Being interfaced with the HRF Data Acquisition System. Being synchronized with other experiment hardware as required. It is possible the CBPD data will be routed through the Data Acquisition System and synchronized in conjunction with any data routed through it. The CBPD shall operate continuously in a temperature ranging from 10 to 40 degrees Centigrade and in humidity ranging from 10 to 70 percent relative, non-condensing. Use a physiocal during run-mode pressure waveform monitoring to observe the plethysmogram and to observe the physiological state of the finger circulation under the cuff.
Internal calibration occurs when needed.
The CBPD shall provide an accuracy of at least 1.5 percent of full scale. Requires 28 Vdc power supply and will have a battery power supply pack to allow a minimum of eight hours of operation. Accommodate three different size finger cuffs.