The Triune, a new silicone tympanostomy tube.

• 1589 Triune tubes were inserted in a two year period and followed every six months to determine the rate of extrusion and the complication rate. • 73% of patients followed up and the average extrusion was at 30 months. • Perforation rate was 2.2%, tube granulomas formed in 2.8%, and 6.2% of tubes became plugged. • 32% of the tubes developed otorrhea. • The conclusion is that the Triune is a medium term tube with a low complication rate. • Graphing the extrusion of 1158 tubes seen in follow up, it is evident that the average extrusion is at approximately 30 months, see below. • 26 of 1158 intubated ears seen in follow up developed perforations, which represents 2.2%. • 32 of 1158 tubes or 2.8% developed granulomas averaging 1.4 mm in size. All resolved with topical antibiotic/steroid drops. • Serous or blood clot plugs developed in 72 tubes or 6.2%. • Otorrhea occurred in 264 ears out of 818 questioned regarding drainage. This represents 32.3%. Results 1. Kay DJ, Nelson M, Rosenfield R. Meta-analysis of tympanostomy tube sequelae. Otolaryngol Head Neck Surg 2001;124:374-80. 2. Goode RL. Long-term middle ear ventilation with T tubes: the perforation problem (editorial). Otolaryngol Head Neck Surg 1996;115:500-1. 3. Mandel EM, Casselbrant ML, Kurs-Lasky M. Acute otorrhea: bacteriology of a common complication of tympanostomy tubes. Ann Otol Rhinol Laryngol 1994; 103:713-718. Bibliography • Triune tubes were inserted into all the author’s patients needing more than brief middle ear ventilation in a private otolaryngology group practice. • Over a two year period from March, 2004 to March of 2006, 1,589 tubes were inserted, 1132 under general anesthesia and 458 under local anesthesia. • Patients were instructed to return in one month and every six months thereafter for microscopic examination. 1158 tubes or 73% were seen in follow up. • Complications were documented in a spreadsheet at every return visit with a mean follow up of 12.3 months. • Tubes were manufactured using a cold extrusion process by Grace Medical in Memphis, Tenn. • Tubes were inserted using micro alligator forceps with the lower jaw in the lumen of the tube. See figure 2 below. Methods and Materials Figure 3-Extrusion Rate • The Triune tube ventilates the middle ear an average of 30 months. • The Triune Tube has a low perforation rate, particularly considering the 30 months to extrusion. • Most likely the low perforation rate is related to the tube’s conformity to the convex shape of the TM. Conclusions The premise of the Triune’s design is that the T-tube causes a high percentage of perforations due to the “tenting” of the tympanic membrane by the perpendicular arms. The Triune tube was intended to conform closely to the convex shape of the tympanic membrane and to stabilize the tube perpendicular to the plane of the TM by using three soft arms that are angled down 15 degrees from perpendicular. The arms are .2mm in thickness, soft, flat, and therefore relatively atraumatic to the TM. See figures below. n=823 n=666 n=373 n=503 n=321 n=201 n=203 0 5 10 15 20 25 30 35 40 45 50 4 8 12 16 20 24 28 P e rc e n ta g e E x tr u d e d Months Post Insertion Extrusion Rate