While accuracy in this extended temperature region is slightly less than in the 0 to 45 degree range, it is adequate for the vast majority of applications encountering temperature extremes. All of these improvements translate into a more consistent, accurate system that is easier to test and is less affected by temperature variations. We were also able to give this transducer system an extended linear range . The extended linear range is intended for axial position measurements. However, the extended linear range is not compatible with installations using intrinsic safety barriers external to the monitoring system.
Sensors are available in two different mounting configurations.
These gold-plated brass connectors are highly robust and are much easier to use than our previous stainless steel connectors. They should only be tightened finger-tight, which means that they will be secure after 2 to 5 clicks. A spring-loaded locking mechanism prevents the connectors from becoming loose, even when subject to high vibration. The gold-plated brass construction and locking mechanism virtually eliminates the possibility of intermittent connections caused by a loosening connection. The 3300 XL 11 mm Transducer System has also achieved compliance with European CE standards with fewer restrictions than previous transducer systems.