Cable connecting I-330 and computer not plugged in securely. Incorrect serial port selected. Readings for EMG, Temp, EDR, etc. Check the setup configuration screen for correct labels and correct scaling. Please consult manual.
Screen appears to freeze after a keystroke.
The display will appear to freeze until that channel number is pressed. Computer freezes when a display is selected and sometimes see 'error code 3'. This occurs most often with serial devices. Check that the cable is securely plugged into the computer and into the CapScan amplifier and that the power unit is plugged into a power strip or the wall and that the CapScan amplifier is turned on. Another reason for this error might be that an incorrect serial port has been selected.