The result is fast, accurate and reliable response with outstanding sensitivity and selectivity. Multiples of the 6-channel controller can be configured in larger enclosures for modular and space efficient expansion of the system. The Model 2000 is the new generation in microprocessor- based gas detection systems emphasizing simplicity of operation and reliability. It was clearly designed with the end user in mind. High, low and trouble LEDs provide visual indication of alarm status, while an integral buzzer provides audible indication of alarms at the controller.
Programmable relays offer added flexibility in area monitoring and reduced installation cost.
LCD prompts you for easy setup of each channel. Push button switches enable the user to address various setup parameters on the LCD and easily program each channel. An optional battery backup with trickle charge circuit is available as well as RS-485. Does not drift - no autozeroing or adjustments required. Maintains calibration for over six months - good practice dictates calibration be checked monthly.