DEPARTMENT UF THE AIR FURCE CENTER REGULATION 55-2 Headquarters Air Force Technical Applications Center Volume IX Patrick AFB FL, 32925-6001 1 April 1988 SPECIFIC STATION REQUIREMENTS FOR DETACHMENT 452 This regulation establishes the procedures for station unique operations and analysis. It applies to all active duty Air Force members assigned to the station. Personnel who violate the specific prohibitions and requirements of this regulation may be prosecuted under the Uniform Code of Military Justice (UCMJ). Distribution lieitea to Dod ana DoD contractors only; to protect information and technical data which advance the state-of-the-art or describe new technology in an area of significant or potentially significant military application, 1 April 1988. Other requests shall be referred to HQ/DOSB. 1. Station Designator. The station designator for Detachment 452 is KSRS (Korean Seismological Research Station). Use KSR for the three element entry preceding the station name on data messages. 2. Timing Standards. JJY or BPM. 3. Routine Calibrations. Perform SPS ana LPS calibrations sequentially, commencing immediately after Ul0CZ and 0130Z respectively. The LPS frequency responses may be performed at the rate of one site per day. 4. Data Reports. Submit special data reports in accordance with Volume I. Submit routine data reports within 6 hours of the end of the reporting period. 5. EDIT tape registration numbers are 5100 through 5199. 6. Training Outage. Training outage (IAW Vol I) is grantea for Thursday of each week from 2300Z through 02602 Friday. 7. SPS Develocorder Presentation: a. Primary Develocoraer: TRACE DATA MAG ASN CHAN DISP ID SCALE DEV SENSE VULTS 1 S$Z1BP36013 000K SPDSOL SPL360 1.0 0.61 2 SZ1BP06U13 1000K sppso2 SPLU60 1.0 0.61 3 SZ1BP12013  1000K SPDSU3 SPL120 1.0 0.61 4 SZ1BP18U13 1000K SPDS04 SPL180 1.0 0.61 5 $Z1BP24013 = =1000K sPoso5 SPL240 1.0 0.61 6 $Z18P3U013  i000K SPDSU6 SPL300 1.0 0.61 7 SZigPG00sy 1000K SPDS%7 $PZ000 1.0 0.61 & $Z1BP03013 1000K SPDSO8 SPLO30 1.0 0.61 9 $21BP28713 1000K SPDS11 SPL287 1.0 0.61 10 $Z1B8P30816 100UK SPUS12 $PQ308 1.0 0.61 11 $Z1%P33815 1000K SPDS13 SPP338 1.0 0.61 12. $Z1175H 250K SPRW20 2.44 13. SN1I75H 250K SPRW21 2.44 14 SE1175H 250K SPRW22 2.44 b. Secondary Develocoraer: TRACE DATA MAG ASN CHAN DISP ID SCALE DEY SENSE VOLTS 1 $Z1108 250K SPRWOS 2.440 2 $Z1114 250K SPRW14 2.440 .3 S$Z1111 250K SPRW11 2.440 Supersedes CENR 55-2, Vol IX, 1 July 1982. No of Printed Pages: 9 OPR: OUSB (MSgt M. P. Clark) Approved by: Co) T. H. Niquette Editor: SSgt D2. M. Pless Distribution: X
Cenk 55~z, Vol IX 1 April 19838 9. Channels transmitted to the TOS: CHANNEL DISP Ib SCALE SPHUU1 SPRWOL 1.0 SPHUUZ SPRRU3 1.0 SPHUG3 SPRWOS 1.0 SPHUG4 SPRWO7 1.0 SPHUOS SPKWOS 1.0 SPHUU6 SPRKWUS 1.0 SPHDG7 SPRW1O 1.0 SPHOOS SPRW12 1.0 SPHDO9 SPRK12 1.0 SPHD20 SPRW13 1.0 SPHD11 SPRW14 1.0 SPHD12 SPRW16 1.0 SPHD13 SPRW17 1.6 SPHD14 SPRUI8 1.0 SPHD15 SPRW19 1.6 SPHU16 SPRW15 1.0 SPHD17 SPRW21 1.0 SPHD18 SPRW22 1.0 LPHDOL LPSC1Z 1.0 LPHUOZ LPSC1N 1.0 LPHDO3 LPSCIE 1.0 LPHDU4 LPSC2Z 1.0 LPHDOS LPSC2N 1.0 LPHDO6 LPSC2E 1.0 LPHDO7 LPSC3Z 1.0 LPHDOS LPSC3N 1.0 LPHOUY LPSC3E 1.0 LPHUD1O LPSC4Z 1.0 LPHD11 LPSC4N 1.0 LPHD12 LPSC4E 1.0 LPHD13 LPSCSZ 1.0 LPHD14 LPSC6Z 1.0 LPHD15 LPSC7Z 1.0 10. STPRK Designator/Channel Identifier Cross Reference: “STPR DESIGNATOR CHANNEL ID INPUT SENSITIVITY SPRWOL SZ1101 4.88* SPRHOZ SZ1ICZ 4.88* SPRWU3 $Z1103 4.88* SPRWO4 $Z1104 4.88* SPRWUS $Z1105 4.8&* SPRWU6 $Z1106 4.88* SPKHU7 SZ1107 4.88* SPRWUS SZ1I0& 4.8&* SPRWUS SZ1109 4.88* SPRW1O $Z1110 4.4u&* SPKW11 SZ1111 4.88* SPRW1Z SZ1112 4.88* SPRW13 $Z1113 4.88* SPKW14 $Z1114 4.88* SPKW15 SZ1116 4.88* SPRW16 $Z1116 4.8&5* SPRW17 $Z1117 4.88* SPRW1S $Z1118 4.88* SPRKW19 SZiI19 4.88* SPRW2Z0 $Z1175H 4.&8* SPRH21 SN1LI75H 4.88* SPRKZZ SEL175H 4.88* SPRW23 SZ1175L 0.0976* SPRW24 SWLI75L U.U976* SPRRZS SELIV75L 0.0976* LPSC1Z (LPSC11) LZI1IA 5.U+ LPSCIN (LPSC12Z) LNLIA §.0+ LPSC1E (LPSC13) LELIA 5.0+ LPSCZZ (LPSC21) LZ1Ib 5 LPSC2n (LPSC22) LNIIB 5.
CENR 55-2, Vol IX NUTE: return voltage at the reference frequency, then multiply by the normalizing factor. The results can then be compared with the values listed in Volume I, to determine if they are within tolerances. 12. STPR CPU Configuration Parameters: a. CPU 1: CUNFIGURATION IDENTIFICATION = Cxxxx-LHL OPERATE] IDENTIFICATION = OPERATEL SITE IDENTIFICATION = 452 LP DATA AND INSTRUMENT TYPE (A,31,36) NUMBER UF SHORT PERIOD ARRAY CHANNELS NUMBER OF SHORT PERIOD OTHER CHANNELS NUMBER OF LONG PERIOD ARRAY CHANNELS NUMBER UF LONG PEKIOD OTHER CHANNELS NUMBER OF SHOKT PERIOD PROCESSES = 16 NUMBER OF LONG PERIUD PROCESSES = 4 uuu 31 19 6 21 Q SHORT PERIOD FREQUENCY FILTER LENGTH = 99 LUNG PERIOD FREQUENCY FILTER LENGTH = 1 AMUUNT UF SHORT PERIOD TIME DELAY REQUIRED = 0 AMOUNT OF LONG PERIUD TIME DELAY REQUIRED = U SP CUURDINATES: 0,0,0 1,0.853,3.836 RRM OODONAHAM AWM wv © Nr Ov »9-150,2.483 13,4.046,0.408 14,2.207,-0.945 15,0.000,0.000 16,-2.531,1.056 17,-3.855,2.817 18,-3.531,4.892 19,-1.545,6.653 LP COORDINATES: 0,0,0 1,5.959,15.085,C 2,12.506,34.911,C 3,23.275,17.865,C 4,16.256,3.039,C 5,0.000,0.C00,C 6,-14.596,13.992,C 7,-7.566,30.371,C SP FREQUENCY FILTER PAnAMETERS: 50 0.0091,-.0001,-.0005,-.0011,-.0016,-.0020,-.0020,~.0017,0.0014,-.0012 ~.0011,-.0013,-.0014,-.0011,-.0004,0.0007,0.0019,0.0027,0.9030,0.0029 0.0027 ,0.0028,0.0034,0.0043,0.0050,0.0049,0.0038,0.0017 ,-.0007,-.0027 -.0037,-.0041,-.0045,-.0065,~-.0103,-.0162,-.0221,-.0266,~-.0273,-.0254 ~.0224, -.0237,~.0315,-.0481, -.0653,-.0731,-.0456,0.0324,0.2035,0.3910 0.2035 ,0.0324,-,0456,-.0731,-.0653,-.0481,-.0315,-.0237,-.0224,-.0254 +=.0273, -.0266,-.0221,-.0162,-.0103,-.0065,-.0045,-.0041,~.0037,-.0027 -.0007 ,0.0017,0.0038,0.0049,0.0050,0.0043,0.0034,0.0028,0.0G27,0.0029 6.0030, 0.0027 ,0.0019,0.0007,-.0004,-.0011,-.0014,~-.0013,-.0011,-.0012 *,0014,-.0017,-.0020,-.0020,-.0016,-.0011,-.0005,-.0001,-.0001 LP FREQUENCY FILTER PARAMETERS: U 0.9999 SP BEAM PARAMETERS: SPL360,0,000,13,8 SPLO60,0,060,13,8 SPL120,0,120,13,8 5 To normalize the frequency response, divide the return voltage at each frequency by the
CENR 55-2, Vol IX 1 April 1988 ~.0319,-.0319,-.0319, -.0319,-.0319,-.0319, -.0319, -.0319,-.0319,-.0319 -.0319,-.0319,-.0319,-.0319,-.0319,-.0319,0.0000,0.0000,0.0000,0.0000 0.0000,0.0000,0.0000,0.0000,0.0000,0.0000,0.0000,0.0000,0.0000,0.0000 0.0000 0.0000 ,06.0000,0.0000,0.0000 SP PROCESSING DELAY = 80 LP PROCESSING DELAY = 15 SECONDS PER RECORD = 3 b. CPU 2: CONFIGURATION IDENTIFICATION = Cxxxx-2HL OPERATE2 IDENTIFICATION = OPERATE2 SITE IDENTIFICATION = 452 LP DATA AND INSTRUMENT TYPE (A,31,36) = 31 NUMBER OF SHORT PERIOD ARRAY CHANNELS = 19 NUMBER OF SHORT PERIOD OTHER CHANNELS = 6 NUMBER OF LONG PERIOD ARRAY CHANNELS = 21 NUMBER OF LONG PERIOD OTHER CHANNELS = 0 NUMBER OF SHORT PERIOD PROCESSES = 16 NUMBER OF LONG PERIOD PROCESSES = NO SP CHAN TO BE TRANSMITTED VIA HSM = 18 NO LP CHAN TO BE TRANSMITTED VIA HSM = 15 *NUMBERK UF CUNTACT SENSOR MONITORS = *NUMBEK OF A/D CHANNEL MONITORS = AMOUNT OF SP EDIT TIME DELAY REQUIRED = 0 AMOUNT OF LP EDIT TIME DELAY REQUIRED = 9 SP COORDINATES: -150,2.483 13,4.046,0.408 14,2.207,-0.945 15,0.000,0.000 16,-2.531,1.056 17,-3.855,2.817 18,-3.531,4.892 19,-1.545,6.653 LP COORDINATES: 0,0,0 5.959,15.085,C 12.506,34.911,C 23.275,17.865,C 16.256,3.039,C ,0.000,0.000,C 37 14.596,13.992,C -7.666,30.371,C CALIBRATION DEFAULT PARAMETERS: 35,1.0,10,1,010000,0.9,1.1,2.93,8 »1.708 708 708 708 708 708 708 708 CALIBRATION DEFAULT PARAMETERS: 88 ,0.04,10,1,013000,0.9,1.1,1.289,7,3 040, .194 -160,1.94 -067,.194 -050,.194 8 00 5, 8, 1. 1 l 1 1 1 1 1 2 3, 4, 5 6 7, SP 0. 1. 0. 0. 1.5 2.0 2.5 3.0 4.0 L 6.3 0. 0 0 0
CENR 55-2, Vol IX 1 April 1988 * Monitors may be acaea/moaifiea at the station's discretion. OFFICIAL JAY J. JAYNES, Colonel, USAF Commander RICHARD E. COOK, SMSgt, USAF Director of Administration SUMMARY OF CHANGES Rewrote in active voice. Deleted references to specific paragraphs in Vol I. Added purpose Statement. Added distribution statement. Aaged CPU 1 & CPU 2 configurations. Changed develocoraer displays. Deleted adaptive processes. Added fixed filter processes. Updated format.