The receive frequency range is scanned and the result displayed as a spectrum on the screen. The measurement parameters such as start frequency, stop frequency, IF bandwidth, measurement time, attenuation and display mode can be freely selected. Single-shot or repetitive scanning can be selected. Two test traces can be displayed at a time. For comparison measurements, waveforms can also be loaded from a floppy disk.
Frequency scanning over the desired frequency range is at maximum speed and with fixed attenuation.
Frequency scanning is quasi-continuous with selectable step size and measurement time as well as automatic attenuation setting. Frequency scanning over a defined data set with up to 400 frequencies. The results, along with all relevant information, can be output in graphical form or as lists on a plotter or printer. In addition, data can be stored on a floppy disk along with the complete receiver setup. Information can be entered via a line editor or, more conveniently, via an optional keyboard.