page_title: Gas chromatography–mass spectrometry
section_title: Instrumentation
hierarchical_section_title: Gas chromatography–mass spectrometry / Instrumentation
caption_attribution_description: English: A gas chromatograph–mass spectrometer (GCMS) with open door - (ThermoQuest Trace 200). The apparatus is the property of CBMiM PAN
context_page_description: Gas chromatography–mass spectrometry is an analytical method that combines the features of gas-chromatography and mass spectrometry to identify different substances within a test sample. Applications of GC-MS include drug detection, fire investigation, environmental analysis, explosives investigation, and identification of unknown samples, including that of material samples obtained from planet Mars during probe missions as early as the 1970s. GC-MS can also be used in airport security to detect substances in luggage or on human beings. Additionally, it can identify trace elements in materials that were previously thought to have disintegrated beyond identification. Like liquid chromatography–mass spectrometry, it allows analysis and detection even of tiny amounts of a substance.
  GC-MS has been regarded as a "gold standard" for forensic substance identification because it is used to perform a 100% specific test, which positively identifies the presence of a particular substance. A nonspecific test merely indicates that any of several in a category of substances is present.
context_section_description: The GC-MS is composed of two major building blocks: the gas chromatograph and the mass spectrometer. The gas chromatograph utilizes a capillary column whose properties regarding molecule separation depend on the column's dimensions (length, diameter, film thickness) as well as the phase properties (e.g. 5% phenyl polysiloxane). The difference in the chemical properties between different molecules in a mixture and their relative affinity for the stationary phase of the column will promote separation of the molecules as the sample travels the length of the column. The molecules are retained by the column and then elute (come off) from the column at different times (called the retention time), and this allows the mass spectrometer downstream to capture, ionize, accelerate, deflect, and detect the ionized molecules separately. The mass spectrometer does this by breaking each molecule into ionized fragments and detecting these fragments using their mass-to-charge ratio.
  These two components, used together, allow a much finer degree of substance identification than either unit used separately. It is not possible to make an accurate identification of a particular molecule by gas chromatography or mass spectrometry alone. The mass spectrometry process normally requires a very pure sample while gas chromatography using a traditional detector (e.g. Flame ionization detector) cannot differentiate between multiple molecules that happen to take the same amount of time to travel through the column (i.e. have the same retention time), which results in two or more molecules that co-elute. Sometimes two different molecules can also have a similar pattern of ionized fragments in a mass spectrometer (mass spectrum). Combining the two processes reduces the possibility of error, as it is extremely unlikely that two different molecules will behave in the same way in both a gas chromatograph and a mass spectrometer. Therefore, when an identifying mass spectrum appears at a characteristic retention time in a GC-MS analysis, it typically increases certainty that the analyte of interest is in the sample.
caption: The insides of the GC-MS, with the column of the gas chromatograph in the oven on the right.
exif: {"Image Orientation": "Horizontal (normal)", "Image XResolution": "180", "Image YResolution": "180", "Image ResolutionUnit": "Pixels/Inch", "Image Software": "Adobe Photoshop CS2 Windows", "Image DateTime": "2013:10:23 18:28:43", "Image ExifOffset": "164", "Thumbnail Compression": "JPEG (old-style)", "Thumbnail XResolution": "72", "Thumbnail YResolution": "72", "Thumbnail ResolutionUnit": "Pixels/Inch", "Thumbnail JPEGInterchangeFormat": "302", "Thumbnail JPEGInterchangeFormatLength": "4463", "EXIF ColorSpace": "Uncalibrated", "EXIF ExifImageWidth": "2016", "EXIF ExifImageLength": "1100"}