Process the sample and view. Only the left or right half of an image can be exposed by turning the TWIN switch to L or R . Utilize this function according to your microscopic purpose. Direct photo procedure Select a visual field, and carry out focusing and astigmatism correction. Optimize image brightness and contrast.
Image disappears from the flicker-free image CRT and is newly formed on the screen.
While photographing, the selected one of the exposure lamps 1,2, and 3 flickers. When scan reaches the lowermost line, the buzzer beeps to notify the end of image recording. Once scan has covered two thirds of the screen after start of recording, its cancellation cannot be accepted.