Document: NUREG-0800
Document ID: 66e7af3b-21c0-444f-8737-0830bb176eba
Document Type: srp
Title: PROCESS AND EFFLUENT RADIOLOGICAL MONITORING INSTRUMENTATION
Source: NUREG-0800
Source URL: https://www.nrc.gov/docs/ML1502/ML15029A182.pdf
Revision Date: 2023-06
Chapter: 11
Section ID: 11.5
CFR Part: 
CFR Title: 

Content:
ptance Criteria,” SECY-05-0197. ADAMS Accession No. ML052770257. 11.5-55 Revision 6 – January 2016 Table 1: Provisions for Monitoring and Sampling Gaseous Process and Waste Streams (*) No. Process Systems(**) In Process Conti In Effluent In Process Grabk In Effluent Grabk Contl ACFj Conti 1. Waste Offgas Holdup Systema NG NG (NG) (NG,H3) (1) 2 Condenser Evacuation Systemb NG (NG) (NG) I (NG,H3) (I) 3. Building Vent and Stack Release Point and Systemsc − − NG H3 (I) 4. Containment Purge Systems and Isolation Condenser Exhaust Systemsd NG NGm (NG) I (NG,I,H3) (I) 5. Auxiliary or Service Building System _ _ (NG) I (NG,H3) (I) 6. Fuel Storage Area Ventilation Systeme (NG) NGm (NG) I (NG,H3) (I) 7. Radwaste Area Vent Systems _ _ (NG) I (NG,H3) (I) 8. Turbine Gland Seal Condenser Vent System _ _ (NG) I (NG,H3) (I) 9. Mechanical Vacuum Pump Exhaust(Hogging System) _ _ (NG) I (NG,H3) (I) 10. Evaporator Vent Systems _ _ (NG) I (NG,H3) (I) 11. Vents from Liquid Radwaste Tanks and Components _ _ (NG) (I) (NG,H3) (I) 12. Steam Generator Systems and Generator _ _ (NG) I (NG,H3) (I) Blowdown and Degasifier Systems Treatment Systems 13. Turbine Building Vent Systems _ _ (NG) I (NG,H3) (I) 14. Pressurizer and Boron Recovery Vent Systems _ _ (NG) I (NG,H3) (I) 15. Ventilation from Waste Compactors, Shredders, Etc.(as Permanently Installed or Mobile Systems) R – I I I (I) 16. Exhausts from compressed air and instrument air systems with potential for radioactive contamination R – I I I (I) (*) For the key to legend, see the notes on the page following Table 2. (**) The listing is not considered to be all inclusive. Applicant should identify process and effluent systems that are unique to the specific type of reactor technology described in the application. 11.5-56 Revision 6 – January 2016 Table 2: Provisions for Monitoring and Sampling Liquid Process and Waste Streams (*) Monitor Provisions Sample Provisions No. Process Systems(**) In Process In Effluent In Process In