COLD CUBE Cold Cube is a heat absorbing and dissipating device that mounts between the lamphouse and Scrambler Input Module. A dichroic is mounted at 45 degrees so that IR passes straight through to a heat dissipating tower, and the visible light is reflected into the microscope. Cold Cube is effective from 400 nm and up. It is pictured with a Zeiss HBO 100W arc lamphouse attached to a Scrambler Input Module. It may also be attached directly to a microscope.
Models are available for all microscopes.
The clear aperture is 23mm. The wheel is black anodized aluminum and is 12mm thick. The wheel is fully supported by both annular and thrust ball bearings. A quiet and positive stepper motor driver is implemented by no slip cog wheels and belt. The unit is supplied with appropriate adapters for direct mounting between your microscope and lamphouse, or inline with a Scrambler system.