The user interface at the controlling computer is the same, whichever link is used. Model 5800's command structure involves 28 commands which execute individually to set the pulser-receiver parameters. All instrument settings may be viewed locally through an eight line by forty character LCD display which features adjustable contrast and back lighting.
All attenuators use relay switched resistors for accuracy and stability.
Four settings are available for each. The Model 5800 is housed in a rugged, attractive rack adaptable instrument cabinet 3.5 inches high. All connections other than the transmit and receive jacks are on the rear panel of the unit. Local control also allows the 5800PR to stand alone 40 Stored setups. With a receiver bandwidth of 35MHz, the Model 5800PR is well suited for general purpose ultrasonic testing in a wide variety of metals, plastics, ceramics, composites, and biological specimens.