The bowls can be operated asynchronously and have totally independent tanks and controls. The disinfection cycle can be operated manually or automatically with soak times being operator controlled. A buzzer signals the completion of the cycle. An LED display gives indication of the process being carried out and the time remaining during an automatic cycle. The unit has two disinfectant storage tanks and a waste storage tank.
Mains water is supplied via a quick release connector.
This supply should encompass a type 'A' air gap tank for compliance with the water bylaws and a filtration system depending on the quality of water required for rinsing. The waste tank is continuously discharged at a convenient point using the integral pumps and controls. An alcohol rinse can be carried out at the touch of a button. Validation of each scope processed is printed out complete with the patient number and scope identification. Clear Perspex lids make the work area easy to use and enables the operator to observe the status in the bowl.