Well, here's your answer. The Chub-E4 has four inputs for reading four different sensors simultaneously, and we'll configure those inputs in any of three different ways. With this thermometer, reading thermocouples, PRTs, and thermistors accurately from the same device is no problem. Au-Pt, and allow you to choose between internal and external reference junction compensation. Typical accuracy for a type J thermocouple using internal reference junction compensation at 600 C is 0.35 C, not including the thermocouple.
Thermocouples connect using standard or miniature terminations.
Measurements are made each second and can be taken simultaneously or sequentially. A special high-speed mode allows measurements to be taken at the rate of 10 per second. If you think three sensor types and four inputs sounds versatile, wait until you see the display panel on the Chub-E4. Displaying measurements in C, F, K, ohms, or millivolts and choosing temperature resolution from 1 to 0.0001 are just the beginning. You can also select any eight data fields to view on-screen.