To adjust the setpoint make sure the numeric display is set to Setpoint , then adjust the setpoint to the desired value. See adjusting setpoint value. If you need to change the setpoint value during your experiment on a routine basis there is a remote setpoint jack on the back of the controller. See Remote Setpoint. Dish is the location where the calibration of the dish thermistor reading circuit can be adjusted.
Reference is the location where the calibration of the thermistor reading circuit can be adjusted.
Adjust Volts is the location where the non-dynamic voltage adjustment is made. This is the energy value sent to the dish when the controller is in Imaging Mode . Adjust Volts must be selected for the display and the Mode switch must be pressed for this value to appear. The display will show a number that is 10 times the actual voltage value being sent to the dish. Adjustments can be made to this value by turning the small screw located in the hole corresponding to Imaging Volts.