In manual motion and acquisition, the device will sample a waveform at a single position. Acquisition is defined by an external pulse, which is assumed to indicate the presence of data at the sensor. The data is presented to the local console and channel access interface. User definable parameters include data array length, presentation, and reduction. Automatic scan and acquire In automatic scan and acquire, the device will sample an array of waveforms from a range of positions.
Acquisition is synchronized to external pulse train and mechanism position information.
The data is presented to the local console and channel access interface. The device is capable of an automatic self-test manually activated by the user. The results of the test are presented to the local console and channel access interface. Bias voltage adjustment For accurate measurement of the wire current the wire must be biased. This bias voltage is adjustable.