Each buffer contains 320 vector and 32 waveform records. Vector Data includes Direct, GAP, 1X and 2X, Amp and Phase, Not 1X for each dynamic monitor channel. Waveform records include both Synchronous and Asynchronous data. Waveform Sampling Waveforms are dc coupled to prevent low frequency roll off. Both Synchronous and Asynchronous data includes 1024 samples.
Asynchronous data is anti-alias filtered.
Waveform data is pre-triggered. SDI support includes 1X, 2X, High Resolution Gap and Not 1X static data. 110 Vac and 0.38 Amps typical at 220 Vac. Auto-configured after power up. Modem allowed between first communications processor and host computer.