Tighten the nut until the nebulizer's liquid input end is held in place. The connector kit is now installed and the Teflon tubing should be held in place against the nebulizer's capillary taper. The nebulizer can now be installed into the instrument and the Teflon tubing either inserted into the sample or rinse solution for natural aspiration or into the end of the peristaltic pump tubing for pumped aspiration. The high pressure fitting easily holds up to 190 psig. Push nut on far enough to expose maria.
Gravity can assist assembly by having sidearm pointing up during installation.
Push O-ring over the maria on the sidearm. Put the front ferrule on the sidearm with the base facing the O-ring and the conical side facing the union body. Screw the union body and nut together firmly. East Warner Ave. They do not need to be removed.