The Dalles daily chronicle. [volume] (The Dalles, Or.) 1890-1948, 1892-06-18
WEATHER GABGERS.
THERMOMETERS ARE MANuFAc-
TuRso WITH GREAT CARE.

Though Quickly, Pass Through Nine
teen PairS of Hands-The Different
Processes of the Instruments.
either delicate scientific operation, or
one of the simplest tasks of the skilled
mechanic, according to the sort of ther-
mometer made With the extremely
sensitive and minutely accurate instru-
ments designed for scientific uses great
care is taken, and they are kept in stock
for months, sometimes for years, to be
compared and recompared with instru-
ments that are known to be trustworthy.
But so much time cannot be spent over
the comparatively cheap thermometers
IN common|use, and these are made rap
| idly, though always carefully. The
method of manufacture has been SO sys-
tematized within a few years that the
very cheapest thermometer should not
vary more than fraction of degree
from the correct point.
Whether the Thermometer is to be
charged with mercury or alcohol; wheth-
er it is to be mounted in frame of
wood, pressed tin or brass, the process
is substantially the same. Mercury is
generally used for scientific instruments,
but most makers prefer alcohol, because
it is cheaper. The alcohol is colored
red with an ariline dye which does not
fade.
The thermometer maker buys his glass
tubes in long strips from the glass fac
tories. The glassblower on the premises
cuts these tubes to the proper lengths,
and with his gas jet and blowpipe makes
the bulb ON the lower end. The bulbs
are then filed with colored alcohol, and
the tubes stand for twenty-four hours.
On the following day another work
man holds each bulb in turn over a gas
jet until the colored fluid, by its expan-
sion, entirely fills the tube 1t then
goes back into the hands of the glass
blower. He closes the upper end and
turns the tip backward to make the lit
tie glass hook which will help keep the
tube in place in the frame.
MARKING THE TUBES
The tubes now rest until some hun
dreds of them, perhaps thousands, are
ready Then the process of gauging be-
gins. There are no marks whatever ON
the tube, and the first guide mark to be
made is the freezing point, 32 degS.
Fahrenheit. This is found by plunging
tho bulbs into melting snow. No other
thermometer is needed for guide, for
melting snow gives invariably the exact
freezing point. This is an unfailing test
for any thermometer whose accuracy
may be suspected. But melting snow is
not always to be had, and little ma
chine resembling a sausage grinder is
brought into use. This machine shaves
block Of ice into particles, which an
swer the purpose as well as snow.
When the bulbs have been long enough
in the melting snow a workman takes
them one by one from their icy bath
seizing each SO that his thumb nail
marks the exact spot to which tho fluid
has fallen. Here he makes scarcely
percepHble mark upon the glass with a
ine file, and goes on to the next
Tho tubes, with the freezing point
marked on each, now go into the hands
of another workman, who plunges them,
bulb down, into vessel filled with
water kept constantly at Ga degs.
A standard thermometer attached to
the inside of this vessel shows that the
temperature of the water is correct.
Another tiny file scratch is put at "a.
Then third workman plunges the
bulbs into another vessel of water kept
constantly at 96. This is marked
Like the others, and the tube is now sup
piled with these guide marks, each
30 degs. from the next. A small tab
is then attached to each tube, ON which
its number is written-for, owing to un-
avoidable variations in the bore of the
tube, each one varies slightly from the
others.
MARKING THE CASES
With its individuality thus established
the tube goes into the hands of mark
er, who fits its bulb and, hook into the
frame it is to occupy, and makes slight
scratches on the frame corresponding to
the 81, "a and 96 degree marks on the
tube. The frame has a number CORE
spending with the number of the tube,
and the tube is laid away In rack amid
thousands.
The frame, whether it be wood, tin
or brass, goes to the gauging room,
where it Is laid upon n steeply sloping
table, exactly in the position marked for
a thermometer of that size. The 80, 81
and 96 degree marks must correspond
with the marks upon the table. II they
do not, the error in marking Is detected
and the frame is sent back for correc-
ton. s
a A long, straight bar of wood or metal
extends diagonally across the table from
the lower right hand corner to the upper
left hand corner. On the right this
works upon pivot and on the left it
rests in q ratchet, which let's it ascend
or descend only one notch at ., time.
That notch marks the exact distance of
2degs. With the three scratches already
made for basis the marker could hardly
make mistake in the degrees If he tried.
The marks made upon the frame or
case are all made by hand with geo.