Also suitable for other applications requiring extremely precise temperature control. High output chrome plated immersion heaters provide efficient heat transfer, and a turbine stirrer circulation system provides temperature uniformity without causing turbulence or entrainment of air bubbles. A continuous heat circuit permits heating of the bath liquid at several times the normal rate. For operation at ambient temperatures or below, circulate tap water or refriderated coolant through the built-in cooling coil. Standard Model and Turntable Model - Standard model accommodates six glass capillary viscometers.
Turntable model accomodates twelve round viscometer holders.
Freely moving turntable permits each viscometer to be positioned at the front of the bath for easy viewing. Bath interior of both models is brightly illuminated by a fluorescent lamp. Control unit is mounted in removable drawer for ease of servicing, if required. Order capillary viscometers, viscometer holers and thermometers separately. Power is also interrupted at any temperature if the PT-RTD sensor probe becomes disconnected or malfunctions.