DEPARTMENT OF THE AIR FORCE CENTER REGULATION 55-2 Headquarters Air Force Technical Applications Center Volume iv Patrick AFB FL 32925-6001 13 July 1987 SPECIFIC STATION REQUIREMENTS FOR BELBASI SEISMIC RESEARCH STATION This regulation establishes the procedures for station unique operations and analysis. It applies to all active duty Air Force members assigned to the station. Personnel who viclate the specific prohibitions and requirements of this regulation may be prosecuted under the Uniform Code of Military Justice (UCMJ). Distribution limited to DoD and DoD contractors only; to protect information and technical data which advance the state-of-the-art or describe new technology in an area of significant or potentially significant military application, 1 July 1987. Other requests shall be referred to HQ/DOSB. 1. Station Designator. The station designator for Belbasi Seismic Research Station is BEAR. 2. Timing Standard. WNV, WHVH, and RTA. 3. Routine Calibrations. Perform SPS and LPS calibrations sequentialiy, commencing immedi - ately after 07302. 4. Edit tape registration numbers are 5900 - 5999. 5. Training Cutage. Outage authorized in CENR 55-2, Yol I is granted for Tuesday cf each week from 1200Z thru 15002. 6. Special Data Reports. Submit speciai data reports in accordance with CENR $5-2, Yol I. ia addition, submit special data reports for all teleseismic signals received with an azimuth between 175 and 205 degrees. Overlays are not available. 7. Summation Channel. Individual vertical array channel(s) may be manually lined out af ane- log summation(s) (but not from processed data) when cultural or wind noise increases trace background on the individual channel(s) to more than twice the background everage of other érray channels. Monitor individual channels linea out hecause of high background to determine when the background has subsided enough to return the channels to summation(s). 8. SPS Develocorder Presentations: a. Primary Develocorder: TRACE DATA MAG 1 SZ18P 36008 2COCK 2 $Z1B?06008 20CCK 3 SZ1BP12008 290CK 4 $Z1BP 18008 2C0CK 5 $Z18P24008 2cccex 6 $218? 36008 eSCCK 7 $Z18P00099 200K 8 £Z18P06513 200CK 9 $Z1BP03320 2000K 10 SZ1ST 1000K if SZLIGIH 250K Supersedes CENR 55-2, Vol I¥, 1 May 1985. No. of Printed Pages: 7 OPR: DOSB (TSgt C. W. Stephens) Approved By: Lt Col D. NW. Adams) Editor: SSgt D. M. Pless Bistribution: X
TRACE 12 13 DATA SNLI61H SE1161iH b. Secondary Deveiccorder. TRACE 1 ve} 10 ll 12 13 NOTE: Do not include SZ1101 DATA $Z1103 SZ1I16 SZ1115 SZ1110 $Z18P09099 $Z18P01013 SPARE SZ1ST SZ116iL SZ1161# SZil61" SNLi61M SELI61M in SZ\ST. 8. L°S Develocorder Presentation. TRACE 1 2 7 * MAG parameters are listed in CENR 55-2, Vol 1. 9. STPR Designator/Channel Identifier Cross Reference. STPR DESIGNATOR SPRWOL SPRWO2 SPRWO3 SPRWO4 SPRAGS DATA LZ1161M LN1I61M LELI61M LZ1161H LNITI61H ny LIOLH LZLi61L CHANHEL 10 SZLI01 SZ1102 S21103 $Z1104 SZ1105 INPUT SENSITI N/A 4,88* 4.88* Yor IY 13 July 1987 MAG 250K 250K 10K 10K 10K lK VITY
CENR 55-2, Yo) STPR DESIGNATOR SPRWO6 SPRWO7 SPRWO8 SPRWO9 SPRYW10 SPRW11 SPRW12 SPRW13 SPRW14 SPRW15 SPRW16 SPRH17 SPRW18 SPRW19 SPRW20 SPRW21 SPRW22 SPRH23 LPSC1Z LPSCIN LPSCIE LPSC2Z LPSC2N LPSC2E SPB369 SP8060 SPB120 SPB180 SPB240 SPB3C0 SPZ000 SPL310 SPU033 SPLO65 IV 13 July 1987 CHANNEL [D SZ1106 SZ1107 $Z1i08 $Z1109 SZ1110 SZ1111 SZ1112 SZ1113 SZ1E14 SZ1115 SZ1LI16 SZ1161H SNII6]H SE1I61H SZII61L SNITO1L SElIG61L SZ1ST LZ1I61H LNLI61H LELI61H LZ1161L UNLIGIL LELI61L $21BF 36008 $Z18P06008 SZ1BP 12008 $Z18P 18008 $218? 24008 SZ1BP 30008 $Z18P00099 $2Z13201013 $Z13293320 $Z13?06513 INPUT SENSITIVITY 4.88* 4.88* 4.88* 4.88* 4.88* 4.88* 0.0976* 0.9976* 0.0976* N/A 10+ 10+ 10+ 1.0+ 1.0+ 1.0+ N/A MSA N/A N/A N/A N/A N/A N/A N/A N/A re)
4 CENR 55-2, Voi IV 13 July 1987 * Volts peak-to-peak for a 100 miliimicron equivalent OF as measured at the output of the SCC or KS36000 filter. + Volts peak-to-peak for a 10 micron equivalent DF as measured at the output of the KS36000 filter. NOTE: Do not include SZ110i and SZ1106 in beam processing. [Initialize STPR operational soft- ware to delete SZIIGi and SZ1106 (SPRWO1 and SPRWO6) from processing using the CHNUSE function. 10. STPR Frequency Response Yoltages and Normalizing Factors: a. Short Perisd: STPR VOLTAGE NORMAL IZING FREQUENCY UAS KS36006 FACTOR *1.0 +. 708 0.854 1 0.5 1.708 9.854 1 0.8 1.7C8 0.854 1 1.5 1.708 0.854 1 2.0 708 0.254 1 2.5 1.758 0.354 1 3.0 1.708 9.856 1 4.0 1.703 0.654 1 dD. Lon: Period: FREQUENCY STPR YOUTAGE NORMALIZING FACTOR *0.040 0.666 1 0.100 4.562 -1429 0.067 0.666 1 0.050 0.666 1 0.033 9.866 1 0.025 0.666 1 0.020 9.666 1 * Reference Frequency NOTE: To normalize the frequency response, multioly tne AGAIN by the nor- malizing factor for each frequency. ine resu'ts can then be compared with the values listed in CENR 55-2, Vol I to determine if the channel is within tolerances. ll. STPR CPU Configuration Parameters: a. CPUL: CONFIGURATION IDENTIFICATION = Cxxxx-ILS OPERATE] IDENTIFICATION = OPERATEL SITE IDENTIFICATION = 301 LP DATA AND INSTRUMENT TYPE (A,31,36) NUMBER OF SHORT PERIOD ARRAY CHANNELS NUMBER OF SHORT PERTUD OTHER CHANNELS NUMBER OF LONG PERIOD ARRAY CHANNELS = 6 NUMBER OF LONG PERIOD OTHER CHANNELS = 2 NUMBER OF SHORT PERICD PROCESSES = 16 Bo wom ou ag poe or
CENR 58-2, Vol [IV 13 July 1987 NUMBER OF LONG PERIOG PROCESSES = 1 SHORT PERIOD FREQUENCY FILTER LENGTH = 99 LONG PERIOD FREQUENCY FILTER LENGTH = 1 AMOUNT OF SHORT PERJUD TIME DELAY REQUIRED = 0 AMOUNT OF LONG PERIOD TIME DELAY REQUIRED = 0 SP COORDINATES: 9,0.806,-i.452 10,-0.261,-1.421 11,-1.162,-1.544 12,-1.257,-0.68 13,-1.874,-0.958 14,-2.253, -0.587 15,-1.826,0.216 16,-0.878,1.544 LP COORDINATES: a 0,0 1,0,0,C 2,9,0,C SP FREQUENCY FILTER PARAMETERS: 0.0001 ,-.0001,-.0005,-.0911,-.0015,-.0020,-.0020,-.0017,0.0014,-.0012 -.0011,-.0013,-.0014,-.0011,-.G004 ,0.0C07 ,0.0019,0.0027 ,0.0030,0.0029 0.0027 ,0.0028 ,0.0034,0.0043,0.0050,0.0049,0.0033,0.0017,-.0007,-.9027 ~ .0037,-.0041,-.0045,-.0065,-.0103,-.0162,-.0221,-.0266,-.0273,-.0254 - 0224, -.0237,-.0315,-.0481,-.0653,-.0731, -.0456,0.0324,0.2035,0.2910 0.2035,0.0324, -,0456,-.0731, -.0653, -.0481,-.0315,-.0237,-.0224,-.9254 -.0273,-.0266,-.0221,-.0162,-.0103,-.0065,-.0045, 0041, ~.0037,-.0027 ~|0007°0.0017 ,0.0038,0.0049 ,0.0050,0.0043,0.0024,0.9028,0.0027,0.0029 0.0030,0.0027 ,¢.0019,0.0007 -.0004,-.0011,-.0014,-.0013,-.0011,-.0012 ~.0014,-.0017,-.0020,-.0020,-.0016,-.0011,-.0005,~.0001,-.0001 LP FREQUENCY FILTER PARAMETERS: re) 0.9999 SP SEAM PARAMETER $PB360,0,000,8. SPB060,0,060, 5P3120,0,12 $PB180,0,18 SPB240,0,24 $PB300,0, 30 0,0 0 wwwwowinn ~~ ee ee BOoooooOn sPz000, , SPLO65.0, 6 0,8 $PU033,0,033, 20. 0,8 $PL010,0,010,13.0,8 LP BEAM PARAMETERS: LPB36Z,1,000,3.5,8 SP PROCESSING DELAY = 60 LP PROCESSING DELAY = i SECONDS PER RECORD = 4 &. 0,8. 0,8. 0,8. 00,8. 0,0 5,1 we b. cPU2: CONFIGURATION IDENTIFICATION = Cxxxx-2LS OPERATE2 IDENTIFICATION = OPERATE2 SITE IDENTIFICATION = 301 LP DATA AND INSTRUMENT TYPE (A,31,36) = A NUMBER OF SHORT PERIOD ARRAY CHANNELS = 16 NUMBER OF SHORT PERIOD OTHER CHANNELS = 7 NUMBER OF LONG PERIOD ARRAY CHANNELS = 5 NUMBER OF LONG PERIOD OTHER CHANNELS = 9 NUNBER OF SHORT PERE) 220CESSES = 19 NUMBER OF LONG PERIOD PROCESSES = 1 NO SP CHANNELS TO 8€ TRANSMITTE) “EA 45.1 = 0
CENR 55-2, Yol i¥ 13 July 1987 NG LP CHANNELS TO 8E& TRANSMITTED VIA HSM = 0 NUMBER OF CONTACT SENSOR MONITORS = 2 NUMBER OF A/D CHANNEL CHANNEL MONITORS = 1 AMOUNT OF SP EDIT TIME DELAY REQUIRED = 0 AMOUNT OF iP EDIT TIME DELAY REQUIRED = O SP COORDINATES: 0,0,0 1,0.0,0.0 2,0.403,0.710 3,0. 735.0. 247 4,0.38,-0.494 5,-0. 546,-0.58 6 ,-0.878,0.093 7,-0.688,0.71 8,1.731,0.803 9,0.806,-1.452 10,-0.261,-1.421 11,-1.162,-1.544 12,-1.257,-0.68 13,-1.874, -0.958 14,-2.253, -0.587 15,-1.826,0.216 16,-0.878,1.544 LP COORDINATES: BRATION DEFAULT PARAMETERS: -000,25,1,0730C0,0.9,1.1,2.928,8 W327 OOO . meen OWwWNs « WON DOW Dr rere rs es pe bee ee ee a ay -708 LIBRATION DEFAULT PARAMETERS: ,0. 04, 10,1,080000,0.9,1-1,3.751,7,3 P CHANNEL CONFIGURATION FOR CALIBRATION SYSTEM: a en er ee OOnNIAN AWN — ee When oO ~ 2 rere pee re rere rere re rere re NOOO DODOMT RUNNFOOMONNY O . 1,14 1,15 1,16 1,19 1,19 1,19 1,19 1,29 1,19 1,24
CENK 55-2, Vol IV 3 juiy 1987 P CHANNEL CONFIGURATION FOR CALIBRATION SYSTEM: Cn ee oe oo oo oe oe sl rl rl i yi ai SP BEAM PARAMETERS: SPB360,0,000,8.0,B SPB060,0,060,8.0 $PB120,0, 120.8. 0 5PB180,0,180,8. 0 SPB240,0,240,8.0 $PB300,0,300,8.0 $P2000,0,0,0,0,8 SPL065,0,065,13.0, SPU033,0,033,20.0, SPLOi0,0,010,1 : LP BEAM PARAMETE S: LPB36Z,1,000,3.5,8 RELAY INDENTIFIERS AND NORMAL STATUS FOR EACH CONTACT SENSOR MONITOR: LOWBAT ,1 ACFAIC,1 IDENTIFIERS AND LIMITS FOR EACH A/D CHANNEL MONITOR: LNPOWR,-9.5,9.5 SECONDS PER RECORD = 1 3.0,8 0.0,8 3.0,8 R OFFICIAL BILLY J. BINGHAM, Colonel, USAF Commander WALLACE L. HUFFAKER, CHSgt, USAF Director of Administration SUMMARY OF CHANGES Rewrote in active voice. Incorporated IMC 85-1. Added purpose statement. Added limited distribution paragraph. Added summation channel paragraph. Deleted references to specific paragraphs to Vol I. Added configuration parameters. SP array coordinates updated.