The PUR and BUSY logic pulses are used to prevent analysis and correct for the reset dead time in the associated ADC or multichannel analyzer. Output impedance is 50 W . Useful for dead-time corrections with an associated ADC or multichannel analyzer. Positive NIM standard logic pulse is selectable as active high or active low via a printed circuit board jumper. Output impedance is 50 W .
Output also present for a pulsed reset preamplifier during reset, and reset overload recovery.
Output pulse is selectable as active high or active low by means of a printed circuit board jumper. Output impedance is 50 W . Used with an associated ADC or multichannel analyzer to prevent analysis of distorted pulses. 1 Effectiveness of the Automatic Pole-Zero feature. See specifications for spectrum broadening and spectrum shift.