Document: NUREG-0800
Document ID: 66e7af3b-21c0-444f-8737-0830bb176eba
Document Type: srp
Title: PROCESS AND EFFLUENT RADIOLOGICAL MONITORING INSTRUMENTATION
Source: NUREG-0800
Source URL: https://www.nrc.gov/docs/ML1502/ML15029A182.pdf
Revision Date: 2023-06
Chapter: 11
Section ID: 11.5
CFR Part: 
CFR Title: 

Content:
lification of radiation detectors in supporting the functions of radiation monitoring systems. The applicant described the functional interdependence and logic in alarming and terminating or diverting process or effluent streams in complying with doses for 11.5-42 Revision 6 – January 2016 members of the public and effluent concentration limits under 10 CFR Part 20, and avoid exceeding the design objectives of Appendix I to 10 CFR Part 50. The staff concludes that the information describing the design and operational features of the PERMISS complies with NRC regulations, consistent with NRC and industry guidance on radiation monitoring and sampling, and the staff confirms that the acceptance criteria of this section have been met. Supporting information on the staff’s evaluation of the applicant’s information is presented in SER Chapter 7. 4. Under the requirements of 10 CFR 50.34(f)(2) on TMI-related requirements , the applicant has included provisions for the sampling and monitoring of process and effluent streams and conducting analysis of samples, including the proposed analytical programs, during postulated accidents in accordance with the requirements of 10 CFR 50.34(f)(2)(viii), 10 CFR 50.34(f)(2)(xiv)(E), 10 CFR 50.34(f)(2)(xvii), 10 CFR 50.34(f)(2)(xxvi), and 10 CFR 50.34(f)(2)(xxvii) using the guidance of RG 1.97 and Appendix 11.5-A to SRP Section 11.5. This conclusion is supported by the reviews and evaluations conducted using the guidance in SRP Section 13.3 and SRP Chapter 7. The applicant has identified the specific revision of RG 1.97 applicable to the application, as described in SRP Chapter 7 and BTP 7-10. Supporting information on the staff’s evaluation and findings of the applicant’s information is presented in SER Chapter 7 and SER Sections 12.3-12.4, 12.5, and 13.3 and summarized here. 5. The review evaluated P&IDs, process flow diagrams, and descriptions of system proposed sampling points for the liquid, gaseous, and solid waste