Lead wire compensation is supported with 5-wire input configurations. Bridge completion and shunt calibration resistors may be connected to user-accessible screw terminals. 0.5 Hz to 100 KHz. All channels are configured with a spreadsheet-like setup screen and include parameters such as name, sensor type, scaling formula, triggering thresholds, DSP settings and optional alarm points. A bar graph display is also available for selected channels.
All data is stored to the PC hard drive.
Any of the newer laptop PCs can be used with the VDAS for maximum flexibility and fast, on-site data analysis.