These columns are flexible and can be wound into coils. They have the advantages of physical strength, flexibility and low reactivity. Column temperature For precise work, column temperature must be controlled to within tenths of a degree. The optimum column temperature is dependant upon the boiling point of the sample. As a rule of thumb, a temperature slightly above the average boiling point of the sample results in an elution time of 2 - 30 minutes.
Minimal temperatures give good resolution, but increase elution times.
If a sample has a wide boiling range, then temperature programming can be useful. There are many detectors which can be used in gas chromatography. Different detectors will give different types of selectivity. A non-selective detector responds to all compounds except the carrier gas, a selective detector responds to a range of compounds with a common physical or chemical property and a specific detector responds to a single chemical compound. Detectors can also be grouped into concentration dependant detectors and mass flow dependant detectors .