Document: NUREG-0800
Document ID: e459fbaa-1cf6-4054-a83a-500a6895feaf
Document Type: srp
Title: PROCESS AND EFFLUENT RADIOLOGICAL MONITORING INSTRUMENTATION AND
Source: NUREG-0800
Source URL: https://www.nrc.gov/docs/ML0523/ML052350110.pdf
Revision Date: 2023-06
Chapter: 11
Section ID: 11.5
CFR Part: 
CFR Title: 

Content:
ncluded in the design for replac- ing or decontaminating detectors without opening the-boundary of the process system or without losing the capability to isolate the system or divert the effluent to a standby treatment system (as appropriate). i. ETSB will review special features, applicable topical reports, and data referenced in the SAR on a case-by-case basis. 2. ETSB reviews the technical specifications proposed by the applicant for process and effluent radiological monitoring and sampling at the OL stage. The reviewer determines that the content and intent of the technical specifications prepared by the applicant are in agreement with the requirements developed as a result of the staff's review. The review will include the evaluation or development of appropriate limiting con- ditions for operation and their bases consistent with the plant design. IV. EVALUATION FINDINGS ETSB verifies that sufficient information has been provided and that the review supports conclusions of the following type, to be included in the staff's safety evaluation report: The staff concludes that the process and effluent radiological monitor- ing instrumentation and sampling systems are acceptable and meet the relevant requirements of 10 CFR Part 20, §20.106, and General Design Criteria 60, 63, and 64. This conclusion is based on the following: 11.5-6 Rev. 3 - July 1981 The process and effluent radiological monitoring and sampling systems include the instrumentation for monitoring and sampling radioactivity, contaminated liquid, gaseous, and solid waste process and effluent streams. Our review included the provisions proposed to sample and monitor all station effluents in accordance with General Design Criterion 64, the provisions proposed to provide automatic termination of effluent releases and assure control over discharges in accordance with General Design Criterion 60, the provisions proposed for sampling and monitoring plant waste process streams for process control in