However, the phase relationship between the bridge signal and excitation voltage has reversed 180. The Validyne Variable Reluctance Pressure Transducer has a number of advantages over typical transducers. This transducer has low symmetrical internal volumes which allow it to accept corrosive liquids and gases. To prevent damage, you must keep dirt particles or congealable fluids out of these cavities. For the best dynamic response, you should couple the transducer closely to any gaseous pressure source OR make sure it is free of entrapped air or gas in liquid systems.
High overpressure can cause a permanent zero shift or diaphragm damage.
High Output Signal - This transducer provides a high output signal with low susceptibility to electrical noise, a recognized advantage of carrier systems. You should operate this transducer through a carrier demodulator designed for it. Replaceable Diaphragm - This transducer has only one pressure sensitive element the diaphragm which you can replace if necessary. It is essential to read the diaphragm replacement instructions before replacing it. Transducer Installation Proper installation of a transducer is vital to its operation.