The information on this site is given in good faith. Colin Medical Instruments will accept no responsibility for any losses arising from errors in this document that is attributed to Republication information or information supplied by others. Mark the appropriate test box as pass if unit continues to function correctly, fail if problems occur. Define the year code as selected on the screen, and printed on the printout. Unit continues to operate without fail at the selected cuff interval.
Unit continues to print upon the completion of each measurement.
Unit continues to output the appropriate RS232 protocol. Clock and date change according to test. When device is turned off and then back on all the above remain true. Select an automated cuff interval of 5 min. Allow the device to operate for a period of one hour.