Design software available as add-on options. Smart electrodes Directly controls Swift Smart Electrodes. Total automation of the measure sequence for a large array of electrodes possible. Measuring resolution max 30 nV, depends on voltage level Screen resolution 4 digits in engineering notation. Actual electrode voltage depends on transmitted current and ground resistivity.
Instrument always starts up in low voltage setting for safety.
Input gain ranging Automatic, always uses full dynamic range of receiver. Max 500 V SP compensation Automatic cancellation of SP voltages during resistivity measurement. Constant and linearly varying SP cancels completely. Measure cycles Running average of measurement displayed after each cycle. Automatic cycle stops when reading errors fall below user set limit or user set max cycles are done.