Assessment of a Novel Dielectric Microsensor for Monitoring Coagulation Factor Therapy in Children with Hemophilia with and without Inhibitors

Abstract Introduction: Monitoring coagulation factor or bypassing agent therapy has important clinical implications for the management of children with hemophilia, especially those with inhibitors. For patients with hemophilia, conventional laboratory-based coagulation tests are time-consuming, labor-intensive, and costly. Moreover, no monitoring assay is currently available for use with bypassing agents in patients with hemophilia and inhibitors, except for thromboelastography or rotational thromboelastometry (ROTEM), both of which are not widely available clinically and are labor-intensive. There is a need to develop an easy-to-use, low-cost, portable global assay of coagulation for monitoring therapies in hemophilia. The aim of this study was to assess the clinical utility of monitoring replacement therapy using a new technique based on a novel dielectric microsensor (ClotChip) that can potentially enable global blood coagulation assessment in a miniaturized, portable measurement platform. Methods: Patients accrued for the study were children with severe hemophilia A & B (n = 6), children with hemophilia and inhibitors (n = 4), and normal, healthy children without coagulopathy (n = 22). Blood samples were obtained by venipuncture into collection tubes containing 3.2% sodium citrate anticoagulant. Samples from patients with hemophilia were collected both prior to the initiation of replacement therapy and 15 minutes after administration of an appropriate coagulation factor VIII or IX for severe hemophilia A or B, respectively, or a bypassing agent (FEIBA or recombinant factor VIIa) for hemophilia with inhibitors. Whole blood samples were then tested with the ClotChip within 2 hours of collection, and coagulation was induced with CaCl2 prior to the start of ClotChip measurements. The ClotChip is based on the fully electrical technique of dielectric spectroscopy (DS), and features a three-dimensional, parallel-plate, capacitive sensing structure integrated into a low-cost (material cost Results: We observed a significant decrease in the ClotChip Tpeak parameter for post-therapy samples compared to baseline (pre-therapy) for children with hemophilia without inhibitors (Fig 1C, n = 6, p = 0.0001) and with inhibitors (Fig 1D, n = 4, p = 0.0024). Interestingly, baseline ClotChip measurements for 2 tolerized patients with inhibitors exhibited Tpeak of 970s and 1,010s, which are within the range of Tpeak for patients without inhibitors (750s - 1,320s), while the baseline ClotChip measurements of 2 non-tolerized patients with inhibitors exhibited higher Tpeak of 1,810s and 2,100s (Fig 1E). Moreover, ClotChip measurements for all samples taken post-therapy exhibited Tpeak that fell within the reference range of Tpeak for normal, healthy children (180s - 720s, Fig 1E). Taken together, these data suggest that the ClotChip Tpeak parameter is sensitive to detection and correction of coagulopathy in children with hemophilia with and without inhibitors, as indicated by Tpeak reaching reference range levels after coagulation factor therapy. Finally, the ClotChip Tpeak exhibited a very strong correlation (Fig 1F, Pearson's r = 0.94) to the clinically important CT parameter of ROTEM. Conclusions: Our studies demonstrate the feasibility of monitoring coagulation factor therapy in children with hemophilia with and without inhibitors, using a novel, microfluidic, dielectric sensor (ClotChip), allowing whole blood assessment of hemostasis in a single disposable sensor. ClotChip has potential to fulfill an important unmet clinical need for monitoring therapy, especially for hemophilia with inhibitors. Further studies are underway to assess the ability of ClotChip to differentiate bleeding phenotypes and monitor outcomes for treatment of bleeds in children with hemophilia. Download : Download high-res image (315KB) Download : Download full-size image Disclosures Ahuja: CSL Behring: Honoraria; Bayer: Honoraria; Shire: Honoraria; Shire: Speakers Bureau. Suster: XaTek, Inc.: Patents & Royalties: Michael Suster is an inventor of intellectual property that has been licensed by Case Western Reserve University to XaTek, Inc., and as such, may receive future royalties. Maji: XaTek, Inc.: Patents & Royalties: Denath Maji is an inventor of intellectual property that has been licensed by Case Western Reserve University to XaTek, Inc., and as such, may receive future royalties. Mohseni: XaTek, Inc.: Patents & Royalties: Pedram Mohseni is an inventor of intellectual property that has been licensed by Case Western Reserve University to XaTek, Inc., and as such, may receive future royalties.