For temperature measurements below 1000 F, the instrument displays to a 0.1 resolution. With a measurement more than 1000 F, the instrument automatically switches to a 1 resolution. Thermocouple types R, S, and B only display in 1 increments. For any applications where a 1 resolution is desired for everything, the unit can be set to display only 1 increments throughout its measuring range. Large LEDs make temperature readings easy to read.
Power is selectable via an internal slide switch.
3-wire, Platinum RTDs, either .00385 or .00392 alpha. Each of these nine are selected via a miniature rotary switch located directly behind the snap-off front lens. No calibration is required to switch between thermocouple types. Another miniature DIP switch, also behind the lens, selects between F or C display. Alarm Setpoint Option .