DEPARTMENT  UF  THE  AIR  FORCE 
Headquarters  Air  Force  Technical  Applications  Center 
Patrick  AFB  FL,  32925-6001 
CENTER  REGULATION  55-2 
Volume  VI 
1  April  1988 
Operations 
SPECIFIC  STATION  REQUIREMENTS  FUR  DETACHMENT  415 
This  regulation  establishes  the  proceaures  for  station  unique  operations  and  analysis. 
Distribution  limited  to  UoU  and  DoO  contractors  only;  to  protect  information  and  technical  data 
which  advance  the  state-of-the-art  or  describe  new  technology  in  an  area  of  significant  or 
potentially  significant  Military  application,  1  April  1988.     Other   requests  shall  be  referred 
to  HQ/UOSB.  f 
1.  Station  Designator.  The  station  designator  for  Detachment  415  is  CMCM.  Use  CHM  CHCH  for  the 
three  element  entry  and  station  designator  on  data  messages.  Mark  CEN  Form  10s,  using  the 
appropriate  color,  with  the  first  two  letters  of  the  station  designator. 
2.  Special  Data  Reports.  Submit  special  data  reports  in  accordance  with  Volume  I. 
3.  Timing  Standard.  Navy  Navigation  Satellite  System  time. 
4.  Routine  Calibrations.  Perform  SPS  and  LPS  calibrations  sequentially  using  the  Central 
Terminal,  commencing  immediately  after  03002.  Use  an   amplitude  factor  4  (lOOmu)  for  the  SPS  anc 
an  amplitude  factor  2  UOu)  for  the  LPS. 
5.  EDIT  tape  registration  numbers  are   5600  through  5699. 
6.  Training  Outage.     Outage   authorized  in  CENR  55-2,  Vol  I  is  granted  for  Wednesday  of  each 
week  from  05002  through  09002.  If  mission  requirements  preclude  using  Wednesday,  Thursday  is 
your  alternate  day. 
7.  Routine  Data  Reporting.  Reference  Volume  I.  Report  data  twice  daily  using  reporting 
periods  of  0000-11592  and  1200-23592. 
8.  SPS  Uevelocoroer  Presentations: 
a.  Primary  Uevelocoroer: 
'RACE 
:  DATA 
MAG 
ASN  CHAN 
DISP  ID 
SCALE 
DEV  SENS  V 
S22BP36013 
20Q0K 
SPDS01 
SPL360 
1.0 
0.195 
S22BP06013 
2000K 
SPDS02 
SPL060 
1.0 
0.195 
S22BP12013 
2000K 
SPUS03 
SPL120 
1.0 
0.195 
S22BP18013 
2000K 
SPDS04 
SPL180 
1.0 
0.195 
S22BP24013 
2000K 
SPDS05 
SPL240 
1.0 
0.195 
S22BP30013 
2000K 
SPDS06 
SPL300 
1.0 
0.195 
S22BP00099 
2000K 
SPDS07# 
SP2000 
1.0 
0.195 
S22BP33713 
20U0K 
SPDS08# 
SPL337 
1.0 
0.195 
S22BP31715 
20U0K 
SPDS09# 
SPP317 
1.0 
0.195 
S22I76H 
250K 
SPDS14# 
SPRU22 
1.0 
0.390 
SN2I76H 
250K 
SPDS15# 
SPRW23 
1.0 
0.390 
SE2I76H 
250K 
SPUS16# 
SPRW24 
1.0 
0.390 
S22I76M 
100K 
SPDS14# 
SPRW22 
1.0 
0.976 
S22I76L 
10K 
SPDS14# 
SPRW22 
1.0* 
0.976 
#  Display  recorded  on  another  aevelocoroer  channel  at  equal  or  different  gain. 
*  Change  display  scale  factor  to  10  for  develocorder  sensitivity  checks. 
b.  Secondary  Develocorder: 
TRACE 
DATA 
MAG 
ASN  CHAN 
DISP  ID 
SCALE 
UEV SENS  V 
* 
S22I16 
bUOK* 
SPDS10 
SPRWlb 
1.0 
0.781 
S22I13 
500K* 
SPUS11 
SPRW13 
1.0 
0.781 
S22I10 
50UK* 
SPUS12 
SPRW10 
1.0 
0.781 
S22I11 
500K* 
SPDS13 
SPRW11 
1.0 
0.781 
S22BP00099 
2U00K# 
SPDS07 
SP2U00 
1.0 
0.195 
S22BP337 
'13 
2000K# 
SPDS08 
SPL337 
1.0 
0.195 
Supersedes  CtNR  bb-2,  Vol  VI,  2  August  1986  (See  signature  page  for  summary  of  changes.) 
No.  of  Printed  Pages;   7 
OPR:  UUSB  (TSgt  C.W.  Stephens) 
Approved  by:  Col  T.H.  Niouette 
Editor  SSgt  D.  M.  Pless 
Distribution:  X
CENR 
56-2, 
,  Vol  VI 
RACE 
DATA 
MAG 
ASN  CHAN 
DISP  ID 
SCALE 
DEV 
SENS  VOLTAGE 
SZ2BP31715 
2000Kf 
SP0S09 
SPP317 
1.0 
0.195 
SPAKE 
SZ2I76M 
100K# 
SPUS14 
SPRW22 
1.0 
0.976 
SN2I76M 
10GK# 
SP0S15 
SPRW23 
1.0 
0.976 
SE217&M 
100KI 
SPUSlb 
SPRW24 
1.0 
0.976 
SZ2I76L 
10K? 
SPDS14 
SPRW22 
l.U** 
0.976 
SN2I76L 
lOKf 
SPUS15 
SPKW23 
1.0** 
0.976 
SE2I76L 
10K# 
SPOSlb 
SPKW24 
1.0** 
0.976 
1  April  1988 
if 
* 
Display  recorded  on  another  develocorder  channel  at  equal  or  different  gain. 
Displays  may  be  changed  at  the  host  station  commander's  discretion.  Notify  HQ/D0S8  of  any 
cnanges. 
Cnange  Jisplay  scale  factor  to  10  for  develocorder  sensitivity  checks. 
9.  LPS  Develocorder  Presentation: 
TRACE 
DATA 
MAG 
ASN  CHAN 
DISP  ID 
SCALE 
DEV  SENS  VI 
LZ5BP3603.5 
100K 
LPDS01 
LPH36Z 
5* 
0.358 
LZ58P0903.5 
100K 
LPDS02 
LPH09Z 
5* 
0.368 
LZ5BP1803.5 
100K 
LPDS03 
LPH18Z 
5* 
0.358 
LZ5BP2703.5 
100K 
LPDS04 
LPH27Z 
5* 
U.358 
LZ5IA(? 
5K 
LPDS08 
LPSC11 
5* 
0.358 
LZ5IA0 
50K 
LPDS05 
LPSC11 
50* 
0.368 
LN5IA0 
50K 
LPDS06 
LPSC12 
SO* 
0.358 
LE5IA§ 
50K 
LPDS07 
LPSC13 
50* 
U.358 
*  Change  display  scale  factor  to  1.0  for  develocorder  sensitivity  checks. 
i?  Should  BB01/LPA  fail,  assign  another  site  that  is  within  operational  tolerances  using  the 
following  precedence:  LPF,  LPB,  LPE,  LPC,  LPD. 
NOTE:  LP  PGain  must  be  set  to  20.0. 
10.  Data  cross-reference  lists: 
CT 
DDS 
STPR 
STPR 
ISENSE 
DEV 
INST 
RTID 
CHANNEL 
GAIN 
CH  ID 
CGAIN 
Mu/CT 
ID 
U01 
SP01 
SOI 
SPRW01 
1.0 
0.16 
SZ2I01 
U02 
SP02 
S02 
SPRW02 
1.0 
0.16 
SZ2I02 
U03 
SP03 
S03 
SPRW03 
1.0 
0.16 
SZ2I03 
U04 
SP04 
S04 
SPRW04 
1.0 
0.16 
SZ2I04 
U05 
SP05 
S05 
SPRW05 
1.0 
0.16 
SZ2I05 
U06 
SP06 
S06 
SPRW06 
1.0 
0.16 
SZ2I06 
U07 
SP07 
S07 
SPRW07 
1.0 
0.16 
SZ2I07 
U08 
SP08 
SOB 
SPRW08 
1.0 
0.16 
SZ2I08 
U09 
SP09 
S09 
SPRH09 
1.0 
0.16 
SZ2I09 
U10 
SP10 
S10 
SPRW10 
1.0 
0.16 
SZ2I10 
Ull 
SP11 
Sll 
SPRW11 
1.0 
0.16 
SZ2I11 
U12 
SP12 
S12 
SPRW12 
1.0 
0.16 
SZ2I12 
U13 
SP13 
S13 
SPRW13 
1.0 
0.16 
SZ2I13 
U14 
SP14 
S14 
SPRW14 
1.0 
0.16 
SZ2I14 
U16 
SP15 
S15 
SPRW15 
1.0 
0.16 
SZ2I15 
U16 
SP16 
S16 
SPRW16 
1.0 
0.16 
SZ2I16 
U17 
SP17 
S17 
SPRW17 
1.0 
0.16 
SZ2I17 
U18 
SP18 
S18 
SPRW18 
1.0 
0.16 
SZ2I18 
KSZ 
BB01 
S19 
SPRW19 
1.0 
0.16 
SZ2I76H 
KSN 
BB01 
S20 
SPRW20 
1.0 
0.16 
SN2I76H 
KSE 
BB01 
S21 
SPRW21 
1.0 
0.16 
SE2I76H 
KSZ 
BB01 
S22 
SPRW22 
1.0 
0.64 
SZ2I76M 
KSN 
BBOi 
S23 
SPRW23 
1.0 
0.64 
SN2I76M 
KSE 
BB01 
S24 
SPRW24 
1.0 
0.64 
SE2I76M 
KSZ 
BBOI 
S25 
SPRW25 
1.0 
5.12 
SZ2I76L 
KSN 
BBOI 
S26 
SPRW26 
1.0 
5.12 
SN2I76L 
KSE 
BBOI 
S27 
12' 
SPRW27 
1.0 
5.12 
SE2I76L 
LPBZ 
LP01 
L01 
— 
LPSC21 
1.0 
0.167 
LZ5IB 
LPBN 
LP01 
L01 
— 
LPSC22 
1.0 
0.167 
LN5IB 
LPBE 
LP01 
L01 
— 
LPSC23 
1.0 
0.167 
LE5IB 
LPCZ 
LP02 
L02 
-- 
LPSC31 
1.0 
0.167 
LZ5IC 
LPCN 
LP02 
L02 
— 
LPSC32 
1.0 
0.167 
LN5IC 
LPCE 
LP02 
L02 
-- 
LPSC33 
1.0 
0.167 
LE5IC 
LPDZ 
LP03 
L03 
— 
LPSC41 
1.0 
0.167 
LZ5ID 
LPDN 
LPU3 
L03 
-- 
LPSC42 
1.0 
0.167 
LN5ID
55-2, Vol  VI 
1  Apri I  1988 
CT 
DOS 
STPR 
STPR 
1SENSE 
OEV 
INST 
KTIU 
CHANNEL 
GAIN 
CH  10 
CGAIN 
Mu/CT 
LPOE 
LP03 
L03 
— 
LPSC43 
1.0 
0.167 
LE51D 
LPEZ 
LP04 
L04 
— 
LPSC51 
1.0 
0.167 
L25IE 
LPEN 
LP04 
L04 
— 
LPSC52 
1.0 
0.167 
LN5IE 
LPEE 
LP04 
LU4 
— 
LPSC53 
1.0 
0.167 
LE5IE 
LPFZ 
LPU5 
LOS 
— 
LPSC61 
1.0 
0.167 
LZ5IF 
LPFN 
LPOb 
LOS 
-- 
LPSC62 
1.0 
0.167 
LN5IF 
LPFE 
LP05 
LU5 
— 
LPSC63 
1.0 
0.167 
LE5IF 
LPA2 
BBOl 
L06 
— 
LPSC11 
1.0 
0.167 
LZ5IA 
LPAN 
bBOl 
L06 
— 
LPSC12 
1.0 
0.167 
LN5IA 
LPAE 
BBOl 
LU6 
-- 
LPSC13 
1.0 
0.167 
LE5IA 
11.  Central  Terminal  Configuration  Parameters: 
a.  General  Site  Configuration  (Menu  Selection  3) 
Site  ID  Number 
Number of  9600  1 BPS  Lines 
Number of  4800  1 BPS  Lines 
Number of  analog  channel: 
Number of  544  Boards 
Number of  SPRTs 
Number of  LPRTs 
Number of  BBKTs 
RT-  Sf 
jecific  Configuration  (Mem i  Sele 
RT 
PORT 
C/V 
TIME 
RTID 
ADDR 
ADDR 
DELAY 
SLOT 
SPU1 
2*** 
V 
SP02 
2*** 
V 
SPU3 
j*** 
V 
SP04 
j^*** 
Y 
SPU5 
j*** 
V 
SPOb 
I**-* 
V 
SPU7 
]**• 
b 
V 
SPU8 
1  **Kr 
V 
SPuy 
|*** 
b 
V 
spiu 
^K** 
b 
Y 
b 
SP11 
l*** 
V 
SP12 
^k-trk 
V 
SP13 
■^■k-k-k 
V 
SP14 
^■kirk 
V 
SPlb 
±**k 
V 
SPlb 
^irtrk 
V 
SP17 
[*rk~k 
V 
SPlb 
j_*** 
V 
LPU1 
O-k-k-k 
V 
LP02 
•j-k-k* 
V 
LPUJ 
2*-** 
V 
LP04 
2**"* 
V 
LP05 
2*** 
b 
V 
BbOl 
j*** 
V 
***  =  Specific  RT  serial  number 
c.  Analog  Channel  Configuration  (Menu  Selection  5) 
Analog 
Channel 
b 
b 
RTID 
GAIN 
**  All  cnannels  are  site  selectable  ** 
a.  First  Message  to  TOS  Contents  (Menu  Selection  6); 
Number  of  SPRTs  in  First  Message  18 
Number  of  LPkTs  in  First  Message  5 
Number  of  BbRTs  in  First  Message    1
CENR  55-2,  Vol  VI 
1  April  1988 
e. 
12  bit  A/D/A  Channel  Win  Assignments  (Menu  Selection  7) 
CHANNEL 
KT1U   i bAlN 
SOI 
SP01 
S02 
SP02 
bU3 
SP03 
SU4 
SPU4 
5U5 
SP05 
SU6 
SPOb 
SU7 
SPU7 
SOB 
SPU8 
S09 
SP09 
S10 
SP1U 
Sll 
SP11 
S12 
SP12 
S13 
SP13 
S14 
SP14 
SIS 
SPlb 
Sib 
SPlb 
S17 
SP17 
S18 
SP18 
S19 
BBU1SZ 
S20 
bBOlSN 
S21 
BBOISE 
S22 
bbuisz 
S23 
bBOiSN 
S24 
bbOlSE 
S2b 
bbuisz 
S2b 
BBOISN 
S27 
BBOISE 
lb  Bit LPUAKTS  O iann< 
CHANNEL 
RTID 
L01 
LPOl 
L02 
LP02 
L03 
LP03 
L04 
LP04 
LU5 
LP05 
LOb 
BBU1 
(Menu  Selection  8) 
g.  Hardware  and  Software  Settings: 
SP  Desired  Gain  Setting 
LP  Uesi rea  Gain  Setting 
Seconds  Datachron  Set  Behind 
Datum  TCG  Time  Setting 
FTS  Receiver  Settings: 
FTS  Filter  Factor 
Reasonableness  Test 
Time 
0.0005   seconds 
0.005 
0.167 
37.001  seconds,   +/■ 
Sync  to  actual   time 
Latitude: 
Longitude:   *From  on-site  documents* 
Elevation: 
Enabled 
12.     STPR  CPU  Configuration  Parameters: 
a.     CPU1: 
CONFIGURATION  IDENTIFICATION  =  Cxxxx-ILL 
0PERATE1  IDENTIFICATION  =  0PERATE1 
SITE  IDENTIFICATION  =  415 
LP  DATA  AND  INSTRUMENT  TYPE  (A, 31, 36)  =  36 
NUMBER  OF  SHORT  PERIOD  ARRAY  CHANNELS  =  18 
NUMBER  OF  SHURT  PERIOD  OTHER  CHANNELS  =  9 
NUMBER  OF  LONG  PERlUD  ARRAY  CHANNELS  =  18 
NUMBER  OF  LONG  PERIOD  OTHER  CHANNELS 
NUMBER  OF  SHURT  PERlUD  PROCESSES  =  9 
NUMBER  OF  LONG  PERlUD  PROCESSES  =  4 
SHUkT  PERlUD  FREQUCNCY  FILTER  LENGTH 
LUNG  PERIOD  FREQUENCY  FILTER  LENGTH 
AMOUNT  UF  SHORT  PERIOD  TIME  DELAY  REQUIRED  =  U 
AMOUNT  UF  LUNG  PERIOD  TIME  DELAY  REQUIRED  =  0 
=  0 
=  9y 
=  1
CENR  bb-*.  Vol  VI       1  April  1988 
SP  CUUROINATES: 
0,0,0 
1,1.314,-4.048 
2,0.022,-2.509 
3,3.149,-3.408 
4,3.428,-6.368 
5,-0.132,-6.183 
6,-0.608,-4.593 
7,-1.687,-2.582 
8,1.195,-1.292 
9,4.685,-1.455 
10,5.853,-3.020 
11,1.474,-9.040 
12,-2.215,-5.658 
13,-4.162,-3.986 
14,-3.457,-2.321 
15,-1.934,-0.587 
16,0.000,0.000 
17,1.691,0.082 
18,3.859,0.450 
LP  COORDINATES: 
0,0,0 
1,0.000,0.000,0 
2,-10.492,21.836,0 
3, 0.440, 39. 318, C 
4,21.458,43.828,0 
5,21.892,24.590,0 
6,15.715,6.814,0 
SP  FREQUENCY  FILTER  PARAMETERS: 
0.0006,0. 0005, -.0002, -.0012, -.0022, -.0026, -.0024, -.0016, -.0007, -.0004 
-.0007, -.0015, -.0022, -.0020, -.0009, 0.0010, 0.0028, 0.0038, 0.0036, 0.0025 
0.0014, 0.0014, 0.0027, 0.0049, 0.0067, 0.0068, 0.0046, 0.0008, -.0031, -.0052 
-.0046, -.0024, -.0010, -.0029, -.0095, -.0192, -.0280, -.0316, -.0276, -.0188 
-.0120, -.0161, -.0354, -.0648, -.0869, -.0777, -.0174, 0.0911, 0.2099, 0.2658 
0.2099, 0.0911, -.0174, -.0777, -.0869, -.0648, -.0354, -.0161, -.0120, -.0188 
-.0276, -.0316, -.0280, -.0192, -.0095, -.0029, -.0010, -.0024, -.0046, -.0052 
-.0031,0.0008,0.0046,0.0068,0.0067,0.0049,0.0027,0.0014,0.0014,0.0025 
0.0036, 0.0038, 0.0028, 0.0010, -.0009, -.0020, -.0022, -.0015, -.0007, -.0004 
-.0007, -.0016, -.0024, -.0026, -.0022, -.0012, -.0002, 0.0005, 0.0006 
LP  FREQUENCY  FILTER  PARAMETERS 
0.9999 
SP  BEAM  PARAMETERS: 
SPL360,0,000,i3.0,B 
SPL060, 0,060, 13. 0,B 
SPL120.0, 120,13.0,8 
SPL180,0,180,13.0,B 
SPL240,0,240,13.0,B 
SPL300, 0,300, 13. 0,B 
SPZOOO.O.O.O.B 
SPL337, 0,337,13.0,8 
SPP317, 0,317, 15. 0,B 
LP  BEAM  PARAMETERS: 
LPH36Z,1,000,3.5,B 
LPH09Z, 1,090, 3.5,3 
LPH18Z, 1,180, 3.5,8 
LPH27Z, 1,270, 3.5,8 
SP  PROCESSING  DELAY  =  80 
LP  PROCESSING  DELAY  =  12 
SECONDS  PER  RECORD  =  3 
b.  CPU2: 
CONFIGURATION  IDENTIFICATION  =  Cxxxx-i!LL 
0PERATE2  IDENTIFICATION  =  0PERATE2 
SITE  IDENTIFICATION  =  4ib 
LP  DATA  AND  INSTRUMENT  TYPE  (A, 31, 3b)  =  36 
NUMBER  OF  SHORT  PERIUD  ARRAY  CHANNELS  =  18 
NUMBER  OF  SHORT  PERIOD  OTHER  CHANNELS  =  9 
NUMBER  UFLONG  PERIOD  ARRAY  CHANNELS  =  18 
NUMBER  OF  LONG  PERIOD  OTHER  CHANNELS  =  0 
NUMBER  OF  SHORT  PERIOD  PROCESSES  =  9 
NUMBER  OF  LONG  PERIUD  PROCESSES  =  4 
*NUMSJEK  OF  CONTACT  SENSOR  MONITORS  =  1 
*NUMBER  UF  A/D  CHANNEL  CHANNtL  MONITORS  =  1
CLKK  Sb-a,  Vol  VI       1  April  1988 
AMOUNT  OF  SP  EDIT  TIME  DELAY  REQUIRED  =  0 
AMOUNT  UF  LP  EDIT  TIME  DELAY  REQUIRED  =  U 
SP  COORDINATES: 
0,0,0 
1,1.314,-4.048 
2,0.022,-2.509 
3,3.149,-3.408 
4,3.428,-6.368 
5,-0.132,-6.183 
6,-0.608,-4.593 
7,-1.687,-2.582 
8,1.195,-1.292 
9,4.685,-1.455 
10,5.853,-3.020 
11,1.474,-9.040 
12,-2.215,-5.658 
13,-4.162,-3.986 
14,-3.457,-2.321 
15,-1.934,-0.587 
16,0.000,0.000 
17,1.691,0.082 
18,3.859,0.450 
LP  COORDINATES: 
0,0,0 
1, 0.000,0. 000, C 
2, -10. 492, 21. 836, C 
3, 0.440, 39. 318, C 
4, 21. 458, 43. 828, C 
5, 21. 892, 24. 590, C 
6, 15. 715, 6. 814, C 
SP  CALIBRATION  DEFAULT  PARAMETERS: 
0.833,1.0,25,0,030000,0.9,1.1,2.928,8 
1.00,1.708 
0.5,1.708 
0.8,1.708 
1.5,1.708 
2.0,1.708 
2.5,1.708 
3.0,1.708 
4.0,1.708 
LP  CALIBRATION  DEFAULT  PARAMETERS: 
.2539,0.04,10,0,030200,0.9,1.1,1.97,7,3 
0.04,. 2243 
O.J, 2. 243 
0.Vi67,.2243 
0.05,. 2243 
0.033,. 2243 
0.025, .2243 
0.020  .2243 
SP  CHANNEL  CONFIGURATION  FOR  CALIBRATION  SYSTEM: 
1,1 
1,2 
1.3 
1.4 
1.5 
1.6 
1,7 
1.8 
1.9 
1,10 
1.11 
1,12 
1,13 
1,14 
1.15 
1,16 
1.17 
•  1,18 
1,24 
1,24 
1,24 
1.24 
1.24 
1.24 
1.24 
1,24 
1.24
CENR  55-2,  Vol  VI       1  April  1968 
SP  BEAM  PARAMETERS: 
SPL360, 0,000, 13.0,8 
SPL060, 0,060, 13.0,8 
SPL12U.0, 120, 13.0,8 
SPL180.0, 180, 13.0,8 
SPL240, 0,240, 13. 0,B 
SPL300, 0,300, 13.0,8 
SPZOOO,0,0,0,B 
SPL337, 0,337, 13. 0,8 
SPP317, 0,317, 15.0,8 
LP  BEAM  PARAMETERS: 
LPH3bZ,l,000,3.5,B 
LPH09Z,1,090,3.5,B 
LPH18Z,I,18Q.3.5,B 
LPH27Z  1  270  3.5  B 
♦RELAY  IDENTIFIERS  AND  NORMAL  STATUS  FOR  EACH  CONTACT  SENSOR  MONITOR: 
TEST1.1 
♦IDENTIFIERS  AND  LIMITS  FOR  EACH  A/D  CHANNEL  MONITOR: 
LNP0WR,5.4,6.6 
SECONDS  PER  RECORD  =  1 
*  Monitors  may  be  added/modi fiea  at  the  station's  discretion. 
OFFICIAL  JAY  J.  JAYNES,  Colonel,  USAF 
Commanaer 
RICHARD  E.  COOK,  SMSgt,  USAF 
Director  of  Administration 
SUMMARY  OF  CHANGES 
Rewrote  in  active  voice.  Changed  STPR  ISENSE  ana  CGAIN,  and  CT  DOS  gain  for  array  and  high  gain 
channels.  Changed  SP  and  LP  CT  gain  settings.  Changed  LP  develocoraer  display  and  channel 
precedence.  Adjusted  SP  array  channel  develocoraer  sensitivity  check  voltages.