Clamp all unused lines. Control unit enters run mode. The status screen will be displayed. This screen displays flow and fluids removed. The settings can be adjusted by touching the soft keys on the bottom of the screen and following the instructions that appear on the screen.
Control unit enters the standby mode.
Circuit can now be changed or therapy discontinued. Press this key if blood is to be returned to the patient at the end of therapy. Hang a liter of normal saline on the lower left corner hook. Clamp red access line and the patient catheter. Disconnect line from patient.