LEPAKTMENT OF THE AIR FUKCE CENTER REGULATIUN 55-2 Headquarters Air Force Technica} Applications Center Volume V Patrick AF¥ FL 32925-6001 2 November 1987 SPECIFIC STATION REQUIKEMENTS FUR DETACHMENT 313 This regulation establishes the proceaures for station unique operations ana analysis. It applies to all active auty Air Force members assignea to the station. Personnel who violate the specitic prohibitions ana requirements of this regulation may be prosecuted under tne Uniform Coae of Military Justice (UCMJ). Distribution limited to LoD ana DoD contractors only; to protect information anda technical aata which advance the state-of-the-art or describe new technology in an area of significant or potentially significant military application, 2 November 1987. Other requests shall be referrea to HQ/00SB. 1. Station Designator. The station aesignator for Detachment 313 is ZEBU. 2. Timing Stanaara. MSF. 3. koutine Calibrations. Perform SPS ana LPS calibrations sequentially commencing immediately after 09002. 4. Eait Tape registration numbers are 5500-5599. 5. Training Outage. Outage authorizea in Volume I is grantea tor Weanesaay of each week from 12002 through 1500Z. 6. Special Data Reports. Submit special aata reports in accoraance with CENR 55-2, Volume I. In adaition, submit a special aata report for all teleseismic signals received with an azimuth between 140 ana 170 degrees. Overlays are not available. 7. Summation Channel. Individual vertical array channel(s) may be manually linea out of analog summation(s) (but not from processea aata) when cultural or wing noise increases trace background on tne individual channel(s) to more than twice the background average of other array channels. Monitor indiviaual channels iinea out because of nigh background to determine when the backgrouna has subsided enough to return the channels to summation(s). & SPS Develocorger Presentations: a. Primary Develocoraer Presentations: TRACE DATA MAG 1 SZ18P36013 2000 2 SZ1BPU6u13 2000 3 SZIBP12013 2u00 4 SZ1BP18U13 2000 5 SZ1bP 24013 20u0 6 SZ1BP30013 2000 7 SZ1BPOL099 2UUU & SZLIOSL 10 9 SZ1SA 1000 10 SZ1Sb 106U Supersedes CENR 55-2, Vol V, 1 duly 1987. No of Printea Pages: 8 OPR: UUSB (TSgt C. W. Stephens) Appreved by: Col T. H. Niquette Eqitor: SSgt vu. M. Pless Distribution: X
2 CENR 55-2, Volume V 2 November 1987 TRACE DATA MAG ll SZISC 1000 12 $Z1165H 25U 13 SN1165H 250 14 SELIO5H 250 NOTE: Partial summations contain the tollowing contributors; SZI1SA - SZ1IO1, 02, 03, 04, 05, 06, and 07. SZ1SB - SZ1101, 04, 06, U7, 12, ana 18, SZ1SC - SZIIOi, 02, 05, U6, 09 and 15. b. Secondary Develocoraer Presentations: TRACE UATA MAG 1 SZ1119 Suu 2 SZLI1U 500 e 3 SZ1116 500 4 SZ1114 * svu 5 SZ1SA luuu 6 SZ1bPUz114 2uuu 7 $Z18PU3017 zuuu 8 SZ1bPUS1 LO 2UUG y SZ14PU5518 2uuU lu SZ16PU7816 zuvU ll SZ16P 15619 zuuu iz SZ1105t ; 5 13 $21165m 5u 14 SN1LO5M 50 15 SE 116514 50 * use Trace 4 wnenever a spare trace is required. 9. LPS UVevelocoraer Presentation: TRACE DATA maG 1 LZ116514 1UK 2 LNII65M 10K 3 LEL1650 10k 4 LZ1165H * 5 LN Lod * 6 LELIOdH * 7 LZLLodL In * AG Tolerances are listea in CENK 55-z, Volume I.
CENR 55-2, Volume V2 November 1987 lu. STPK Designator/Channel lIaentifier Cross keference: STPK DESIGNATUR CHANNEL ID INPUT SENSITIVITY SPRWOL Ss2llul 4.68* SPKWUZ SZ110z 4.88* SPRWUS SZ11u3 4.66* SPKWU4 " §Z11u4 4.86* SPRWUS SZ1Iu5 4.84* SPKWUO $Z1106 4.85* SPKWU7 sZlilu7 4 .0s* SPKWUS SZ11U& 4.88* SPRWOY Sal luy 4.8u* SPKW1U Ssz11lu 4.88* SPRW11 SZ1I11 4.6u* SPKW1Z SZ1I12 4.88* SPRW13 $Z1113 4.06* SPKW14 SZ1114 4.48* SPRW15S SZ1115 4 .86* SPRW16 $Z1116 4.8&* SPKW1L7 SZ1117 4.85* SPRKLS SZ1118 4.&b* SPRWLY SZ1I1y 4.68* SPKWZU SZ11 05H 4.88* SPRW21 SNLT65H 4.8B* SPRW2Z SELIO5H 4.88* SPRWZ3 SZ11T65L U.UY¥70* SPKWZ4 SNLT65L 0.0976* SPRWZD SELIO5L U.UY¥76* SPRWZ6 SZ1SA N/A SPRW27 $21SB N/A SPRWZ8 SZ1SC N/A LPSC12 LZ1165H 10+ LPSC1N LNLI65H 10+ LPSCIE LELIO5H lu+ LPSC2Z LZ1165L 1.U+ LPSC2N LNLI6SL 1.Q+ LPSC2E LELI@SL 1.0+
4 CENR 55-2, Volume V2 November 1957 STPR DESIGNATUK CHANNEL Iv INPUT SENSITIVITY SPL36U SZiBP 36013 N/A SPLU6O SZ1BPUbULS N/A SPL120 $Z1BP 12013 N/A SPL1&0 $Z18P 18013 N/A SPL24uU SZ1BP24U13 N/A SPL300 $Z16P3U013 N/A SPZUUU SZ1bPUUUS9 N/A SPNU21 SZ16P02114 N/A SPRUSU $Z1BPU3U17 N/A SPQU51 SZ1BP05116 N/A SPsub5 SZ16PU55iz8 N/A SPQU78 SZ1BPU7816 N/A SPD137 $Z1BP13749 N/A sPT156 SZ1BP15619 N/A SPVz54 $Z1BP25421 N/A SPYZ8U SZ1BP2Z40Z6 N/A LPH362 LZ1BP 3603.5 N/A * - Volts peak-to-peak for a 10G millimicron equivalent UF as measurea at the output of the SCC or KSsbUUU filter. + - Volts peak-to-peak for a 10 micron equivalent DF as measurea at the output of tne KS360U0 filter. ll. STPR Frequency Response Voltages and Normalizing Factors: a. Snort Perioa: FREQUENCY STPK VULTAGE NUKMALIZING FACTUK UAS KS360U00 UAS AND KS36UUU *L.u 1.708 -054 1 G.5 1.708 -854 1 Q.& 1.708 654 1 1.5 1.708 -854 1 2.U 1.708 -854 1 2.5 1.708 -854 1 3.U 1.70% -654 1 4.U 1.708 -54 1
CENR 55-2, Volume V 2 November 1987 b. tong Perioa: \ FREQUENCY STPR VOLTAGE *U.0400 0.666 U.1UUU 4.602 U.Lo07 U.666 U.USUU U.666 U.0333 0.666 U.UZ5U 0.606 U.U2UU U.o66 * - xeference Frequency NOTE: NORMALIZING FACTOR 1 0.1429 ro To normalize tne Frequency Kesponse, divide the return voltage of each frequency by the return voltage at tne reference frequency, then multiply by the normalizing factor. The results can then be compared with the values listea in CENR 55-2, Vol I to determine if they are within tolerances. 12. STPR CPU Configuration Parameters: a. CPU 1: CUNFIGURATIUN IDENTIFICATIUN = Cxxxx-1LS OPERATE] IVENTIFICATION = UPERATEL SITE IDENTIFICATION = 313 LP DATA AND INSTRUMENT TYPE (A,31,36) NUMBER UF SHORT PEKIUD ARRAY CHANNELS NUMBER OF SHUKT PERIUD UTHER CHANNELS NUMBEK OF LUNG PERIUU ARRAY CHANNELS NUMBER UF ILUNG PERIUD UTHER CHANNELS NUMBER UF SHURT PEKIUD PRUCESSES = lo NUMBER UF LONG PERIOD PROCESSES = 1 SHORT PERIOD FREQUENCY FILTER LENGTH = 99 LONG PERIOD FREGUENCY FILTER LENGTH = 1 AMOUNT UF SHURT PERIOD TIME UELAY KEQUIRED = U AMUUNT UF LUNG PERIUD TIME DELAY REQUIRED = 0 SP CUURDINATES: 0,0,0 1,1.854,-0.556 2,3.281,0.000 3,3.138,-1.297 4,2.282,-2.224 §,-0.143,-0.927 6,1.284,-0.185 7,1.426,1.297 8,3.708,1.668 9,5.848,0.185 10,5.705,-1.853 11,3.994,-3.150 12,1.997,-4.977 13,0.000,-2.780 14,-1.949,-2.224 15,-2.853,-0.741 16,-2.425,1.483 17,-0.713,2.965 18,1.569,3.706 19,5.848,2.780 LP COORDINATES: nou Lom Ce p> oO Cc
CENR 55-2, Volume VY 2 November 19&7 SP FREQUENCY FILTER PARAMETERS: SU y.UUU7 0.0005 ,0.0002,-.0061, -.G004,-.0007,-.0010,-.0012,-.0015,-.G016 -.0017,-.0017,-.0016,-.0014,-.00i1,-.0007,-.0001,0.0005,0.0011,0.0018 9.0024 ,0.0031,0.0036 0.0040 ,0.0042,0.0042,0.0640,0.0035,0.0026,0.0014 -.0001,-.0020, -.0043,-.0068,-.0097,-.0129,-.0163, -.0198,-.0235,-.0272 -.0308, -.0343,-.0377,-.0407,-.0434,-.0457,-.0476,-.0490,-.0498,0.9499 -.0498, -.0490,-,U476,-.0457 ,-.0434,-.0407,-.0377,~.0343, -.0308,-.0272 -.0235,-.0198,-.0163,-.0129, -.0097, -.0068,-.0043,-.0020,-.0001,0.0014 0.0026 ,0.0035,0.0040 ,0.0042,0.0042,0.0040,0.0036 0.0031 ,0.0024 ,0.0018 0.0011,0.0005,-.0001,-.0007,-.0011,-.0014,-.0016,-.0017,-.0017,-.0016 -.0015, -.0012,-.0010,-.0007,-.0004,-.0001,0.0002,0.0005,0.G007 LP FREQUENCY FILTER PARAMETERS: 0 0.9999 SP BEAM PARAMETERS: SPL360,0,000,13.0,8 SPL06U,0,060,13.0, SPL120,0,120,13.0 SPL1&U,0,180,13.0 SPL240,0, 240,13. SPL300,0,300,13. SPZ000,0,0,0,8 sPN021,0,021,13. SPRO30,0,030,16. SPyU51,0,051,15. 7. 5. 0 0 §PS055,0,055,1 SPQU78,0,078,1 $P0137,0,137,8.5 SPT156,0, 156,18. SP¥254,0,254,21. SPY26U,0,280,25.6 LP BEAM PARAMETERS: LPH36Z,1,000,3.5,8 SP PRUCESSING DELAY LP PRUCESSING DELAY SECONDS PER KECURD = 3 8B 8 B 69 1 CPu 2: CUNFIGURATIUN IDENTIFICATIUN = Cxxxx-2LS UPERATE2 IGENTIFICATIUN = OPERATEZ SITE IDENTIFICATIUN = 313 LP DATA AND INSTRUMENT TYPE (A,31,36) = A NUMBER UF SHORT PERIUU ARRAY CHANNELS = 19 NUMBER CF SHURT PERIUY OTHER CHANNELS = 9 NUMBER UF LUNG PEKIGU ARKAY CHANNELS = 6 NUMBER UF LUNG PERIGD UTHER CHANNELS = 0 NUMBER UF SHUKT PEKIGD PRUCESSES = 16 NUMBER UF LUNG PERIUD PROCESSES = 1 NO UF SP CHANNELS TU BE TRANSMITTED VIA HSia NG UF LP CHANNELS TU BE TRANSMITTED VIA HSin NUMBER UF CUNTACT SENSUR MUNITORS = 4 NUMBER UF A/D CHANNEL MUNITURS = 1 AMOUNT UF SP EDIT TIME DELAY REQUIRED AMUUNT UF LP EUIT TIME DELAY REQUIRED SP CUUKUINATES: 0,0,0 1,1.854,-0.556 2,3.281,0.000 3,3.138,-1.297 4,2.282,-2.224 5,-0.143,-0.927 6,1.284,-0.185 7,1.426,1.297 8,3.708,1.668 9,5.848,0.185 10,5.705,-1.853 11,3.994,-3.150 12,1.997,-4.077 13,0.000,-2.780 14,-1.949,-2.224 wou Cc wou Cc
CENR 55-2, Volume V2 November 1987 15,-2.853,-0.741 16,-2.425,1.483 17,-0.713,2.965 18,1.569,3.706 19,5.848,2.780 LP CUURDINATES: 0,0,0 1,0,0,C 2,0,0,C SP CALIBRATION DEFAULT PARAMETERS: -833,1.000,10,1,090900,0.9,1.1,2.928,8 -0,1.708 1.708 1.708 1.708 1 1 1 oOoro 793 - 708 - 708 -0,1.708 P CALIBRATIUN DEFAULT PARAMETERS: -333,0.04,10,1,093000,0.9,1.1,3.750,7,3 -040,0.666 -100,4.662 -067,0.666 -050,0.666 -033,0.666 -025,0.666 -020,0.666 P CHANNEL CONFIGURATIUN FUR CALIBRATION SYSTEM: we ew we . rr rt hs OO NI ON BON) — NNO WMH O we ewe ee ee ww ~ Me RR ee Ree NODDOOOGOKT AWNHNe ° 2 1,18 1,19 1,20 1,20 1,20 1,20 1,20 1,20 1,21 1,21 L,2l LP CHANNEL CONFIGURATION FOR CALISRATIUN SYSTEM: SP BEAM PAKAMETERS: SPL360 ,0,000,13.0, SPLU60,0,060,13.0, SPL120,0,120,13.0 SPL160,0,180,13.0 SPL240 ,0,240,13.0 3.0 B B B B B SPL3U0,0,300,1 B , ’ , ’
8 CENR 55-2, Volume Y 2 November 1987 $P2006,0,0,0,8 SPNO21,0,021,1 SPRO30,0,030,1 SPQU51,0,051,1 1 1 1 SPS055,0,055, SPQU78,0,078, $P0137,0,137, SPT156,0,156, SPV254,0,254,2 SPY280,0,280,25.6 LP BEAM PARAMETERS: LPH36Z,1,000,3.5,B RELAY IDENTIFIERS AND NORMAL STATUS FUK EACH CONTACT SENSGR MONITUR: ACFAIL,1 BATTLU,O RDOOR,! MTDOOR , 0 IDENTIFIERS AND LIMITS FUR EACH A/D CHANNEL MUNITOR: LNPOWR ,5.4,6.6 SECONDS PEx XECURD = 1 3.8,B 6.9,8 5.7,8 7.6,B $.7,B -5, 8.5,B 1.1,8 B B UFFICIAL SUMMARY UF CHANGES Upaatea channel data tables and processes to incorporate expanded SP array. Changed Develocoraer displays.