Check that the cable is securely plugged into the computer and into the CapScan amplifier and that the power unit is plugged into a power strip or the wall and that the CapScan amplifier is turned on. Another reason for this error might be that an incorrect serial port has been selected. Readings in scope and other displays are 0. This happens with internal cards connected to the Prism 5 or CapScan amplifiers. Most often it is an incorrect setup, either IRQ or address, of the card.
Program runs very slowly.
Then run VEEG3. DOS Prism 5 CapScan Readings 0 or 45. This happens with internal cards connected to the Prism 5 amplifier. Most often it is an incorrect setup, either IRQ or address, of the card. UniComp Windows Erratic and erroneous readings in Temp and other channels.