Winchester Arm 10. They either exceeded the normal range of values for estuarine waters or showed no variation. Temperature data were within the normal range of values so was not deleted. Time stamp was off so 15 minutes was subtracted to return to whole and half hours. However, no post-deployment calibration was recorded, so data were not deleted.
Low salinities are consistent with high rainfall events.
All pH data were deleted, as they were too low, possibly due to poor calibration. DO data showed less variation than expected but because they were within expected range and no calibration information was available, the data were not deleted. All pH data were deleted see Mar. 12 above. Temp, specific conductivity, and depth were all within normal ranges. The original .csv file was corrupted and a new .csv was exported from the .dat file in EcoWatch.