The 3810, along with the 2800 series family of standard and custom probes, uses an advanced capacitive gaging technology to provide exceptional resolution, large operating ranges and large standoff distances. Precision alignment Options Operating ranges Operating ranges can be user specified to optimize resolution for a specific application. Probes ADE Technologies has developed a new, lower cost family of high performance capacitive sensors providing a greater temperature stability and measurement linearity. Measurement ranges from 20 microns to 2 millimeters are available in standard products. Custom probe configurations are available to met unique applications requirements.
Actual resolution depends on probe selection, bandwidth and operating range.
Resolution improves with larger sensor area, reduced bandwidth and smaller range. Other selections from 10 kHz to 5 kHz. Measurement resolution using Model 2804 2mm active area with operating range 50 microns, standoff 100 microns, 1kHz bandwidth. Analog Output Single, preset 10 volts for full scale range. Drive Synchronizationt For multi-unit applications.