A pair of velcro straps are included so the user can easily attach the thermocouples to pipes at two different locations. He can also choose 1 degree resolution or 0.1 degree resolution. The ST3 Thermometer from Fieldpiece has solved the temperature compensation problem that is common in many instruments of this type. We didn't introduce a product in this category until we solved the temperature compensation problem. Now you can trust your measurement, even when the ambient temperatures are changing fast.
Many competitive products require several minutes to stabilize when the ambient temperature changes.
The ST3 uses a custom ceramic component to ensure that the two points in the k-type thermocouple circuit that have to be at the same temperature are in sync. When the ambient temperature changes fast, those two points will remain at the same temperature, even as they are both changing. The new ST3 Thermometer is field-calibratable using the ice bucket calibration method. The field technician first stabilizes a large cup of ice water. One probe is then immersed into the water and stabilized.