Acquire single set of correlated BLM reading on REL event. Data update to CLC at 0.5 Hz. Post-Mortem Analysis Move all BLM data buffered in MADCs up to console after a beam abort for analysis. For all 400 BLMs this is about 13 Mbytes of data. Upload can take up to 5 minutes.
Analysis includes finding peaks, displaying evolution ring-wide and for a single channel.
As above, move all buffered BLM data to the console for analysis without stopping data acquisition. The MADC systems that buffer BLM data will be set up to run scan group 0 continuously at 720 Hz. This scan group will be the source of BLM data for all of the above uses except the DigitalScope application. Scan group 1 will be available to users of the DigitalScope application. A user will be permitted to change the MADC settings for this scan group to retrieve the desired data.