where 0,998203 is the density of water at 20 °C.4.3. Method of measurement (3)

4.3.1. Method using twin-pan balance

Fill the pycnometer with the prepared test sample, following the procedure described in 4.2.1 above.

Let p" be the weight in grams required to establish equilibrium at t °C.

Mass of liquid contained in the pycnometer = p + m p".

Apparent density at t °C:

Calculate the density at 20 °C using one of the correction tables given later, in accordance with the nature of the liquid being measured: dry wine (Table II), natural or concentrated must (Table III), sweet wine (Table IV).

The 20 °C/20 °C specific gravity of the wine is calculated by dividing its density at 20 °C by 0,998203.4.3.2. Method using a single-pan balance (3)

Weigh the tare bottle and let its mass be T.Calculate dT = T1 T0.Mass of the empty pycnometer at the time of measurement = P m + dT.

Weigh the pycnometer filled with the prepared test sample, following the procedure described in 4.2.1 above. Let its mass att °C be P2.

Mass of liquid contained in pycnometer at t °C = P2 (P m + dT)

Apparent density at t °C:

Calculate the density at 20 °C of the liquid under test (dry wine, natural or concentrated must, or sweet wine) as indicated in 4.3.1 above.

The 20 °C/20 °C specific gravity is calculated by dividing the density at 20 °C by 0,998203.

4.3.3. The repeatability of the density measurementsfor dry and semi-sweet wines: r = 0,00010,

and for sweet wines: r = 0,00018.

4.3.4. The reproducibility of the density measurements:for dry and semi-sweet wines: R = 0,00037,and for sweet wines: R= 0,00045.5. USUAL METHODS

5.1. Hydrometry

5.1.1. Apparatus

5.1.1.1. Hydrometer

Hydrometers shall conform to ISO Standards as regards their dimensions and graduation.

They shall have a cylindrical bulb, and a stem of circular cross-section with a minimum diameter of 3 mm. For dry wines, they shall be graduated from 0,983 to 1,003 with graduation marks every 0,0010 and 0,0002. Each mark at 0,0010 shall be separated from the next corresponding mark by at least 5 mm. For measuring the density of de-alcoholized wines, sweet wines and musts, a set of five hydrometers graduated from 1,000 to 1,030, 1,030 to 1,060, 1,060 to 1,090, 1,090 to 1,120 and 1,120 to 1,150 shall be used. These hydrometers shall be graduated in densities at 20 °C with marks at least every 0,0010 and 0,0005, each mark at 0,0010 being separated by at least 3 mm from the next corresponding mark.

These hydrometers shall be graduated so as to be read 'at the top of the meniscus'. An indication of the graduation in density or specific gravity at 20 °C and of the reading at the top of the meniscus shall be carried either on the graduated scale or on a slip of paper enclosed in the bulb. These hydrometers shall be calibrated by a government department.

5.1.1.2. A calibrated thermometer graduated in minimum intervals of 0,5 °C.

5.1.1.3. A measuring cylinder of 36 mm internal diameter and 320 mm height, held vertical by supporting levelling screws.5.1.2. Procedure

5.1.2.1. Method of measurement

Pour 250 ml of the test sample, prepared as in 5.1.1.3 above, into the measuring cylinder and introduce the thermometer and hydrometer into it. Read the temperature one minute after mixing the sample well to produce a uniform temperature. Remove the thermometer and read the apparent density at t °C from the hydrometer after a further minute.

The apparent density at t °C is then corrected to 20 °C by using the tables applicable to dry wines (Table V), musts (Table VI) or wines containing sugars (Table VII).

The 20 °C/20 °C specific gravity is obtained by dividing the density at 20 °C by 0,998203.5.2. Densimetry using a hydrostatic balance

5.2.1. Apparatus

5.2.1.1. Hydrostatic balance with a maximum capacity of at least 100 g and a sensitivity of 0,1 mg.

Identical Pyrex glass floats with a volume of at least 20 ml are fixed under both pans by suspending them with a thread of diameter not exceeding 0,1 mm.

The float suspended under the right-hand pan must be capable of being introduced into a measuring cylinder bearing a mark indicating the level. The measuring cylinder must have an internal diameter at least 6 mm larger than that of the float. The float must be capable of being contained completely within the volume of the measuring cylinder located below the mark; the surface of the liquid to be measured must be penetrated only by the supporting thread. The temperature of the liquid in the measuring cylinder is measured by a thermometer graduated in steps of 0,2 °C.

5.2.1.2. A single-pan hydrostatic balance may also be used.5.2.2. Procedure

5.2.2.1. Standardization of a hydrostatic balance

With the two floats in air, establish balance by placing weights in the right-hand pan. Record the mass p of these added weights.

Fill the measuring cylinder with pure water up to the mark and read the temperature t °C after shaking and allowing to stand for two or three minutes.Re-establish balance by placing weights in the right-hand pan. Let the mass of these bep2.

The volume of the float at 20 °C is given by:V20 °C = (p2 p) (F + 0,0012)where F is the factor given in Table I for the temperature t °C.

p and V20 °C are the characteristics of the float.5.2.2.2. Method of measurement

The right-hand float is immersed in the measuring cylinder filled with wine (or must) up to the mark. The temperature t °C of the wine (or must) is read and the massp" needed to re-establish balance is recorded.

The apparent density rt is given byThis density is corrected to 20 °C by using one of the Tables II, III or IV if the float is of Pyrex glass.

6. EXAMPLE OF THE CALCULATION OF THE DENSITY AT 20 °C AND THE 20 °C/20 °C SPECIFIC GRAVITY (REFERENCE METHOD)

6.1. Pycnometry using a twin-pan balance

6.1.1. Standardization of the pycnometer:

1. Weighing of clean dry pycnometer:Tare = pycnometer + p

p = 104,9454 g

2. Weighing of pycnometer filled with water at t °C:Tare = pycnometer + water + p2

p2 = 1,2396 g att = 20,5 °C

3. Calculation of mass of air contained in the pycnometer:m = 0,0012 (p p2)

m = 0,0012 (104,9454 1,2396)

m = 0,1244 g

4. Characteristic values to be retained:Tare of empty pycnometer, p + m:

p + m = 104,9454 + 0,1244

p + m = 105,0698 gVolume at 20 °C = F (p + m p2)t °CF20,5 °C = 1,001900

V20 °C = (105,0698 1,2396) × 1,001900

V20 °C = 104,0275 mlMass of water at 20 °C =

M20 °C = V20 °C · 0,998203

M20 °C = 103,8405 g6.1.2. Determination of the density and specific gravity at 20 °C/20 °C of a dry wine

r" = 1,2622 at 17,80 °Cr17,80 °C = 0,99788 g/ml

Table II enables r20 °C to be calculated from rt °C using the relationship:

For t = 17,80 °C and for an alcoholic strength of 11 % vol. c= 0,54.6.2. Pycnometry using a single-pan balance

6.2.1. Standardization of the pycnometer:

1. Weight of clean dry pycnometer:P = 67,7913 g

2. Weight of pycnometer filled with water at t °C:P1 = 169,2715 at 21,65 °C

3. Mass of air contained in the pycnometer:m = 0,0012 (P1 P)

m = 0,0012 × 101,4802

m = 0,1218 g

4. Characteristic values to be retained:Tare of empty pycnometer, P m:

P m= 67,7913 0,1218

P m = 67,6695 gVolume at 20 °C = [P1 (P m)] Ft °C

F21,65 °C = 1,002140

V20 °C = 1,002140 (169,2715 67,6695)

V20 °C = 101,8194 mlMass of water at 20 °C:

M20 °C = V20 °C × 0,998203

M20 °C = 101,6364 gMass of tare bottle, To:

To = 171,9160 g6.2.2. Determination of the density and specific gravity at 20 °C of a dry wine

T1 = 171,9178 g

dT = 171,9178 171,9160 = 0,0018 g

P m + dT = 67,6695 + 0,0018 = 67,6713 g

P2 = 169,2799 at 18 °Cr18 °C = 0,99793 g/ml

Table II enables r20 °C to be calculated from rt °C using the relationship:

For t= 18 °C and an alcoholic strength of 11 % vol. c = 0,49.

TABLE I F factors by which the mass of water contained in the Pyrex pycnometer at t °C has to be multiplied in order to calculate the pycnometer volume at 20 °C

TABLE II Temperature corrections c to the density of alcohol-free dry wines measured with a Pyrex glass pycnometer at t °C to relate the result to 20 °C

TABLE III Temperature corrections cto the density of natural musts and of concentrated musts measured with a Pyrex glass pycnometer at t °C to relate the result to 20 °C

TABLE IV Temperature corrections c to the density of wines of 13 % vol and above containing residual sugar measured with a Pyrex glass pycnometer at t °C to relate the result to 20 °C

TABLE V Temperature corrections c to the density of dry wines and alcohol-free dry wines measured with an ordinary glass pycnometer or hydrometer at t °C to relate the result to 20 °C

TABLE VI Temperature corrections cto the density of natural musts and of concentrated musts measured with a pycnometer or hydrometer of ordinary glass att °C to relate the result to 20 °C

TABLE VII Temperature corrections cto the density of wines of 13 % vol and above containing residual sugar measured with a hydrometer or pycnometer of ordinary glass at t °C to relate the result to 20 °C

2. EVALUATION BY REFRACTOMETRY OF THE SUGAR CONCENTRATION IN GRAPE MUSTS, CONCENTRATED GRAPE MUSTS AND RECTIFIED CONCENTRATED GRAPE MUSTS 1. PRINCIPLE OF THE METHOD

The refractive index at 20 °C, expressed either as an absolute value or as a percentage by mass of sucrose, is given in the appropriate table to provide a means of obtaining the sugar concentration in grams per litre and in grams per kilogram for grape musts, concentrated grape musts and rectified concentrated grape musts.2. APPARATUS

2.1. Abbé refractometer

The refractometer used must be fitted with a scale giving:- either percentage by mass of sucrose to 0,1 %,- or refractive indices to four decimal places.

The refractometer must be equipped with a thermometer having a scale extending at least from +15 °C to +25 °C and with an arrangement for circulating water enabling measurements to be made at a temperature of 20 ± 5 °C.

The operating instructions for this instrument must be strictly adhered to, particularly with regard to calibration and the light source.3. PREPARATION OF THE SAMPLE

3.1. Must and concentrated must

Pass the must, if necessary, through a dry gauze folded into four and, after discarding the first drops of the filtrate, carry out the determination on the filtered product.3.2. Rectified concentrated must

Depending on the concentration, use either the rectified concentrated must itself or a solution obtained by making up 200 g of rectified concentrated must to 500 g with water, all weighings being carried out accurately.4. PROCEDURE

Bring the sample to a temperature close to 20 °C. Place a small test sample on the lower prism of the refractometer, taking care (because the prisms are pressed firmly against each other) that this test sample covers the glass surface uniformly. Carry out the measurement in accordance with the operating instructions of the instrument used.

Read off the percentage by mass of sucrose to within 0,1 % or read the refractive index to four decimal places.

Carry out at least two determinations on the same prepared sample.

Note the temperature t °C.5. CALCULATION

5.1.Temperature correction

5.1.1. Instruments graduated in percentage by mass of sucrose: use Table I to obtain the temperature correction.

5.1.2. Instruments graduated in refractive index: find the index measured at t °C in Table II to obtain (column 1) the corresponding value of the percentage by mass of sucrose at t °C. This value is corrected for temperature and expressed as a concentration at 20 °C by means of Table I.5.2. Sugar concentration in must and concentrated must

Find the percentage by mass of sucrose at 20 °C in Table II and read from the same row the sugar concentration in grams per litre and grams per kilogram. The sugar concentration is expressed in terms of invert sugar to one decimal place.5.3. Sugar concentration in rectified concentrated must

Find the percentage by mass of sucrose at 20 °C in Table III and read from the same row the sugar concentration in grams per litre and grams per kilogram. The sugar concentration is expressed in terms of invert sugar to one decimal place.

If the measurement was made on diluted rectified concentrated must, multiply the result by the dilution factor.5.4. Refractive index of must, concentrated must and rectified concentrated must

Find the percentage by mass of sucrose at 20 °C in Table II and read from the same row the refractive index at 20 °C. This index is expressed to four decimal places.

TABLE I Correction to be made when the percentage by mass of sucrose has been determined at a temperature different from 20 °C

TABLE II Table giving the sugar concentration (4) in must and concentrated must in grams per litre and grams per kilogram, determined by means of a refractometer graduated either in percentage by mass of sucrose at 20 °C or in refractive index at 20 °C. The density at 20 °C is also given

TABLE III Table giving the sugar (5) concentration in rectified concentrated must in grams per litre and grams per kilogram, determined by means of a refractometer graduated either in percentage by mass of sucrose at 20 °C or in refractive index at 20 °C. The density at 20 °C is also given

3. ALCOHOLIC STRENGTH BY VOLUME 1. DEFINITION

The alcoholic strength by volume is the number of litres of ethanol contained in 100 litres of wine, both volumes being measured at a temperature of 20 °C. It is expressed by the symbol '% vol'.

Note:Homologues of ethanol, together with the ethanol and ethanol homologues in ethyl esters, are included in the alcoholic strength since they occur in the distillate.2. PRINCIPLE OF METHODS

2.1. Distillation of wine made alkaline by a suspension of calcium hydroxide. Measurement of the alcoholic strength of the distillate.

2.2. Reference method: measurement of the density of the distillate with a pycnometer.2.3. Usual methods:

2.3.1. Measurement of the alcoholic strength of the distillate with a hydrometer.

2.3.2. Measurement of the alcoholic strength of the distillate by densimetry using a hydrostatic balance.

2.3.3. Measurement of the alcoholic strength of the distillate by refractometry.

Note:To obtain the alcoholic strength from the density of the distillate, use Tables I, II and III in Appendix II to this section of the Annex. These have been calculated from the International Tables of Alcoholic Strength published in 1972 by the International Legal Metrology Organization in its Recommendation 22 and adopted by the OIV (General Assembly, 1974).Table I gives the general formula relating the alcoholic strength by volume and the density of alcohol-water mixtures as a function of temperature.3. METHOD OF OBTAINING DISTILLATE

3.1. Apparatus

3.1.1. Distillation apparatus, consisting of:- a round-bottomed 1-litre flask with ground-glass joints,- a rectifying column about 20 cm in height or any apparatus to prevent splashing,- a source of heat; any pyrolysis of extracted matter must be prevented by a suitable arrangement,- a condenser terminated by a drawn-out tube taking the distillate to the bottom of a graduated receiving flask containing several ml of water.

3.1.2. Steam-distillation apparatus consisting of:1. a steam-generator;2. a steam pipe;3. a rectifying column;4. a condenser.

Any type of distillation or steam-distillation apparatus may be used provided that it satisfies the following test:Distil an ethanol-water mixture with an alcoholic strength of 10 % vol five times in succession. The distillate should have an alcoholic strength of at least 9,9 % vol after the fifth distillation, i.e. the loss of alcohol during each distillation should not be more than 0,02 % vol.3.2. Reagents

3.2.1. A 2 M suspension of calcium hydroxide, obtained by carefully pouring 1 litre of water at 60 to 70 °C on to 120 g of quicklime (CaO).3.3. Preparation of sample

Remove the bulk of any carbon dioxide from young and sparkling wines by stirring 250 to 300 ml of the wine in a 500-ml flask.3.4. Procedure

Measure out 200 ml of the wine using a graduated flask.

Record the temperature of the wine.

Transfer the wine to the distillation flask and introduce the steam-pipe of the steam-distillation apparatus. Rinse the graduated flask four times with successive 5-ml washings of water added to the flask or the steam-pipe. Add 10 ml of (3.2.1) calcium hydroxide and several pieces of inert porous material (pumice, etc).

Collect the distillate in the 200-ml graduated flask used to measure the wine.

Collect a volume of about three-quarters of the initial volume if distillation is used and a volume of 198 to 199 ml of distillate if steam distillation is used. Make up to 200 ml with distilled water, keeping the distillate at a temperature within 2 °C of the initial temperatures.

Mix with great care, using a circular motion.

Note:In the case of wines containing particularly large concentrations of ammonium ions, the distillate may be redistilled under the conditions described above, but replacing the suspension of calcium hydroxide with 1 M sulphuric acid diluted to 10 parts in 100 (vol/vol).4. REFERENCE METHOD

Measurement of the alcoholic strength of the distillate using a pycnometer.4.1. Apparatus

4.1.1. Use the standardized pycnometer as described in the chapter 'Density and specific gravity' (Annex, chapter 1).4.2. Procedure

Measure the apparent density of the distillate (3.4) at t °C as described in the chapter 'Density and specific gravity' (Annex, chapter 1, sections 4.3.1 and 4.3.2). Let this density be rt.4.3. Expression of results

4.3.1. Method of calculation

Find the alcoholic strength at 20 °C using Table I. In the horizontal line of this table corresponding to the temperature T (expressed as a whole number) immediately below t °C, find the smallest density greater than rt. Use the tabular difference just below this density to calculate the density r at this temperature T.

On the line of the temperatureT, find the density r immediately above r2 and calculate the difference between the densities rand r2. Divide this difference by the tabular difference just to the right of the density r2.The quotient gives the decimal part of the alcoholic strength, while the whole number part of this strength is shown at the head of the column in which the density r2 is located.

An example of the calculation of an alcoholic strength is given in Appendix I to this chapter of the Annex.

Note:This temperature correction has been incorporated in a computer program and might possibly be carried out automatically.4.3.2. Repeatability, r: r = 0,10 % vol.4.3.3. Reproducibility, R: R = 0,19 % vol.5. USUAL METHODS

5.1. Hydrometry

5.1.1. Apparatus

5.1.1.1. Alcoholmeter.

The alcoholmeter must conform to the specifications for class I or class II equipment defined in International Recommendation No 44, Alcoholmeters and Alcohol Hydrometers, of the ILMO (International Legal Metrology Organization).

5.1.1.2. Thermometer graduated in degrees and in 0,1 °C from 0 to 40 °C certified to within 1/20th degree.

5.1.1.3. Measuring cylinder of diameter 36 mm and height 320 mm held vertically by supporting levelling screws.5.1.2.Procedure

Pour the distillate (3.4) into the measuring cylinder. Ensure that the cylinder is kept vertical. Insert the thermometer and alcoholmeter. Read the temperature on the thermometer one minute after stirring to equilibrate the temperature of the measuring cylinder, the thermometer, the alcoholmeter and the distillate. Remove the thermometer and read the apparent alcoholic strength after one minute. Take at least three readings using a magnifying glass. Correct the apparent strength measured at t °C for the effect of temperature using Table II.

The temperature of the liquid must differ very little from ambient temperature (at most, by 5 °C).5.2. Densimetry using a hydrostatic balance

5.2.1. Apparatus

5.2.1.1. The hydrostatic balance is used as described in the chapter 'Density and specific gravity'.5.2.2. Procedure

The apparent density of the distillate at t °C is measured as described in the chapter 'Density and specific gravity', section 5.2.2.

5.2.3. Expression of results

Find the alcoholic strength at 20 °C by following the method described in 4.3.1, using Table I if the float is of Pyrex glass and Table III if it is of ordinary glass.5.3.Refractometry

5.3.1. Apparatus

5.3.1.1. Refractometer enabling refractive indices to be measured in the range 1,330 to 1,346. Depending on the type of equipment, measurements are made:- either at 20 °C with a suitable instrument,- or at ambient temperature t °C by an instrument fitted with a thermometer enabling the temperature to be determined to within at least 0,05 °C. A table giving temperature corrections will be provided with the instrument.5.3.2. Procedure

The refractive index of the wine distillate obtained as in 3.3 above is measured by following the procedure prescribed for the type of instrument used.5.3.3. Expression of results

Table IV is used to find the alcoholic strength corresponding to the refractive index at 20 °C.

Note:Table IV gives the alcoholic strengths corresponding to refractive indices for both pure alcohol-water mixtures and for wine distillates. In the case of wine distillates, it takes into account the presence of impurities in the distillate (mainly higher alcohols). The presence of methanol lowers the refractive index and thus the alcoholic strength.

6. EXAMPLE OF THE CALCULATION OF THE ALCOHOLIC STRENGTH OF A WINE

6.1. Measurement by pycnometer on a twin-pan balance

6.1.1. The constants of the pycnometer have been determined and calculated as described in chapter 1, 'Density and specific gravity', section 6.1.1.6.1.2. Weighing of pycnometer filled with distillateNumeric examplet °C = 18,90 °C

Tare = pycnometer + distillate att °C + p" t °C corrected = 18,70 °C

p" = 2,8074 g

p + m p" = mass of distillate at t °C 105,0698 2,8074 = 102,2624 g

Apparent density at t °C

6.1.3. Calculation of alcoholic strength

Refer to the table of apparent densities of water-alcohol mixtures at different temperatures, as indicated above

On the line 18 °C of the table of apparent densities, the smallest density greater than the observed density of 0,983076 is 0,98398 in column 11 % vol

The density at 18 °C is: (98307,6 + 0,7 × 22) 10 5 = 0,98323

0,98398 0,98323 = 0,00075

The decimal portion of the % vol of alcoholic strength is 75/114 = 0,65

The alcoholic strength is 11,65 % vol6.2. Measurement by pycnometer on a single pan balance

6.2.1. The constants of the pycnometer have been determined and calculated as described in chapter 1, 'Density and specific gravity', section 6.2.1.

6.2.2. Weighing of the pycnometer filled withdistillate

Weight of tare bottle at the time of

measurement in grams: T1 = 171,9178Pycnometer filled with distillate

at 20,50 °C in grams: P2 = 167,8438Variation in the buoyancy of air: dT = 171,9178 171,9160

= +0,0018Mass of the distillate at 20,50 °C: Lt = 167,8438 (67,6695 + 0,0018)

= 100,1725Apparent density of the distillate:6.2.3. Calculation of alcoholic strength

Refer to the table of apparent densities of water-alcohol mixtures at different temperatures, as indicated above

On the line 20 °C of the table of apparent densities, the smallest density greater than observed density of 0,983825 is 0,98471 in column 10 % vol

The density at 20 °C is: (98382,5 + 0,5 × 24) 10 5 = 0,983945

0,98471 0,983945 = 0,000765

The decimal portion of the % vol of alcoholic strength is 76,5/119 = 0,64

The alcoholic strength is 10,64 % vol

FORMULA FROM WHICH TABLES OF ALCOHOLIC STRENGHTS OF ETHANOL-WATER MIXTURES ARE CALCULATEDThe density r in kilograms per metre cubed (kg/m³) of an ethanol-water mixture at temperature t in degrees Celsius is given by the formula below as a function of:

- the alcoholic strength by weight p expressed as a decimal (6),- the temperature t in °C (EIPT 68),- the numerical coefficients below.The formula is valid for temperatures between 20 and +40 °C.

Numerical coefficients in the formula

TABLE I INTERNATIONAL ALCOHOLIC STRENGTH AT 20 °CTable of apparent densities of ethanol-water mixtures - Pyrex pycnometerDensities at t °C corrected for air buoyancy

TABLE II INTERNATIONAL ALCOHOLIC STRENGTH AT 20 °CTable of corrections to be applied to the apparent alcoholic strength to correct for the effect of temperatureAdd or subtract from the apparent alcoholic strength at t °C (ordinary glass alcoholmeter) the correction indicated below

TABLE III INTERNATIONAL ALCOHOLIC STRENGTH AT 20 °CTable of apparent densities of ethanol-water mixtures - Ordinary glass apparatusDensities at t °C corrected for air buoyancy

TABLE IV Table giving the refractive indices of pure ethanol-water mixtures and distillates at 20 °C and the corresponding alcoholic strengths at 20 °C

4. TOTAL DRY EXTRACT

Total dry matter1. DEFINITION

The total dry extract or the total dry matter includes all matter which is non-volatile under specified physical conditions. These physical conditions must be such that the matter forming the extract undergoes as little alteration as possible while the test is being carried out.

The sugar-free extract is the difference between the total dry extract and the total sugars.

The reduced extract is the difference between the total dry extract and the total sugars in excess of 1 g/l, potassium sulphate in excess of 1 g/l, any mannitol present and any other chemical substances which may have been added to the wine.

The residual extract is the sugar-free extract less the fixed acidity expressed as tartaric acid.

The extract is expressed in grams per litre and it should be determined to within the nearest 0,5 g.2. PRINCIPLE OF THE METHOD

Single method: measurement by a densimeter

The total dry extract is calculated indirectly from the specific gravity of the must and, for wine, from the specific gravity of the alcohol-free wine.

This dry extract is expressed in terms of the quantity of sucrose which, when dissolved in water and made up to a volume of one litre, gives a solution of the same specific gravity as the must or the alcohol-free wine. This quantity is shown in Table I.3. METHOD OF CALCULATION

The 20/20 specific gravity drof the 'alcohol-free wine' is calculated using the following formula:dr = dv da + 1,000where:dv = specific gravity of the wine at 20 °C (corrected for volatile acidity) (7),da = specific gravity at 20 °C of a water-alcohol mixture of the same alcoholic strength as the wine.

dr may also be calculated from the densities at 20 °C, rv of the wine and ra of the water-alcohol mixture of the same alcoholic strength by the formula:dr = 1,0018 (rv ra) + 1,000where the coefficient 1,0018 approximates to 1 when rv is below 1,05, which is most often the case.4. EXPRESSION OF RESULTS

Table I should be used for calculating the total dry extract in g/l from the 20/20 specific gravity drof the alcohol-free wine or from the specific gravity d20 °C

20 °C of the must.

The total dry extract is expressed in g/l to one decimal place.

TABLE I for the calculation of the content total dry extract (g/l)

Interpolation table

5. REDUCING SUGARS 1. DEFINITION

Reducing sugars comprise all the sugars exhibiting ketonic and aldehydic functions and are determined by their reducing action on an alkaline solution of a copper salt.2. PRINCIPLE OF THE METHODS

2.1. Clarification

2.1.1. Reference method:after neutralization and removal of alcohol, the wine is passed through an ion exchange column in which its anions are exchanged for acetate ions, followed by clarification with neutral lead acetate.

2.1.2. Usual methods: the wine is treated with one of the following reagents:

2.1.2.1. Neutral lead acetate;

2.1.2.2. Zinc 2-hexacyanoferrate.2.2. Determination

2.2.1. Single method: the clarified wine or must is reacted with a specific quantity of an alkaline copper salt solution and the excess copper ions are then determined iodometrically.3. CLARIFICATION

The sugar content of the liquid in which sugar is to be determined must lie between 0,5 and 5 g/l.

Dry wines should not be diluted during clarification; sweet wines should be diluted during clarification in order to bring the sugar level to within the limits prescribed in the following table:

DescriptionSugar content

(g/l)DensityDilution

(%)Musts and mistelles>125>1,0381Sweet wines, whether fortified or not25 to 1251,005 to 1,0384Semi-sweet wines5 to 250,997 to 1,00520Dry wines< 5< 0,997No dilution3.1. Reference method

3.1.1. Reagents

3.1.1.1. 1 M solution hydrochloric acid (HCl);

3.1.1.2. 1 M solution sodium hydroxide (NaOH);

3.1.1.3. 4 M solution acetic acid (CH3COOH);

3.1.1.4. 2 M solution sodium hydroxide (NaOH);

3.1.1.5. Anion exchange resin (Dowex 3 (20-50 mesh) or equivalent resin).

Preparation of the anion exchange resin column

Place a small plug of glass wool and 15 ml of the anion exchange resin (3.1.1.5) in the bottom of the burette.

Before the resin is used, subject it to two complete cycles of regeneration by passing alternately the 1 M solutions of hydrochloric acid (3.1.1.1) and sodium hydroxide (3.1.1.2) through it. After rinsing with 50 ml of distilled water, transfer the resin to a beaker, add 50 ml of the 4 M solution of acetic acid (3.1.1.3) and stir for five minutes. Refill the burette with resin and pour 100 ml of the 4 M acetic acid solution (3.1.1.3) through the column. (It is preferable to have a stock of the resin stored in a bottle filled with this 4 M acetic acid solution.) Wash the column with distilled water until the effluent is neutral.

Regeneration of the resin

Pour 150 ml of a 2 M sodium hydroxide solution through the resin to remove acids and most of the pigments fixed to the resin. Rinse with 100 ml of water, and then pour 100 ml of 4 M acetic acid solution through it. Wash the column until the effluent is neutral.

3.1.1.6. Neutral lead acetate solution (approximately saturated)Neutral lead acetate [Pb (CH3 COO)2 · 3 H2O], 250 g;

very hot water to 500 ml;

stir until dissolved.

3.1.1.7. Calcium carbonate (Ca CO3)3.1.2. Procedure

3.1.2.1. Dry wines

Place 50 ml of the wine in a beaker having a diameter of about 10 to 12 cm together with ½ (n 0,5) ml of 1 M sodium hydroxide solution (3.1.1.2) (n being the volume of 0,1 M sodium hydroxide solution used for determining total acidity in 10 ml of wine), and evaporate over a boiling water bath in a stream of hot air until the liquid is reduced to about 20 ml.

Pour this liquid through an anion exchange resin column in acetate form (3.1.1.5) at a rate of 3 ml every two minutes. Collect the effluent in a 100 ml volumetric flask. Wash the vessel and column six times using 10 ml of distilled water each time. Stirring all the time, add 2,5 ml of saturated lead acetate solution (3.1.1.6) and 0,5 g of calcium carbonate (3.1.1.7) to the effluent: shake several times and allow to stand for at least 15 minutes. Make up to the mark with water. Filter.

1 ml of this filtrate corresponds to 0,5 ml of wine.

3.1.2.2. Musts, mistelles, sweet and semi-sweet wines:

The dilutions below are given for guidance.

1. Musts and mistelles: prepare a 10 % solution of the liquid to be analysed and take 10 ml of the diluted sample.

2. Sweet wines, whether fortified or not,having a density between 1,005 and 1,038: prepare a 20 % solution of the liquid to be analysed and take 20 ml of the diluted sample.

3. Semi-sweet wines having a density at 20 °C between 0,997 and 1,005: take 20 ml of the undiluted wine.Allow the abovementioned volume of wine or must to flow through an anion exchange column in acetate form at a rate of 3 ml every two minutes. Collect the effluent in a 100 ml volumetric flask, and rinse the column with water until about 90 ml of the effluent is obtained. Add 0,5 g calcium carbonate and 1 ml saturated lead acetate solution to the effluent. Stir and allow to stand for 15 minutes, stirring occasionally. Make up to the mark with water. Filter.

In case:1. 1 ml of filtrate corresponds to 0,01 ml of must or mistelle.2. 1 ml of filtrate corresponds to 0,04 ml of sweet wine.3. 1 ml of filtrate corresponds to 0,20 ml of semi-sweet wine.

3.2. Usual methods

3.2.1. Clarification by neutral lead acetate

3.2.1.1. Reagents

Solution of neutral lead acetate (approximately saturated) (see 3.1.1.6).Calcium carbonate.3.2.1.2. Procedure

3.2.1.2.1. Dry wines: place 50 ml of the wine in a 100 ml volumetric flask; add ½ (n 0,5) ml of a 1 M solution of sodium hydroxide (3.1.1.2) (where n is the volume of a 0,1 M solution of sodium hydroxide used to determine the total acidity in 10 ml of wine). Add, while stirring, 2,5 ml of saturated lead acetate solution (3.1.1.6) and 0,5 g calcium carbonate (3.1.1.7). Shake several times and allow to stand for at least 15 minutes. Make up to the mark with water. Filter.1 ml of the filtrate corresponds to 0,5 ml of the wine.3.2.1.2.2. Musts, mistelles, sweet and semi-sweet wines: into a 100 ml volumetric flask, place the following volumes of wine (or must or mistelle), the dilutions being given for guidance:

1. Musts and mistelles: prepare a 10 % solution of the liquid to be analysed and take 10 ml of the diluted sample.

2. Sweet wines, whether fortified or not, having a density between 1,005 and 1,038: prepare a 20 % solution of the liquid to be analysed and take 20 ml of the diluted sample.

3. Semi-sweet wineshaving a density between 0,997 and 1,005: take 20 ml of the undiluted wine.Add 0,5 g calcium carbonate, about 60 ml water and 0,5, 1 or 2 ml of saturated lead acetate solution. Stir and leave to stand for at least 15 minutes, stirring occasionally. Make up to the mark with water. Filter.

In case:1. 1 ml of filtrate corresponds to 0,01 ml of must or mistelle.2. 1 ml of filtrate corresponds to 0,04 ml of sweet wine.3. 1 ml of filtrate corresponds to 0,20 ml of semi-sweet wine.3.2.2. Clarification by zinc 2-hexacyanoferrate

This clarification process should be used only for white wines, lightly coloured sweet wines and musts.3.2.2.1. Reagents

3.2.2.1.1. Solution I, potassium 2-hexacyanoferrate:

Potassium 2-hexacyanoferrate (K4 Fe (CN)6 · 3 H2O), 150 g;

Water to 1 000 ml.

3.2.2.1.2. Solution II, zinc sulphate:

Zinc sulphate (Zn SO4 · 7 H2O), 300 g;

Water to 1 000 ml.

3.2.2.2. Procedure

Into a 100 ml volumetric flask, place the following volumes of wine (or must or mistelle), the dilutions being given for guidance:

1.Musts and mistelles: prepare a 10 % solution of the liquid to be analysed and take 10 ml of the diluted sample.

2. Sweet wines, whether fortified or not, having a density between 1,005 and 1,038: prepare a 20 % solution of the liquid to be analysed and take 20 ml of the diluted sample.

3. Semi-sweet wineshaving a density between 0,997 and 1,005: take 20 ml of the undiluted wine.

4. Dry wines: take 50 ml of undiluted wine.Add 5 ml of solution I, potassium 2-hexacyanoferrate (3.2.2.1.1), and 5 ml of solution II, zinc sulphate (3.2.2.1.2). Mix. Make up to the mark with water. Wait 10 minutes. Filter.

In case:1. 1 ml of filtrate corresponds to 0,01 ml of must or mistelle.2. 1 ml of filtrate corresponds to 0,04 ml of sweet wine.3. 1 ml of filtrate corresponds to 0,20 ml of semi-sweet wine.4. 1 ml of filtrate corresponds to 0,50 ml of dry wine.

4. DETERMINATION OF SUGARS

4.1. Reagents

4.1.1 Alkaline copper salt solution:copper sulphate, pure, CuSO4 · 5H2O 25 g

citric acid (C6H8O7 · H2O) 50 g

crystalline sodium carbonate, Na2CO3 · 10H2O 388 g

water to 1 000 mlDissolve the copper sulphate in 100 ml of water, the citric acid in 300 ml of water and the sodium carbonate in 300 to 400 ml of hot water. Mix the citric acid and sodium carbonate solutions. Add the copper sulphate solution and make up to one litre.4.1.2. 30 % potassium iodide solution:potassium iodide (KI) 30 g

water to 100 mlStore in a coloured glass bottle.4.1.3. 25 % sulphuric acid:concentrated sulphuric acid, (H2SO4) r20 = 1,84 g/ml 25 g

water to 100 mlAdd the acid slowly to the water, allow to cool and make up to 100 ml with water.4.1.4. 5 g/l starch solution:

Mix 5 g of starch in with about 500 ml of water. Bring to the boil, stirring all the time, and boil for 10 minutes. Add 200 g of sodium chloride (NaCl). Allow to cool and then make up to one litre with water.Sodium thiosulphate, 0,1 M solutionInvert sugar solution, 5 g/l, to be used for checking the method of determination:

Place the following into a 200 ml volumetric flask:pure dry sucrose (C12H22O11) 4,75 g

water, approximately 100 ml

concentrated hydrochloric acid (HCl) (r20 = 1,16 - 1,19 g/ml) 5 mlHeat the flask in a water-bath maintained at 60 °C until the temperature of the solution reaches 50 °C; then keep the flask and solution at 50 °C for 15 minutes. Allow the flask to cool naturally for 30 minutes and then immerse it in a cold water-bath. Transfer the solution to a one-litre volumetric flask and make up to one litre. This solution keeps satisfactorily for a month. When it is to be used, neutralize the test sample (the solution being approximately 0,06 M acid) with sodium hydroxide solution.4.2. Procedure

Mix 25 ml of the alkaline copper salt solution, 15 ml water and 10 ml of the clarified solution in a 300 ml conical flask. This volume of sugar solution must not contain more than 60 mg of invert sugar.

Add a few small pieces of pumice stone. Fit a reflux condenser to the flask and bring the mixture to the boil within two minutes. Keep the mixture boiling for exactly 10 minutes.

Cool the flask immediately in cold running water. When completely cool, add 10 ml of 30 % potassium iodide solution (4.1.2), 25 ml of 25 % sulphuric acid (4.1.3) and 2 ml of starch solution (4.1.4).

Titrate with 0,1 M sodium thiosulphate solution (4.1.5) Let nbe the number of ml used.

Also carry out a blank titration in which the 10 ml of sugar solution is replaced by 10 ml of distilled water. Let n2 be the number of ml of sodium thiosulphate used.4.3. Expression of results

4.3.1. Calculations

The quantity of sugar, expressed as invert sugar, contained in the test sample is given in the table below as a function of the number (n2 - n) of ml of sodium thiosulphate used.

The sugar content of the wine is to be expressed in grams of invert sugar per litre to one decimal place, account being taken of the dilution made during clarification and of the volume of the test sample.4.3.2. Repeatability

r = 0,015 xixi = concentration of inverted sugar in g/l per sample4.3.3. Reproducibility

R = 0,058 xixi = concentration of inverted sugar in g/l per sample

6. SUCROSE 1. PRINCIPLE OF METHODS

II. For qualitative testing by thin-layer chromatography: sucrose is separated from other sugars using thin-layer chromatography on plate coated with cellulose. The developing agent is urea-hydrochloric acid at 105 ° C.

II. For testing and determination by high-performance liquid chromatography: the sucrose is separated in a column of alkylamine-bonded silica and detected by refractometry. The result is quantified by reference to an external standard analysed under the same conditions.

Note: Authentication of a must or of a wine may be checked by the method using NMR of deuterium described for detecting the enrichment of musts, rectified concentrated musts and wines.For testing and determination of sucrose, chromatography in gaseous phase may also be used, as described in chapter 42, point (f).2. QUALITATIVE TESTING BY THIN-LAYER CHROMATOGRAPHY

2.1. Equipment

2.1.1. Chromatograhic plates covered with a desired thickness of cellulose powder (e.g. MN 300) (20 × 20).

2.1.2. Chromatography tank.

2.1.3. Micrometric syringe or micropipette.

2.1.4. Oven with regulation to 105 ± 2 °C.2.2. Reagents

2.2.1. Decolourizing charcoal.

2.2.2. Mobile phase:Dichloromethane - glacial acetic acid (p20 - 1,05 g/ml)- ethanol - methanol - water (50:25:9:6:10).

2.2.3. Developing agentUrea 5 g

Hydrochloric acid 2 M 20 ml

Ethanol 100 ml

2.2.4. Reference solutionsGlucose 35 g

Fructose 35 g

Sucrose 0,5 g

Distilled water1 000 ml2.3. Procedure

2.3.1. Preparation of sample

When the must or wine is strongly coloured, decolourize it by treating it with activated charcoal.

For rectified concentrated musts, use the solution with a sugar concentration of 25 % by mass (25 ° Brix) prepared as described in the chapter 'pH of wine and must', section 4.1.2, and dilute it with water to a quarter of its concentration by making 25 ml up to 100 ml in a volumetric flask.2.3.2. Obtaining the chromatogram

Place on a parallel line 2,5 cm from the bottom edge of the plate:- 10 µl of sample- 10 µl of standard.

Place the plate in the tank, previously saturated with the vapour from the mobile phase. Allow the mobile phase to migrate to within 1 cm of the top of the plate. Remove the plate and dry it in a current of warm air. Repeat the migration two more times, drying the plate each time. Spray the plate uniformly with 15 ml of colouring agent and place in the oven at 105 °C for approximately five minutes.2.4. Results

Saccharose and fructose appear as a deep blue spot on a white background: glucose gives a less intense green spot.3. TESTING AND DETERMINATION BY HIGH-PERFORMANCE LIQUID CHROMATOGRAPHY

The chromatographic conditions are given for guidance3.1. Equipment

3.1.1. High-performance liquid chromatograph equipped with:

1. 10 µl loop injector,2. a detector: a differential refractometer or an interferometer refractometer,3. an alkylamine-bonded silica column (length 25 cm, internal diameter 4 mm),4. a guard column filled with the same phase,5. an arrangement for insulating the guard column and analytical columns or for maintaining their temperature (30 ° C),6. a recorder and, if required, an integrator,7. mobile phase flow rate: 1 ml/min.3.1.2. Arrangement for membrane filtration (0,45 µm).3.2. Reagents

3.2.1. Doubly distilled water.

3.2.2. HPLC quality acetonitrile (CH3CN).

3.2.3. Mobile phase: acetonitrile-water, previously subjected to membrane filtration (0,45 µm), (80:20 v/v).This mobile phase must be outgassed before being used.

3.2.4. Standard solution: 1,2 g/l aqueous sucrose solution. Filter using a 0,45 µm membrane filter.

3.3. Procedure

3.3.1. Preparation of sample:

- For wines and musts: filter using a 0,45 µm membrane filter.- For rectified concentrated musts: use the solution obtained by diluting the rectified concentrated must to 40 % (m/v) as described in the chapter 'Total acidity', section 5.1.2., and filtering it using a 0,45 µm membrane filter.3.3.2.Chromatographic determination

Inject in turn into the chromatograph 10 µl of the standard solution and 10 µm of the sample prepared as described in 3.3.1. Repeat these injections in the same order.Record the chromatogram.The retention time of the sucrose is approximately 10 minutes.3.4. Calculations

For the calculation, use the average of two results for the standard solution and the sample.3.4.1. For wines and musts:calculate the concentration in g/l.3.4.2. For rectified concentrated musts: let C be the sucrose concentration in g/l of the 40 % (m/v) solution of rectified concentrated must. The sucrose concentration in g/kg of the rectified concentrated must is then: 2,5C.3.5. Expression of results

The sucrose concentration in wines, musts and rectified concentrated musts is expressed in grams per litre for wines and musts and in grams per kilogram for rectified concentrated musts, each to one place of decimals.

7. GLUCOSE AND FRUCTOSE 1. DEFINITION

Glucose and fructose may be determined individually by an enzymatic method, with the sole aim of calculating the glucose/fructose ratio.2. PRINCIPLE OF THE METHOD

Glucose and fructose are phosphorylated by adenosine triphosphate (ATP) during an enzymatic reaction catalysed by hexokinase (HK), and produce glucose 6-phosphate (G6P) and fructose 6-phosphate (F6P):

glucose + ATP G6P + ADP

fructose + ATP F6P + ADP

The glucose 6-phosphate is first oxidized to gluconate 6-phosphate by nicotinamide adenine dinucleotide phosphate (NADP) in the presence of the enzyme glucose 6-phosphate dehydrogenase (G6PDH). The quantity of reduced nicotinamide adenine dinucleotide phosphate (NADPH) produced corresponds to that of glucose 6-phosphate and thus to that of glucose.

G6P + NADP+ gluconate 6-phosphate + NADPH + H+

The reduced nicotinamide adenine dinucleotide phosphate is determined from its absorption at 340 nm.

At the end of this reaction, the fructose 6-phosphate is converted into glucose 6-phosphate by the action of phosphoglucose isomerase (PGI):

The glucose 6-phosphate again reacts with the nicotinamide adenine dinucleotide phosphate to give gluconate 6-phosphate and reduced nicotinamide adenine dinucleotide phosphate, and the latter is then determined.3. APPARATUS

- A spectrophotometer enabling measurements to be made at 340 nm, the wavelength at which absorption by NADPH is at a maximum. Absolute measurements are involved (i.e. calibration plots are not used but standardization is made using the extinction coefficient of NADPH), so that the wavelength scales of and absorbence values obtained from the apparatus must be checked.If not available, a spectrophotometer using a source with a discontinuous spectrum which enables measurements to be made at 334 nm or at 365 nm may be used.- Glass cells with optical path lengths of 1 cm or single-use cells.- Pipettes for use with enzymatic test solutions, 0,02, 0,05, 0,1, 0,2 ml.4. REAGENTS

4.1. Solution 1:buffer solution (0,3 M triethanolamine, pH 7,6, 4 × 10 3 M in Mg 2+): dissolve 11,2 g triethanolamine hydrochloride ((C2H5)3N · HCl) and 0,2 g MgSO4 · 7H2O in 150 ml of doubly distilled water, add about 4 ml of 5 M sodium hydroxide (NaOH) solution to obtain a pH value of 7,6 and make up to 200 ml.

This buffer solution may be kept for four weeks at +4 °C.

4.2. Solution 2: nicotinamide adenine dinucleotide phosphate solution (about 11,5 × 10 3 M): dissolve 50 mg disodium nicotinamide adenine dinucleotide phosphate in 5 ml of doubly distilled water.

This solution may be kept for four weeks at +4 °C.

4.3. Solution 3: adenosine 52-triphosphate solution (about 81 × 10 3 M): dissolve 250 mg disodium adenosine 52-triphosphate and 250 mg sodium hydrogencarbonate (NaHCO3) in 5 ml of doubly distilled water.

This solution may be kept for four weeks at +4 °C.

4.4. Solution 4: hexokinase/glucose 6-phosphate dehydrogenase: mix 0,5 ml hexokinase (2 mg of protein/ml or 280 U/ml) with 0,5 ml glucose 6-phosphate dehydrogenase (1 mg of protein/ml).

This mixture may be kept for a year at about +4 °C.

4.5. Solution 5: phosphoglucose isomerase (2 mg of protein/ml or 700 U/ml). The suspension is used undiluted.

This may be kept for a year at about +4 °C.

Note:All solutions used above are available commercially.5. PROCEDURE

5.1. Preparation of sample

Depending on the estimated amount of glucose + fructose per litre, dilute the sample as follows:

Measurement

at 340 and 334 nmMeasurement

at 365 nmDilution with waterDilution factor Fup to 0,4 g/l 0,8 g/l--up to 4,0 g/l 8,0 g/l1 + 9 10up to 10,0 g/l20,0 g/l1 + 24 25up to 20,0 g/l40,0 g/l1 + 49 50up to 40,0 g/l80,0 g/l1 + 99 100above 40,0 g/l80,0 g/l1 + 9991 000

5.2. Determination

With the spectrophotometer adjusted to the 340 nm wavelength, make measurements using air (no cell in the optical path) or water as reference.Temperature between 20 and 25 °C.Into two cells with 1 cm optical paths, place the following:

Reference cell Sample cellSolution 1 (4.1) (taken to 20 °C) 2,50 ml 2,50 ml

Solution 2 (4.2) 0,10 ml0,10 ml

Solution 3 (4.3) 0,10 ml 0,10 ml

Sample to be measured 0,20 ml

Doubly distilled water 0,20 ml

Mix, and after about three minutes read off the absorbence of the solutions (A1). Start the reaction by adding:Solution 4 (4.4) 0,02 ml 0,02 ml

Mix; wait 15 minutes; read off the absorbence and check that the reaction has stopped after a further two minutes (A2).

Add immediately:Solution 5 (4.5) 0,02 ml 0,02 ml

Mix; read off the absorbence after 10 minutes and check that the reaction has stopped after a further two minutes (A3).

Calculate the differences in the absorbences:A2 A1 corresponding to glucose,A3 A2 corresponding to fructose,for the reference and sample cells.

Calculate the differences in absorbence for the reference cell (DAR) and the sample cell (DAS) and then obtain:for glucose: DAG = DAS DARfor fructose: DAF = DAS DAR

Note:The time needed for the completion of enzyme activity may vary from one batch to another. The above value is given only for guidance and it is recommended that it be determined for each batch.5.3.Expression of results

5.3.1. Calculation

The general formula for calculating the concentrations is:where V = volume of the test solution (ml)v = volume of the sample (ml)M = molecular mass of the substance to be determinedd = optical path in the cell (cm)e = absorption coefficient of NADPH at 340 nm (= 6,3 mM 1 × l × cm 1)and V = 2,92 ml for the determination of glucoseV = 2,94 ml for the determination of fructosev = 0,20 mlM = 180d = 1

so that:For glucose: C (g/l) = 0,417 DAG

For fructose: C (g/l) = 0,420 DAF

If the sample was diluted during its preparation, multiply the result by the dilution factor F.

Note:If the measurements are made at 334 or 365 nm, then the following expressions are obtained:- measurement at 334 nm: e = 6,2 (mmole 1 × 1 × cm 1)For glucose: C (g/l) = 0,425 DAG

For fructose: C (g/l) = 0,428 DAF- measurement at 365 nm: e = 3,4 (mmole × 1 1 × cm 1)For glucose: C (g/l) = 0,773 DAG

For fructose: C (g/l) = 0,778 DAF5.3.2. Repeatability (r)

r = 0,056 xi5.3.3. Reproducibility (R)

R = 0,12 + 0,076 xi

xi = concentration of glucose or fructose in g/l.

8. DETECTING ENRICHMENT OF GRAPE MUSTS, CONCENTRATED GRAPE MUSTS, RECTIFIED CONCENTRATED GRAPE MUSTS AND WINES BY APPLICATION OF NUCLEAR MAGNETIC RESONANCE OF DEUTERIUM (SNIF-NMR/RMN-FINS) 1. DEFINITION

The deuterium contained in the sugars and the water in grape must will be redistributed after fermentation in molecules I, II, III and IV of the wine:CH2D CH2 OH CH3 CHD OH

I IICH3 CH2 OD HOD

III IV

The addition of exogenous sugar (sugaring in the dry) before the must ferments will have an effect on the distribution of the deuterium.

As compared with the figures for parameters for a natural control wine from the same region, the enrichment of an exogenous sugar will lead to the following variations:

Parameters

Wine(D/H)I(D/H)II(D/H)QWR- Natural - Enriched: - beet sugar - cane sugar - maize sugar

(D/H)I : Isotope ratio associated with molecule I(D/H)II : Isotope ratio associated with molecule II (D/H)QW : Isotope ratio of the water in the wine

R = 2(D/H)II/(D/H)I expresses the relative distribution of deuterium in molecules I and II; R is measured directly from the h-intensities of the signals and thenR = 3hII/hI.

(D/H)I mainly characterizes the vegetable species which synthesized the sugar and to a lesser extent the geographical location of the place of harvest (type of water used during photosynthesis).

(D/H)II represents the climatology of the place of production of the grapes (type of rainwater and weather conditions) and to a lesser extent the sugar concentration of the original must.

(D/H)QW represents the climatology of the place of production and the sugar content of the original must.2. PRINCIPLE

The parameters defined above (R,(D/H)I, (D/H)II) are determined by nuclear magnetic resonance of the deuterium in the ethanol extracted from the wine or from the fermentation products of the must, the concentrated must or the rectified concentrated must obtained under given conditions; they may be supplemented by determining the isotope ratio of the water extracted from the wine, (D/H)QW and by determining the ratio13C/12C in the ethanol.

Pending the creation of a Community data bank proceed as follows:In the case of wines, control samples taken in the regions must be accompanied by natural control samples (at least three) of the same origin (geographical place and vintage); three series of such samples are to be taken.In the case of musts, concentrated musts and rectified concentrated musts, three series of control samples are to be made up of natural musts of the same origin (geographical place and vintage).

For checks on products prepared within their own territory and pending the establishment of a Community data bank, the Member States may on a transitional basis use a national data bank.3. PREPARATION OF THE SAMPLE FOR ANALYSIS

3.1. Extraction of ethanol and water in the wine

Note:Any method for ethanol extraction can be used as long as 98 to 98,5 % of the total alcohol in the wine is recovered in a distillate which contains 92 to 93 % mas (95 % vol).3.1.1. Apparatus and reagents

Apparatus for extracting ethanol (Figure 1) comprising:- electric heating mantle with voltage regulator,- one-litre round-bottom flask with ground glass neck joint,- Cadiot column with rotating band (moving part in Teflon),- 125 ml conical flasks with ground glass neck joints,- 125 and 60 ml bottles with plastic stoppers.Reagents for the determination of water by the Karl Fischer method (e.g. Merck 9241 and 9243).3.1.2. Procedure

3.1.2.1. Determine the alcoholic strength of the wine (tv) to better than the nearest 0,05 % vol.3.1.2.2. Extraction of the ethanol

Introduce a homogeneous sample of 500 ml of wine of alcoholic strength tv into the flask in the distillation apparatus with a constant reflux ratio of about 0,9. Place a 125 ml ground conical flask, calibrated beforehand, to receive the distillate. Collect the boiling liquid between 78,0 and 78,2 °C, i.e. approximately 40 to 60 ml. If the temperature exceeds 78,5 °C, discontinue collection for five minutes.

When the temperature returns to 78 °C, recommence collecting the distillate until 78,5 °C and repeat this operation until the temperature, after discontinuing collection and operating within a closed circuit, remains constant. Complete distillation lasts approximately five hours. This procedure enables between 98 and 98,5 % of the total alcohol in the wine to be recovered in a distillate with a strength of between 92 and 93 % mas (95 % vol), a strength for which the NMR conditions described in section 4 have been established.

The ethanol collected is weighed.

A homogeneous 60 ml sample of the residues is kept in a 60 ml flask and represents the water in the wine. Its isotope ratio may be determined if required.

Note:If a spectrometer fitted with a 10 mm probe is available (cf. section 4), a homogeneous test sample of 300 ml of wine is sufficient.3.1.2.3. Determination of the alcoholic strength of the alcohol extracted

The water content (p2 g) is determined by the Karl Fischer method using a sample of about 0,5 ml of alcohol of exactly known mass p.

Figure 1Distillation apparatus for the extraction of ethanol

The strength by mass of the alcohol is given by3.2. Fermentation of musts, concentrated musts and rectified concentrated musts

3.2.1. Apparatus and reagents

Tartaric acidDIFCO Bacto Yeast Nitrogen Base without amino acidsActive dry yeasts (Saccharomyces cerevisae)

If the isotope ratio of the must is known the yeast can be reactivated prior to use for 15 minutes in a minimum amount of lukewarm non-distilled water, so that the isotope ratio is similar to that of the must.

If the isotope ratio of the must is not known it is better to use fresh/direct.

Fermentation vessel of a capacity of 1,5 litres fitted with a device to keep it airtight and to condense alcohol vapour, since no loss of ethanol during fermentation must be tolerated. The rate of conversion of fermentable sugars into ethanol should be greater than 98 %.3.2.2. Procedure

3.2.2.1. Musts

- Fresh mustsPlace one litre of must, whose concentration of fermentable sugars has been previously determined, in the fermentation vessel. Add 1 g of dry yeast reactivated beforehand. Insert device to keep it airtight. Allow fermentation to proceed at around 20 °C until the sugar is used up. After determining the alcoholic strength of the fermentation product and calculating the rate of conversion of sugars into alcohol, the fermented liquid is centrifuged and distilled to extract the ethanol.

- Musts with fermentation prevented by the addition of sulphur dioxideDe-sulphite a quantity of must slightly in excess of one litre (i.e. 1,2 litre) by bubbling nitrogen through the must in a water bath at 70 to 80 °C under reflux until the total sulphur dioxide content is less than 200 mg/l. Take care to see that the must is not concentrated through evaporation of water by using effective cooling. Place 1 litre of de-sulphited must in the fermentation vessel and continue as described for fresh must.Note:If potassium metabisulphite is used to sulphite the must, 0,25 ml of sulphuric acid (r20 = 1,84 g/ml) per gram of metabisulphite used per litre of must should be added to the must before de-sulphiting.3.2.2.2. Concentrated musts

Place V ml of concentrated must containing a known amount of sugar (approximately 170 g) into the fermentation vessel. Top up to one litre with (1 000 V) ml of water from the normal water supply of same isotope ratio as natural must samples. Add (3.2.1) dry yeasts (1 g) and 3 g of DIFCO Bacto Yeast Nitrogen Base without amino acids. Homogenize and proceed as before.3.2.2.3. Rectified concentrated musts

Proceed as described in 3.2.2.2, topping up to one litre with (1 000 V) ml of water from the normal water supply of same isotope ratio but also containing 3 g dissolved tartaric acid.

Note:Retain 50 ml of sample of must or sulphur dioxide treated must or concentrated must or rectified concentrated must with a view to the possible extraction of the water and the determination of its isotope ratio (D/H)WQ. The extraction of the water contained in the must may be very simply carried out by azotropic distillation using toluene.

3.3. Preparation of alcohol sample for NMR measurement

3.3.1. Reagents

N, N-tetramethyl urea (TMU); use a sample of standard TMU with a given, monitored isotope ratio D/H. This sample may be supplied by:Directorate-General for Science, Research and Development,

Community Bureau of References,

200 rue de la Loi, B-1049 Brussels.3.3.2. Procedure

- 15 mm diameter NMR probe:in a previously weighed bottle, collect 7 ml alcohol obtained as in 3.1.2 and weigh it to the nearest 0,1 mg (mA); then take a 3 ml sample of the internal standard (TMU) and weigh to the nearest 0,1 mg (mst). Homogenize by shaking.

- 10 mm diameter NMR probe:

3,2 ml of alcohol and 1,3 ml TMU are sufficient.Depending on the type of spectrometer and probe used (cf. section 4), add a sufficient quantity of hexafluorobenzene as a field-frequency stabilization substance (lock):

Spectrometer10 mm probe15 mm probe7,05 T150 µl200 µl9,4 T 35 µl 50 µl

3.4. Preparation of a water sample for the NMR measurement, for the purpose of a possible determination of its isotope ratio3.4.1. Reagents

N, N-tetramethyl urea (TMU) : see 3.3.1.

3.4.2. Procedure

Place 3 ml of water obtained as in 3.1.2 or 3.2 (note) into a tared flask and weigh to the nearest 0,1 mg (m2E). Place 4 ml of internal standard (TMU) and weigh to the nearest 0,1 mg (m2st). Homogenize by shaking.

Note:If the laboratory has a mass spectrometer for determining isotope ratios, the measurement may be carried out on this instrument to reduce the load on the NMR spectrometer. It is necessary to standardize the ratio Trv (5.2) for each series of wines examined.4. RECORDING OF ²H NMR SPECTRA OF THE ALCOHOL AND THE WATER

Determination of isotope parameters.4.1. Apparatus

- NMR spectrometer fitted with a specific 'deuterium' probe tuned to the characteristic frequency Vo of the field Bo (e.g. for Bo = 7,05 T, Vo = 46,05 MHz and for Bo = 9,4 T, Vo = 61,4 MHz) having a proton decoupling channel (B2) and field-frequency stabilization channel (lock) at the fluorine frequency.

The resolution measured on the spectrum, transformed without exponential multiplication (i.e. LB = 0) (Figure 2b) and expressed by the half-width of the methyl and methylene signals of ethanol and the methyl signal of TMU, must be less than 0,5 Hz. The sensitivity, measured with an exponential multiplying factor LB equal to 2 (Figure 2a) must be greater than or equal to 150 for the methyl signal of ethanol of alcoholic strength 95 % vol (93,5 % mas).

Under these conditions, the confidence interval for the measurement of the signal height, calculated for a 97,5 % probability (one-sided test) and 10 repetitions of the spectrum, is 0,35 %.- Automatic sample changer (possibly)- Data-processing software- 15 mm or 10 mm sample tubes according to spectrometer performance.4.2. Standardization of spectrometer and checks

4.2.1. Standardization

Carry out customary standardization for homogeneity and sensitivity according to the manufacturer's specifications.4.2.2. Checking the validity of the standardization

Use standard ethanols, designated by the letters C, V and B, having isotope concentrations that are different but accurately standardized. They carry the following meaning:- C: alcohol from cane sugar or maize,- V: wine spirit,- B: beet alcohol.These samples are supplied by the Community Bureau of References.

Following the procedure described in 4.3, determine the isotope values of these alcohols, denoting them Cmeas, Vmeas, Bmeas (see 5.3).

Compare them with the given corresponding standard values, denoted by a superscript Cst, Bst, Vst (see 5.3).

Figure 2a²H NMR spectrum of an ethanol from wine with an internal standard (TMU: N, N-tetramethylurea)

Figure 2b²H spectrum of ethanol taken under the same conditions as those of Figure 2a, but without exponential multiplication (LB = 0)

The standard deviation for repeatability obtained on an average of 10 repetitions of each spectrum must be less than 0,01 for the ratio R and less than 0,3 ppm for (D/H)I and (D/H)II.

The average values obtained for the various isotopic parameters (R, (D/H)I, (D/H)II) must be within the corresponding standard deviation of repeatability given for those parameters for the three standard alcohols by the Community Bureau of References. If they are not, carry out the checks again.4.3. Conditions for obtaining NMR spectra

Place a sample of alcohol prepared as in 3.3 (or the water sample, prepared as in 3.4) in a 15-mm or 10-mm tube and introduce it into the probe.

The conditions for obtaining NMR spectra are as follows:- a constant probe temperature (e.g. 302 K);- acquisition time of at least 6,8 s for 1 200 Hz spectral width (16K memory) (i.e. about 20 ppm at 61,4 MHz or 27 ppm at 46,1 MHz);- 90° pulse;- adjustment of acquisition time: its value must be of the same order as the dwell time;- parabolic detection: fix the offset 01 between the OD and CHD reference signals for ethanol and between the HOD and TMU reference signals for water;- determine the value of the decoupling offset 02 from the proton spectrum measured by the decoupling coil on the same tube. Good decoupling is obtained when 02 is located in the middle of the frequency interval existing between the CH3- and CH2- groups. Use the wide band decoupling mode.

For each spectrum, carry out a number of accumulations NS sufficient to obtain the signal-to-noise ratio given in 4.1 and repeat this set of NS accumulations NE = 10 times. The values of NS depend on the types of spectrometer and probe used (cf. section 4). Examples of the possible choices are:

Spectrometer10 mm probe15 mm probe7,05 TNS = 304NS = 2009,4 TNS = 200NS = 128

5.EXPRESSION OF RESULTS

5.1. Ethanol

For each of the 10 spectra (see NMR spectrum for ethanol, Figure 2a), determine:- mst and mA, see 3.3.2.- tD, see 3.1.2.3.- (D/H)st = isotope ratio of internal standard (TMU) indicated on the bottle supplied by the Community Bureau of References.

The use of peak heights instead of peak area, which is less precise, supposes that peak width at half height is identical and is a reasonable approximation if applicable (Figure 2b).5.2. Water

When the isotope ratio of water is determined by NMR from the water-TMU mixture, the following relationship is used:with- m2st and m2E, see 3.4.2.- (D/H)st = isotope ratio of the internal standard (TMU) indicated on the bottle supplied by the Community Bureau of References.

5.3. For each of the isotope parameters, calculate the average of 10 determinations and the confidence interval.

Optional software (e.g. SNIF-NMR) suitable for the spectrometer computer enables such calculations to be carried out on-line.

Note:If, after standardization of the spectrometer, there is a systematic difference between the average values obtained for the characteristic isotopes of the standard alcohols (4.2.2) and the values indicated by the Community Bureau of References, to within the standard deviation, the following corrections may be applied to obtain the true value for any sample X.

The interpolation will be made by taking the values for the standard sample which straddle that of the sample X.

Let (D/H)iXmeas be the measured value and (D/H)iXcorr be the corrected value. This will give:(D/H)iXcorr = (D/H)iBst + a [ (D/H)iXmeas (D/H)iBmeas ]where

Example:Standard samples supplied and standardized by the Community Bureau of References:(D/H)IVst = 102,0 ppm (D/H)IBst = 91,95 ppmStandard samples measured by the laboratory:(D/H)IVmeas = 102,8 ppm (D/H)IBmeas = 93,0 ppmSuspect uncorrected sample: (D/H)IXmeas = 100,2 ppma = 1,0255 and (D/H)IXcorr. = 99,3 ppm are calculated.6. INTERPRETATION OF RESULTS

Compare the value RX obtained for the R ratio of the suspect sample with the ratios obtained for the control wines. If RX differs by more than two standard deviations from the average R T value obtained for the control wine, adulteration may be assumed.6.1. Addition of beet sugar, cane sugar or maize glucose

6.1.1. Wines

RX higher than RT: beet sugar is assumed to have been added.

RX less than RT: cane sugar or maize sugar is assumed to have been added.Note that (D/H)XI and (D/H)WQX are increased.Consider (D/H)XI:- Beet sugar is assumed to have been added: (D/H)XI of the suspect sample is lower than (D/H)IT, the average value obtained from the control samples, by more than one standard deviation- Cane sugar or maize sugar is assumed to have been added:

(D/H)IX is greater than (D/H)IT by more than one standard deviation- Calculation of enrichment E expressed in % vol of ethanol:- Addition of beet sugar:where(D/H)IB = isotope ratio for the location I of the beet alcohol;(D/H)IB = 92,5 (8) tV = alcoholic strength of the analysed wine (X).- Addition of cane sugar or maize sugar:

where(D/H)IC = isotope ratio for the location I of the cane sugar or maize sugar;(D/H)IC = 110,5 (9) tV = alcoholic strength of the analysed wine (X)6.1.2.Musts, concentrated musts and rectified concentrated masts

The values of the isotopic parameters for the alcohol extracted as described in 3.1 from the fermented product obtained (3.2) from must, concentrated must and rectified concentrated must are examined according to the instructions given in 6 under 'Interpretation of results' (6.1.1) and compared with the alcohol extracted from the fermentation product of musts.

The enrichment, E % vol, expresses the volume of alcohol added to the fermented product. Knowing the dilution that may have been carried out prior to fermentation (concentrated musts and rectified concentrated musts), assuming that 16,83 g of sugar yield 1 % vol of alcohol, calculate the amount of sugar (mass) added per litre of must, concentrated must or rectified concentrated must.6.2. Addition of a mixture of beet sugar and cane sugar or maize glucose

The isotope ratios (D/H)I and R are changed less than when only one type of sugar is added.

(D/H)II is higher, as is (D/H)WQ.These additions may be confirmed by determining the13C/12C ratio of the ethanol by mass spectrometry; in that case the ratio is higher.

9. ASH CONTENT 1. DEFINITION

The ash content is defined to be all those products remaining after igniting the residue left from evaporation of the wine. The ignition is carried out in such a way that all the cations (excluding the ammonium cation) are converted into carbonates or other anhydrous inorganic salts.2. PRINCIPLE OF THE METHOD

The wine extract is ignited at a temperature between 500 and 550 °C until complete combustion (oxidation) of organic material has been achieved.3. APPARATUS

3.1. boiling water-bath;

3.2. balance sensitive to 0,1 mg;

3.3. hot-plate or infra-red evaporator;

3.4. temperature-controlled electric muffle furnace;

3.5. desiccator;

3.6. flat-bottomed platinum dish 70 mm in diameter and 25 mm in height.4. PROCEDURE

Pipette 20 ml of wine into the previously tared platinum dish (original weight Po g). Evaporate on the boiling water-bath, and heat the residue on the hot-plate at 200 °C or under the infra-red evaporator until carbonization begins. When no more fumes are produced, place the dish in the electric muffle furnace maintained at 525 ± 25 °C. After 15 minutes of carbonization, remove the dish from the furnace, add 5 ml of distilled water, evaporate on the water-bath or under the infra-red evaporator, and again heat the residue to 525 °C for 10 minutes.

If combustion (oxidation) of the carbonized particles is not complete, repeat the operations of washing the carbonized particles, evaporation of water and ignition.

For wines with a high sugar content, it is advantageous to add a few drops of pure vegetable oil to the extract before the first ashing to prevent excessive foaming.

After cooling in the desiccator, the dish is weighed (P1 g).

The weight of the ash in the sample (20 ml) is then P = (P1- Po) g.5. EXPRESSION OF RESULTS

5.1. Method of calculation

The weight P of the ash in grams per litre will be given to two decimal places by the expression: P= 50p

10. ALKALINITY OF THE ASH 1. DEFINITION

The alkalinity of the ash is defined as the sum of cations, other than the ammonium ion, combined with the organic acids in the wine.2. PRINCIPLE OF THE METHOD

The ash is dissolved in a known (excess) amount of a hot standardized acid solution; the excess is determined by titration using methyl orange as an indicator.3. REAGENTS AND APPARATUS

3.1. 0,05 M sulphuric acid solution (H2SO4);

3.2. 0,1 M sodium hydroxide solution (NaOH);

3.3. methyl orange, 0,1 % solution in distilled water;

3.4. boiling water-bath.4. PROCEDURE

Add 10 ml of the 0,05 M sulphuric acid solution (3.1) to the ash from 20 ml of wine contained in the platinum dish. Place the dish on the boiling water-bath for about 15 minutes, breaking up and agitating the residue with a glass rod to speed up the dissolution. Add two drops of methyl orange solution and titrate the excess sulphuric acid against 0,1 M sodium hydroxide (3.2) until the colour of the indicator changes to yellow.5. EXPRESSION OF RESULTS

Method of calculation

The alkalinity of the ash, expressed in milliequivalents per litre to one decimal place, is given by

A = 5 (10 n)where n ml is the volume of 0,1 M sodium hydroxide used.

11. CHLORIDES 1. PRINCIPLE

Chlorides are determined directly in the wine by potentiometry using an Ag/AgCl electrode.2. APPARATUS

2.1. pH/mV meter graduated at intervals of at least 2 mV.

2.2. Magnetic stirrer.

2.3. Ag/AcCl electrode with a saturated solution of nitrate potassium as electrolyte.

2.4. Microburette graduated in 1/100 ml.

2.5. Chronometer.3. REAGENTS

3.1. Standard chloride solution: 2,1027 g of potassium chloride, KCl (max. 0,005 % Br), dried before use by leaving in a desiccator for several days, are diluted in distilled water and made up to one litre. 1 ml of this solution contains 1 mg Cl .

3.2. Silver nitrate titrating solution: 4,7912 g of analytical grade silver nitrate, AgNO3 are diluted in a 10 % (v/v) alcohol solution and made up to one litre. 1 ml of this solution corresponds to 1 mg Cl .

3.3. Nitric acid, of at least 65 % purity (r20 = 1,40 g/ml).4. PROCEDURE

4.1. 5,0 ml of standard chloride solution are measured into a 150 ml cylindrical vessel placed on a magnetic stirrer, diluted with distilled water to approximately 100 ml and acidified with 1,0 ml of nitric acid (at least 65 %). After immersing the electrode, titrate by adding the silver nitrate titrating solution with the microburette, with moderate stirring. Begin by adding 1,00 ml for the first 4 ml and read the corresponding millivolt values. Add the next 2 ml in fractions of 0,20 ml. Finally, continue the addition in fractions of 1 ml until a total of 10 ml has been added. After each addition, wait for approximately 30 seconds before reading the corresponding millivolts. Transfer the values thus obtained onto graph paper against the corresponding millilitres of titrating solution and determine the potential of the equivalence point on the basis of the singular point on the curve obtained.

4.2. 5 ml of the standard chloride solution are measured into a 150 ml cylindrical vessel with 95 ml of distilled water and 1 ml of nitric acid (at least 65 %). Immerse the electrode and titre, whilst stirring, until the potential of the equivalence point is obtained. This determination is repeated until a good degree of agreement in the results is obtained. This check must be carried out before each series of measurements of chlorides in the samples.

4.3. 50 ml of wine for analysis are measured into a 150 ml cylindrical vessel. Add 50 ml of distilled water and 1 ml of nitric acid (at least 65 %) and titrate using the procedure describedin 4.2.5. EXPRESSION OF RESULTS

5.1. Calculations

If n represents the number of millilitres of silver nitrate titrating solution, the chloride content in the tested liquid is:

20 × n expressed as milligrams of Cl per litre,0,5633 × n expressed as milliequivalents per litre,32,9 ×n expressed as milligrams of sodium chloride per litre.5.2. Repeatability (r):

r = 1,2 mg Cl per litrer = 0,03 meg per litrer = 2,0 mg NaCl per litre5.3. Reproducibility (R):

R = 4,1 mg Cl per litreR = 0,12 meg per litreR = 6,8 mg NaCl per litre6. Note: For very precise determination.

Refer to the complete titration curve obtained during determination of the test liquid with the silver nitrate solution.

(a) Measure 50 ml of the wine to be analysed into a 150 ml cylindrical vessel. Add 50 ml of distilled water and 1 ml of nitric acid (at least 65 %). Titrate using the silver nitrate solution, adding 0,5 ml at a time and recording the corresponding potential in millivolts. Derive from this first titration the approximate volume of silver nitrate solution required.

(b) Recommence determination in the same conditions. Begin by adding 0,5 ml of titrating solution at a time until the volume added is 1,5 to 2 ml less that the volume determined in (a). Hereafter add 0,2 ml at a time. Continue to add the solution beyond the approximately located equivalence point in a symmetrical manner, i.e. by adding 0,2 ml and then 0,5 ml at a time.

The end point of the measurement and the exact volume of silver nitrate consumed are obtained:- either by drawing the curve and determining the equivalence point,- or by the following calculation:Where:V = volume of titrating solution at equivalence point;V2 = volume of titrating solution before the largest potential change;D Vi = constant volume of the increments of titrating solution, i.e. 0,2 ml;DD El = second difference in potential before the largest potential change;DD E2 = second difference in potential after the largest potential change.

Example:

Volume of AgNO3 titrating solutionE potential in mVDifference D ESecond difference DD E 0204 4 0,2208 0 4 0,4212 2 6 0,6218 0 6 0,8224 0 6 1,0230 2 8 1,2238 4 12 1,4250 10 22 1,6272 22 44 1,8316 10 34 2,0350 8 26 2,2376 6 20 2,4396

In this example, the end point of the titration is between 1,6 and 1,8 ml: the largest potential change (D E = 44 mV) occurs in this interval. The volume of silver nitrate titrating solution consumed to measure the chlorides in the test sample is:

12. SULPHATES 1. PRINCIPLE

1.1. Reference method

Precipitation of barium sulphate and weighing. The barium phosphate precipitated in the same conditions is eliminated by washing the precipitate in hydrochloric acid.

In the case of musts or wine rich in sulphur dioxide, prior de-sulphiting by boiling in an airtight vessel is recommended.1.2. Quick test method

Wines are classified into several categories using the so-called limits method, based on the precipitation of barium sulphate using a barium ion titrant.2. REFERENCE METHOD

2.1. Reagents

2.1.1. 2 M solution of hydrochloric acid.

2.1.2. Barium chloride solution of 200 g/l of BaCl2· 2H2O.2.2. Procedure

2.2.1. General procedure:

Measure 40 ml of the analysis sample into a 50 ml centrifuge tube; add 2 ml of 2 M hydrochloric acid and 2 ml of barium chloride solution at 200 g/l. Stir with a glass stirrer; rinse the stirrer with a little distilled water and leave to stand for five minutes. Centrifuge for five minutes, then carefully decant the supernatant liquid.

Next wash the barium sulphate precipitate as follows: add 10 ml of 2 M hydrochloric acid, place the precipitate in suspension and centrifuge for five minutes, then carefully decant the supernatant liquid. Repeat the washing procedure twice in the same conditions using 15 ml distilled water each time.

Quantitatively transfer the precipitate, by rinsing with distilled water, into a tared platinum capsule and place over a water bath at 100 °C until fully evaporated. The dried precipitate is calcined several times briefly over a flame until a white residue is obtained. Leave to cool in a desiccator and weigh.

Let m= the mass in milligrams of barium sulphate obtained.2.2.2. Special procedure: sulphited must and wine with a high sulphur dioxide content.

Beforehand, eliminate the sulphur dioxide.

Measure 25 ml of water and 1 ml of pure hydrochloric acid (r 20 = 1,15 to 1,18 g/ml) into a 500 ml conical flask equipped with a dropping funnel and an outlet tube. Boil the solution to remove the air and introduce 100 ml of wine through the dropping funnel. Continue boiling until the volume of liquid in the flask has been reduced to about 75 ml and quantitatively transfer it, after cooling, to a 100 ml volumetric flask. Make up to mark with water. Determine the sulphates in a 40 ml sample as indicated in 2.2.1.

2.3. Expression of results

2.3.1. Calculations:

The sulphate content, expressed in milligrams per litre of potassium sulphate, K2SO4 is:18,67 × m

The sulphate content in musts or wine is expressed in milligrams per litre of potassium sulphate, with no decimal point.2.3.2. Repeatability

up to 1 000 mg/l: r = 27 mg/labout 1 500 mg/l: r = 41 mg/l2.3.3. Reproducibility

up to 1 000 mg/l: R = 51 mg/labout 1 500 mg/l: R = 81 mg/l3. QUICK TEST METHOD

3.1. Reagents

3.1.1. Barium chloride titrating solution

2,804 g of barium chloride, BaCl2·2H2O and 10 ml of hydrochloric acid (r20 = 1,15 to 1,18 g/ml) are dissolved in enough water to obtain one litre of solution, 1 ml of this solution precipitates a quantity of sulphate ions equivalent to 2 mg of potassium sulphate.

3.1.2. Sulphuric acid (r20 = 1,84 g/ml), 1/10 (m/v) solution.3.2. Procedure

Measure 10 ml of musts or wine into three test tubes; add to No 1: 3,5 ml, to No 2: 5 ml and to No 3: 10 ml of the barium chloride solution. Shake and bring to the boil; leave to stand for one to two hours. The liquid in each tube is decanted, filtered and divided into two parts. To one part, add several drops of the dilute sulphuric acid solution, to the other some drops of the barium chloride solution. Examine each tube and note whether the liquid is clear or cloudy. Interpretation of these observations is given in the table below.

WineBarium chlorideFiltered wine +diluted sulphuric acidbarium chloride solutionFirst test(ml)(ml)cloudyclear103,5(less than 0,7 g K2SO4/l)

clear cloudy

(more than 0,7 g K2SO4/l)Second test105cloudyclear(less than 1 g K2SO4/l)

clear cloudy

(more than 1 g K2SO4/l)Third test1010cloudyclear(less than 2 g K2SO4/l)

clear cloudy

(more than 2 g K2SO4/l)

13. TOTAL ACIDITY 1. DEFINITION

The total acidity of the wine is the sum of its titratable acidities when it is titrated to pH 7 against a standard alkaline solution.

Carbon dioxide is not included in the total acidity.2. PRINCIPLE OF THE METHOD

Potentiometric titration or titration with bromothymol blue as an indicator and comparison with an end-point colour standard.3. REAGENTS

3.1. Buffer solution pH 7,0:

- monopotassium phosphate, (KH2PO4) 107,3 g- 1 M sodium hydroxide (NaOH) solution 500 ml- water to1 000 ml

Alternatively, ready made buffer solutions are available commercially.

3.2. 0,1 M sodium hydroxide (NaOH) solution.

3.3. 4 g/l bromothymol blue indicator solution:

- bromothymol blue (C27H28Br2O5S) 4 g- neutral ethanol, 96 % vol 200 mlDissolve and add:- water free of CO2 200 ml- 1 M sodium hydroxide solution sufficient to produce blue-green colour (pH 7) 7,5 ml approximately- water to1 000 ml4. APPARATUS

4.1. Water vacuum pump.

4.2. 500 ml vacuum flask.

4.3. Potentiometer with scale graduated in pH values, and electrodes. The glass electrode must be kept in distilled water. The calomel/saturated potassium chloride electrode must be kept in a saturated potassium chloride solution. A combined electrode is most frequently used: it should be kept in distilled water.

4.4. Measuring cylinders 50 ml (wine), 100 ml (rectified concentrated must).5. PROCEDURE

5.1. Preparation of sample:

5.1.1. Wines

Elimination of carbon dioxide. Place about 50 ml of wine in a vacuum flask; apply vacuum to the flask with the water pump for one to two minutes, whilst shaking continuously.

5.1.2. Rectified concentrated musts

Introduce 200 g of accurately weighed rectified concentrated must. Make up to the mark with 500 ml water. Homogenize.5.2. Potentiometric titration

5.2.1. Calibration of pH meter

The pH meter is now calibrated for use at 20 °C, according to the manufacturer's instructions, with the pH 7,00 buffer solution at 20 °C.5.2.2. Method of measurement

Into a measuring cylinder (4.4), introduce a volume of the sample, prepared as described in 5.1, equal to 10 ml in the case of wine and 50 ml in the case of rectified concentrated must. Add about 10 ml of distilled water and then add the 0,1 M sodium hydroxide solution (3.2) from the burette until the pH is equal to 7 at 20 °C. The sodium hydroxide must be added slowly and the solution stirred continuously. Let n ml be the volume of 0,1 M NaOH added.5.3. Titration with indicator (bromothymol blue)

5.3.1. Preliminary test: end-point colour determination.

Into a measuring cylinder (4.4) place 25 ml of boiled distilled water, 1 ml of bromothymol blue solution (3.3) and a volume prepared as in (5.1) equal to 10 ml in the case of wine and 50 ml in the case of rectified concentrated must. Add the 0,1 M sodium hydroxide solution (3.2) until the colour changes to blue-green. Then add 5 ml of the pH 7 buffer solution (3.7).5.3.2. Measurement

Into a measuring cylinder (4.4) place 30 ml of boiled distilled water, 1 ml of bromothymol blue solution (3.3) and a volume of the sample, prepared as described in 5.1, equal to 10 ml in the case of wine and 50 ml in the case of rectified concentrated must. Add 0,1 M sodium hydroxide solution (3.2) until the same colour is obtained as in the preliminary test above (5.3.1). Let n ml be the volume of 0,1 M sodium hydroxide added.6. EXPRESSION OF RESULTS

6.1. Method of calculation

6.1.1. Wines

The total acidity expressed in milliequivalents per litre is given by:

A = 10n.It is recorded to one decimal placeThe total acidity expressed in grams of tartaric acid per litre is given by:

A2 = 0,075A

It is recorded to one decimal place.6.1.2. Rectified concentrated musts

- The total acidity expressed in milliequivalents per kilogram of rectified concentrated must is given by a = 5n.

-The total acidity expressed in milliequivalents per kilogram of total sugars is given byP = % concentration (m/m) of total sugars.

It is recorded to one decimal place.6.2. Repeatability (r) for titration with the indicator:

r = 0,9 meq/litrer = 0,07 g tartaric acid/litrefor white, rosé and red wines.6.3. Reproducibility (R) for titration with the indicator (5.3):

For white and rosé wines:R = 3,6 meq/litreR = 0,3 g tartaric acid/litre

For red wines:R = 5,1 meq/litreR = 0,4 g tartaric acid/litre

14. VOLATILE ACIDITY 1. DEFINITION

The volatile acidity is formed from the acids of the acetic series present in wine in the free state and combined as a salt.2. PRINCIPLE OF THE METHOD

Titration of the volatile acids separated from the wine by steam distillation and titration of the distillate.

Carbon dioxide is first removed from the wine.

The acidity of free and combined sulphur dioxide distilled under these conditions should be deducted from the acidity of the distillate.

The acidity of any sorbic acid which may have been added to the wine must also be deducted.

Note: Part of the salicylic acid used in some countries to stabilize the wines before analysis is present in the distillate. This must be determined and deducted from the acidity. The method of determination is given in section 7 of this chapter.3. REAGENTS

3.1. Crystalline tartaric acid (C4H6O6).

3.2. 0,1 M sodium hydroxide solution (NaOH).

3.3. 1 % phenolphthalein solution in 96 % vol neutral alcohol.

3.4. Hydrochloric acid (r20 = 1,18 to 1,19 g/ml) diluted 1/4 (v/v).

3.5. 0,005 M iodine (I2) solution.

3.6. Crystalline potassium iodide (KI).

3.7. 5 g/l starch solution.Mix 5 g of starch with about 500 ml of water. Bring to the boil, stirring continuously and boil for 10 minutes. Add 200 g sodium chloride. When cool, make up to one litre.

3.8. Saturated solution of sodium borate (Na2B4O7 · 10H2O), i.e. about 55 g/l at 20 °C.4. APPARATUS

4.1. Steam distillation apparatus consisting of:1. a steam generator; the steam must be free of carbon dioxide;2. a flask with steam pipe;3. a distillation column;4. a condenser.This equipment must pass the following three tests:(a) Place 20 ml of boiled water in the flask. Collect 250 ml of the distillate and add to it 0,1 ml of 0,1 M sodium hydroxide solution (3.2) and two drops of the phenolphthalein solution (3.3). The pink colouration must be stable for at least 10 seconds (i.e. steam to be free of carbon dioxide).(b) Place 20 ml of a 0,1 M acetic acid solution in the flask. Collect 250 ml of the distillate. Titrate with the 0,1 M sodium hydroxide solution (3.2): the volume of this used must be at least 19,9 ml (i.e. at least 99,5 % of the acetic acid entrained with the steam).(c) Place 20 ml of 1 M lactic acid solution in the flask. Collect 250 ml of the distillate and titrate the acid with the 0,1 M sodium hydroxide solution (3.2).The volume of sodium hydroxide solution added must be less tha