CENR  55-2 
Volume  II 
1  July  1987 
SPECIFIC  STATION  REQUIREMENTS  FOR  EL  079 
This  regulation  establishes  the  procedures  for  station  unique  operations  and  analysis. 
Distribution  limited  to  DoD  and  DoD  contractors  only;  to  protect  information  and  technical  data 
which  advance  the  state-of-the-art  or  describe  new  technology  in  an  area  of  significant  or 
potentially  significant  military  application,  1  July  1987.  Other  requests  shall  be  referred  to 
HQ/DOSB. 
1.  Operating  Concept.  Normally,  all  operations  and  maintenance  requirements  are  accomplished 
during  the  attended  hours  as  determined  by  contractor  management.  All  days  are  considered  duty 
days.  A  complete  equipment  check  is  required  at  the  beginning  and  end  of  the  attended  period. 
2.  Station  Designator.  The  station  designator  for  Equipment  Location  079  is  JOEY. 
3.  Timing  Standard  is  WWV. 
4.  Routine  Calibrations.  Commence  SP  calibrations  immediately  after  1330Z  followed  by  the  LP 
CALs.  Perform  the  LPS  frequency  responses  at  the  rate  of  one  site  per  day. 
5.  Edit  tape  registration  numbers  are   5000  through  5099. 
6.  Outage  authorized  in  Volume  I,  is  granted  for  Monday  of  each  week  from  1500Z  through  1800/1. 
7.  Analysis  Requirements: 
a.  The  station  is  exempt  from  routine  analysis  and  data  reporting  with  the  following 
exceptions: 
(1)  Transmit  data  reports  covering  periods  requested  by  the  GSOC.  Include  in  this 
report  all  events  extending  into,  or  continuing  out  of,  the  requested  period. 
(2)  If  data  request  covers  more  than  one  ZULU  day,  a  new  computer  function  data  line 
(BBBBBB  JJJ  JOEY  (date)  CMM  PART  ONE)  will  pj-ecede  each  day's  data.  If  data  are  requested  over 
an  extended  period  of  time,  each  data  reporting  period  will  cover  12  hours  (0001  -  1200Z  and 
1200  -  2400Z). 
b.  In  addition  to  the  above  requirement,  maintain  a  continuous  capability  to  respond  to 
review  requests.  Establish  analysis  and  reporting  exercises  to  insure  analysis  proficiency  of 
personnel. 
c.  Provide  selected  analysis  periods  to  HQ  for  review  to  evaluate  the  station's  analysis 
and  reporting  capability.  Procedures  are  as  follows: 
(1)  Analyze  1600  -  2000Z  on  the  15th  day  of  each  month. 
(2;  Prepare  a  message  (do  not  transmit)  using  correct  format  as  specified  in  Volume  I, 
and  forward  with  the  appropriate  station  log.  Do  not  complete  address  elements. 
(3)  It  is  not  the  intent  of  this  program  to  limit  the  station's  analysis  and  reporting 
exercises  to  1  day  a  month.  Accomplish  analysis  and  reporting  training  on  a  continuing  basis, 
and  this  program  may  be  used  to  complement  that  training. 
8.  SPS  Develocorder  Presentations: 
a.  Primary: 
TRACE  DATA  MAG 
1  SZ1BP29216  2000K 
2  SZ1BP31516  2000K 
3  SZ1BP33716  2000K 
Supersedes  CENR  55-2,  Volume  II,  18  May  1981. 
No  of  Printed  Pages:  11 
OPR:  DOSB 
Distribution:  X
CENR 
Vol 
II 
1  July  1987 
TRACE 
f)ATA 
MAG 
SZ18P36016 
2000K 
SZ1BP02216 
20C0K 
SZ1BP04516 
2000K 
SZ1BP00099 
2000K 
S 
SZ1BP15716 
2000K 
SZ1BP30314 
2000K 
SZ0AP30314 
2000K 
SZ0AP02216 
2000K 
SZ1BP21124 
2000K 
SZ1I32H 
500K 
b. 
Sec ondary: 
TRACE 
OATA 
MAG 
SZ1I19 
500K 
SZ1I13 
500K 
SZ1I10 
50CK 
SZ1I16 
500K 
SZ0AP30314 
2000K 
SZ0AP02216 
2000K 
SPARE 
- 
SZ1I32M 
50K 
SZ1I32L 
5K 
SZ1I32H 
500K 
SN1I32H 
500K 
SE1I32H 
500K 
9.  LPS Develocorder Presentatfi on: 
TRACE 
DATA 
MAG 
LZ13P3603.5 
100K 
LZ13P0903.5 
100K 
LZ1BP1803.5 
100K 
LZ1BP2703.5 
100K 
LZ1ICH 
50K* 
LN1ICH 
50K* 
LE1ICH 
50K* 
LZ1ICL 
*  Norma 
illy record data from LP site  C  on traces  5,  6,  7  and  8. 
Should site  C fail ,  substitute 
any  operable  LP  site.
CENR  55-2, ,  Vol  II   1  July  1987 
10.   STPR Designator/Channel  Identifier Cross  Reference 
STPR  DESIG iNATGR 
CHANNEL  ID 
SPRW01 
SZ1I01 
SPRW02 
SZ1I02 
SPRW03 
SZ1I03 
SPRW04 
SZ1I04 
SPRW05 
SZ1I05 
SPRW06 
SZ1I06 
SPRW07 
SZ1I07 
SPRW08 
SZ1I08 
SPRW09 
SZ1I09 
SPRW10 
SZ1I10 
SPRW11 
SZ1I11 
SPRW12 
SZ1I12 
SPRW13 
SZ1I13 
SPRW14 
SZ1I14 
SPRW15 
SZ1I15 
SPRW16 
SZ1I16 
SPRW17 
SZ1I17 
SPRW18 
SZ1I18 
SPRW19 
SZ1I19 
SPRW20 
SZ1I32H 
SPRW21 
SN1I32H 
SPRW22 
SE1I32H 
SPRW23 
SZ1I32M 
SPRW24 
SZ1I32L 
LPSC1Z 
LZ1IA 
LPSC1N 
LN1IA 
LPSC1E 
LE1IA 
LPSC2Z 
LZ1IB 
LPSC2N 
LN1IB 
INPUT  SENSITIVITY 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
4.88* 
.488* 
.0488* 
5.7735 
5.7735 
5.7735 
5.7735 
5.7735
STPR  DESIGNATOR 
LPSC2E 
LPSC3Z 
LPSC3N 
LPSC3E 
LPSC4Z 
LPSC4N 
LPSC4E 
LPSC5Z 
LPSC5M 
LPSC5E 
SPZOOO 
SPQ360 
SPQ022 
SPQ045 
SPQ157 
SPQ292 
SPQ315 
SPQ337 
SPQ02A 
SPN30A 
SPN303 
SPY211 
LPH36Z 
LPH09Z 
LPH18Z 
LPH27Z 
CHANNEL  ID 
LE1IB 
LZ1IC 
LN1IC 
LE1IC 
LZ1ID 
LN1ID 
LE1ID 
LZ1IE 
LN1IE 
LE1IE 
SZ1BP00099 
SZ1BP36016 
SZ1BP02216 
SZ1BP04516 
SZ1BP15716 
SZ1BP29215 
SZ1BP31516 
SZ1BP33716 
SZ0AP02216 
SZ0AP30314 
SZ1BP30314 
SZ1BP21124 
LZ1BP3603.5 
LZ1BP0903.5 
LZ1BP1803.5 
LZ1BP2703.5 
CENR  55-2,  Vol  II   1  July  1987 
INPUT  SENSITIVITY 
5.7735 
5.7735 
5.7735 
5.7735 
5.7735 
5.7735 
5.7735 
5.7735 
5.7735 
5.7735 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
*  Volts  peak-to-peak  for  100  millimicron  equivalent  OF  measured  at  the  output  of  the  SCC. 
+  Input  sensitivities  for  the  triaxial  sensors  are   measured  at  the  output  of   the  SCC. 
11.  Channels  to  be  transmitted  to  the  GSOC: 
CHANNEL  ID 
SZ1I32H 
SN1I32H 
SE1I32H 
SZ1I32M 
STPR  DESIGNATOR 
SPRW20 
SPRW21 
SPRW22 
SPRW23
CENR  55-2,  Vol  II   1  July  1987 
CHANNEL  ID 
SZ1I32L 
SZ1BP00099 
SZ1BP360L6 
SZ1BP02216 
SZ1BP04516 
SZ1BP15716 
SZ1BP29216 
SZ1BP21124 
SZ1BP31516 
SZ1BP33716 
SZ1I09 
SZ1I11 
SZ1I14 
SZ1I17 
LZ1BP3603.5 
LZ1BP0903.5 
LZ1BP1803.5 
LZ1BP2703.5 
LZ1IC 
LN1IC 
LE1IC 
LZ1IA 
LN1IA 
LE1IA 
LZ1IB 
LN1IB 
LE1IB 
LZ1ID 
LZ1IE 
STPR  DESIGNATOR 
SPRW24 
SPZOOO 
SPQ360 
SPQ022 
SPQ045 
SPQ157 
SPQ292 
SPY211 
SPQ315 
SPQ337 
SPRW09 
SPRW11 
SPRW14 
SPRW17 
LPH36Z 
LPH09Z 
LPH18Z 
LPH27Z 
LPSC3Z 
LPSC3N 
LPSC3E 
LPSC1Z 
LPSC1N 
LPSC1E 
LPSC2Z 
LPSC2N 
LPSC2E 
LPSC4Z 
LPSC5Z 
12.  STPR  Frequency  Response  Voltages  and  Normalizing  Factors: 
a.  Short  Period: 
FREQ  STPR  VOLTAGE 
1.0  1.708 
0.5  1.708 
0.8  1.708 
NORMALIZING  FACTOR
FREO 
STPR  VOLTAGE 
1.5 
1.708 
2.0 
1.708 
2.5 
1.708 
3.0 
1.708 
4.0 
1.708 
b. 
Long 
Peri od: 
FREQ 
STPR  Y0LTA6E 
.0400 
1.053 
.1000 
5.265 
.0667 
1.053 
.0500 
1.053 
.0333 
1.053 
.0250 
1.053 
.0200 
1.053 
*  Reference Frequency 
CENR  55-2,  Vol  II  1  July  1987 
NORMALIZING  FACTOR 
NORMALIZING  FACTOR 
.2 
NOTE:  To  normalize  the  Frequency  Response,  divide  the  return  voltage  of  each  frequency  by  the 
return  voltage  at  the  reference  frequency,  then  multiply  by  the  normalizing  factor.  The 
results  can  then  be  compared  to  the  table  in  paragraph  13  to  determine  if  they  are  within 
tolerances. 
13.  Frequency  Response  Parameters  for  TRIAX  W/30237  Telemetry  Amplifiers. 
Frequency  (Hz) 
.0200 
.0250 
.0333 
.0400 
.0500 
.0667 
.1000 
Period  (Sec) 
Minimum 
1.977 
1.762 
1.300 
1.000 
.4849 
.1318 
.0144 
Nominal 
2.471 
2.073 
1.444 
1.000 
.5388 
.1648 
.0192 
Maximum 
2.965 
2.384 
1.588 
1.000 
.5927 
.1978 
.0240 
14.  Ground  Motion  Table. 
TRIAXIAL  SENSOR 
30237  TELE  AMP 
Period  (Sec) 
GtT 
.8354 
.4854 
.3269 
.2385 
H 
.1831 
.1457
CENR  55-2,  Vol  II   1  July  1987 
TRIAXIAL SENSOR 
30237  TELE  AMP 
Period  (Sec) 
fitT 
.1169 
.0962 
.0807 
.0688 
.0594 
.0545 
.0502 
.0464 
.0430 
.0400 
.0384 
.0370 
.0356 
.0344 
.0332 
.0328 
.0324 
.0321 
.0318 
.0315 
.0313 
.0312 
.0310 
.0309 
.0309 
.0309 
.0309 
.0309 
.0310 
.0311 
.0313 
.0315 
.0317 
.0320 
.0324 
15.  STPR  CPU  Configuration  Parameters: 
a.  CPU1: 
CONFIGURATION  IDENTIFICATION  =  Cxxxx-IHS 
0PERATE1  IDENTIFICATION  =  0PERATE1 
SITE  IDENTIFICATION  =  079 
LP  DATA  AND  INSTRUMENT  TYPE  (A, 3 1,36)  =  A 
NUMBER  OF  SHORT  PERIOD  ARRAY  CHANNELS  =  19 
NUMBER  OF  SHORT  PERIOD  OTHER  CHANNELS  «  5 
NUMBER  OF  LONG  PERIOD  ARRAY  CHANNELS  =  15 
NUMBER  OF  LONG  PERIOD  OTHER  CHANNELS  =  0 
NUMBER  OF  SHORT  PERIOD  PROCESSES  =  12 
NUMBER  OF  LONG  PERIOO  PROCESSES  =  4 
SHORT  PERIOD  FREQUENCY  FILTER  LENGTH  =  99 
LONG  PERIOD  FREQUENCY  FILTER  LENGTH  =  99 
AMOUNT  OF  SHORT  PERIOD  TIME  DELAY  REQUIRED  =  0 
AMOUNT  OF  LONG  PERIOD  TIME  DELAY  REQUIRED  =  0 
SP  COORDINATES: 
0,3.844,-5.502 
1,0.0,0.0 
2,2.531,1.993 
3,3.147,-1.565 
4,0.366,-3.091 
5,-1.996,-1.859 
6,-2.714,1.402 
7,-0.240,3.050 
8,2.314,5.211
CENR  55-2,  Vol  II   i  July  1987 
9,5.151,4.057 
10,5.744,0.745 
11,6.132,-2.807 
12,3.646,-4.351 
13,0.899,-6.438 
14,-1.913,-5.061 
15,-4.869,-3.704 
16,-5.235,-0.399 
17,-5.896,2.605 
18,-3.202,4.087 
19,-0.892,6.271 
LP  COORDINATES: 
0,3.844,-5.502 
1,0. 082,9. 607, T 
2,15.324, 10. 269, T 
3, 3. 843, -5. 519, T 
4, -14. 816, 7. 298, T 
5,-2. 117, 23. 664, T 
SP  FREQUENCY  FILTER  PARAMETERS: 
0.0001, -.0001, -.0005, -.0011, -.0016, -.0020, -.0020, -.0017, 0.0014, -.0012 
-.0011, -.0013, -.0014, -.0011, -.0004, 0.0007, 0.0019, 0.0027, 0.0030, 0.0029 
0.0027, 0.0028, 0.0034, 0.0043, 0.0050, 0.0049, 0.0038, 0.0017, -.0007, -.0027 
-.0037, -.0041, -.0045, -.0065, -.0103, -.0162, -.0221, -.0266, -.0273, -.0254 
-.0224, -.0237, -.0315,-. 0481, -.0653, -.0731, -.0456, 0.0324, 0.2035, 0.3910 
0.2035, 0.0324, -.0456, -.0731, -.0653, -.0431, -.0315, -.0237, -.0224, -.0254 
-.0273, -.0266, -.0221, -.0162, -.0103, -.0065, -.0045, -.0041, -.0037, -.0027 
-.0007,0.0017,0.0038,0.0049,0.0050,0.0043,0.0034,0.0028,0.0027,0.0029 
0.0030, 0.0027, 0.0019, 0.0007, -.0004, -.0011, -.0014, -.0013, -.0011, -.0012 
-.0014, -.0017, -.0020, -.0020, -.0016, -.0011, -.0005, -.0001, -.0001 
LP  FREQUENCY  FILTER  PARAMETERS: 
0.0020,0.0023,0.0026,0.0028,0.0028,0.0025,0.0020,0.0016,0.0013,0.0011 
0.0012, 0.0013, 0.0012, 0.0007, -.0001, -.0012, -.0023, -.0032, -.0037, -.0040 
-.0040, -.0043, -.0051, -.0066, -.0084, -.0104, -.0119, -.0127, -.0126, -.0122 
-.0120, -.0128, -.0148, -.0180, -.0213, -.0237, -.0239, -.0218, -.0179, -.0148 
-.0143, -.0197, -.0299, -.0435, -.0515, -.0456, -.0078, 0.0697, 0.2150, 0.3650 
0.2150, 0.0697, -,0078, -.0456, -.0515, -.0435, -.0299, -.0197, -.0143, -.0148 
-.0179, -.0218, -.0239, -.0237, -.0213, -.0180, -.0148, -.0128, -.0120, -.0122 
-.0126, -.0127, -.0119, -.0104, -.0084, -.0066, -.0051, -.0043, -.0040, -.0040 
-.0037, -.0032, -.0023, -.0012, -.0001, 0.0007, 0.0012, 0.0013, 0.0012, 0.0011 
0.0013,0.0016,0.0020,0.0025,0.0028,0.0028,0.0026,0.0023,0.0020 
SP  BEAM  PARAMETERS: 
SPQ02A,0,22,16,A,31,0.08,1 
SPN30A, 0,303, 14, A, 31, 0.08,1 
SPZOOO.O.O.O.B 
SPQ360, 0,0, 16,8 
SPQ022, 0,22,16,8 
SPQ045, 0,45,16,8 
SPQ157, 0,157, 16, B 
SPY211, 0,211, 24, B 
SPQ292, 0,292,16,8 
SPN303, 0,303, 14, B 
SPQ315,0,315,16,B 
SPQ337, 0,337,16,8 
LP  BEAM  PARAMETERS: 
LPH36,1,0,3.5,B 
LPH09,1,90,3.5,B 
LPH18, 1,180,3.5,8 
LPH27,1,270,3.5,B 
SPQ02A  CONSTRAINTS: 
0,0,0,0,0,0,0,0,0,0 
0,0,0,0,0,1,0,0,0,0 
0,0,0,0,0,0,0,0,0,0
CENR  55-2,  Vol  II   1  July  1987 
SPM3QA  CONSTRAINTS: 
0,0,0,0,0,0,0,0,0,0 
0,0,0,0,0,1,0,0,0,0 
0,0,0,0,0,0,0,0,0,0 
SP  PROCESSING  DELAY  =  100 
LP  PROCESSING  DELAY  =  58 
SECONDS  PER  RECORD  =  3 
b.  CPU2: 
CONFIGURATION  IDENTIFICATION  =  Cxxxx-2HS 
0PERATE2  IDENTIFICATION  =  0PERATE2 
SITE  IDENTIFICATION  *  079 
LP  DATA  AND  INSTRUMENT  TYPE  (A, 31, 36)  =  A 
NUMBER  OF  SHORT  PERIOD  ARRAY  CHANNELS  =  19 
NUMBER  OF  SHORT  PERIOD  OTHER  CHANNELS  =  5 
NUMBER  OF  LONG  PERIOD  ARRAY  CHANNELS  =  15 
NUMBER  OF  LONG  PERIOD  OTHER  CHANNELS  =  0 
NUMBER  OF  SHORT  PERIOD  PROCESSES  =  12 
NUMBER  OF  LONG  PERIOD  PROCESSES  =  4 
NO  S?   CHANNELS  TO  BE  TRANSMITTED  VIA  HSM  =  18 
NO  LP  CHANNELS  TO  BE  TRANSMITTED  VIA  HSM  =  15 
NUMBER  OF  CONTACT  SENSOR  MONITORS  =  1 
NUMBER  OF  A/D  CHANNEL  CHANNEL  MONITORS  =  1 
AMOUNT  OF  SP  EDIT  TIME  DELAY  REQUIRED  =  0 
AMOUNT  OF  LP  EDIT  TIME  DELAY  REQUIRED  =  0 
SP  COORDINATES: 
0,3.844,-5.502 
1,0.0,0.0 
2,2.531,1.993 
3,3.147,-1.565 
4,0.366,-3.091 
5,-1.996,-1.859 
6,-2.714,1.402 
7,-0.240,3.050 
8,2.314,5.211 
9,5.151,4.057 
10,5.744,0.745 
11,6.132,-2.807 
12,3.646,-4.351 
13,0.899,-6.438 
14,-1.913,-5.061 
15,-4.869,-3.704 
16,-5.235,-0.399 
17,-5.896,2.605 
18,-3.202,4.087 
19,-0.892,6.271 
LP  COORDINATES: 
0,3.844,-5.502 
1, 0.082, 9. 607, T 
2,15.324, 10.269, T 
3, 3.843, -5. 519, T 
4,-14.816, 7. 298.T 
5,-2.117,23.664,T 
SP  CALIBRATION  DEFAULT  PARAMETERS: 
0.833,1.0,25,1,133000,0.9,1.1,2.931,8 
1.0,1.708 
0.5,1.708 
0.8,1.708 
1.5,1.708 
2.0,1.708 
2.5,1.708 
3.0,1.708 
4.0,1.708 
LP  CALIBRATION  DEFAULT  PARAMETERS: 
1.2156,0.04,10,1,140000,0.9,1.1,2.375,7,3 
0.040,0.528 
0.100,5.264 
0.067,1.0528 
0.050,1.0528
CENR  55-2,  Vol  II   1  July  1937 
0.033,1.0528 
0.025,1.0528 
0.020,1.0528 
SP  CHANNEL  CONFIGURATION  FOR  CALIBRATION  SYSTEM: 
1,1 
2,1 
9,1 
1,2 
1,3 
1.4 
2,2 
2,3 
2,4 
9,2 
9.3 
9,4 
9,5 
1,5 
1,6 
1,7 
1,8 
2,5 
2,6 
9,6 
9,7 
9,8 
9,6 
9,6 
LP  CHANNEL  CONFIGURATION  FOR  CALIBRATION  SYSTEM: 
5,1 
5,1 
5.1 
5,2 
5,2 
5,2 
9.1 
9,1 
9,1 
2,1 
2,1 
2,1 
5,3 
5,3 
5,3 
SP  BEAM  PARAMETERS: 
SPQ02A,0,22,16,A,31,0.08,1 
SPN30A, 0,303, 14, A,31, 0.08,1 
SPZOOO, 0,0,0,3 
SPQ360,0,0,16,B 
SPQ022,0,22,16,B 
SPQ045, 0,45,16,8 
SPQ157, 0,157, 16, B 
SPY211, 0,211, 24, B 
SPQ292,0, 292, 16,3 
SPM303, 0,303, 14, B 
SPQ315,0,315,16,B 
SPQ337, 0,337, 16,B 
LP  BEAM  PARAMETERS: 
LPH36, 1,0,3.5, B 
LPH09,1,90,3.5,B 
LPH18,1,180,3.5,B 
LPH27, 1,270, 3.5,8 
HIGH  SPEED  MODEM  CONFIGURATION: 
SPRW20,SPRW21,SPRW22,SPRW23,SPRW24,SPZOOO,SPQ360,SPQ022,SPQ045,SPQ157 
SPQ292,SPY211,SPQ315,SPQ337,SPRW09,SPRW11,SPRM14,SPRW17,LPH36Z,LPH09Z 
LPH18Z,LPH27Z,LPSC3Z,LPSC3N,LPSC3E,LPSC1Z,LPSC1N,LPSC1E,LPSC2Z,LPSC2N 
LPSC2E , LPSC4Z , LPSC4Z , LPSC5Z
CENR  55-2,  Vol  II   1  July  1987  11 
RELAY  IDENTIFIERS  AND  NORMAL  STATUS  FOR  EACH  CONTACT  SENSOR  MONITOR: 
C NT AC  1.1 
IDENTIFIERS  AND  LIMITS  FOR  EACH  A/D  CHANNEL  MONITOR: 
LNPOWR, 5.4,6.6 
SECONDS  PER  RECORD  =  1 
OFFICIAL 
SUMMARY  OF  CHANGES 
Incorporated  IMC  86-1  and  87-1.  Added  purpose  statement.  Added  limited  distribution  state- 
ment. Deleted  references  to  specific  paragraphs  in  Vol  I.  Added  TRIAX  frequency  response 
parameters  and  ground  motion  table.  Added  configuration  parameters  for  CPU  1  and  CPU  2.