DEPARTMENT  OF  THE  AIR  FORCE 
Headquarters  Air  Force  Technical  Applications  Center 
Patrick  AFB  FL,  32925-6001 
CENTER  REGULATION  55-2 
Volume  IX 
I  April  1988 
SPECIFIC  STATION  REQUIREMENTS  FOR  DETACHMENT  452 
This  regulation  establishes  the  procedures  for  station  unique  operations  and  analysis.  It 
applies  to  all  active  duty  Air  Force  members  assigned  to  the  station.  Personnel  who  violate  the 
specific  prohibitions  and  requirements  of  this  regulation  may  be  prosecuted  under  the  Uniform 
Code  of  Military  Justice  (UCMJ). 
Distribution  limited  to  DoO  and  DoD  contractors  only;  to  protect  information  and  technical  data 
which  advance  the  state-of-the-art  or  describe  new  technology  in  an  area  of  significant  or 
potentially  significant  military  application,  1  April  1988.  Other  requests  shall  be  referred  to 
HQ/DOSB. 
1.  Station  Designator.  The  station  designator  for  Detachment  452  is  KSRS  (Korean  Seismological 
Research  Station).  Use  KSR  for  the  three  element  entry  preceding  the  station  name  on  data 
messages. 
2.  Timing  Standards.  JJY  or  BPM. 
3.  Routine  Calibrations.  Perform  SPS  and  LPS  calibrations  sequentially,  commencing  immediately 
after  010CZ  and  0130Z  respectively.  The  LPS  frequency  responses  may  be  performed  at  the  rate  of 
one  site  per  day. 
4.  Data  Reports.   Submit  special  data  reports  in  accordance  with  Volume  I.  Submit 
routine  data  reports  within  6  hours  of  the  end  of  the  reporting  period. 
5.  EDIT  tape  registration  numbers  are  5100  through  5199. 
6.  Training  Outage.  Training  outage  (IAW  Vol  I)  is  granted  for  Thursday  of  each  week  from 
2300Z  through  0200Z  Friday. 
7.  SPS  Develocorder  Presentation: 
a.  Primary  Develocorder: 
TRACE    DATA       MAG     ASN  CHAN   DISP  ID 
SZ1BP36013 
SZ1BP06013 
SZ1BP12013 
SZ1BP18013 
SZ1BP24013 
SZ1BP30013 
SZ1BP00099 
SZ1BP030I3 
SZ1BP28713 
SZ1BP30816 
SZ1BP33815 
SZ1I75H 
SN1I75H 
SE1I75H 
1000K 
1000K 
1000K 
1000K 
I000K 
1000K 
1000K 
1000K 
1000K 
1000K 
1000K 
250K 
250K 
250K 
SPDS01 
SPDS02 
SPDS03 
SPDS04 
SPDS05 
SPDS06 
SPDS'J7 
SPDS08 
SPDS11 
SPDS12 
SPDS13 
b.  Secondary  Develocorder: 
TRACE    DATA       MAG     ASN  CHAN 
1  SZ1I08 
2  SZ1I14 
3  SZ1I11 
250K 
250K 
250K 
Supersedes  CENR  '55-2',  Vol  IX,  1  July  1982. 
No  of  Printed  Pages:  9 
OPR:  OOSB  (MSgt  M.  P.  Clark) 
Approved  by:  Col  T.  H.  Niquette 
Editor:  SSgt  D.  M.  PI  ess 
Distribution:  X 
SPRW08 
SPRW14 
SPRW11 
SCALE 
SPL360  1.0 
SPL060  1.0 
SPL120  1.0 
SPL180  1.0 
SPL240  1.0 
SPL300  1.0 
SPZ0O0  1.0 
SPL030  1.0 
SPL287  1.0 
SPQ308  1.0 
SPP338  1.0 
SPRW20 
SPRW21 
SPRW22 
DISP  ID    SCALE 
DEV  SENSE 
VOLTS 
0.61 
0.61 
0.61 
0.61 
0.51 
0.61 
0.61 
0.61 
0.61 
0.61 
0.61 
2.44 
2.44 
2.44 
DEV  SENSE 
VOLTS 
2.440 
2.440 
2.440
CtNk  5£>-2,  Vol  IX     1  April  1983 
9.  Channels  transmitted  to  the  TOS: 
10. 
CHANNEL 
U1SP  ID 
SCALE 
SPHUU1 
SPRW01 
1.0 
SPH002 
SPKWU3 
1.0 
SPHUG3 
SPRW05 
1.0 
SPHU04 
SPRW07 
1.0 
SPHU05 
SPRWOb 
1.0 
SPHU06 
SPRW09 
1.0 
SPHD07 
SPRW10 
1.0 
SPHDOb 
SPRW11 
1.0 
SPHDQ9 
SPRW12 
1.0 
SPH01U 
SPRW13 
1.0 
SPHOil 
SPRW14 
1.0 
SPHD12 
SPRW16 
1.0 
SPHD13 
SPRW17 
1.0 
SPHU14 
SPRW18 
1.0 
SPHU15 
SPRW19 
1.0 
SPHD16 
SPRW15 
1.0 
SPHD17 
SPRW21 
1.0 
SPHD18 
SPRW22 
1.0 
LPHD01 
LPSC1Z 
1.0 
LPHDO^ 
LPSC1N 
1.0 
LPHD03 
LPSC1E 
1.0 
LPHDU4 
LPSC22 
1.0 
LPH005 
LPSC2N 
1.0 
LPHU06 
LPSC2E 
1.0 
LPHD07 
LPSC3Z 
1.0 
LPHDOb 
LPSC3N 
1.0 
LPHD09 
LPSC3E 
1.0 
LPHD10 
LPSC4Z 
1.0 
LPHD11 
LPSC4N 
1.0 
LPHD12 
LPSC4E 
1.0 
LPHD13 
LPSC5Z 
1.0 
LPHD14 
LPSC6Z 
1.0 
LPHD15 
LPSC7Z 
1.0 
STPK  Designator/Channel  Identifier  Cross  Reference: 
STPR  DESIGNATOR    CHANNEL  ID     INPUT  SENSITIVITY 
SPRW01 
SZ1I01 
4. 88* 
SPRW02 
SZ1I02 
4.bb* 
SPRW03 
SZ1I03 
4.88* 
SPRW04 
SZ1I04 
4.8b* 
SPRWOb 
SZ1I05 
4.88* 
SPRWOb 
SZ1I06 
4.88* 
SPRW07 
SZ1I07 
4.88* 
SPRWOb 
SZ1I08 
4.88* 
SPRW09 
SZ1I09 
4.88* 
SPRW10 
SZ1I10 
4.88* 
SPRW11 
SZ1I11 
4.88* 
SPRW12 
SZ1U2 
4.88* 
SPRW13 
SZ1I13 
4.88* 
SPRW14 
SZ1I14 
4.8b* 
SPKW15 
SZ1I15 
4.bb* 
SPRW16 
SZ1I16 
4.bb* 
SPRW17 
SZ1I17 
4.88* 
SPRWlb 
SZlllb 
4.88* 
SPRW19 
SZ1I19 
4.88* 
SPRW20 
SZ1I75H 
4.88* 
SPRW21 
SN1I75H 
4.88* 
SPRW22 
SE1I75H 
4.88* 
SPRW23 
SZ1I75L 
0.0976 
SPRW24 
SN1I75L 
0.0976 
SPRW2b 
SE1I75L 
0.0976 
LPSC1Z (LPSC11) 
LZ1IA 
5.0+ 
LPSC1N (LPSC12) 
LN1IA 
5.0+ 
LPSC1E (LPSC13) 
LE1IA 
5.0+ 
LPSC2Z (LPSC21) 
U1IB 
5.0+ 
LPSC2N (LPSC22) 
LN1IB 
5.0+
CENR  55-2,  Vol   IX  1  April   1988  5 
NOTE:     To  normalize  the  frequency  response,  divide  the  return  voltage  at  each  frequency  by  the 
return  voltage  at  the  reference  frequency,  then  multiply  by  the  normalizing  factor.     The 
results  can  then  be  compared  with  the  values  listed  in  Volume  I,  to  determine  if  they  are 
within  tolerances. 
12.     STPR  CPU  Configuration  Parameters: 
a.     CPU  1: 
CONFIGURATION  IDENTIFICATION  =  Cxxxx-iHL 
0PERATE1  IDENTIFICATION  =  0PERATE1 
SITE  IDENTIFICATION  =  452 
LP  DATA  AND  INSTRUMENT  TYPE  (A,31,36)  =  31 
NUMBER  OF  SHORT  PERIOD  ARRAY  CHANNELS  =  19 
NUMBER  OF  SHORT  PERIOD  OTHER  CHANNELS  =  6 
NUMBER  OF  LONG  PERIOD  ARRAY  CHANNELS  =  21 
NUMBER  OF  LONG  PERIOD  OTHER  CHANNELS  =  0 
NUMBER  OF  SHORT  PERIOD  PROCESSES  =  16 
NUMBER  OF  LONG  PERIOD  PROCESSES  =  4 
SHORT  PERIOD  FREQUENCY  FILTER  LENGTH  =  99 
LUNG  PERIOD  FREQUENCY  FILTER  LENGTH  =  1 
AMOUNT  OF  SHORT  PERIOD  TIME  DELAY  REQUIRED  =  0 
AMOUNT  OF  LONG  PERIOD  TIME  DELAY  REQUIRED  =  U 
SP  COORDINATES: 
0,0,0 
1,0.853,3.836 
2,1.663,5.912 
3,2.943,4.021 
4,3.208,1.612 
5,0.706,1.705 
6,-1.104,3.188 
7,-1.030,5.041 
8,0.338,8.136 
9,3.355,7.635 
10,4.046,6.004 
11,6.106,4.299 
12,5.150,2.483 
13,4.046,0.408 
14,2.207,-0.945 
15,0.000,0.000 
16,-2.531,1.056 
17,-3.855,2.817 
18,-3.531,4.892 
19,-1.545,6.653 
LP  COORDINATES: 
0,0,0 
1,5.959,15.085,C 
2,12.506,34.911,C 
3,23.275,17.865,C 
A, 16. 256, 3. 039, C 
5,0.000,0.C00,C 
6, -14. 596,13. 992, C 
7, -7. 566, 30. 371, C 
SP  FREQUENCY  FILTER  PAaAMETERS: 
0.0001, -.0001,-. 0005, -.0011, -.0016, -.0020, -.0020, -.0017, 0.0014, -.0012 
-.0011,-. 0013, -.0014, -.0011, -.0004, 0.0007,0. 0019, 0.0027, 0.0030, 0.0029 
0.0027, 0.0028, 0.0034, 0.0043, 0.0050, 0.0049, 0.0038,0.001/, -.0007, -.0027 
-.0037, -.0041, -.0045, -.0065, -.0103, -.0162, -.0221, -.0266, -.0273, -.0254 
-.0224, -.0237, -.0315, -.0481, -.0653, -.0731, -.0456,0. 0324, 0.2035, 0.3910 
0.2035, 0.0324,-, 0456, -.0731, -.0653, -.0481, -.0315, -.0237, -.0224, -.0254 
.-.0273, -.0266,-0221, -.0162, -.0103, -.0065, -.0045, -.0041, -.0037, -.0027 
-.0007,0.0017,0.0038,0.0049,0.0050,0.0043,0.0034,0.0028,0.0027,0.0029 
0.0030, 0.0027,0.0019, 0.0007, -.0004, -.0011, -.0014,-. 0013, -.0011, -.0012 
*.0014,-. 0017, -.0020, -.0020, -.0016, -.0011, -.0005, -.0001, -.0001 
LP  FREQUENCY  FILTER  PARAMETERS: 
0.9999 
SP  BEAM  PARAMETERS: 
SPL360, 0,000,13,8 
SPL060, 0,060, 13, B 
SPL120, 0,120,13,8
CENR  55-2,  Vol  IX     1  April  1988 
-.0319, -.0319, -.0319, -.0319, -.0319, -.0319, -.0319, -.0319, -.0319, -.0319 
-.0319, -.0319, -.0319, -.0319, -.0319, -.0319, 0.0000, 0.0000, 0.0000, 0.0000 
0.0000,0.0000,0.0000,0.0000,0.0000,0.0000,0.0000,0.0000,0.0000,0.0000 
0.0000,0.0000,0.0000,0.0000,0.0000 
SP  PROCESSING  DELAY  =  80 
LP  PROCESSING  DELAY  =  15 
SECONDS  PER  RECORD  =  3 
b.  CPU  2: 
CONFIGURATION  IDENTIFICATION  =  Cxxxx-2HL 
0PERATE2  IDENTIFICATION  =  0PERATE2 
SITE  IDENTIFICATION  =  452 
LP  DATA  AND  INSTRUMENT  TYPE  (A, 31, 36)  =  31 
NUMBER  OF  SHORT  PERIOD  ARRAY  CHANNELS  =  19 
NUMBER  OF  SHORT  PERIOD  OTHER  CHANNELS  =  6 
NUMBER  OF  LONG  PERIOD  ARRAY  CHANNELS  =  21 
NUMBER  OF  LONG  PERIOD  OTHER  CHANNELS  =  0 
NUMBER  OF  SHORT  PERIOD  PROCESSES  =  16 
NUMBER  OF  LONG  PERIOD  PROCESSES  =  4 
NO  SP  CHAN  TO  BE  TRANSMITTED  YIA  HSM  =  18 
NO  LP  CHAN  TO  BE  TRANSMITTED  VIA  HSM  =  15 
♦NUMBER  OF  CONTACT  SENSOR  MONITORS  =  6 
♦NUMBER  OF  A/D  CHANNEL  MONITORS  =  3 
AMOUNT  OF  SP  EDIT  TIME  DELAY  REQUIRED  =  0 
AMOUNT  OF  LP  EDIT  TIME  DELAY  REQUIRED  =  0 
SP  COORDINATES: 
0,0,0 
1,0.853,3.836 
2,1.663,5.912 
3,2.943,4.021 
4,3.208,1.612 
5,0.706,1.705 
6,-1.104,3.188 
7,-1.030,5.041 
8,0.338,8.136 
9,3.355,7.635 
10,4.046,6.004 
11,6.106,4.299 
12,5.150,2.483 
13,4.046,0.408 
14,2.207,-0.945 
15,0.000,0.000 
16,-2.531,1.056 
17,-3.855,2.817 
18,-3.531,4.892 
19,-1.545,6.653 
LP  COORDINATES: 
0,0,0 
1,5.959, 15. 085, C 
2, 12. 506, 34. 911,  C 
3,23.275, 17. 865, C 
4, 16. 256, 3. 039, C 
5, 0.000,0. 000, C 
6, -14. 596, 13. 992, C 
7,-7.666,30.371,C 
SP  CALIBRATION  DEFAULT  PARAMETERS: 
0.835,1.0,10,1,010000,0.9,1.1,2.93,8 
1.00,1.708 
0.5,1.708 
0.8,1.708 
1.5,1.708 
2.0,1.708 
2.5,1.708 
3.0,1.708 
4.0,1.708 
LP  CALIBRATION  DEFAULT  PARAMETERS: 
0.388,0.04,10,1,013000,0.9,1.1,1.289,7,3 
0.040, .194 
0.100,1.94 
0.067,. 194 
0.050, .194
CENR  55-2,  Vol    IX  1  April    1988 
*    Monitors  may  be  aaaea/modifiea  at  the  station's  discretion. 
OFFICIAL  "  JAY  J.  JAYNES,  Colonel,   USAF 
Commander 
RICHARD  E.  COOK,  SMSgt,  USAF 
Director  of  Administration 
SUMMARY  OF  CHANGES 
Rewrote  in  active  voice.  Deleted  references  to  specific  paragraphs  in  Vol  I.  Added  purpose 
statement.  Added  distribution  statement.  Added  CPU  1  &  CPU  2  configurations.  Changed 
develocoraer  displays.  Deleted  adaptive  processes.  Added  fixed  filter  processes.  Updated 
format.