DEPARTMENT  OF  THE  AIR  FORCE 
Headquarters  Air  Force  Technical  Applications  Center 
Patrick  AFB  FL  32925-6001 
CENTER  REGULATION  55-2 
Volume  IV 
13  July  1987 
SPECIFIC  STATION  REQUIREMENTS  FOR  BELBASI  SEISMIC  RESEARCH  STATION 
This  regulation  establishes  the  procedures  for  station  unique  operations  and  analysis.   It 
applies  to  all  active  duty  Air  Force  members  assigned  to  the  station.  Personnel  who  violate 
the  specific  prohibitions  and  requirements  of  this  regulation  may  be  prosecuted  under  the 
Uniform  Code  of  Military  Justice  (UCMJ). 
Distribution  limited  to  DoD  and  DoO  contractors  only;  to  protect  information  and  technical  data 
which  advance  the  state-of-the-art  or  describe  new  technology  in  an  area  of  significant  or 
potentially  significant  military  application,  1  July  1987.  Other  requests  shall  be  referred  to 
HQ/DOSB. 
1.  Station  Designator.  The  station  designator  for  Belbasi  Seismic  Research  Station  is  BEAR. 
2.  Timing  Standard.  WWV,  WWVH,  and  RTA. 
3.  Routine  Calibrations.  Perform  SPS  and  LPS  calibrations  sequentially,  commencing  immedi- 
ately after  0730Z. 
4.  Edit  tape  registration  numbers  are   5900  -  5999. 
5.  Training  Outage.  Outage  authorized  in  CENR  55-2,  Vol  I  is  granted  for  Tuesday  cf  each  week 
from  1200Z  thru  15002. 
6.  Special  Data  Reports.  Submit  special  data  reports  in  accordance  with  CENR  55-2,  Vol  I.   In 
addition,  submit  special  data  reports  for  all  teleseismic  signals  received  with  an  azimuth 
between  175  and  205  degrees.  Overlays  are   not  available. 
7.  Summation  Channel.  Individual  vertical  array   channel (s)  may  be  manually  lined  out  of  ana- 
log summation(s)  (but  not  from  processed  data)  when  cultural  or  wind  noise  increases  trace 
background  on  the  individual  channel (s)  to  more  than  twice  the  background  average  of  other 
array  channels.  Monitor  individual  channels  linea  out  h<=>rause  of  high  background  to  determine 
when  the  background  has  subsided  enough  to  return  the  channels  to  summation! s) . 
8.  SPS  Develocorder  Presentations: 
a.  Primary  Develocorder: 
TRACE 
DATA 
SZ1BP36008 
SZ1BP06008 
SZ1BP12003 
SZ1BP18008 
SZ13P24008 
SZ13P30003 
SZ13P00099 
SZ1BP06513 
SZ1BP03320 
SZ1ST 
SZ1I61H 
MAG 
2C0CK 
20CCK 
200GK 
2C0CK 
2CCCX 
2C0CK 
2Q0CK 
20C0K 
1000K 
2  5  OK 
Supersedes  CENR  55-2,  Vol  IV,  1  May  1985. 
No.  of  Printed  Pages:  7 
OPR:   DOSB  (TSgt  C.  W.  Stephens) 
Approved  By:   Lt  Col  D.  W.  Adams) 
Editor:  SSgt  D.  H.  Pless 
Distribution:   X
TRACE 
DATA 
SN1I61H 
SE1I61H 
Secondary  Qevelccorder. 
TRACE 
DATA 
SZ1I03 
SZ1I16 
SZ1115 
SZ1I10 
SZ1BP00099 
SZ1BP01013 
SPARE 
SZ1ST 
g 
SZ1I61L 
SZ1I61H 
SZ1I61M 
SN1I61M 
SE1I61M 
NOTE:   Do  not  include  SZ1I01  in  SZ1ST. 
3.  LPS  Develocorder  Presentation. 
TRACE  DATA 
1  LZ1I61M 
2  LN1I61M 
3  LE1I61M 
4  LZ1I51H 
5  LN1I61H 
6  LE1I61H 
7  LZ1I61L 
*  MAG  parameters  are   listed  in  CEMR  55-2,  Vol  I. 
9.  STPR  Designator/Channel  Identifier  Cross  Reference. 
STPR  DESIGNATOR  CHANNEL  ID 
SPRW01  SZH01 
SPRW02  SZ1I02 
SPRW03  SZ1I03 
SPRU04  SZ1I04 
SPRV/05  SZ1I05 
CENR  55-2,  Vol  IV 
MAG 
250K 
250K 
MAG 
500K 
500K 
500K 
300K 
2000K 
2000K 
I000K 
5K 
250K 
50K 
50K 
50K 
13  July  1987 
MAG 
10K 
10K 
10K 
IK 
INPUT  SENSITIVITY 
N/A 
4.88* 
4.38* 
4.38* 
4.38*
CENR  55-2,  Vol  IV   13  July  1937 
STPR  DESIGNATOR 
SPRW06 
SPRW07 
SPRW08 
SPRU09 
SPRUIO 
SPRW11 
SPRW12 
SPRW13 
SPRW14 
SPRW1S 
SPRW16 
SPRW17 
SPRW18 
SPRW19 
SPRW20 
SPRW21 
SPRW22 
SPRW23 
LPSC1Z 
LPSC1N 
LPSC1E 
LPSC2Z 
LPSC2N 
LPSC2E 
SPB360 
SPB060 
SPB120 
SPB180 
SPB240 
SPB3CO 
SPZOOO 
SPLOIO 
SPU033 
SPL065 
CHANNEL  ID 
SZ1I06 
SZ1IQ7 
SZ1I08 
SZ1I09 
SZ1I10 
SZ1I11 
SZ1I12 
SZ1I13 
SZ1I14 
SZ1I15 
SZ1I16 
SZ1I61H 
SN1I61H 
SE1I61H 
SZ1I61L 
SN1I61L 
SE1I61L 
SZ1ST 
LZ1I61H 
IN1I61H 
LE1I61H 
LZ1I61L 
LN1I611 
LE1I61L 
SZ1BP36003 
SZ1BP06008 
SZ1BP12008 
SZ1BP18008 
SZ1BP24008 
SZ1BP30003 
SZ1BP00099 
SZ13P01G13 
SZl'iP03320 
SZ1UP06513 
INPUT  SENSITIVITY 
4.88* 
4.88* 
4.38* 
4.88* 
4.88* 
4.88* 
4.88* 
4.38* 
4.88* 
4.88* 
4.68* 
4.88* 
4.88' 
4.88* 
0.0976* 
0.0976* 
0.0976* 
N/A 
iO  + 
10  + 
10+ 
1.0+ 
1.0+ 
1.0+ 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A
CENR   55-2,    Vol    IV        13  July    1987 
*  Volts  peak-to-peak   for  a   100  millimicron  equivalent  OF  as  measured  at   the  output  of   the  5CC 
or  KS3600O   filter. 
+  Volts  peak-to-peak   for  a   10  micron  equivalent  DF  as  measured  at   the  output  of   the  KS36000 
filter. 
NOTE:     Do  not   include  SZ1I01  and  SZ1I06   in  beam  processing.      Initialize  STPR  operational    soft- 
ware  to  delete  SZ1I01   and  SZ1I06   (SPRW01  and  SPRW06)    from  processing  using   the  CHNUSE   function. 
10.      STPR  Frequency  Response  Voltages  and  Normalizing  Factors: 
a.     Short  Period: 
FREQUENCY 
*1.0 
0.5 
0.8 
1.5 
2.0 
2.5 
3.0 
4.0 
STPR  VOLTAGE 
UAS      KS36000 
: .  708 
0.854 
1.708 
0.854 
1.7C8 
0.854 
1.703 
0.854 
1 .  708 
0.854 
1.708 
0.354 
1.708 
0  .  3  s  4 
1.703 
0.354 
NORMALIZING 
FACTOR 
b.  Lon;:  Period: 
FREQUENCY 
*0.040 
0.100 
0.067 
0.050 
0.033 
0.025 
0.020 
*  Reference  Frequency 
;t?r  voltage 
0.666 
4.562 
0.666 
0.566 
0.666 
0.666 
0.666 
NORMALIZING  FACTOR 
.1429 
NOTE:  To  normalize  the  frequency  resoonse,  multiply  tne  RGAIN  by  the  nor- 
malizing factor  for  each  frequency.  The  results  can  thon  be  compared  with  the 
values  listed  in  CENR  55-2,  Vol  I  to  determine  if  the  channel  is  within 
tolerances. 
11.  STPR  CPU  Configuration  Parameters: 
a.  CPU1: 
CONFIGURATION  IDENTIFICATION  =  Cxxxx-lLS 
OPERATEl  IDENTIFICATION  =  0PERATE1 
SITE  IDENTIFICATION  =  301 
LP  DATA  AND  INSTRUMENT  TYPE  (A, 31, 36)  =  A 
1  r 
NUMBER  OF  SHORT  PERIOD  ARRAY  CHANNELS 
NUMBER  OF  SHORT  PERIOD  OTHER  CHANNELS  --    ', 
NUMBER  OF  LONG  PERIOD  ARRAY  CHANNELS  ~-    5 
NUMBER  OF  LONG  PERIOD  OTHER  CHANNELS  -  0 
NUMBER  OF  SHORT  PERIOD  PROCESSES  =  10 
in
CENR  55-2,  Vol  !V   13  July  1987 
NUMBER  OF  LONG  PERIOD  PROCESSES  =  1 
SHORT  PERIOD  FREQUENCY  FILTER  LENGTH  =  99 
LONG  PERIOD  FREQUENCY  FILTER  LENGTH  =  1 
AMOUNT  OF  SHORT  PERIOD  TIME  DELAY  REQUIRED  =  0 
AMOUNT  OF  LONG  PERIOD  TIME  DELAY  REQUIRED  =  0 
SP  COORDINATES: 
0,0,0 
1,0.0,0.0 
2,0.403,0.710 
3,0.735,0.247 
4,0.38,-0.494 
5,-0.546,-0.68 
6, -0.878,0. 093 
7,-0.688,0.71 
8,1.731,0.803 
9,0.806,-1.452 
10,-0.261,-1.421 
11,-1.162,-1.544 
12,-1.257,-0.68 
13,-1.874,-0.958 
14,-2.253,-0.587 
15,-1.826,0.216 
16,-0.878,1.544 
LP  COORDINATES: 
0,0,0 
1,0, O.C 
2.0.0.C 
SP  FREQUENCY  FILTER  PARAMETERS: 
0. 0001, -.0001, -.0005,-. 001 1,-. 0016, -.0020, -.0020, -.0017, 0.0014, -.0012 
-.0011, -.0013, -.0014, -.0011, -.0004, 0.0007,0. 0019, 0.0027, 0.0030, 0.0029 
0.0027, 0.0028, 0.0034, 0.0043, 0.0050, 0.0049, 0.0033, 0.0017, -.0007, -.0027 
-.0037, -.0041, -.0045, -.0065, -.0103, -.0162, -.0221, -.0266, -.0273, -.0254 
-.0224, -.0237, -.0315, -.0481, -.0653, -.0731, -.0456, 0.0324, 0.2035, 0.3910 
0.2035, 0.0324, -.0456, -.0731, -.0653, -.0481, -.0315, -.0237, -.0224, -.0254 
-.0273, -.0266, -.0221, -.0162, -.0103, -.0065, -.0045, -.0041, -.0037, -.0027 
-.0007,0.0017,0.0038,0.0049,0.0050,0.0043,0.0034,0.0023,0.0027,0.0029 
0.0030,0. 0027, Z. 0019, 0.00fi7  -. 0004,-. 0011 , -.0014, -.0013, -.0011 ,-.0012 
-.0014, -.0017, -.0020, -.0020, -.0016, -.0011, -.0005, -.0001, -.0001 
LP  FREQUENCY  FILTER  PARAMETERS: 
0.9999 
SP  BEAM  PARAMETERS: 
SPB360, 0,000,8.0,3 
SPB060,0,060,8.0,3 
SP3120, 0,120, 8.0,8 
SPB180,0,180,8.0,B 
SPB240,0,240,8.0,B 
SPB300,0,300,8.0,B 
SPZOOO,0,0,0,0,B 
SPL065, 0,065,13.0,6 
SPU033, 0,033, 20. 0,B 
SPL010, 0,010, 13. 0,B 
LP  BFAM  PARAMETERS: 
LPB36Z,1,0C0,3.5,3 
SP   PROCESSING  DELAY   =  60 
LP   PROCESSING  DELAY   =   i 
SECONDS  PER  RECORD  =  4 
b.      CPU2: 
CONFIGURATION   IDENTIFICATION   =  Cxxxx-2LS 
0PERATE2   IDENTIFICATION   =  0PERATE2 
SITE    IDENTIFICATION   =   301 
LP   DATA  AND   INSTRUMENT   TYPE    (A, 31, 36)    =  A 
NUMBER  OF   SHORT  PERIOD   ARRAY  CHANNELS   =    15 
NUMBER  OF   SHORT  PERIOD   OTHER  CHANNELS   =   7 
NUMBER  OF   LONG  PERIOD  ARRAY  CHANNELS   =   5 
NUMBER  OF   LONG  PERIOD  OTHER   CHANNELS   =   0 
NUMBER  OF   SHORT  PEHIOO   ACCESSES   =    10 
NUMBER  OF   LONG  PERIOD   PROCESSES   =   1 
NO   SP   CHANNELS   TO  BE    TRANSMITTED    /I A    JS.l   -   0
CENR  55-2,  Vol  IV   13  July  1987 
NO  LP  CHANNELS  TO  8E  TRANSMITTED  VIA  HSM  =  0 
NUMBER  OF  CONTACT  SENSOR  MONITORS  =  2 
NUMBER  OF  A/D  CHANNEL  CHANNEL  MONITORS  =  1 
AMOUNT  OF  SP  EDIT  TIME  DELAY  REQUIRED  =  0 
AMOUNT  OF  LP  EDIT  TIME  DELAY  REQUIRED  =  0 
SP  COORDINATES: 
0,0.0 
1,0.0,0.0 
2,0.403,0.710 
3,0.735,0.247 
4,0.38,-0.494 
5,-0.546,-0.58 
6,-0.878,0.093 
7,-0.688,0.71 
8,1.731,0.803 
9,0.806,-1.452 
10,-0.261,-1.421 
ll,-1 .162,-1.544 
12,-1.257,-0.68 
13,-1.874,-0.958 
14,-2.253,-0.587 
15,-1.826,0.216 
16,-0.878,1.544 
LP  COORDINATES: 
0,0,0 
1.0, 0,C 
2,0.0,C 
SP   CALIBRATION   DEFAULT  PARAMETERS: 
0.833,1.000,25,1,073000,0.9,1.1,2.928,8 
1.00,1.708 
0.5,1.708 
0.8,1.708 
1.5,1.708 
2.0,1.708 
2.5,1.708 
3.0,1.708 
4.0,1.708 
LP  CALIBRATION   DEFAULT  PARAMETERS: 
1.333,0.04,10,1,080000,0.9,1.1,3.751,7,3 
0.040,0.666 
0.100,4.666 
0.067,0.666 
0.050,0.666 
0.033,0.666 
0.025,0.666 
0.020,0.666 
SP  CHANNEL  CONFIGURATION  FOR  CALIBRATION  SYSTEM: 
1.1 
1.2 
1,3 
1,4 
1,5 
1.6 
1.7 
1,8 
1,9 
1,10 
1,U 
1.12 
1,13 
1,14 
1,15 
1,16 
1,19 
1,19 
1,19 
1.19 
1,19 
1,19 
1,24
CENH  55-2,  Vol  IV   13  July  i987 
.P  CHANNEL  CONFIGURATION  FOR  CALIBRATION  SYSTEM: 
SP  BEAM  PARAMETERS: 
SPB360,0,000,8.0,B 
SPB06O.O, 060, 8.0,8 
SPB120,0,120,8.0,B 
SPB180,0,180,8.0,B 
SPB240,0,240,3.0,B 
SPB300, 0,300,8.0,8 
SPZOOO, 0,0, 0,0,8 
SPL065, 0,065, 13. 0,B 
SPU033, 0,033, 20. O.B 
SPL010, 0,010,13.0,8 
LP  BEAM  PARAMETERS: 
LPB36Z,1,000,3.5,B 
RELAY  INDENTIFIERS  AND  NORMAL  STATUS  FOR  EACH  CONTACT  SENSOR  MONITOR: 
LOUBAT.l 
ACFAIC.l 
IDENTIFIERS  AND  LIMITS  FOR  EACH  A/D  CHANNEL  MONITOR: 
LNP0WR,-9.5,9.5 
SECONDS  PER  RECORD  =  1 
OFFICIAL 
BILLY  J.  BINGHAM,  Colonel,  USAF 
Commander 
WALLACE  L.  HUFFAKER,  CMSgt,  USAF 
Director  of  Administration 
SUMMARY  OF  CHANGES 
Rewrote  in  active  voice.   Incorporated  IMC  85-1.  Added  purpose  statement.  Added  limited 
distribution  paragraph.  Added  summation  channel  paragraph.  Deleted  references  to  specific 
paragraphs  to  Vol  I.  Added  configuration  parameters.  SP  array  coordinates  updated.