Daily logs of this campaign are available. Ultimately, we want to determine which piece of hardware excites the detectors and the amplitude of the noisy features so produced. The HRC Amp D is defective and was ignored. All on side 1.
Detectors start at room temperature.
Turn on the TECs and start taking a sequence of 16 HRC or 42 WFC subarrays as it cools down. So, the TECs are turned off just before the housing temperature reaches this limit and the detectors are allowed to warm up. See the associated logs in the binder to correlate the image number with temperature range for the HRC and WFC sets. It is read out 42 consecutive times. For each frame, evaluate the median and standard deviation in the trailing physical overscan.