A converter in the potentiostat modules allows accurate control over a wide range of potentials in 1 millivolt steps. The current capabilities of the potentiostat modules allow use with larger electrodes. All critical signals are independently shielded in the high performance cables to each cell which are six feet long. 1281 Multiplexers together to a single Solartron potentiostat. It consists of a metal enclosure to which two or three model 1281 male input connectors are attached in a manner so as to mate with the desired number of multiplexers that are stacked one upon the other.
Internal wiring is kept as short as possible so as to minimize stray capacitance.
Connections Internal wiring is provided to appropriate connectors for the desired type of potentiostats. BNC type jacks are provided for the users cables that attach to the potentiostat. This configuration results in the lowest possible stray capacitances for the input signals to the multiplexers. Separate sense signals and driven shields are maintained where possible, depending on the type potentiostat used and the available signals. Connections provided - Working, counter, RE1 and RE2 BNC connectors to connect with BNC type 50 ohm cables to the Solartron 12xx series Electrochemical Interface.