Document ID: EPA-HQ-OAR-2018-0815-0028
Agency: epa
Document Type: Rule
Title: Test Methods and Performance Specifications for Air Emission Sources
Posted Date: 2020-10-07T04:00Z

[Federal Register Volume 85, Number 195 (Wednesday, October 7, 2020)]
[Rules and Regulations]
[Pages 63394-63422]
From the Federal Register Online via the Government Publishing Office [www.gpo.gov]
[FR Doc No: 2020-18824]

[[Page 63393]]

Vol. 85

Wednesday,

No. 195

October 7, 2020

Part III

Environmental Protection Agency

-----------------------------------------------------------------------

40 CFR Parts 51, 60, 61, and 63

Test Methods and Performance Specifications for Air Emission Sources; 
Final Rule

  Federal Register / Vol. 85, No. 195 / Wednesday, October 7, 2020 / 
Rules and Regulations  

[[Page 63394]]

-----------------------------------------------------------------------

ENVIRONMENTAL PROTECTION AGENCY

40 CFR Parts 51, 60, 61, and 63

[EPA-HQ-OAR-2018-0815; FRL-10012-11-OAR]
RIN 2060-AU39

Test Methods and Performance Specifications for Air Emission 
Sources

AGENCY: Environmental Protection Agency (EPA).

ACTION: Final rule.

-----------------------------------------------------------------------

SUMMARY: This action corrects and updates regulations for source 
testing of emissions. These revisions include corrections to inaccurate 
testing provisions, updates to outdated procedures, and approved 
alternative procedures that will provide flexibility to testers. These 
revisions will improve the quality of data and will not impose any new 
substantive requirements on source owners or operators.

DATES: The final rule is effective on December 7, 2020. The 
incorporation by reference of certain materials listed in the rule is 
approved by the Director of the Federal Register as of December 7, 
2020]. The incorporation by reference of certain other materials listed 
in the rule was approved by the Director of the Federal Register as of 
July 6, 2006.

ADDRESSES: The EPA has established a docket for this action under 
Docket ID No. EPA-HQ-OAR-2018-0815. All documents in the docket are 
listed on the http://www.regulations.gov website. Although listed in 
the index, some information is not publicly available, e.g., 
confidential business information or other information whose disclosure 
is restricted by statute. Certain other material, such as copyrighted 
material, is not placed on the internet and will be publicly available 
only in hard copy. Publicly available docket materials are available 
electronically through http://www.regulations.gov.

FOR FURTHER INFORMATION CONTACT: Mrs. Lula H. Melton, Office of Air 
Quality Planning and Standards, Air Quality Assessment Division (E143-
02), Environmental Protection Agency, Research Triangle Park, NC 27711; 
telephone number: (919) 541-2910; fax number: (919) 541-0516; email 
address: melton.lula@epa.gov.

SUPPLEMENTARY INFORMATION: 
    The supplementary information in this preamble is organized as 
follows:

Table of Contents

I. General Information
    A. Does this action apply to me?
    B. What action is the Agency taking?
    C. Judicial Review
II. Background
III. Incorporation by Reference
IV. Summary of Amendments
    A. Method 201A of Appendix M of Part 51
    B. General Provisions (Subpart A) of Part 60
    C. Standards of Performance for New Residential Wood Heaters 
(Subpart AAA) of Part 60
    D. Standards of Performance for Municipal Solid Waste Landfills 
That Commenced Construction, Reconstruction, or Modification After 
July 17, 2014 (Subpart XXX) of Part 60
    E. Standards of Performance for Commercial and Industrial Solid 
Waste Incineration Units (Subpart CCCC) of Part 60
    F. Emission Guidelines and Compliance Times for Commercial and 
Industrial Solid Waste Incineration Units (Subpart DDDD) of Part 60
    G. Standards of Performance for Stationary Spark Ignition 
Internal Combustion Engines (Subpart JJJJ) of Part 60
    H. Standards of Performance for Stationary Combustion Turbines 
(Subpart KKKK) of Part 60
    I. Standards of Performance for New Residential Wood Heaters, 
New Residential Hydronic Heaters and Forced-Air Furnaces (Subpart 
QQQQ) of Part 60
    J. Method 4 of Appendix A-3 of Part 60
    K. Method 5 of Appendix A-3 of Part 60
    L. Method 7C of Appendix A-4 of Part 60
    M. Method 7E of Appendix A-4 of Part 60
    N. Method 12 of Appendix A-5 of Part 60
    O. Method 16B of Appendix A-6 of Part 60
    P. Method 16C of Appendix A-6 of Part 60
    Q. Method 24 of Appendix A-7 of Part 60
    R. Method 25C of Appendix A-7 of Part 60
    S. Method 26 of Appendix A-8 of Part 60
    T. Method 26A of Appendix A-8 of Part 60
    U. Performance Specification 4B of Appendix B of Part 60
    V. Performance Specification 5 of Appendix B of Part 60
    W. Performance Specification 6 of Appendix B of Part 60
    X. Performance Specification 8 of Appendix B of Part 60
    Y. Performance Specification 9 of Appendix B of Part 60
    Z. Performance Specification 18 of Appendix B of Part 60
    AA. Procedure 1 of Appendix F of Part 60
    BB. Appendix B to Part 61--Test Methods
    CC. Method 107 of Appendix B of Part 61
    DD. General Provisions (Subpart A) of Part 63
    EE. Portland Cement Manufacturing (Subpart LLL) of Part 63
    FF. Method 301 of Appendix A of Part 63
    GG. Method 308 of Appendix A of Part 63
    HH. Method 311 of Appendix A of Part 63
    II. Method 315 of Appendix A of Part 63
    JJ. Method 316 of Appendix A of Part 63
    KK. Method 323 of Appendix A of Part 63
V. Public Comments on the Proposed Rule
VI. Statutory and Executive Order Reviews
    A. Executive Order 12866: Regulatory Planning and Review and 
Executive Order 13563: Improving
    Regulation and Regulatory Review
    B. Executive Order 13771: Reducing Regulations and Controlling 
Regulatory Costs
    C. Paperwork Reduction Act (PRA)
    D. Regulatory Flexibility Act (RFA)
    E. Unfunded Mandates Reform Act (UMRA)
    F. Executive Order 13132: Federalism
    G. Executive Order 13175: Consultation and Coordination With 
Indian Tribal Governments
    H. Executive Order 13045: Protection of Children From 
Environmental Health Risks and Safety Risks
    I. Executive Order 13211: Actions Concerning Regulations That 
Significantly Affect Energy Supply, Distribution or Use
    J. National Technology Transfer and Advancement Act (NTTAA) and 
1 CFR Part 51
    K. Executive Order 12898: Federal Actions To Address 
Environmental Justice in Minority Populations and Low-Income 
Populations
    L. Congressional Review Act (CRA)

I. General Information

A. Does this action apply to me?

    The revisions promulgated in this final rule apply to industries 
that are subject to the current provisions of 40 Code of Federal 
Regulations (CFR) parts 51, 60, 61, and 63. We did not list all of the 
specific affected industries or their North American Industry 
Classification System (NAICS) codes herein since there are many 
affected sources in numerous NAICS categories. If you have any 
questions regarding the applicability of this action to a particular 
entity, consult either the air permitting authority for the entity or 
your EPA Regional representative as listed in 40 CFR 63.13.

B. What action is the Agency taking?

    We are promulgating corrections and updates to regulations for 
source testing of emissions. More specifically, we are correcting 
typographical and technical errors, updating testing procedures, and 
adding alternative equipment and methods the Agency has deemed 
acceptable to use.

C. Judicial Review

    Under section 307(b)(1) of the Clean Air Act (CAA), judicial review 
of this final rule is available by filing a petition for review in the 
United States Court of Appeals for the District of Columbia Circuit by 
December 7, 2020. Under section 307(d)(7)(B) of the CAA, only an 
objection to this final rule that was raised with reasonable 
specificity during the period for public comment can be raised during 
judicial review. Moreover, under section 307(b)(2) of the CAA, the 
requirements that are the

[[Page 63395]]

subject of this final rule may not be challenged later in civil or 
criminal proceedings brought by the EPA to enforce these requirements.

II. Background

    The EPA catalogs errors, corrections, and approved alternatives to 
test methods, performance specifications, and associated regulations in 
40 CFR parts 51, 60, 61, and 63 and updates and revises these 
provisions periodically. The most recent revisions to testing 
regulations for air emission sources were proposed in the Federal 
Register on December 13, 2019 (84 FR 68069). The public comment period 
ended February 11, 2020, and 18 comment letters were received from the 
public; 13 of the comment letters were relevant, and the other 5 
comment letters were considered beyond the scope of the proposed rule. 
This final rule was developed based on public comments that the agency 
received on the proposed rulemaking.

III. Incorporation by Reference

    Consistent with the proposal, EPA has incorporated by reference 
various consensus standards. Specifically, the EPA has incorporated 
ASTM D 2369-10, which covers volatile organic content of coatings, in 
Method 24. In addition, in response to comments the EPA has 
incorporated ASTM D5623-16 and ASTM D7039-15a in subpart KKKK of part 
60, which involves procedures for determining the sulfur content of 
liquid fuels. These standards were developed and adopted by ASTM 
International and may be obtained from http://www.astm.org or from the 
ASTM at 100 Barr Harbor Drive, P.O. Box C700, West Conshohocken, PA 
19428-2959.
    The EPA has incorporated by reference SW-846 Method 6010D and SW-
846 Method 6020B in Method 12. Method 6010D covers inductively coupled 
plasma-atomic emission spectrometry (ICP-AES) analysis, and Method 
6020B covers inductively coupled plasma-mass spectrometry (ICP-MS) 
analysis. These methods may be obtained from https://www.epa.gov or 
from the U.S. Environmental Protection Agency, 1200 Pennsylvania Avenue 
NW, Washington, DE 20460.
    The EPA has incorporated by reference Gas Processors Association 
(GPA) 2140 and GPA 2261 in subpart KKKK of part 60, which involve 
procedures for determining the sulfur content of gaseous fuels. The EPA 
also incorporated by reference GPA 2166 and GPA 2174 in subpart KKKK of 
part 60, which involve procedures for obtaining samples from gaseous 
and liquid fuels, respectively. These GPA standards were developed and 
adopted by the Gas Processors Association and may be obtained from 
https://gpamidstream.org/ or from the Gas Processors Association, 6526 
East 60th Street, Tulsa, OK 74145.
    The EPA has incorporated by reference International Organization 
for Standardization (ISO) 10715 in subpart KKKK of part 60. This 
standard involves procedures for obtaining samples from gaseous fuels. 
This standard was developed by the International Organization for 
Standardization and may be obtained from https://www.iso.org/home.html 
or from the IHS Inc., 15 Inverness Way East, Englewood, CO 80112.
    The EPA incorporated by reference American Petroleum Institute 
(API) Manual of Petroleum Measurement Standards, Chapter 14--Natural 
Gas Fluids Measurement, Section 1--Collecting and Handling of Natural 
Gas Samples for Custody Transfer (MPMS 14.1) in subpart KKKK of part 
60. This standard involves procedures for manually obtaining sampling 
from gaseous fuels. This standard was developed by the American 
Petroleum Institute and may be obtained from https://api.org/ or from 
the American Petroleum Institute, 1220 L Street NW, Washington, DC 
20005.
    ASTM D4057-5 (Reapproved 2000), ASTM D4177-95 (Reapproved 2000), 
ASTM D5287-97 (Reapproved 2002), ASTM D6348-03, ASTM D6784-02 
(Reapproved 2008), and ASME PTC 19.10-1981 were previously approved for 
incorporation by reference, and no changes were proposed.
    The EPA updated the ASTM standards referenced in Method 311, but 
these standards are not incorporated by reference. The EPA did not 
update the ASTM standards referenced in Performance Standard 18, which 
are not incorporated by reference.

IV. Summary of Amendments

A. Method 201A of Appendix M of Part 51

    Consistent with our proposal, in Method 201A, section 1.2, the 
erroneous gas filtration temperature limit of 30 [deg]C is revised to 
29.4 [deg]C. In section 1.6, the erroneous word ``recommended'' is 
corrected to ``required.'' Section 6.2.1(d) is revised to allow 
polystyrene petri dishes as an alternative to polyethylene due to the 
lack of commercially available polyethylene petri dishes. The 
polystyrene petri dishes offer similar chemical resistivity to acids 
and inorganics as polyethene and have been shown to transfer extreme 
low residual gravimetric mass to filters when used in ambient air 
applications. In section 8.6.6, the erroneous stack temperature of 
 10 [deg]C is revised to  28 [deg]C. In section 
17.0, the erroneous caption for Figure 7 is corrected from ``Minimum 
Number of Traverse Points for Preliminary Method 4 Traverse'' to 
``Maximum Number of Required Traverse Points,'' and the erroneous y-
axis label is corrected from ``Minimum Number of Traverse Points'' to 
``Maximum Number of Traverse Points.''

B. General Provisions (Subpart A) of Part 60

    Consistent with our proposal, in the General Provisions, 40 CFR 
60.17(h) is revised to add ASTM D2369-10 to the list of incorporations 
by reference and to re-number the remaining consensus standards that 
are incorporated by reference in alpha-numeric order.
    In 40 CFR 60.17(j) is revised to add SW-846-6010D and SW-846-6020B 
to the list of incorporations by reference and to re-number the 
remaining standards that are incorporated by reference in alpha-numeric 
order.
    In 40 CFR 60.17(k) is revised to add GPA Standards 2166-17 and 
2174-14 to the list of incorporations by reference and to re-number the 
remaining GPA standards that are incorporated by reference in alpha-
numeric order.
    In 40 CFR 60.17(l) is revised to add ISO 10715:1997 to the list of 
incorporations by reference.

C. Standards of Performance for New Residential Wood Heaters (Subpart 
AAA) of Part 60

    In 40 CFR 60.534(h), the language is amended based on comments 
received in response to an Advance Notice of Proposed Rulemaking 
(ANPRM), for Standards of Performance for New Residential Wood Heaters, 
New Residential Hydronic Heaters and Forced-Air Furnaces (83 FR 61585, 
November 30, 2018). Several commenters stated that the final clause of 
these existing paragraphs would create loopholes that allow 
manufacturers and test labs to withhold critical testing data. The EPA 
recognizes that this provision was not intended to create an avenue for 
omissions and is clarifying these communications and their reporting.

D. Standards of Performance for Municipal Solid Waste Landfills That 
Commenced Construction, Reconstruction, or Modification After July 17, 
2014 (Subpart XXX) of Part 60

    In 40 CFR 60.766(a)(3), the text for calibration of temperature 
measurement is revised to provide clarity and

[[Page 63396]]

improve the consistency of implementation, as proposed.

E. Standards of Performance for Commercial and Industrial Solid Waste 
Incineration Units (Subpart CCCC) of Part 60

    Consistent with our proposal, Subpart CCCC of Part 60 is revised to 
clarify that (1) initial and annual performance testing for particulate 
matter (PM) for waste-burning kilns and energy recovery units (ERU) is 
to be conducted using Method 5 or Method 29 of Appendix A of Part 60; 
(2) the required particulate matter continuous parameter monitoring 
system (PM CPMS) is used to demonstrate continuing compliance with the 
PM emission limit; and (3) heat input information must be reported for 
each ERU. The current language in 40 CFR 60.2110(i), (i)(1)(iii) and 
60.2145(b), when read together, make it clear that for purposes of 
demonstrating compliance with the PM emission limit, there must be 
initial testing and subsequently, annually and for ongoing continuous 
demonstration of compliance, that data from the compliant performance 
test in turn must be used to set an operating limit for the PM CPMS.
    Paragraphs 60.2110(i)(1) and 60.2145(j) are revised to clarify that 
the PM CPMS coupled with an operating limit is used for continuing 
compliance demonstration with the PM emission limit. Paragraphs 
60.2110(i)(1)(iii) and (i)(2) are revised to include Method 29 as an 
alternative to Method 5 to measure PM in determining compliance with 
the PM emission limit. Paragraph 60.2145(j) is also revised to add PM 
to the list of pollutants for which performance tests are conducted 
annually. Paragraph (p) is added to 40 CFR 60.2210 to require that 
annual reports include the annual heat input and average annual heat 
input rate of all fuels being burned in ERUs in order to verify which 
subcategory of ERU applies.
    The required annual performance test timeframe is changed from 
``between 11 and 13 calendar months following the previous performance 
test'' to ``no later than 13 calendar months following the previous 
performance test'' in paragraphs 60.2145(y)(3) and 60.2150. The current 
2-month testing range can present operational and testing challenges 
for facilities that have multiple commercial and industrial solid waste 
incineration (CISWI) units. In addition, this revision is consistent 
with other rules, such as the National Emission Standards for Hazardous 
Air Pollutants from Hazardous Waste Combustors, that maybe applicable 
to CISWI units.
    Tables 6 (Emission Limitations for Energy Recovery Units) and 7 
(Emission Limitations for Waste-Burning Kilns) are revised to clarify 
the performance test method for PM. The fourth column of the 
``Particulate matter (filterable)'' row of Table 6 is revised to remove 
the requirement to use a PM CPMS as the performance test method for 
large ERU. The fourth column of the ``Particulate matter (filterable)'' 
row of Table 7 is revised to remove the requirement to use a PM CPMS 
and to instead specify Methods 5 and 29 as alternatives for measuring 
PM to determine compliance with the PM limit. The third column of the 
``Particulate matter (filterable)'' row of Table 7 is changed from a 
30-day rolling average to specify a 3-run average with a minimum sample 
volume of 2 dry standard cubic meters (dscm) per run.

F. Emission Guidelines and Compliance Times for Commercial and 
Industrial Solid Waste Incineration Units (Subpart DDDD) of Part 60

    Consistent with our proposal, subpart DDDD of part 60 is revised to 
clarify that (1) initial and annual performance testing for PM for 
waste-burning kilns and ERU is to be conducted using Method 5 or Method 
29 of Appendix A of part 60; (2) the required PM CPMS is used to 
demonstrate continuing compliance with the PM emission limit; and (3) 
heat input information must be reported for ERU. The current language 
in 40 CFR 60.2675(i) and (i)(1)(iii) and 60.2710(b), when read 
together, makes it clear that for purposes of demonstrating compliance 
for PM, performance testing must be used initially and then annually 
while for purposes of ongoing continuous demonstration of compliance, 
data from the compliant performance test is in turn to be used to set 
an operating limit for the PM CPMS.
    Paragraphs 60.2675(i)(1) and 60.2710(j) are revised to clarify that 
the PM CPMS is used for continuing compliance demonstration with the PM 
emission limit. Paragraph 60.2710(j) is also revised to clarify that PM 
performance tests are conducted annually and 40 CFR 60.2675(i)(1)(iii) 
and (i)(2) are revised to include Method 29 as an alternative to Method 
5 to measure PM in determining compliance with the PM emission limit.
    Also, the required annual performance test timeframe is changed 
from ``between 11 and 13 calendar months following the previous 
performance test'' to ``no later than 13 calendar months following the 
previous performance test'' in 40 CFR 60.2710(y)(3) and 60.2715. The 
current 2-month testing range can present operational and testing 
challenges for facilities that have multiple CISWI units. Additionally, 
we note that this revision is consistent with other rules, such as the 
National Emission Standards for Hazardous Air Pollutants from Hazardous 
Waste Combustors that might be applicable to CISWI units.
    Tables 7 (Emission Limitations for Energy Recovery Units) and 8 
(Emission Limitations That Apply to Waste-Burning Kilns) are revised to 
clarify the performance test method for PM. The fourth column of the 
``Particulate matter filterable'' row of Table 7 is revised to remove 
the requirement to use a PM CPMS as the performance test method for 
large ERU. The fourth column of the ``Particulate matter filterable'' 
row of Table 8 is revised to specify Methods 5 and 29 as alternatives 
for measuring PM to determine compliance with the PM emission limit. 
The third column of the ``Particulate matter filterable'' row of Table 
8 is changed from a 30-day rolling average to specify a 3-run average 
with a minimum sample volume of 1 dscm per run.

G. Standards of Performance for Stationary Spark Ignition Internal 
Combustion Engines (Subpart JJJJ) of Part 60

    In Table 2 of subpart JJJJ, text is added to clarify that when 
stack gas flowrate measurements are necessary, they must be made at the 
same time as pollutant concentration measurements unless the option in 
Method 1A is applicable and is being used. This revision is consistent 
with our proposal.

H. Standards of Performance for Stationary Combustion Turbines (Subpart 
KKKK) of Part 60

    As explained at proposal, in 2006, the EPA promulgated the 
combustion turbine criteria pollutant NSPS, subpart KKKK of 40 CFR part 
60 (71 FR 38482, July 6, 2006). This rule, which includes a sulfur 
dioxide (SO2) emissions standard for all fuels, such as 
natural gas, also made provisions to minimize the compliance burden for 
owners/operators of combustion turbines burning natural gas and/or low 
sulfur distillate oil. At the time, the Agency recognized that any 
SO2 testing requirements for owners/operators of combustion 
turbines burning natural gas would result in compliance costs without 
any associated environmental benefit.
    As explained at proposal, the initial and subsequent performance 
tests required in 40 CFR 60.4415 may be satisfied by fuel analyses 
performed by the facility, a contractor, the fuel vendor, or any other 
qualified agency as

[[Page 63397]]

described in 40 CFR 60.4415(a)(1). However, the allowed fuel sample and 
sulfur content measurement methods are not typically used by fuel 
vendors and, as a result, tariff sheets cannot be used without approval 
of an alternate method. We further explained that owner/operators of 
the combustion turbines were now conducting sampling and testing using 
a limited number of test methods, which is a burden that was not 
intended in the original rulemaking.
    To align the rule requirements with the original intent of subpart 
KKKK, the EPA proposed and solicited comment on additional sampling and 
sulfur content measurement methods in order to provide flexibility to 
the regulatory community for purposes of satisfying the SO2 
performance testing requirements. Commenters supported both test 
methods the EPA specifically proposed and test methods the EPA 
solicited comments on as additional compliance options. Commenters also 
stated that the EPA should align the performance testing requirements 
in 40 CFR 60.4415 with the monitoring requirements in 40 CFR 60.4365 
and allow the use of a fuel tariff sheet or contract to satisfy the 
performance testing requirements. Commenters further requested that the 
EPA should allow for the use of the fuel sampling procedures specified 
in section 2.3.1.4 or 2.3.2.4 of appendix D to part 75 to demonstrate 
compliance with the SO2 performance testing requirements. 
The EPA did not receive any comments opposing the proposed amendments.
    In this action, 40 CFR 60.4415(a) is amended, as proposed, to 
include GPA 2166 and ISO 10715 for manual sampling of gaseous fuels and 
GPA 2174 for manual sampling of liquid fuels. In addition, in response 
to comments supporting the EPA's solicitation for comment on additional 
test methods, 40 CFR 60.4415(a) is amended to include API MPMS 14.1 for 
manual sampling of gaseous fuels. In response to comments supporting 
the EPA's solicitation for comment for determining the sulfur content 
of liquid fuels, 40 CFR 60.4415(a) is amended to include ASTM D5623 and 
ASTM D7039. In response to comments supporting the EPA's solicitation 
for comment for determining the sulfur content of gaseous fuels, 40 CFR 
60.4415(a) is amended to include GPA 2140 and GPA 2261. The EPA has 
determined that these additional test methods will provide additional 
flexibility to the regulated community without any emissions increase.
    In addition, in response to comments, the EPA is amending 40 CFR 
60.4415(a) to allow for the use of a purchase contract, tariff sheet, 
or transportation contract for the fuel as an option for demonstrating 
compliance with the SO2 performance testing requirements. 
Also, in response to comments, 40 CFR 60.4415(a) is amended to allow 
for the use of the fuel sampling procedures specified in section 
2.3.1.4 or 2.3.2.4 of appendix D to part 75 to demonstrate compliance 
with the SO2 performance testing requirements. These 
amendments will align the performance testing requirements with the 
monitoring requirements in 40 CFR 60.4365 and are consistent with the 
original intent, including the estimated regulatory burden, of the 
rule. Therefore, the EPA considers these options sufficient to 
demonstrate compliance with subpart KKKK. The Agency notes that this 
approach is consistent with the SO2 performance testing 
requirements in other NSPS (e.g., 40 CFR 60.49b(r) in subpart Db).

I. Standard of Performance for New Residential Wood Heaters, New 
Residential Hydronic Heaters and Forced-Air Furnaces (Subpart QQQQ) of 
Part 60

    In subpart QQQQ, in 40 CFR 60.5476(i), the language is amended 
based on comments received in response to an ANPRM for Standards of 
Performance for New Residential Wood Heaters, New Residential Hydronic 
Heaters and Forced-Air Furnaces (83 FR 61585, November 30, 2018). 
Several commenters stated that the final clause of these existing 
paragraphs would create loopholes that would likely allow manufacturers 
and test labs to withhold critical testing data. The EPA recognizes 
that this provision was not intended to create an avenue for omissions 
and has now clarified these communications and their reporting.

J. Method 4 of Appendix A-3 of Part 60

    In Method 4, the erroneous leak check procedures in section 8.1.3 
are corrected. In response to comments, section 8.1.3.2.1 is revised to 
remove the erroneous probe nozzle language, and section 8.1.3.2.2 is 
revised to remove the erroneous reference to section 8.1.3.2.1. The 
erroneous section 8.1.4.2 is corrected, and in the table in section 
9.1, the erroneous reference to section 8.1.1.4 is replaced with 
section 8.1.3.2.2.
    Method 4 is revised to standardize the constants between Methods 4 
and 5, and more significant digits are added to constants to remove 
rounding and truncation errors. Also, the option for volumetric 
determination of the liquid content is deleted to remove the 
unnecessary density conversion. We believe most method users have moved 
to gravimetric measurement of the liquid contents in order to reduce 
testing costs and increase the accuracy of liquid measurement. 
Revisions occur in various sections (2.1, 6.1.5, 11.1, 11.2, 12.1.1, 
12.1.2, 12.1.3, 12.2.1, and 12.2.2) and Figures 4-4 and 4-5. Also, in 
response to comments, the language in section 8.1.2.1 is revised to be 
consistent with our decision to disallow the option for volumetric 
moisture measurement.

K. Method 5 of Appendix A-3 of Part 60

    In Method 5, sections 6.2.4 and 8.1.2 are revised to allow 
polystyrene petri dishes as an alternative to polyethylene due to the 
lack of commercially available polyethylene petri dishes. The 
polystyrene petri dishes offer similar chemical resistivity to acids 
and inorganics as polyethene and have been shown to transfer extreme 
low residual gravimetric mass to the filters when used in ambient air 
applications.
    Method 5 is also revised to standardize the constants between 
Methods 4 and 5, and more significant digits are added to constants to 
remove rounding and truncation errors. Also, the option for volumetric 
determination of the liquid content is deleted to remove the 
unnecessary density conversion. We believe most method users have moved 
to gravimetric measurement of the liquid contents to lower the cost and 
increase the accuracy of the liquid measurement. Revisions occur in 
various sections (6.1.1.8, 6.2.5, 8.1.2.1, 8.7.6.4, 12.1, 12.3, 12.4, 
12.11.1, 12.11.2, 16.1.1.4, and 16.2.3.3) and in Figure 5-6. All these 
revisions are consistent with the proposal.

L. Method 7C of Appendix A-4 of Part 60

    In Method 7C, in section 7.2.11, the erroneous chemical compound, 
sodium sulfite is corrected to sodium nitrite, as proposed.

M. Method 7E of Appendix A-4 of Part 60

    In Method 7E, section 8.5 is revised to ensure that the specified 
bias and calibration error checks are performed consistently. The 
results of the post-run system bias and calibration error checks are 
used to validate the run, as well as to correct the results of each 
individual test run for bias found in the sampling system. The more 
frequently these checks are performed, the more accurate the bias 
adjusted data will be. All these revisions are consistent with the 
proposal.

[[Page 63398]]

N. Method 12 of Appendix A-5 of Part 60

    In Method 12, sections 7.1.2, 8.7.1.6, 8.7.3.1, and 8.7.3.6 are 
revised to remove references regarding the use of silicone grease, 
which is no longer allowed when conducting Method 5, and section 12.3 
is revised to correctly refer to the title of section 12.4 of Method 5.
    Sections 8.7.3.3 and 12.1 are revised based on a public comment to 
be consistent with the revision to eliminate the option for volumetric 
determination of the liquid content of impingers in Method 5. The 
language in section 8.7.3.3 is revised, and ``[rho]w = 
Density of water, 0.9982 g/ml (0.002201 lb/ml)'' is removed from 
section 12.1.
    Section 16.1 allows measurements of PM emissions in conjunction 
with the lead measurement but does not currently provide enough detail 
to ensure proper PM measurement. Revisions to section 16.1 provide 
testers with necessary procedures to execute PM and lead emissions 
measurements using one sampling train.
    Sections 16.3, 16.4.1, 16.4.2, 16.5, 16.5.1, and 16.5.2 are revised 
to specify appropriate EPA analytical methods, as well as supporting 
quality assurance procedures, as part of allowed alternatives for the 
use of inductively coupled plasma-atomic emission spectrometry (ICP-
AES) and inductively coupled plasma-mass spectrometry (ICP-MS) for 
sample analysis. Section 16.0 currently allows three alternatives to 
the atomic absorption analysis otherwise required in Method 12; 
specifically, ICP-AES in section 16.4, ICP-MS in section 16.5, and cold 
vapor atomic fluorescence spectrometry (CVAFS) in section 16.6. 
Regarding options to use ICP-AES and ICP-MS for analysis of lead, 
sections 16.4 and 16.5 currently do not include either specifics for 
applying these candidate analytical techniques, or procedures for 
assessing data quality. The revisions provide the needed specificity by 
referencing existing EPA methods for ICP-AES and ICP-MS along with 
supporting quality assurance requirements. The option to use CVAFS to 
measure lead (section 16.6) is removed since CVAFS for lead is not 
generally available and there is no existing EPA method for conducting 
it. These revisions are consistent with the proposal.

O. Method 16B of Appendix A-6 of Part 60

    In Method 16B, in section 2.1, the erroneous phrase ``an integrated 
gas sample'' is corrected to ``a gas sample.'' In sections 6.1 and 8.2, 
the reference to section 8.4.1 is changed to 8.3.1 since section 8.4.1 
is renumbered to 8.3.1. The text in section 8.3, ``Analysis. Inject 
aliquots of the sample into the GC/FPD analyzer for analysis. Determine 
the concentration of SO2 directly from the calibration 
curves or from the equation for the least-squares line.'' is moved to 
section 11.1 to be consistent with EPA test method formatting. Sections 
8.4, 8.4.1, and 8.4.2 are renumbered to 8.3, 8.3.1, and 8.3.2, 
respectively, since the text in section 8.3 is moved to section 11.1. 
In section 11.1, the sentence ``Sample collection and analysis are 
concurrent for this method (see section 8.3).'' is deleted. Section 
11.2 is added so that a uniform set of analysis results would be 
obtained over the test period. These revisions are consistent with the 
proposal.

P. Method 16C of Appendix A-6 of Part 60

    In Method 16C, in section 13.1, ``gas concentration'' is replaced 
with ``span'' for clarity, as proposed.

Q. Method 24 of Appendix A-7 of Part 60

    In Method 24, section 6.2, ASTM D 2369-10, which is the most recent 
version of ASTM D 2369, is added as proposed.

R. Method 25C of Appendix A-7 of Part 60

    We proposed to change the correction of non-methane organic 
compounds (NMOC) within the method. Currently, NMOC is to be corrected 
by using either nitrogen or oxygen content. The correction is through 
use of nitrogen unless the nitrogen content exceeds a threshold of 20 
percent. When the nitrogen threshold is above 20 percent, the 
correction is through use of oxygen. We considered multiple options for 
revisions, based on data provided by industry. These options and data 
are available in the docket for this rulemaking, docket ID EPA-HQ-OAR-
2018-0815. The revisions to the correction that we considered are for 
when only oxygen is used as a NMOC correction, setting a rainfall 
threshold in lieu of a nitrogen percent threshold, and requiring a 
methane measurement and using methane only as the correction. We 
provided amendatory text for each option in docket ID EPA-HQ-OAR-2018-
0815. Based on comments we received on proposed options, we are 
finalizing Option 3 with revisions to the ambient air ratio quality 
assurance to alleviate the sampling issues in arid areas. Therefore, 
sections 8.4.2, 9.1, 12.5, 12.5.1, and 12.5.2 are revised.

S. Method 26 of Appendix A-8 of Part 60

    In Method 26, in section 8.1.2, the misspelled word 
``undereporting'' in the next to the last sentence is corrected to 
``under reporting,'' as proposed.

T. Method 26A of Appendix A-8 of Part 60

    In Method 26A, section 6.1.3, a reference to section 6.1.1.7 of 
Method 5 is added to make the filter temperature sensor placement 
consistent with the requirements in Method 5. Also, in section 6.1.3, 
the requirement that the filter temperature sensor must be encased in 
glass or Teflon is added because of the reactive nature of the halogen 
acids. In section 8.1.5, the misspelled word ``undereporting'' is 
corrected to ``under reporting.'' These revisions are consistent with 
the proposal.

U. Performance Specification 4B of Appendix B of Part 60

    In Performance Specification 4B, the response time in section 4.5 
is changed from ``must not exceed 2 minutes'' to ``must not exceed 240 
seconds'' to be consistent with the response time in Performance 
Specification 4A, as proposed.

V. Performance Specification 5 of Appendix B of Part 60

    In Performance Specification 5, section 5.0, the erroneous term 
``users manual'' is replaced with ``user's manual,'' and in the note in 
section 8.1, the sentence ``For Method 16B, you must analyze a minimum 
of three aliquots spaced evenly over the test period.'' is added to 
provide consistency with the number of aliquots analyzed in Method 16B, 
which may be used as the reference method. This revision is consistent 
with the proposal. In addition, the typo, ``space'' in the first 
sentence in the note in section 8.1 is corrected to ``spaced''.

W. Performance Specification 6 of Appendix B of Part 60

    In Performance Specification 6, section 13.1 is revised to clarify 
that the calibration drift test period for the analyzers associated 
with the measurement of flow rate should be the same as that for the 
pollutant analyzer that is part of the continuous emission rate 
monitoring system (CERMS), as proposed. Section 13.2 is revised for 
clarity and to be consistent with the requirements in Performance 
Specification 2, as proposed, and the erroneous reference to 
Performance Specification 1 is corrected to Performance Specification 2 
in response

[[Page 63399]]

to a public comment we received on the proposal.

X. Performance Specification 8 of Appendix B of Part 60

    In Performance Specification 8, a new section 8.3 is added to 
require that an instrument drift check be performed as described in 
Performance Specification 2, and the existing sections 8.3, 8.4, and 
8.5 are re-numbered as 8.4, 8.5, and 8.6, respectively. These revisions 
are consistent with the proposal.

Y. Performance Specification 9 of Appendix B of Part 60

    In Performance Specification 9, the quality control and performance 
audit sections are clarified. In section 7.2, a requirement that 
performance audit gas must be an independent certified gas cylinder or 
cylinder mixture certified by the supplier to be accurate to two 
percent of the tagged value supplied with the cylinder is added.
    In section 8.3, an incorrect reference concerning quality control 
requirements that pertain to the 7-day drift test is clarified and 
corrected, and an incorrect reference to the error calculation equation 
is corrected. In section 8.4, a requirement to ensure that performance 
audit samples challenge the entire sampling system including the sample 
transport lines is added, and quality control requirements that must be 
met for performance audit tests are specified by adding references to 
sections 13.3 and 13.4.
    In section 10.1, the erroneous word ``initial'' is deleted from the 
title, ``Initial Multi-Point Calibration,'' and the quality control 
requirements that must be met for multi-point calibrations are 
specified by referencing sections 13.1 and 13.2 in addition to 13.3. 
Sections 10.1 and 10.2 are clarified such that calibrations may be 
performed at the instrument rather than through the entire sampling 
system. The inadvertently omitted word, ``by'' is inserted in the 
sentence in section 10.2 that reads, ``The average instrument response 
shall not vary more than 10 percent from the certified concentration 
value of the cylinder for each analyte.''
    In section 13.1, language is clarified to ensure that every time a 
triplicate injection is performed, the calibration error must be less 
than or equal to 10 percent of the calibration gas value. In section 
13.2, language is clarified to specify that the linear regression 
correlation coefficient must be determined to evaluate the calibration 
curve for instrument response every time the continuous emission 
monitoring system (CEMS) response is evaluated over multiple 
concentration levels. Section 13.4 is added to describe the quality 
control requirements for the initial and periodic performance audit 
test sample. These revisions are consistent with the proposal.

Z. Performance Specification 18 of Appendix B of Part 60

    In Performance Specification 18, section 2.3 is revised to clarify 
that Method 321 is only applicable to Portland cement plants. Also, in 
section 11.9.1, the reference to Method 321 is deleted because Method 
321 is specific to Portland cement plants, and it is already specified 
in the applicable regulations. These revisions are consistent with the 
proposal.

AA. Procedure 1 of Appendix F of Part 60

    In Procedure 1, section 5.2.3(2), the criteria for cylinder gas 
audits (CGAs) as applicable to diluent monitors is specified for 
clarity, as proposed.

BB. Appendix B to Part 61--Test Methods

    In the index to Appendix B to Part 61, the inadvertently omitted 
Method 114--Test Methods for Measuring Radionuclide Emissions from 
Stationary Sources and Method 115--Monitoring for Radon-222 Emissions 
are added in response to a comment on the proposed rulemaking.

CC. Method 107 of Appendix B of Part 61

    In Method 107, the erroneous Equation 107-3 is corrected by adding 
the omitted plus (+) sign, as proposed.

DD. General Provisions (Subpart A) of Part 63

    In the General Provisions of Part 63, in 40 CFR 63.2, consistent 
with the proposal, the definition of alternative test method is revised 
to exclude ``that is not a test method in this chapter and'' because 
this clarifies that use of methods other than those required by a 
specific subpart requires the alternative test method review and 
approval process.

EE. Portland Cement Manufacturing (Subpart LLL) of Part 63

    In subpart LLL, the units of measurement in Equations 12, 13, 17, 
18, and 19 are revised to add clarity and consistency. Equations 12 and 
13 are corrected so that the operating limit units of measurement is 
calculated correctly. The calculation of the operating limit is 
established by a relationship of the total hydrocarbons (THC) CEMS 
signal to the organic HAPs compliance concentration. As explained at 
proposal, in Table 1 in Part 63, Subpart LLL, the THC and organic HAP 
emissions limits units are in ppmvd corrected to 7 percent oxygen. 
Therefore, the average organic HAP values in equation 12 need to be in 
ppmvd, corrected to 7 percent oxygen, instead of ppmvw. The THC CEMS 
monitor units of measure are ppmvw, as propane and the variables are 
updated to reflect this. The variables in Equations 13 and 19 reference 
variables in Equations 12 and 18, respectively. Those variables are 
updated for consistency between the equations.
    The units of measurement in Equation 17 should be the monitoring 
system's units of measure. It is possible for those systems to be on 
either a wet or a dry basis. Currently, the equation is only on a wet 
basis, even though it should be on the basis of the monitor (wet or 
dry). The changes to the units of measure from ppmvw to ppmv takes 
either possibility into account. For Equations 17 and 18, the operating 
limit units of measure are changed to the units of the CEMS monitor, 
ppmv. These revisions are consistent with the proposal.

FF. Method 301 of Appendix A of Part 63

    In Method 301, section 11.1.3, the erroneous SD in Equation 301-13 
is replaced with SDd, consistent with the proposal.

GG. Method 308 of Appendix A of Part 63

    In Method 308, section 12.4, erroneous Equation 308-3 is corrected, 
and in section 12.5, erroneous Equation 308-5 is corrected, consistent 
with the proposal.

HH. Method 311 of Appendix A of Part 63

    In Method 311, in sections 1.1 and 17, the ASTM is updated. 
Specifically, in section 1.1, ASTM D4747-87 is updated to D4747-02, and 
ASTM D4827-93 is updated to D4827-03. Also, in section 1.1, Provisional 
Standard Test Method, PS 9-94 is replaced with D5910-05. In section 17, 
ASTM D4457-85 is updated to ASTM D4457-02, and ASTM D4827-93 is updated 
to ASTM D4827-03. These updates are consistent with the proposal.

II. Method 315 of Appendix A of Part 63

    In Method 315, in Figure 315-1, an omission is corrected by adding 
a ``not to exceed'' blank criteria for filters used in this test 
procedure. The blank criteria were derived from evaluation of blank and 
spiked filters used to prepare Method 315 audit samples. We set the 
allowable blank correction for filters

[[Page 63400]]

based on the greater of two criteria. The first criterion requires the 
blank to be at least 10 times the measured filter blanks from the audit 
study. The second criterion requires the blank to be at least 5 times 
the resolution of the analytical balance required in Method 315. The 
``not to exceed'' value is, therefore, based on the second criterion 
(balance resolution) because it is the higher of the two criteria. 
These revisions are consistent with the proposal.

JJ. Method 316 of Appendix A of Part 63

    In Method 316, section 1.0, the erroneous positive exponents are 
corrected to negative exponents. Also, the title of section 1.0, 
``Introduction,'' is changed to ``Scope and Application'' to be 
consistent with the Environmental Monitoring Management Council (EMMC) 
format for test methods. These revisions are consistent with the 
proposal.

KK. Method 323 of Appendix A of Part 63

    In the title of Method 323, the misspelled word ``Derivitization'' 
is corrected to ``Derivatization,'' and in section 2.0, the misspelled 
word ``colorietrically'' is corrected to ``colorimetrically.'' These 
revisions are consistent with the proposal.

V. Public Comments on the Proposed Rule

    Eighteen comment letters were received from the public on the 
proposed rulemaking; 13 of the comment letters were relevant, and the 
other five comment letters are considered beyond the scope of the 
proposed rulemaking. The public comments and the agency's responses are 
summarized in the Response to Comments document located in the docket 
for this rule. See the ADDRESSES section of this preamble.

VI. Statutory and Executive Order Reviews

    Additional information about these statutes and Executive Orders 
can be found at http://www2.epa.gov/laws-regulations/laws-and-executive-orders.

A. Executive Order 12866: Regulatory Planning and Review and Executive 
Order 13563: Improving Regulation and Regulatory Review

    This action is not a significant regulatory action and was, 
therefore, not submitted to the Office of Management and Budget (OMB) 
for review.

B. Executive Order 13771: Reducing Regulations and Controlling 
Regulatory Costs

    This action is considered an Executive Order 13771 deregulatory 
action. This final rule provides meaningful burden reduction by 
updating and clarifying test methods and performance specifications, 
thereby improving data quality and by providing source testers 
flexibility by incorporating approved alternative procedures.

C. Paperwork Reduction Act (PRA)

    This action does not impose any information collection burden under 
the PRA. The revisions make corrections and updates to existing testing 
methodology and clarify testing requirements.

D. Regulatory Flexibility Act (RFA)

    I certify that this action will not have a significant economic 
impact on a substantial number of small entities under the RFA. In 
making this determination, the impact of concern is any significant 
adverse economic impact on small entities. An agency may certify that a 
rule will not have a significant economic impact on a substantial 
number of small entities if the rule relieves regulatory burden, has no 
net burden or otherwise has a positive economic effect on the small 
entities subject to the rule. This action will not impose emission 
measurement requirements beyond those specified in the current 
regulations, nor does it change any emission standard. We have, 
therefore, concluded that this action will have no net regulatory 
burden for all directly regulated small entities.

E. Unfunded Mandates Reform Act (UMRA)

    This action does not contain any unfunded mandate as described in 
UMRA, 2 U.S.C. 1531-1538, and does not significantly or uniquely affect 
small governments. The action imposes no enforceable duty on any state, 
local or tribal governments or the private sector.

F. Executive Order 13132: Federalism

    This action does not have federalism implications. It will not have 
substantial direct effects on the states, on the relationship between 
the national government and the states, or on the distribution of power 
and responsibilities among the various levels of government.

G. Executive Order 13175: Consultation and Coordination With Indian 
Tribal Governments

    This action does not have tribal implications, as specified in 
Executive Order 13175. This action simply corrects and updates existing 
testing regulations. Thus, Executive Order 13175 does not apply to this 
action.

H. Executive Order 13045: Protection of Children From Environmental 
Health Risks and Safety Risks

    The EPA interprets Executive Order 13045 as applying only to those 
regulatory actions that concern environmental health or safety risks 
that the EPA has reason to believe may disproportionately affect 
children, per the definition of ``covered regulatory action'' in 
section 2-202 of the Executive Order. This action is not subject to 
Executive Order 13045 because it does not concern an environmental 
health risk or safety risk.

I. Executive Order 13211: Actions Concerning Regulations That 
Significantly Affect Energy Supply, Distribution or Use

    This action is not subject to Executive Order 13211, because it is 
not a significant regulatory action under Executive Order 12866.

J. National Technology Transfer and Advancement Act (NTTAA) and 1 CFR 
Part 51

    This action involves technical standards. The EPA used ASTM D 2369 
in Method 24. The ASTM D 2369 standard covers volatile content of 
coatings. The EPA used (but is not incorporating by reference) ASTM D 
4457, ASTM D 4827, and ASTM D 5910 in Method 311. These ASTM standards 
cover procedures to identify and quantify hazardous air pollutants in 
paints and coatings. The EPA used ASTM D 5623 and ASTM D 7039 in 
subpart KKKK of Part 60. The ASTM D 5623 standard covers the 
determination of sulfur compounds in light petroleum liquids, and the 
ASTM D 7039 standard covers the determination of sulfur in gasoline and 
diesel fuel. The ASTM standards were developed and adopted by the 
American Society for Testing and Materials and may be obtained from 
http://www.astm.org or from the ASTM at 100 Barr Harbor Drive, P.O. Box 
C700, West Conshohocken, PA 19428-2959.
    The EPA used SW-846-6010D and SW-846-6020B in Method 12. Method SW-
846-6010D covers inductively coupled plasma-atomic emission 
spectrometry (ICP-AES) analysis, and Method SW-846-6020B covers 
inductively coupled plasma-mass

[[Page 63401]]

spectrometry (ICP-MS) analysis. These standards were developed and 
adopted by the Environmental Protection Agency and may be obtained from 
http://www.epa.gov or from the U.S. Environmental Protection Agency at 
1200 Pennsylvania Avenue NW, Washington, DC 20460.
    The EPA used API Manual of Petroleum Measurement Standards, Chapter 
14--Natural Gas Fluids Measurement (Section 1) in Subpart KKKK of Part 
60. This API standard involves the collecting and handling of natural 
gas samples for custody transfer. This API standard was developed and 
adopted by the American Petroleum Institute and may be obtained from 
https://www.api.org/ or from the American Petroleum Institute at 1220 L 
Street NW, Washington, DC 20005.
    The EPA used GPA 2166 in Subpart KKKK of Part 60, which involves 
procedures for obtaining samples from gaseous fuels. The EPA used GPA 
2174 in Subpart KKKK of Part 60, which involves procedures for 
obtaining samples from liquid fuels. The EPA used GPA 2140 in subpart 
KKKK of Part 60, which involves liquefied petroleum gas specifications 
and test methods. The EPA used GPA 2261 in subpart KKKK of Part 60, 
which is a procedure for analyzing natural gas and similar gaseous 
mixtures. These GPA standards were developed and adopted by the GPA 
Midstream Association and may be obtained from https://www.gpamidstream.org/ or from the GPA Midstream Association, Sixty 
Sixty American Plaza, Suite 700, Tulsa, OK 74135.
    The EPA used ISO 10715 in subpart KKKK of Part 60. This standard 
involves procedures for obtaining samples from gaseous fuels. This 
standard was developed by the International Organization for 
Standardization and may be obtained from https://www.iso.org/home.html 
or from the ISH Inc., 15 Inverness Way East, Englewood, CO 80112.
    Multiple ASTM and GPA standards were previously approved on July 6, 
2006, and are already included in the regulatory text. Therefore, the 
current the IBR is unchanged in this rule for the following methods: 
ASTM D129-00, ASTM D1072-90 (Reapproved 1999); ASTM D1266-98 
(Reapproved 2003)[egr],1; ASTM D1552-03, ASTM D2622-05, ASTM D3246-05, 
ASTM D4057-95 (Reapproved 2000), ASTM D4084-05, ASTM D4177-95 
(Reapproved 2000); ASTM D4294-03, ASTM D4468-85 (Reapproved 2000); ASTM 
D4810-88 (Reapproved 1999); ASTM D5287-97 (Reapproved 2002); ASTM 
D5453-05, ASTM D6228-98 (Reapproved 2003); ASTM D6667-04, and GPA 2377-
86.

K. Executive Order 12898: Federal Actions To Address Environmental 
Justice in Minority Populations and Low-Income Populations

    The EPA believes that this action is not subject to Executive Order 
12898 (59 FR 7629, February 16, 1994) because it does not establish an 
environmental health or safety standard. This action is a technical 
correction to previously promulgated regulatory actions and does not 
have an impact on human health or the environment.

L. Congressional Review Act (CRA)

    This action is subject to the CRA, and the EPA will submit a rule 
report to each house of the Congress and to the Comptroller General of 
the United States. This action is not a ``major rule'' as defined by 5 
U.S.C. 804(2).

List of Subjects

40 CFR Part 51

    Environmental protection, Air pollution control, Performance 
specifications, Test methods and procedures.

40 CFR Part 60

    Environmental protection, Air pollution control, Incorporation by 
reference, Performance specifications, Test methods and procedures.

40 CFR Parts 61 and 63

    Environmental protection, Air pollution control, Incorporation by 
reference, Performance specifications, Test methods and procedures.

Andrew Wheeler,
Administrator.

    For the reasons set forth in the preamble, the EPA amends 40 CFR 
parts 51, 60, 61, and 63 as follows:

PART 51--REQUIREMENTS FOR PREPARATION, ADOPTION, AND SUBMITTAL OF 
IMPLEMENTATION PLANS

0
1. The authority citation for part 51 continues to read as follows:

    Authority:  23 U.S.C. 101; 42 U.S.C. 7401-7671q.

0
2. In appendix M to part 51, in Method 201A, revise sections ``1.2'', 
``1.6'', ``6.2.1(d)'', and ``8.6.6'' and ``Figure 7'' to read as 
follows:

Appendix M to Part 51--Recommended Test Methods for State 
Implementation Plans

* * * * *

Method 201A--Determination of PM10 and PM2.5 
Emissions From Stationary Sources (Constant Sampling Rate Procedure)

* * * * *
    1.2 Applicability. This method addresses the equipment, 
preparation, and analysis necessary to measure filterable PM. You 
can use this method to measure filterable PM from stationary sources 
only. Filterable PM is collected in stack with this method (i.e., 
the method measures materials that are solid or liquid at stack 
conditions). If the gas filtration temperature exceeds 29.4 [deg]C 
(85 [deg]F), then you may use the procedures in this method to 
measure only filterable PM (material that does not pass through a 
filter or a cyclone/filter combination). If the gas filtration 
temperature exceeds 29.4 [deg]C (85 [deg]F), and you must measure 
both the filterable and condensable (material that condenses after 
passing through a filter) components of total primary (direct) PM 
emissions to the atmosphere, then you must combine the procedures in 
this method with the procedures in Method 202 of appendix M to this 
part for measuring condensable PM. However, if the gas filtration 
temperature never exceeds 29.4 [deg]C (85 [deg]F), then use of 
Method 202 of appendix M to this part is not required to measure 
total primary PM.
* * * * *
    1.6 Conditions. You can use this method to obtain particle 
sizing at 10 micrometers and or 2.5 micrometers if you sample within 
80 and 120 percent of isokinetic flow. You can also use this method 
to obtain total filterable particulate if you sample within 90 to 
110 percent of isokinetic flow, the number of sampling points is the 
same as required by Method 5 of appendix A-3 to part 60 or Method 17 
of appendix A-6 to part 60, and the filter temperature is within an 
acceptable range for these methods. For Method 5, the acceptable 
range for the filter temperature is generally 120 [deg]C (248 
[deg]F) unless a higher or lower temperature is specified. The 
acceptable range varies depending on the source, control technology 
and applicable rule or permit condition. To satisfy Method 5 
criteria, you may need to remove the in-stack filter and use an out-
of-stack filter and recover the PM in the probe between the 
PM2.5 particle sizer and the filter. In addition, to 
satisfy Method 5 and Method 17 criteria, you may need to sample from 
more than 12 traverse points. Be aware that this method determines 
in-stack PM10 and PM2.5 filterable emissions 
by sampling from a required maximum of 12 sample points, at a 
constant flow rate through the train (the constant flow is necessary 
to maintain the size cuts of the cyclones), and with a filter that 
is at the stack temperature. In contrast, Method 5 or Method 17 
trains are operated isokinetically with varying flow rates through 
the train. Method 5 and Method 17 require sampling from as many as 
24 sample points. Method 5 uses an out-of-stack filter that is 
maintained at a constant temperature of 120 [deg]C (248 [deg]F). 
Further, to use this method in place of Method 5 or Method 17, you 
must extend the sampling time so that you collect the minimum mass 
necessary for weighing each portion of this sampling train. Also, if 
you are using this method as an alternative to a test method 
specified in a regulatory

[[Page 63402]]

requirement (e.g., a requirement to conduct a compliance or 
performance test), then you must receive approval from the authority 
that established the regulatory requirement before you conduct the 
test.
* * * * *
    6.2.1 * * *
    (d) Petri dishes. For filter samples; glass, polystyrene, or 
polyethylene, unless otherwise specified by the Administrator.
* * * * *
    8.6.6 Sampling Head. You must preheat the combined sampling head 
to the stack temperature of the gas stream at the test location 
(28 [deg]C, 50 [deg]F). This will heat the 
sampling head and prevent moisture from condensing from the sample 
gas stream.
* * * * *
    17.0 * * *
BILLING CODE 6560-50-P
[GRAPHIC] [TIFF OMITTED] TR07OC20.003

BILLING CODE 6560-50-C

PART 60--STANDARDS OF PERFORMANCE FOR NEW STATIONARY SOURCES

0
3. The authority citation for part 60 continues to read as follows:

    Authority:  42 U.S.C. 7401 et seq.

0
4. Amend Sec.  60.17 by:
0
a. Removing the text ``appendix A-8 to part 60: Method 24,'' and add in 
its place, ``appendix A-7 to part 60: Method 24,'' everywhere it 
appears;
0
b. Revising the last sentence in paragraph (a);
0
c. Redesignating paragraph (e)(2) as (e)(3) and adding a new paragraph 
(e)(2);
0
d. Redesignating paragraphs (h)(192) through (209) as (h)(195) through 
(212), (h)(174) through (191) as (h)(176) through (193), and (h)(95) 
through (173) as (h)(96) through (174), respectively;
0
e. Adding new paragraphs (h)(95), (175), and (194);
0
f. Adding paragraphs (j)(3) and (4);
0
g. Revising paragraph (k) introductory text;
0
h. Redesignating paragraphs (k)(2) and (3) as paragraphs (k)(5) and (6) 
and redesignating paragraph (k)(1) as paragraph (k)(3), respectively;
0
i. Adding new paragraphs (k)(1), (2), and (4);
0
j. Revising newly redesignated paragraph (k)(5); and
0
k. Adding paragraph (l)(2).
    The revisions and additions read as follows:

Sec.  60.17  Incorporations by reference.

    (a) * * * For information on the availability of this material at 
NARA,

[[Page 63403]]

email fedreg.legal@nara.gov, or go to www.archives.gov/federal-register/cfr/ibr-locations.html.
* * * * *
    (e) * * *
    (2) API Manual of Petroleum Measurement Standards, Chapter 14--
Natural Gas Fluids Measurement, Section 1--Collecting and Handling of 
Natural Gas Samples for Custody Transfer, 7th Edition, May 2016, IBR 
approved for Sec.  60.4415(a).
* * * * *
    (h) * * *
    (95) ASTM D2369-10 (Reapproved 2015)e1, Standard Test Method for 
Volatile Content of Coatings, (Approved June 1, 2015); IBR approved for 
appendix A-7 to part 60: Method 24, Section 6.2.
* * * * *
    (175) ASTM D5623-19, Standard Test Method for Sulfur Compounds in 
Light Petroleum Liquids by Gas Chromatography and Sulfur Selective 
Detection, (Approved July 1, 2019); IBR approved for Sec.  60.4415(a).
* * * * *
    (194) ASTM D7039-15a, Standard Test Method for Sulfur in Gasoline, 
Diesel Fuel, Jet Fuel, Kerosine, Boideisel, Biodiesel Blends, and 
Gasoline-Ethanol Blends by Monochromatic Wavelength Dispersive X-ray 
Fluorescence Spectrometry, (Approved July 1, 2015); IBR approved for 
Sec.  60.4415(a).
* * * * *
    (j) * * *
    (3) SW-846-6010D, Inductively Coupled Plasma-Optical Emission 
Spectrometry, Revision 5, July 2018, in EPA Publication No. SW-846, 
Test Methods for Evaluating Solid Waste, Physical/Chemical Methods, 
Third Edition, IBR approved for appendix A-5 to part 60: Method 12.
    (4) SW-846-6020B, Inductively Coupled Plasma-Mass Spectrometry, 
Revision 2, July 2014, in EPA Publication No. SW-846, Test Methods for 
Evaluating Solid Waste, Physical/Chemical Methods, Third Edition, IBR 
approved for appendix A-5 to part 60: Method 12.
    (k) GPA Midstream Association (formerly known as Gas Processors 
Association), Sixty Sixty American Plaza, Suite 700, Tulsa, OK 74135.
    Note 1 to paragraph (k): Material in this paragraph that is no 
longer available from GPA may be available through the reseller HIS 
Markit, 15 Inverness Way East, P.O. Box 1154, Englewood, CO 80150-1154, 
https://global.ihs.com/. For material that is out-of-print, contact 
EPA's Air and Radiation Docket and Information Center, Room 3334, 1301 
Constitution Ave. NW, Washington, DC 20460 or a-and-rdocket@epa.gov.
    (1) GPA Midstream Standard 2140-17 (GPA 2140-17), Liquefied 
Petroleum Gas Specifications and Test Methods, (Revised 2017), IBR 
approved for Sec.  60.4415(a).
    (2) GPA Midstream Standard 2166-17 (GPA 2166-17), Obtaining Natural 
Gas Samples for Analysis by Gas Chromatography, (Reaffirmed 2017), IBR 
approved for Sec.  60.4415(a).
* * * * *
    (4) GPA Standard 2174-14 (GPA 2174-14), Obtaining Liquid 
Hydrocarbon Samples for Analysis by Gas Chromatography, (Revised 2014), 
IBR approved for Sec.  60.4415(a).
    (5) GPA Standard 2261-19 (GPA 2261-19), Analysis for Natural Gas 
and Similar Gaseous Mixtures by Gas Chromatography, (Revised 2019), IBR 
approved for Sec.  60.4415(a).
* * * * *
    (l) * * *
    (2) ISO 10715:1997(E), Natural gas--Sampling guidelines, (First 
Edition, June 1, 1997), IBR approved for Sec.  60.4415(a)
* * * * *

Subpart AAA--Standards of Performance for New Residential Wood 
Heaters

0
5. Amend Sec.  60.534 by revising paragraph (h) to read as follows:

Sec.  60.534  What test methods and procedures must I use to determine 
compliance with the standards and requirements for certification?

* * * * *
    (h) The approved test laboratory must allow the manufacturer, the 
manufacturer's approved third-party certifier, the EPA and delegated 
state regulatory agencies to observe certification testing. However, 
manufacturers must not involve themselves in the conduct of the test 
after the pretest burn has begun. Communications between the 
manufacturer and laboratory or third-party certifier personnel 
regarding operation of the wood heater must be limited to written 
communications transmitted prior to the first pretest burn of the 
certification test series. During certification tests, the manufacturer 
may communicate with the third-party certifier, and only in writing, to 
notify them that the manufacturer has observed a deviation from proper 
test procedures by the laboratory. All communications must be included 
in the test documentation required to be submitted pursuant to Sec.  
60.533(b)(5) and must be consistent with instructions provided in the 
owner's manual required under Sec.  60.536(g).

Subpart XXX--Standards of Performance for Municipal Solid Waste 
Landfills That Commenced Construction, Reconstruction, or 
Modification After July 17, 2014

0
6. Amend Sec.  60.766 by revising paragraph (a)(3) to read as follows:

Sec.  60.766  Monitoring of operations.

* * * * *
    (a) * * *
    (3) Monitor temperature of the landfill gas on a monthly basis as 
provided in 60.765(a)(5). The temperature measuring device must be 
calibrated annually using the procedure in 40 CFR part 60, appendix A-
1, Method 2, section 10.3 such that a minimum of two temperature 
points, bracket within 10 percent of all landfill absolute temperature 
measurements or two fixed points of ice bath and boiling water, 
corrected for barometric pressure, are used.
* * * * *

Subpart CCCC--Standards of Performance for Commercial and 
Industrial Solid Waste Incineration Units

0
7. Amend Sec.  60.2110 by revising paragraphs (i) introductory text, 
(i)(1), and (i)(2) introductory text to read as follows:

Sec.  60.2110  What operating limits must I meet and by when?

* * * * *
    (i) If you use a PM CPMS to demonstrate continuing compliance, you 
must establish your PM CPMS operating limit and determine compliance 
with it according to paragraphs (i)(1) through (5) of this section:
    (1) Determine your operating limit as the average PM CPMS output 
value recorded during the performance test or at a PM CPMS output value 
corresponding to 75 percent of the emission limit if your PM 
performance test demonstrates compliance below 75 percent of the 
emission limit. You must verify an existing or establish a new 
operating limit after each repeated performance test. You must repeat 
the performance test annually and reassess and adjust the site-specific 
operating limit in accordance with the results of the performance test:
    (i) Your PM CPMS must provide a 4-20 milliamp output, or digital 
equivalent, and the establishment of its relationship to manual 
reference

[[Page 63404]]

method measurements must be determined in units of milliamps;
    (ii) Your PM CPMS operating range must be capable of reading PM 
concentrations from zero to a level equivalent to at least two times 
your allowable emission limit. If your PM CPMS is an auto-ranging 
instrument capable of multiple scales, the primary range of the 
instrument must be capable of reading PM concentration from zero to a 
level equivalent to two times your allowable emission limit; and
    (iii) During the initial performance test or any such subsequent 
performance test that demonstrates compliance with the PM limit, record 
and average all milliamp output values, or their digital equivalent, 
from the PM CPMS for the periods corresponding to the compliance test 
runs (e.g., average all your PM CPMS output values for three 
corresponding Method 5 or Method 29 test runs).
    (2) If the average of your three PM performance test runs are below 
75 percent of your PM emissions limit, you must calculate an operating 
limit by establishing a relationship of PM CPMS signal to PM 
concentration using the PM CPMS instrument zero, the average PM CPMS 
output values corresponding to the three compliance test runs, and the 
average PM concentration from the Method 5 or Method 29 performance 
test with the procedures in (i)(1) through (5) of this section:
* * * * *

0
8. Amend Sec.  60.2145 by revising paragraphs (j) introductory text and 
(y)(3) to read as follows:

Sec.  60.2145  How do I demonstrate continuous compliance with the 
emission limitations and the operating limits?

* * * * *
    (j) For waste-burning kilns, you must conduct an annual performance 
test for particulate matter, cadmium, lead, carbon monoxide, dioxins/
furans and hydrogen chloride as listed in Table 7 of this subpart, 
unless you choose to demonstrate initial and continuous compliance 
using CEMS, as allowed in paragraph (u) of this section. If you do not 
use an acid gas wet scrubber or dry scrubber, you must determine 
compliance with the hydrogen chloride emissions limit using a HCl CEMS 
according to the requirements in paragraph (j)(1) of this section. You 
must determine compliance with the mercury emissions limit using a 
mercury CEMS or an integrated sorbent trap monitoring system according 
to paragraph (j)(2) of this section. You must determine compliance with 
nitrogen oxides and sulfur dioxide using CEMS. You must determine 
continuing compliance with the particulate matter emissions limit using 
a PM CPMS according to paragraph (x) of this section.
* * * * *
    (y) * * *
    (3) For purposes of determining the combined emissions from kilns 
equipped with an alkali bypass or that exhaust kiln gases to a coal 
mill that exhausts through a separate stack, instead of installing a 
CEMS or PM CPMS on the alkali bypass stack or in-line coal mill stack, 
the results of the initial and subsequent performance test can be used 
to demonstrate compliance with the relevant emissions limit. A 
performance test must be conducted on an annual basis (no later than 13 
calendar months following the previous performance test).

0
9. Revise Sec.  60.2150 to read as follows:

Sec.  60.2150  By what date must I conduct the annual performance test?

    You must conduct annual performance tests no later than 13 calendar 
months following the previous performance test.

0
10. Amend Sec.  60.2210 by revising the introductory text and adding 
paragraph (p) to read as follows:

Sec.  60.2210  What information must I include in my annual report?

    The annual report required under Sec.  60.2205 must include the 
items listed in paragraphs (a) through (p) of this section. If you have 
a deviation from the operating limits or the emission limitations, you 
must also submit deviation reports as specified in Sec. Sec.  60.2215, 
60.2220, and 60.2225:
* * * * *
    (p) For energy recovery units, include the annual heat input and 
average annual heat input rate of all fuels being burned in the unit to 
verify which subcategory of energy recovery unit applies.

0
11. Table 6 to subpart CCCC of part 60 is revised to read as follows:

 Table 6 to Subpart CCCC of Part 60--Emission Limitations for Energy Recovery Units That Commenced Construction
            After June 4, 2010, or That Commenced Reconstruction or Modification After August 7, 2013
----------------------------------------------------------------------------------------------------------------
                               You must meet this emission limitation
                                                 \1\                         Using this        And determining
    For the air pollutant    ------------------------------------------  averaging time \2\    compliance using
                                   Liquid/gas             Solids                               this method \2\
----------------------------------------------------------------------------------------------------------------
Cadmium.....................  0.023 milligrams     Biomass--0.0014      3-run average        Performance test
                               per dry standard     milligrams per dry   (collect a minimum   (Method 29 at 40
                               cubic meter.         standard cubic       volume of 4 dry      CFR part 60,
                                                    meter. Coal--        standard cubic       appendix A-8). Use
                                                    0.0017 milligrams    meters per run).     ICPMS for the
                                                    per dry standard                          analytical finish.
                                                    cubic meter.
Carbon monoxide.............  35 parts per         Biomass--240 parts   3-run average (1     Performance test
                               million dry volume.  per million dry      hour minimum         (Method 10 at 40
                                                    volume. Coal--95     sample time per      CFR part 60,
                                                    parts per million    run).                appendix A-4).
                                                    dry volume.
Dioxin/furans (Total Mass     No Total Mass Basis  Biomass--0.52        3-run average        Performance test
 Basis).                       limit, must meet     nanograms per dry    (collect a minimum   (Method 23 at 40
                               the toxic            standard cubic       volume of 4 dry      CFR part 60,
                               equivalency basis    meter. Coal--5.1     standard cubic       appendix A-7).
                               limit below.         nanograms per dry    meters).
                                                    standard cubic
                                                    meter.
Dioxins/furans (toxic         0.093 nanograms per  Biomass--0.076       3-run average        Performance test
 equivalency basis).           dry standard cubic   nanograms per dry    (collect a minimum   (Method 23 of
                               meter.               standard cubic       volume of 4 dry      appendix A-7 of
                                                    meter.\3\ Coal--     standard cubic       this part).
                                                    0.075 nanograms      meters per run).
                                                    per dry standard
                                                    cubic meter.
Fugitive ash................  Visible emissions    Three 1-hour         Visible emission     Fugitive ash.
                               for no more than 5   observation          test (Method 22 at
                               percent of the       periods.             40 CFR part 60,
                               hourly observation                        appendix A-7).
                               period.
Hydrogen chloride...........  14 parts per         Biomass--0.20 parts  3-run average (For   Performance test
                               million dry volume.  per million dry      Method 26, collect   (Method 26 or 26A
                                                    volume. Coal--58     a minimum volume     at 40 CFR part 60,
                                                    parts per million    of 360 liters per    appendix A-8).
                                                    dry volume.          run. For Method
                                                                         26A, collect a
                                                                         minimum volume of
                                                                         3 dry standard
                                                                         cubic meters per
                                                                         run).

[[Page 63405]]

 
Lead........................  0.096 milligrams     Biomass--0.014       3-run average        Performance test
                               per dry standard     milligrams per dry   (collect a minimum   (Method 29 at 40
                               cubic meter.         standard cubic       volume of 4 dry      CFR part 60,
                                                    meter. Coal--0.057   standard cubic       appendix A-8). Use
                                                    milligrams per dry   meters per run).     ICPMS for the
                                                    standard cubic                            analytical finish.
                                                    meter.
Mercury.....................  0.00056 milligrams   Biomass--0.0022      3-run average        Performance test
                               per dry standard     milligrams per dry   (collect enough      (Method 29 or 30B
                               cubic meter.         standard cubic       volume to meet an    at 40 CFR part 60,
                                                    meter. Coal--0.013   in-stack detection   appendix A-8) or
                                                    milligrams per dry   limit data quality   ASTM D6784-02
                                                    standard cubic       objective of 0.03    (Reapproved
                                                    meter.               ug/dscm).            2008).\3\
Nitrogen oxides.............  76 parts per         Biomass--290 parts   3-run average (for   Performance test
                               million dry volume.  per million dry      Method 7E, 1 hour    (Method 7 or 7E at
                                                    volume. Coal--460    minimum sample       40 CFR part 60,
                                                    parts per million    time per run).       appendix A-4).
                                                    dry volume.
Particulate matter            110 milligrams per   Biomass--5.1         3-run average        Performance test
 (filterable).                 dry standard cubic   milligrams per dry   (collect a minimum   (Method 5 or 29 at
                               meter.               standard cubic       volume of 1 dry      40 CFR part 60,
                                                    meter. Coal--130     standard cubic       appendix A-3 or
                                                    milligrams per dry   meter per run).      appendix A-8).
                                                    standard cubic
                                                    meter.
Sulfur dioxide..............  720 parts per        Biomass--7.3 parts   3-run average (for   Performance test
                               million dry volume.  per million dry      Method 6, collect    (Method 6 or 6C at
                                                    volume. Coal--850    a minimum of 60      40 CFR part 60,
                                                    parts per million    liters, for Method   appendix A-4).
                                                    dry volume.          6C, 1 hour minimum
                                                                         sample time per
                                                                         run).
----------------------------------------------------------------------------------------------------------------
\1\ All emission limitations are measured at 7 percent oxygen, dry basis at standard conditions. For dioxins/
  furans, you must meet either the Total Mass Basis limit or the toxic equivalency basis limit.
\2\ In lieu of performance testing, you may use a CEMS or, for mercury, an integrated sorbent trap monitoring
  system to demonstrate initial and continuing compliance with an emissions limit, as long as you comply with
  the CEMS or integrated sorbent trap monitoring system requirements applicable to the specific pollutant in
  Sec.  Sec.   60.2145 and 60.2165. As prescribed in Sec.   60.2145(u), if you use a CEMS or an integrated
  sorbent trap monitoring system to demonstrate compliance with an emissions limit, your averaging time is a 30-
  day rolling average of 1-hour arithmetic average emission concentrations.
\3\ Incorporated by reference, see Sec.   60.17.

0
12. Table 7 to subpart CCCC of part 60 is revised to read as follows:

  Table 7 to Subpart CCCC of Part 60--Emission Limitations for Waste-Burning Kilns That Commenced Construction
                   After June 4, 2010, or Reconstruction or Modification After August 7, 2013
----------------------------------------------------------------------------------------------------------------
                                                                                            And determining
     For the air pollutant          You must meet this      Using this averaging time    compliance using this
                                  emission limitation \1\              \2\                    method 2, 3
----------------------------------------------------------------------------------------------------------------
Cadmium.......................  0.0014 milligrams per dry   3-run average (collect a   Performance test (Method
                                 standard cubic meter.       minimum volume of 4 dry    29 at 40 CFR part 60,
                                                             standard cubic meters      appendix A-8). Use ICPMS
                                                             per run).                  for the analytical
                                                                                        finish.
Carbon monoxide...............  90 (long kilns)/190         3-run average (1 hour      Performance test (Method
                                 (preheater/precalciner)     minimum sample time per    10 at 40 CFR part 60,
                                 parts per million dry       run).                      appendix A-4).
                                 volume.
Dioxins/furans (total mass      0.51 nanograms per dry      3-run average (collect a   Performance test (Method
 basis).                         standard cubic meter.       minimum volume of 4 dry    23 at 40 CFR part 60,
                                                             standard cubic meters      appendix A-7).
                                                             per run).
Dioxins/furans (toxic           0.075 nanograms per dry     3-run average (collect a   Performance test (Method
 equivalency basis).             standard cubic meter.       minimum volume of 4 dry    23 at 40 CFR part 60,
                                                             standard cubic meters).    appendix A-7).
Hydrogen chloride.............  3.0 parts per million dry   3-run average (1 hour      If a wet scrubber or dry
                                 volume.                     minimum sample time per    scrubber is used,
                                                             run) or 30-day rolling     performance test (Method
                                                             average if HCl CEMS is     321 at 40 CFR part 63,
                                                             being used.                appendix A). If a wet
                                                                                        scrubber or dry scrubber
                                                                                        is not used, HCl CEMS as
                                                                                        specified in Sec.
                                                                                        60.2145(j).
Lead..........................  0.014 milligrams per dry    3-run average (collect a   Performance test (Method
                                 standard cubic meter.       minimum volume of 4 dry    29 at 40 CFR part 60,
                                                             standard cubic meters).    appendix A-8). Use ICPMS
                                                                                        for the analytical
                                                                                        finish.
Mercury.......................  0.0037 milligrams per dry   30-day rolling average...  Mercury CEMS or
                                 standard cubic meter. Or                               integrated sorbent trap
                                 21 pounds/million tons of                              monitoring system
                                 clinker \3\.                                           (performance
                                                                                        specification 12A or
                                                                                        12B, respectively, of
                                                                                        appendix B and procedure
                                                                                        5 of appendix F of this
                                                                                        part), as specified in
                                                                                        Sec.   60.2145(j).
Nitrogen oxides...............  200 parts per million dry   30-day rolling average...  Nitrogen oxides CEMS
                                 volume.                                                (performance
                                                                                        specification 2 of
                                                                                        appendix B and procedure
                                                                                        1 of appendix F of this
                                                                                        part).
Particulate matter              4.9 milligrams per dry      3-run average (collect a   Performance test (Method
 (filterable).                   standard cubic meter.       minimum volume of 2 dry    5 or 29 at 40 CFR part
                                                             standard cubic meters).    60, appendix A-3 or
                                                                                        appendix-8).
Sulfur dioxide................  28 parts per million dry    30-day rolling average...  Sulfur dioxide CEMS
                                 volume.                                                (performance
                                                                                        specification 2 of
                                                                                        appendix B and procedure
                                                                                        1 of appendix F of this
                                                                                        part).
----------------------------------------------------------------------------------------------------------------
\1\ All emission limitations are measured at 7 percent oxygen (except for CEMS and integrated sorbent trap
  monitoring system data during startup and shutdown), dry basis at standard conditions. For dioxins/furans, you
  must meet either the Total Mass Basis limit or the toxic equivalency basis limit.
\2\ In lieu of performance testing, you may use a CEMS or, for mercury, an integrated sorbent trap monitoring
  system, to demonstrate initial and continuing compliance with an emissions limit, as long as you comply with
  the CEMS or integrated sorbent trap monitoring system requirements applicable to the specific pollutant in
  Sec.  Sec.   60.2145 and 60.2165. As prescribed in Sec.   60.2145(u), if you use a CEMS or integrated sorbent
  trap monitoring system to demonstrate compliance with an emissions limit, your averaging time is a 30-day
  rolling average of 1-hour arithmetic average emission concentrations.
\3\ Alkali bypass and in-line coal mill stacks are subject to performance testing only, as specified in Sec.
  60.2145(y)(3). They are not subject to the CEMS, integrated sorbent trap monitoring system, or CPMS
  requirements that otherwise may apply to the main kiln exhaust.

[[Page 63406]]

Subpart DDDD--Emission Guidelines and Compliance Times for 
Commercial and Industrial Solid Waste Incineration Units

0
13. Amend Sec.  60.2675 by revising the introductory text to paragraphs 
(i) introductory text, (i)(1), and (i)(2) introductory text to read as 
follows:

Sec.  60.2675  What operating limits must I meet and by when?

* * * * *
    (i) If you use a PM CPMS to demonstrate continuing compliance, you 
must establish your PM CPMS operating limit and determine compliance 
with it according to paragraphs (i)(1) through (5) of this section:
    (1) During the initial performance test or any such subsequent 
performance test that demonstrates compliance with the PM limit, record 
all hourly average output values (milliamps, or the digital signal 
equivalent) from the PM CPMS for the periods corresponding to the test 
runs (e.g., three 1-hour average PM CPMS output values for three 1-hour 
test runs):
    (i) Your PM CPMS must provide a 4-20 milliamp output, or the 
digital signal equivalent, and the establishment of its relationship to 
manual reference method measurements must be determined in units of 
milliamps or digital bits;
    (ii) Your PM CPMS operating range must be capable of reading PM 
concentrations from zero to a level equivalent to at least two times 
your allowable emission limit. If your PM CPMS is an auto-ranging 
instrument capable of multiple scales, the primary range of the 
instrument must be capable of reading PM concentration from zero to a 
level equivalent to two times your allowable emission limit; and
    (iii) During the initial performance test or any such subsequent 
performance test that demonstrates compliance with the PM limit, record 
and average all milliamp output values, or their digital equivalent, 
from the PM CPMS for the periods corresponding to the compliance test 
runs (e.g., average all your PM CPMS output values for the three 
corresponding Method 5 or Method 29 p.m. test runs).
    (2) If the average of your three PM performance test runs are below 
75 percent of your PM emission limit, you must calculate an operating 
limit by establishing a relationship of PM CPMS signal to PM 
concentration using the PM CPMS instrument zero, the average PM CPMS 
output values corresponding to the three compliance test runs, and the 
average PM concentration from the Method 5 or Method 29 performance 
test with the procedures in (i)(1)through (5) of this section:
* * * * *

0
14. Amend Sec.  60.2710 by revising paragraphs (j) introductory text 
and (y)(3) to read as follows:

Sec.  60.2710  How do I demonstrate continuous compliance with the 
amended emission limitations and the operating limits?

* * * * *
    (j) For waste-burning kilns, you must conduct an annual performance 
test for the pollutants (except mercury and hydrogen chloride if no 
acid gas wet scrubber or dry scrubber is used) listed in Table 8 of 
this subpart, unless you choose to demonstrate initial and continuous 
compliance using CEMS, as allowed in paragraph (u) of this section. If 
you do not use an acid gas wet scrubber or dry scrubber, you must 
determine compliance with the hydrogen chloride emissions limit using a 
HCl CEMS according to the requirements in paragraph (j)(1) of this 
section. You must determine compliance with the mercury emissions limit 
using a mercury CEMS or an integrated sorbent trap monitoring system 
according to paragraph (j)(2) of this section. You must determine 
continuing compliance with particulate matter using a PM CPMS according 
to paragraph (x) of this section.
* * * * *
    (y) * * *
    (3) For purposes of determining the combined emissions from kilns 
equipped with an alkali bypass or that exhaust kiln gases to a coal 
mill that exhausts through a separate stack, instead of installing a 
CEMS or PM CPMS on the alkali bypass stack or in-line coal mill stack, 
the results of the initial and subsequent performance test can be used 
to demonstrate compliance with the relevant emissions limit. A 
performance test must be conducted on an annual basis (no later than 13 
calendar months following the previous performance test).

0
15. Revise Sec.  60.2715 to read as follows:

Sec.  60.2715  By what date must I conduct the annual performance test?

    You must conduct annual performance tests no later than 13 calendar 
months following the previous performance test.
0
16. Table 7 to subpart DDDD of part 60 is revised to read as follows:

 Table 7 to Subpart DDDD of Part 60--Model Rule--Emission Limitations That Apply to Energy Recovery Units After
                                                  May 20, 2011
                                     [Date to be specified in state plan] 1
----------------------------------------------------------------------------------------------------------------
                               You must meet this emission limitation
                                                 \2\                         Using this        And determining
    For the air pollutant    ------------------------------------------  averaging time \3\    compliance using
                                   Liquid/gas             Solids                               this method \3\
----------------------------------------------------------------------------------------------------------------
Cadmium.....................  0.023 milligrams     Biomass--0.0014      3-run average        Performance test
                               per dry standard     milligrams per dry   (collect a minimum   (Method 29 at 40
                               cubic meter.         standard cubic       volume of 2 dry      CFR part 60,
                                                    meter. Coal--        standard cubic       appendix A-8). Use
                                                    0.0017 milligrams    meters).             ICPMS for the
                                                    per dry standard                          analytical finish.
                                                    cubic meter.
Carbon monoxide.............  35 parts per         Biomass--260 parts   3-run average (1     Performance test
                               million dry volume.  per million dry      hour minimum         (Method 10 at 40
                                                    volume. Coal--95     sample time per      CFR part 60,
                                                    parts per million    run).                appendix A-4).
                                                    dry volume.
Dioxins/furans (total mass    2.9 nanograms per    Biomass--0.52        3-run average        Performance test
 basis).                       dry standard cubic   nanograms per dry    (collect a minimum   (Method 23 at 40
                               meter.               standard cubic       volume of 4 dry      CFR part 60,
                                                    meter. Coal--5.1     standard cubic       appendix A-7).
                                                    nanograms per dry    meter).
                                                    standard cubic
                                                    meter.
Dioxins/furans (toxic         0.32 nanograms per   Biomass--0.12        3-run average        Performance test
 equivalency basis).           dry standard cubic   nanograms per dry    (collect a minimum   (Method 23 at 40
                               meter.               standard cubic       volume of 4 dry      CFR part 60,
                                                    meter. Coal--0.075   standard cubic       appendix A-7).
                                                    nanograms per dry    meters).
                                                    standard cubic
                                                    meter.
Hydrogen chloride...........  14 parts per         Biomass--0.20 parts  3-run average (for   Performance test
                               million dry volume.  per million dry      Method 26, collect   (Method 26 or 26A
                                                    volume. Coal--58     a minimum of 120     at 40 CFR part 60,
                                                    parts per million    liters; for Method   appendix A-8).
                                                    dry volume.          26A, collect a
                                                                         minimum volume of
                                                                         1 dry standard
                                                                         cubic meter).

[[Page 63407]]

 
Lead........................  0.096 milligrams     Biomass--0.014       3-run average        Performance test
                               per dry standard     milligrams per dry   (collect a minimum   (Method 29 at 40
                               cubic meter.         standard cubic       volume of 2 dry      CFR part 60,
                                                    meter. Coal--0.057   standard cubic       appendix A-8). Use
                                                    milligrams per dry   meters).             ICPMS for the
                                                    standard cubic                            analytical finish.
                                                    meter.
Mercury.....................  0.0024 milligrams    Biomass--0.0022      3-run average (For   Performance test
                               per dry standard     milligrams per dry   Method 29 and ASTM   (Method 29 or 30B
                               cubic meter.         standard cubic       D6784-02             at 40 CFR part 60,
                                                    meter. Coal--0.013   (Reapproved 2008)    appendix A-8) or
                                                    milligrams per dry   \4\, collect a       ASTM D6784-02
                                                    standard cubic       minimum volume of    (Reapproved
                                                    meter.               2 dry standard       2008).\4\
                                                                         cubic meters per
                                                                         run. For Method
                                                                         30B, collect a
                                                                         minimum sample as
                                                                         specified in
                                                                         Method 30B at 40
                                                                         CFR part 60,
                                                                         appendix A).
Nitrogen oxides.............  76 parts per         Biomass--290 parts   3-run average (for   Performance test
                               million dry volume.  per million dry      Method 7E, 1 hour    (Method 7 or 7E at
                                                    volume. Coal--460    minimum sample       40 CFR part 60,
                                                    parts per million    time per run).       appendix A-4).
                                                    dry volume.
Particulate matter            110 milligrams per   Biomass--11          3-run average        Performance test
 filterable.                   dry standard cubic   milligrams per dry   (collect a minimum   (Method 5 or 29 at
                               meter.               standard cubic       volume of 1 dry      40 CFR part 60,
                                                    meter. Coal--130     standard cubic       appendix A-3 or
                                                    milligrams per dry   meter).              appendix A-8).
                                                    standard cubic
                                                    meter.
Sulfur dioxide..............  720 parts per        Biomass--7.3 parts   3-run average (1     Performance test
                               million dry volume.  per million dry      hour minimum         (Method 6 or 6c at
                                                    volume. Coal--850    sample time per      40 CFR part 60,
                                                    parts per million    run).                appendix A-4).
                                                    dry volume.
Fugitive ash................  Visible emissions    Visible emissions    Three 1-hour         Visible emission
                               for no more than 5   for no more than 5   observation          test (Method 22 at
                               percent of the       percent of the       periods.             40 CFR part 60,
                               hourly observation   hourly observation                        appendix A-7).
                               period.              period.
----------------------------------------------------------------------------------------------------------------
\1\ The date specified in the state plan can be no later than 3 years after the effective date of approval of a
  revised state plan or February 7, 2018.
\2\ All emission limitations (except for opacity) are measured at 7 percent oxygen, dry basis at standard
  conditions. For dioxins/furans, you must meet either the total mass basis limit or the toxic equivalency basis
  limit.
\3\ In lieu of performance testing, you may use a CEMS or, for mercury, an integrated sorbent trap monitoring
  system, to demonstrate initial and continuing compliance with an emissions limit, as long as you comply with
  the CEMS or integrated sorbent trap monitoring system requirements applicable to the specific pollutant in
  Sec.  Sec.   60.2710 and 60.2730. As prescribed in Sec.   60.2710(u), if you use a CEMS or integrated sorbent
  trap monitoring system to demonstrate compliance with an emissions limit, your averaging time is a 30-day
  rolling average of 1-hour arithmetic average emission concentrations.
\4\ Incorporated by reference, see Sec.   60.17.

0
17. Table 8 to subpart DDDD of part 60 is revised to read as follows:

Table 8 to Subpart DDDD of Part 60--Model Rule--Emission Limitations That Apply to Waste-Burning Kilns After May
                                                    20, 2011
                                     [Date to be specified in state plan] 1
----------------------------------------------------------------------------------------------------------------
                                                                                            And determining
     For the air pollutant          You must meet this      Using this averaging time    compliance using this
                                  emission limitation \2\              \3\                     method 3 4
----------------------------------------------------------------------------------------------------------------
Cadmium.......................  0.0014 milligrams per dry   3-run average (collect a   Performance test (Method
                                 standard cubic meter.       minimum volume of 2 dry    29 at 40 CFR part 60,
                                                             standard cubic meters).    appendix A-8).
Carbon monoxide...............  110 (long kilns)/790        3-run average (1 hour      Performance test (Method
                                 (preheater/precalciner)     minimum sample time per    10 at 40 CFR part 60,
                                 parts per million dry       run).                      appendix A-4).
                                 volume.
Dioxins/furans (total mass      1.3 nanograms per dry       3-run average (collect a   Performance test (Method
 basis).                         standard cubic meter.       minimum volume of 4 dry    23 at 40 CFR part 60,
                                                             standard cubic meters).    appendix A-7).
Dioxins/furans (toxic           0.075 nanograms per dry     3-run average (collect a   Performance test (Method
 equivalency basis).             standard cubic meter.       minimum volume of 4 dry    23 at 40 CFR part 60,
                                                             standard cubic meters).    appendix A-7).
Hydrogen chloride.............  3.0 parts per million dry   3-run average (collect a   If a wet scrubber or dry
                                 volume.                     minimum volume of 1 dry    scrubber is used,
                                                             standard cubic meter),     performance test (Method
                                                             or 30-day rolling          321 at 40 CFR part 63,
                                                             average if HCl CEMS is     appendix A of this
                                                             being used.                part). If a wet scrubber
                                                                                        or dry scrubber is not
                                                                                        used, HCl CEMS as
                                                                                        specified in Sec.
                                                                                        60.2710(j).
Lead..........................  0.014 milligrams per dry    3-run average (collect a   Performance test (Method
                                 standard cubic meter.       minimum volume of 2 dry    29 at 40 CFR part 60,
                                                             standard cubic meters).    appendix A-8).
Mercury.......................  0.011 milligrams per dry    30-day rolling average...  Mercury CEMS or
                                 standard cubic meter. Or                               integrated sorbent trap
                                 58 pounds/million tons of                              monitoring system
                                 clinker.                                               (performance
                                                                                        specification 12A or
                                                                                        12B, respectively, of
                                                                                        appendix B and procedure
                                                                                        5 of appendix F of this
                                                                                        part), as specified in
                                                                                        Sec.   60.2710(j).
Nitrogen oxides...............  630 parts per million dry   3-run average (for Method  Performance test (Method
                                 volume.                     7E, 1 hour minimum         7 or 7E at 40 CFR part
                                                             sample time per run).      60, appendix A-4).
Particulate matter filterable.  13.5 milligrams per dry     3-run average (collect a   Performance test (Method
                                 standard cubic meter.       minimum volume of 1 dry    5 or 29 at 40 CFR part
                                                             standard cubic meter).     60, appendix A-3 or
                                                                                        appendix-8).

[[Page 63408]]

 
Sulfur dioxide................  600 parts per million dry   3-run average (for Method  Performance test (Method
                                 volume.                     6, collect a minimum of    6 or 6c at 40 CFR part
                                                             20 liters; for Method      60, appendix A-4).
                                                             6C, 1 hour minimum
                                                             sample time per run).
----------------------------------------------------------------------------------------------------------------
\1\ The date specified in the state plan can be no later than 3 years after the effective date of approval of a
  revised state plan or February 7, 2018.
\2\ All emission limitations are measured at 7 percent oxygen (except for CEMS and integrated sorbent trap
  monitoring system data during startup and shutdown), dry basis at standard conditions. For dioxins/furans, you
  must meet either the total mass basis limit or the toxic equivalency basis limit.
\3\ In lieu of performance testing, you may use a CEMS or, for mercury, an integrated sorbent trap monitoring
  system, to demonstrate initial and continuing compliance with an emissions limit, as long as you comply with
  the CEMS or integrated sorbent trap monitoring system requirements applicable to the specific pollutant in
  Sec.  Sec.   60.2710 and 60.2730. As prescribed in Sec.   60.2710(u), if you use a CEMS or integrated sorbent
  trap monitoring system to demonstrate compliance with an emissions limit, your averaging time is a 30-day
  rolling average of 1-hour arithmetic average emission concentrations.
\4\ Alkali bypass and in-line coal mill stacks are subject to performance testing only, as specified in Sec.
  60.2710(y)(3). They are not subject to the CEMS, integrated sorbent trap monitoring system, or CPMS
  requirements that otherwise may apply to the main kiln exhaust.

Subpart JJJJ--Standards of Performance for Stationary Spark 
Ignition Internal Combustion Engines

0
18. Table 2 to subpart JJJJ of part 60 is revised to read as follows:
    As stated in Sec.  60.4244, you must comply with the following 
requirements for performance tests within 10 percent of 100 percent 
peak (or the highest achievable) load].

                     Table 2 to Subpart JJJJ of Part 60--Requirements for Performance Tests
----------------------------------------------------------------------------------------------------------------
                                                                                               According to the
            For each              Complying with the       You must              Using             following
                                     requirement to                                              requirements
----------------------------------------------------------------------------------------------------------------
1. Stationary SI internal         a. Limit the        i. Select the       (1) Method 1 or 1A  (a) Alternatively,
 combustion engine demonstrating   concentration of    sampling port       of 40 CFR part      for NOX, O2, and
 compliance according to Sec.      NOX in the          location and the    60, appendix A-1,   moisture
 60.4244.                          stationary SI       number/location     if measuring flow   measurement,
                                   internal            of traverse         rate.               ducts <=6 inches
                                   combustion engine   points at the                           in diameter may
                                   exhaust.            exhaust of the                          be sampled at a
                                                       stationary                              single point
                                                       internal                                located at the
                                                       combustion                              duct centroid and
                                                       engine;                                 ducts >6 and <=12
                                                                                               inches in
                                                                                               diameter may be
                                                                                               sampled at 3
                                                                                               traverse points
                                                                                               located at 16.7,
                                                                                               50.0, and 83.3%
                                                                                               of the
                                                                                               measurement line
                                                                                               (`3-point long
                                                                                               line'). If the
                                                                                               duct is >12
                                                                                               inches in
                                                                                               diameter and the
                                                                                               sampling port
                                                                                               location meets
                                                                                               the two and half-
                                                                                               diameter
                                                                                               criterion of
                                                                                               Section 11.1.1 of
                                                                                               Method 1 of 40
                                                                                               CFR part 60,
                                                                                               Appendix A, the
                                                                                               duct may be
                                                                                               sampled at `3-
                                                                                               point long line';
                                                                                               otherwise,
                                                                                               conduct the
                                                                                               stratification
                                                                                               testing and
                                                                                               select sampling
                                                                                               points according
                                                                                               to Section 8.1.2
                                                                                               of Method 7E of
                                                                                               40 CFR part 60,
                                                                                               Appendix A.
                                                      ii. Determine the   (2) Method 3, 3A,   (b) Measurements
                                                       O2 concentration    or 3B \b\ of 40     to determine O2
                                                       of the stationary   CFR part 60,        concentration
                                                       internal            appendix A-2 or     must be made at
                                                       combustion engine   ASTM Method D6522-  the same time as
                                                       exhaust at the      00 (Reapproved      the measurements
                                                       sampling port       2005) a d.          for NOX
                                                       location;                               concentration.
                                                      iii. If necessary,  (3) Method 2 or 2C  (c) Measurements
                                                       determine the       of 40 CFR part      to determine the
                                                       exhaust flowrate    60, appendix A-1    exhaust flowrate
                                                       of the stationary   or Method 19 of     must be made (1)
                                                       internal            40 CFR part 60,     at the same time
                                                       combustion engine   appendix A-7.       as the
                                                       exhaust;                                measurement for
                                                                                               NOX concentration
                                                                                               or, alternatively
                                                                                               (2) according to
                                                                                               the option in
                                                                                               Section 11.1.2 of
                                                                                               Method 1A of 40
                                                                                               CFR part 60,
                                                                                               Appendix A-1, if
                                                                                               applicable.
                                                      iv. If necessary,   (4) Method 4 of 40  (d) Measurements
                                                       measure moisture    CFR part 60,        to determine
                                                       content of the      appendix A-3,       moisture must be
                                                       stationary          Method 320 of 40    made at the same
                                                       internal            CFR part 63,        time as the
                                                       combustion engine   appendix A,\e\ or   measurement for
                                                       exhaust at the      ASTM Method D6348-  NOX
                                                       sampling port       03 d e.             concentration.
                                                       location; and
                                                      v. Measure NOX at   (5) Method 7E of    (e) Results of
                                                       the exhaust of      40 CFR part 60,     this test consist
                                                       the stationary      appendix A-4,       of the average of
                                                       internal            ASTM Method D6522-  the three 1-hour
                                                       combustion          00 (Reapproved      or longer runs.
                                                       engine; if using    2005),a d Method
                                                       a control device,   320 of 40 CFR
                                                       the sampling site   part 63, appendix
                                                       must be located     A,\e\ or ASTM
                                                       at the outlet of    Method D6348-03 d
                                                       the control         e.
                                                       device

[[Page 63409]]

 
                                  b. Limit the        i. Select the       (1) Method 1 or 1A  (a) Alternatively,
                                   concentration of    sampling port       of 40 CFR part      for CO, O2, and
                                   CO in the           location and the    60, appendix A-1,   moisture
                                   stationary SI       number/location     if measuring flow   measurement,
                                   internal            of traverse         rate.               ducts <=6 inches
                                   combustion engine   points at the                           in diameter may
                                   exhaust.            exhaust of the                          be sampled at a
                                                       stationary                              single point
                                                       internal                                located at the
                                                       combustion                              duct centroid and
                                                       engine;                                 ducts >6 and <=12
                                                                                               inches in
                                                                                               diameter may be
                                                                                               sampled at 3
                                                                                               traverse points
                                                                                               located at 16.7,
                                                                                               50.0, and 83.3%
                                                                                               of the
                                                                                               measurement line
                                                                                               (`3-point long
                                                                                               line'). If the
                                                                                               duct is >12
                                                                                               inches in
                                                                                               diameter and the
                                                                                               sampling port
                                                                                               location meets
                                                                                               the two and half-
                                                                                               diameter
                                                                                               criterion of
                                                                                               Section 11.1.1 of
                                                                                               Method 1 of 40
                                                                                               CFR part 60,
                                                                                               Appendix A, the
                                                                                               duct may be
                                                                                               sampled at `3-
                                                                                               point long line';
                                                                                               otherwise,
                                                                                               conduct the
                                                                                               stratification
                                                                                               testing and
                                                                                               select sampling
                                                                                               points according
                                                                                               to Section 8.1.2
                                                                                               of Method 7E of
                                                                                               40 CFR part 60,
                                                                                               Appendix A.
                                                      ii. Determine the   (2) Method 3, 3A,   (b) Measurements
                                                       O2 concentration    or 3B \b\ of 40     to determine O2
                                                       of the stationary   CFR part 60,        concentration
                                                       internal            appendix A-2 or     must be made at
                                                       combustion engine   ASTM Method D6522-  the same time as
                                                       exhaust at the      00 (Reapproved      the measurements
                                                       sampling port       2005) a d.          for CO
                                                       location;                               concentration.
                                                      iii. If necessary,  (3) Method 2 or 2C  (c) Measurements
                                                       determine the       of 40 CFR 60,       to determine the
                                                       exhaust flowrate    appendix A-1 or     exhaust flowrate
                                                       of the stationary   Method 19 of 40     must be made (1)
                                                       internal            CFR part 60,        at the same time
                                                       combustion engine   appendix A-7.       as the
                                                       exhaust;                                measurement for
                                                                                               CO concentration
                                                                                               or, alternatively
                                                                                               (2) according to
                                                                                               the option in
                                                                                               Section 11.1.2 of
                                                                                               Method 1A of 40
                                                                                               CFR part 60,
                                                                                               Appendix A-1, if
                                                                                               applicable.
                                                      iv. If necessary,   (4) Method 4 of 40  (d) Measurements
                                                       measure moisture    CFR part 60,        to determine
                                                       content of the      appendix A-3,       moisture must be
                                                       stationary          Method 320 of 40    made at the same
                                                       internal            CFR part 63,        time as the
                                                       combustion engine   appendix A,\e\ or   measurement for
                                                       exhaust at the      ASTM Method D6348-  CO concentration.
                                                       sampling port       03 d e.
                                                       location; and
                                                      v. Measure CO at    (5) Method 10 of    (e) Results of
                                                       the exhaust of      40 CFR part 60,     this test consist
                                                       the stationary      appendix A4, ASTM   of the average of
                                                       internal            Method D6522-00     the three 1-hour
                                                       combustion          (Reapproved         or longer runs.
                                                       engine; if using    2005),a d e
                                                       a control device,   Method 320 of 40
                                                       the sampling site   CFR part 63,
                                                       must be located     appendix A,\e\ or
                                                       at the outlet of    ASTM Method D6348-
                                                       the control         03 d e.
                                                       device
                                  c. Limit the        i. Select the       (1) Method 1 or 1A  (a) Alternatively,
                                   concentration of    sampling port       of 40 CFR part      for VOC, O2, and
                                   VOC in the          location and the    60, appendix A-1,   moisture
                                   stationary SI       number/location     if measuring flow   measurement,
                                   internal            of traverse         rate.               ducts <=6 inches
                                   combustion engine   points at the                           in diameter may
                                   exhaust.            exhaust of the                          be sampled at a
                                                       stationary                              single point
                                                       internal                                located at the
                                                       combustion                              duct centroid and
                                                       engine;                                 ducts >6 and <=12
                                                                                               inches in
                                                                                               diameter may be
                                                                                               sampled at 3
                                                                                               traverse points
                                                                                               located at 16.7,
                                                                                               50.0, and 83.3%
                                                                                               of the
                                                                                               measurement line
                                                                                               (`3-point long
                                                                                               line'). If the
                                                                                               duct is >12
                                                                                               inches in
                                                                                               diameter and the
                                                                                               sampling port
                                                                                               location meets
                                                                                               the two and half-
                                                                                               diameter
                                                                                               criterion of
                                                                                               Section 11.1.1 of
                                                                                               Method 1 of 40
                                                                                               CFR part 60,
                                                                                               Appendix A, the
                                                                                               duct may be
                                                                                               sampled at `3-
                                                                                               point long line';
                                                                                               otherwise,
                                                                                               conduct the
                                                                                               stratification
                                                                                               testing and
                                                                                               select sampling
                                                                                               points according
                                                                                               to Section 8.1.2
                                                                                               of Method 7E of
                                                                                               40 CFR part 60,
                                                                                               Appendix A.
                                                      ii. Determine the   (2) Method 3, 3A,   (b) Measurements
                                                       O2 concentration    or 3B \b\ of 40     to determine O2
                                                       of the stationary   CFR part 60,        concentration
                                                       internal            appendix A-2 or     must be made at
                                                       combustion engine   ASTM Method D6522-  the same time as
                                                       exhaust at the      00 (Reapproved      the measurements
                                                       sampling port       2005) a d.          for VOC
                                                       location;                               concentration.
                                                      iii. If necessary,  (3) Method 2 or 2C  (c) Measurements
                                                       determine the       of 40 CFR 60,       to determine the
                                                       exhaust flowrate    appendix A-1 or     exhaust flowrate
                                                       of the stationary   Method 19 of 40     must be made (1)
                                                       internal            CFR part 60,        at the same time
                                                       combustion engine   appendix A-7.       as the
                                                       exhaust;                                measurement for
                                                                                               VOC concentration
                                                                                               or, alternatively
                                                                                               (2) according to
                                                                                               the option in
                                                                                               Section 11.1.2 of
                                                                                               Method 1A of 40
                                                                                               CFR part 60,
                                                                                               Appendix A-1, if
                                                                                               applicable.

[[Page 63410]]

 
                                                      iv. If necessary,   (4) Method 4 of 40  (d) Measurements
                                                       measure moisture    CFR part 60,        to determine
                                                       content of the      appendix A-3,       moisture must be
                                                       stationary          Method 320 of 40    made at the same
                                                       internal            CFR part 63,        time as the
                                                       combustion engine   appendix A,\e\ or   measurement for
                                                       exhaust at the      ASTM Method D6348-  VOC
                                                       sampling port       03 d e.             concentration.
                                                       location; and
                                                      v. Measure VOC at   (5) Methods 25A     (e) Results of
                                                       the exhaust of      and 18 of 40 CFR    this test consist
                                                       the stationary      part 60,            of the average of
                                                       internal            appendices A-6      the three 1-hour
                                                       combustion          and A-7, Method     or longer runs.
                                                       engine; if using    25A with the use
                                                       a control device,   of a hydrocarbon
                                                       the sampling site   cutter as
                                                       must be located     described in 40
                                                       at the outlet of    CFR 1065.265,
                                                       the control         Method 18 of 40
                                                       device              CFR part 60,
                                                                           appendix A-6,c e
                                                                           Method 320 of 40
                                                                           CFR part 63,
                                                                           appendix A,\e\ or
                                                                           ASTM Method D6348-
                                                                           03 d e.
----------------------------------------------------------------------------------------------------------------
\a\ Also, you may petition the Administrator for approval to use alternative methods for portable analyzer.
\b\ You may use ASME PTC 19.10-1981, Flue and Exhaust Gas Analyses, for measuring the O2 content of the exhaust
  gas as an alternative to EPA Method 3B. AMSE PTC 19.10-1981 incorporated by reference, see 40 CFR 60.17
\c\ You may use EPA Method 18 of 40 CFR part 60, appendix A-6, provided that you conduct an adequate pre-survey
  test prior to the emissions test, such as the one described in OTM 11 on EPA's website (http://www.epa.gov/ttn/emc/prelim/otm11.pdf).
\d\ Incorporated by reference; see 40 CFR 60.17.
\e\ You must meet the requirements in Sec.   60.4245(d).

Subpart KKKK--Standards of Performance for Stationary Combustion 
Turbines

0
19. Amend Sec.  60.4415 by revising paragraph (a) introductory text, 
redesignating paragraphs (a)(1) through (3) as paragraphs (a)(2) 
through (4), adding new paragraph (a)(1), and revising the newly 
redesignated paragraph (a)(2) to read as follows:

Sec.  60.4415  How do I conduct the initial and subsequent performance 
tests for sulfur?

    (a) You must conduct an initial performance test, as required in 
Sec.  60.8. Subsequent SO2 performance tests shall be 
conducted on an annual basis (no more than 14 calendar months following 
the previous performance test). There are four methodologies that you 
may use to conduct the performance tests.
    (1) The use of a current, valid purchase contract, tariff sheet, or 
transportation contract for the fuel specifying the maximum total 
sulfur content of all fuels combusted in the affected facility. 
Alternately, the fuel sampling data specified in section 2.3.1.4 or 
2.3.2.4 of appendix D to part 75 of this chapter may be used.
    (2) Periodically determine the sulfur content of the fuel combusted 
in the turbine, a representative fuel sample may be collected either by 
an automatic sampling system or manually. For automatic sampling, 
follow ASTM D5287 (incorporated by reference, see Sec.  60.17) for 
gaseous fuels or ASTM D4177 (incorporated by reference, see Sec.  
60.17) for liquid fuels. For manual sampling of gaseous fuels, follow 
API Manual of Petroleum Measurement Standards, Chapter 14, Section 1, 
GPA 2166, or ISO 10715 (all incorporated by reference, see Sec.  
60.17). For manual sampling of liquid fuels, follow GPA 2174 or the 
procedures for manual pipeline sampling in section 14 of ASTM D4057 
(both incorporated by reference, see Sec.  60.17). The fuel analyses of 
this section may be performed either by you, a service contractor 
retained by you, the fuel vendor, or any other qualified agency. 
Analyze the samples for the total sulfur content of the fuel using:
    (i) For liquid fuels, ASTM D129, or alternatively D1266, D1552, 
D2622, D4294, D5453, D5623, or D7039 (all incorporated by reference, 
see Sec.  60.17); or
    (ii) For gaseous fuels, ASTM D1072, or alternatively D3246, D4084, 
D4468, D4810, D6228, D6667, or GPA 2140, 2261, or 2377 (all 
incorporated by reference, see Sec.  60.17).
* * * * *

Subpart QQQQ--Standards of Performance for New Residential Hydronic 
Heaters and Forced-Air Furnaces

0
20. Amend Sec.  60.5476 by revising paragraph (i) to read as follows:

Sec.  60.5476  What test methods and procedures must I use to determine 
compliance with the standards and requirements for certification?

* * * * *
    (i) The approved test laboratory must allow the manufacturer, the 
manufacturer's approved third-party certifier, the EPA and delegated 
state regulatory agencies to observe certification testing. However, 
manufacturers must not involve themselves in the conduct of the test 
after the pretest burn has begun. Communications between the 
manufacturer and laboratory or third-party certifier personnel 
regarding operation of the central heater must be limited to written 
communications transmitted prior to the first pretest burn of the 
certification test series. During certification tests, the manufacturer 
may communicate with the third-party certifier, and only in writing to 
notify them that the manufacturer has observed a deviation from proper 
test procedures by the laboratory. All communications must be included 
in the test documentation required to be submitted pursuant to Sec.  
60.5475(b)(5) and must be consistent with instructions provided in the 
owner's manual required under Sec.  60.5478(f).

0
21. Amend Appendix A-3 to part 60 by:
0
a. In Method 4, revising sections ``2.1'', ``6.1.5'', ``8.1.2.1'', 
``8.1.3'', ``8.1.3.2.1'', ``8.1.3.2.2'', ``8.1.4.2'', ``9.1'', 
``11.1'', ``11.2'', ``12.1.1'', ``12.1.2'', ``12.1.3'', ``12.2.1'', and 
``12.2.2'' and ``Figure 4-4'' and ``Figure 4-5''; and
0
b. In Method 5, revising sections ``6.1.1.8'', ``6.2.4'', ``6.2.5'', 
``8.1.2'', ``8.7.6.4'', ``12.1'', ``12.3'', ``12.4'', ``12.11.1'', 
``12.11.2'', ``16.1.1.4'', and ``16.2.3.3'' and ``Figure 5-6''.
    The revisions read as follows:

Appendix A-3 to Part 60--Test Methods 4 Through 5I

* * * * *

Method 4--Determination of Moisture Content in Stack Gases

* * * * *
    2.1 A gas sample is extracted at a constant rate from the 
source; moisture is removed from the sample stream and determined 
gravimetrically.
* * * * *
    6.1.5 Barometer and Balance. Same as Method 5, sections 6.1.2 
and 6.2.5, respectively.
* * * * *

[[Page 63411]]

    8.1.2.1 Transfer water into the first two impingers, leave the 
third impinger empty and add silica gel to the fourth impinger. 
Weigh the impingers before sampling and record the weight to the 
nearest 0.5g at a minimum.
* * * * *
    8.1.3 Leak-Check Procedures.
    8.1.3.1 Leak Check of Metering System Shown in Figure 4-1. That 
portion of the sampling train from the pump to the orifice meter 
should be leak-checked prior to initial use and after each shipment. 
Leakage after the pump will result in less volume being recorded 
than is actually sampled. The following procedure is suggested (see 
Figure 5-2 of Method 5): Close the main valve on the meter box. 
Insert a one-hole rubber stopper with rubber tubing attached into 
the orifice exhaust pipe. Disconnect and vent the low side of the 
orifice manometer. Close off the low side orifice tap. Pressurize 
the system to 13 to 18 cm (5 to 7 in.) water column by blowing into 
the rubber tubing. Pinch off the tubing and observe the manometer 
for one minute. A loss of pressure on the manometer indicates a leak 
in the meter box; leaks, if present, must be corrected.
    8.1.3.2 Pretest Leak Check. A pretest leak check of the sampling 
train is recommended, but not required. If the pretest leak check is 
conducted, the following procedure should be used.
    8.1.3.2.1 After the sampling train has been assembled, turn on 
and set the filter and probe heating systems to the desired 
operating temperatures. Allow time for the temperatures to 
stabilize.
    8.1.3.2.2 Leak-check the train by first plugging the inlet to 
the filter holder and pulling a 380 mm (15 in.) Hg vacuum. Then 
connect the probe to the train, and leak-check at approximately 25 
mm (1 in.) Hg vacuum; alternatively, the probe may be leak-checked 
with the rest of the sampling train, in one step, at 380 mm (15 in.) 
Hg vacuum. Leakage rates in excess of 4 percent of the average 
sampling rate or 0.00057 m\3\/min (0.020 cfm), whichever is less, 
are unacceptable.
    8.1.3.2.3 Start the pump with the bypass valve fully open and 
the coarse adjust valve completely closed. Partially open the coarse 
adjust valve, and slowly close the bypass valve until the desired 
vacuum is reached. Do not reverse the direction of the bypass valve, 
as this will cause water to back up into the filter holder. If the 
desired vacuum is exceeded, either leak-check at this higher vacuum, 
or end the leak check and start over.
    8.1.3.2.4 When the leak check is completed, first slowly remove 
the plug from the inlet to the probe, filter holder, and immediately 
turn off the vacuum pump. This prevents the water in the impingers 
from being forced backward into the filter holder and the silica gel 
from being entrained backward into the third impinger.
    8.1.3.3 Leak Checks During Sample Run. If, during the sampling 
run, a component (e.g., filter assembly or impinger) change becomes 
necessary, a leak check shall be conducted immediately before the 
change is made. The leak check shall be done according to the 
procedure outlined in section 8.1.3.2 above, except that it shall be 
done at a vacuum equal to or greater than the maximum value recorded 
up to that point in the test. If the leakage rate is found to be no 
greater than 0.00057 m\3\/min (0.020 cfm) or 4 percent of the 
average sampling rate (whichever is less), the results are 
acceptable, and no correction will need to be applied to the total 
volume of dry gas metered; if, however, a higher leakage rate is 
obtained, either record the leakage rate and plan to correct the 
sample volume as shown in section 12.3 of Method 5, or void the 
sample run.
    Note: Immediately after component changes, leak checks are 
optional. If such leak checks are done, the procedure outlined in 
section 8.1.3.2 above should be used.
    8.1.3.4 Post-Test Leak Check. A leak check of the sampling train 
is mandatory at the conclusion of each sampling run. The leak check 
shall be performed in accordance with the procedures outlined in 
section 8.1.3.2, except that it shall be conducted at a vacuum equal 
to or greater than the maximum value reached during the sampling 
run. If the leakage rate is found to be no greater than 0.00057 m\3\ 
min (0.020 cfm) or 4 percent of the average sampling rate (whichever 
is less), the results are acceptable, and no correction need be 
applied to the total volume of dry gas metered. If, however, a 
higher leakage rate is obtained, either record the leakage rate and 
correct the sample volume as shown in section 12.3 of Method 5 or 
void the sampling run.
* * * * *
    8.1.4.2 At the end of the sample run, close the coarse adjust 
valve, remove the probe and nozzle from the stack, turn off the 
pump, record the final DGM meter reading, and conduct a post-test 
leak check, as outlined in section 8.1.3.4.
* * * * *
    9.1 Miscellaneous Quality Control Measures.

----------------------------------------------------------------------------------------------------------------
                  Section                         Quality control measure                    Effect
----------------------------------------------------------------------------------------------------------------
Section 8.1.3.2.2..........................  Leak rate of the sampling system   Ensures the accuracy of the
                                              cannot exceed four percent of      volume of gas sampled.
                                              the average sampling rate or       (Reference Method).
                                              0.00057 m\3\/min (0.020 cfm).
Section 8.2.1..............................  Leak rate of the sampling system   Ensures the accuracy of the
                                              cannot exceed two percent of the   volume of gas sampled.
                                              average sampling rate.             (Approximation Method).
----------------------------------------------------------------------------------------------------------------

* * * * *
    11.1 Reference Method. Weigh the impingers after sampling and 
record the difference in weight to the nearest 0.5 g at a minimum. 
Determine the increase in weight of the silica gel (or silica gel 
plus impinger) to the nearest 0.5 g at a minimum. Record this 
information (see example data sheet, Figure 4-5), and calculate the 
moisture content, as described in section 12.0.
    11.2 Approximation Method. Weigh the contents of the two 
impingers, and measure the weight to the nearest 0.5 g.
* * * * *
    12.1.1 Nomenclature.
    Bws = Proportion of water vapor, by volume, in the 
gas stream.
    Mw = Molecular weight of water, 18.015 g/g-mole 
(18.015 lb/lb-mole).
    Pm = Absolute pressure (for this method, same as 
barometric pressure) at the dry gas meter, mm Hg (in. Hg).
    Pstd = Standard absolute pressure, 760 mm Hg (29.92 
in. Hg).
    R = Ideal gas constant, 0.06236 (mm Hg)(m\3\)/(g-mole)([deg]K) 
for metric units and 21.85 (in. Hg)(ft\3\)/(lb-mole) ([deg]R) for 
English units.
    Tm = Absolute temperature at meter, [deg]K ([deg]R).
    Tstd = Standard absolute temperature, 293.15 [deg]K 
(527.67 [deg]R).
    Vf = Final weight of condenser water plus impinger, 
g.
    Vi = Initial weight, if any, of condenser water plus 
impinger, g.
    Vm = Dry gas volume measured by dry gas meter, dcm 
(dcf).
    Vm(std) = Dry gas volume measured by the dry gas 
meter, corrected to standard conditions, dscm (dscf).
    Vwc(std) = Volume of water vapor condensed, corrected 
to standard conditions, scm (scf).
    Vwsg(std) = Volume of water vapor collected in silica 
gel, corrected to standard conditions, scm (scf).
    Wf = Final weight of silica gel or silica gel plus 
impinger, g.
    Wi = Initial weight of silica gel or silica gel plus 
impinger, g.
    Y = Dry gas meter calibration factor.
    [Delta]Vm = Incremental dry gas volume measured by 
dry gas meter at each traverse point, dcm (dcf).
    12.1.2 Volume of Water Vapor Condensed.

[[Page 63412]]

[GRAPHIC] [TIFF OMITTED] TR07OC20.004

Where:

K1 = 0.001335 m\3\/g for metric units,
= 0.04716 ft\3\/g for English units.

    12.1.3 * * *

K3 = 0.001335 m\3\/g for metric units,
= 0.04716 ft\3\/g for English units.
* * * * *
    12.2.1 Nomenclature.
    Bwm = Approximate proportion by volume of water vapor 
in the gas stream leaving the second impinger, 0.025.
    Bws = Water vapor in the gas stream, proportion by 
volume.
    Mw = Molecular weight of water, 18.015 g/g-mole 
(18.015 lb/lb-mole).
    Pm = Absolute pressure (for this method, same as 
barometric pressure) at the dry gas meter, mm Hg (in. Hg).
    Pstd = Standard absolute pressure, 760 mm Hg (29.92 
in. Hg).
    R = Ideal gas constant, 0.06236 [(mm Hg)(m\3\)]/[(g-mole)(K)] 
for metric units and 21.85 [(in. Hg)(ft\3\)]/[(lb-mole)([deg]R)] for 
English units.
    Tm = Absolute temperature at meter, [deg]K ([deg]R).
    Tstd = Standard absolute temperature, 293.15 [deg]K 
(527.67 [deg]R).
    Vf = Final weight of condenser water plus impinger, 
g.
    Vi = Initial weight, if any, of condenser water plus 
impinger, g.
    Vm = Dry gas volume measured by dry gas meter, dcm 
(dcf).
    Vm(std) = Dry gas volume measured by dry gas meter, 
corrected to standard conditions, dscm (dscf).
    Vwc(std) = Volume of water vapor condensed, corrected 
to standard conditions, scm (scf).
    Y = Dry gas meter calibration factor.
    12.2.2 Volume of Water Vapor Collected.
    [GRAPHIC] [TIFF OMITTED] TR07OC20.005
    
    K5 = 0.001335 m\3\/g for metric units,
    = 0.04716 ft\3\/g for English units.
* * * * *
BILLING CODE 6560-50-P
[GRAPHIC] [TIFF OMITTED] TR07OC20.006

BILLING CODE 6560-50-C

[[Page 63413]]

Method 5--Determination of Particulate Matter Emissions From Stationary 
Sources

* * * * *
    6.1.1.8 Condenser. The following system shall be used to 
determine the stack gas moisture content: Four impingers connected 
in series with leak-free ground glass fittings or any similar leak-
free noncontaminating fittings. The first, third, and fourth 
impingers shall be of the Greenburg-Smith design, modified by 
replacing the tip with a 1.3 cm (\1/2\ in.) ID glass tube extending 
to about 1.3 cm (\1/2\ in.) from the bottom of the flask. The second 
impinger shall be of the Greenburg-Smith design with the standard 
tip. Modifications (e.g., using flexible connections between the 
impingers, using materials other than glass, or using flexible 
vacuum lines to connect the filter holder to the condenser) may be 
used, subject to the approval of the Administrator. The first and 
second impingers shall contain known quantities of water (Section 
8.3.1), the third shall be empty, and the fourth shall contain a 
known weight of silica gel, or equivalent desiccant. A temperature 
sensor, capable of measuring temperature to within 1 [deg]C (2 
[deg]F) shall be placed at the outlet of the fourth impinger for 
monitoring purposes. Alternatively, any system that cools the sample 
gas stream and allows measurement of the water condensed and 
moisture leaving the condenser, each to within 0.5 g may be used, 
subject to the approval of the Administrator. An acceptable 
technique involves the measurement of condensed water either 
gravimetrically and the determination of the moisture leaving the 
condenser by: (1) Monitoring the temperature and pressure at the 
exit of the condenser and using Dalton's law of partial pressures; 
or (2) passing the sample gas stream through a tared silica gel (or 
equivalent desiccant) trap with exit gases kept below 20 [deg]C (68 
[deg]F) and determining the weight gain. If means other than silica 
gel are used to determine the amount of moisture leaving the 
condenser, it is recommended that silica gel (or equivalent) still 
be used between the condenser system and pump to prevent moisture 
condensation in the pump and metering devices and to avoid the need 
to make corrections for moisture in the metered volume.
    Note: If a determination of the PM collected in the impingers is 
desired in addition to moisture content, the impinger system 
described above shall be used, without modification. Individual 
States or control agencies requiring this information shall be 
contacted as to the sample recovery and analysis of the impinger 
contents.
* * * * *
    6.2.4 Petri dishes. For filter samples; glass, polystyrene, or 
polyethylene, unless otherwise specified by the Administrator.
    6.2.5 Balance. To measure condensed water to within 0.5 g at a 
minimum.
* * * * *
    8.1.2 Check filters visually against light for irregularities, 
flaws, or pinhole leaks. Label filters of the proper diameter on the 
back side near the edge using numbering machine ink. As an 
alternative, label the shipping containers (glass, polystyrene or 
polyethylene petri dishes), and keep each filter in its identified 
container at all times except during sampling.
* * * * *
    8.7.6.4 Impinger Water. Treat the impingers as follows: Make a 
notation of any color or film in the liquid catch. Measure the 
liquid that is in the first three impingers by weighing it to within 
0.5 g at a minimum by using a balance. Record the weight of liquid 
present. This information is required to calculate the moisture 
content of the effluent gas. Discard the liquid after measuring and 
recording the weight, unless analysis of the impinger catch is 
required (see Note, section 6.1.1.8). If a different type of 
condenser is used, measure the amount of moisture condensed 
gravimetrically.
* * * * *
    12.1 Nomenclature.
    An = Cross-sectional area of nozzle, m\2\ (ft\2\).
    Bws = Water vapor in the gas stream, proportion by 
volume.
    Ca = Acetone blank residue concentration, mg/mg.
    cs = Concentration of particulate matter in stack 
gas, dry basis, corrected to standard conditions, g/dscm (gr/dscf).
    I = Percent of isokinetic sampling.
    L1 = Individual leakage rate observed during the 
leak-check conducted prior to the first component change, m\3\/min 
(ft\3\/min)
    La = Maximum acceptable leakage rate for either a 
pretest leak-check or for a leak-check following a component change; 
equal to 0.00057 m\3\/min (0.020 cfm) or 4 percent of the average 
sampling rate, whichever is less.
    Li = Individual leakage rate observed during the 
leak-check conducted prior to the ``i\th\'' component change (i = 1, 
2, 3 . . . n), m\3\/min (cfm).
    Lp = Leakage rate observed during the post-test leak-
check, m\3\/min (cfm).
    ma = Mass of residue of acetone after evaporation, 
mg.
    mn = Total amount of particulate matter collected, 
mg.
    Mw = Molecular weight of water, 18.015 g/g-mole 
(18.015 lb/lb-mole).
    Pbar = Barometric pressure at the sampling site, mm 
Hg (in. Hg).
    Ps = Absolute stack gas pressure, mm Hg (in. Hg).
    Pstd = Standard absolute pressure, 760 mm Hg (29.92 
in. Hg).
    R = Ideal gas constant, 0.06236 ((mm Hg)(m\3\))/((K)(g-mole)) 
{21.85 ((in. Hg) (ft\3\))/(([deg]R) (lb-mole)){time} .
    Tm = Absolute average DGM temperature (see Figure 5-
3), K ([deg]R).
    Ts = Absolute average stack gas temperature (see 
Figure 5-3), K ([deg]R).
    Tstd = Standard absolute temperature, 293.15 K 
(527.67 [deg]R).
    Va = Volume of acetone blank, ml.
    Vaw = Volume of acetone used in wash, ml.
    V1c = Total volume of liquid collected in impingers 
and silica gel (see Figure 5-6), g.
    Vm = Volume of gas sample as measured by dry gas 
meter, dcm (dcf).
    Vm(std) = Volume of gas sample measured by the dry 
gas meter, corrected to standard conditions, dscm (dscf).
    Vw(std) = Volume of water vapor in the gas sample, 
corrected to standard conditions, scm (scf).
    Vs = Stack gas velocity, calculated by Method 2, 
Equation 2-7, using data obtained from Method 5, m/sec (ft/sec).
    Wa = Weight of residue in acetone wash, mg.
    Y = Dry gas meter calibration factor.
    [Delta]H = Average pressure differential across the orifice 
meter (see Figure 5-4), mm H2O (in. H2O).
    [rho]a = Density of acetone, mg/ml (see label on 
bottle).
    [thgr] = Total sampling time, min.
    [thgr]1 = Sampling time interval, from the beginning 
of a run until the first component change, min.
    [thgr]i = Sampling time interval, between two 
successive component changes, beginning with the interval between 
the first and second changes, min.
    [thgr]p = Sampling time interval, from the final 
(nth) component change until the end of the sampling run, 
min.
    13.6 = Specific gravity of mercury.
    60 = Sec/min.
    100 = Conversion to percent.
* * * * *
    12.3 * * *
    K1 = 0.38572 [deg]K/mm Hg for metric units, = 17.636 
[deg]R/in. Hg for English units.
* * * * *
    12.4 Volume of Water Vapor Condensed
    [GRAPHIC] [TIFF OMITTED] TR07OC20.007
    

Where:

K2 = 0.001335 m\3\/g for metric units, = 0.04716 ft\3\/g 
for English units.
* * * * *
    12.11.1 * * *

Where:

K4 = 0.003456 ((mm Hg)(m\3\))/((ml)([deg]K)) for metric 
units,
= 0.002668 ((in. Hg)(ft\3\))/((ml)([deg]R)) for English units.
* * * * *
    12.11.2 * * *

Where:

K5 = 4.3209 for metric units, = 0.09450 for English 
units.
* * * * *
    16.1.1.4 * * *

Where:

K1 = 0.38572 [deg]K/mm Hg for metric units, = 17.636 
[deg]R/in. Hg for English units.
Tadj = 273.15 [deg]C for metric units = 459.67 [deg]F for 
English units.
* * * * *
    16.2.3.3 * * *

Where:

K1 = 0.38572 [deg]K/mm Hg for metric units, = 17.636 
[deg]R/in. Hg for English units.
* * * * *
    18.0 * * *

Plant _ _ _
Date
Run No.
Filter No.

Amount liquid lost during transport, mg
Acetone blank volume, ml

[[Page 63414]]

Acetone blank concentration, mg/mg (Equation 5-4)
Acetone wash blank, mg (Equation 5-5)
[GRAPHIC] [TIFF OMITTED] TR07OC20.008

* * * * *

0
22. Amend Appendix A-4 to part 60 in Method 7C by revising section 
7.2.11 and in Method 7E by revising section 8.5 introductory text to 
read as follows:

Appendix A-4 to Part 60--Test Methods 6 Through 10B

* * * * *

Method 7C--Determination of Nitrogen Oxide Emissions From Stationary 
Sources--Alkaline--Permanganate/Colorimetric Method

* * * * *
    7.2.11 Sodium Nitrite (NaNO2) Standard Solution, 
Nominal Concentration, 1000 [micro]g NO2-/ml. Desiccate 
NaNO2 overnight. Accurately weigh 1.4 to 1.6 g of 
NaNO2 (assay of 97 percent NaNO2 or greater), 
dissolve in water, and dilute to 1 liter. Calculate the exact 
NO2-concentration using Equation 7C-1 in section 12.2. 
This solution is stable for at least 6 months under laboratory 
conditions.
* * * * *

Method 7E--Determination of Nitrogen Oxide Emissions From Stationary 
Sources (Instrumental Analyzer Procedure)

* * * * *
    8.5 Post-Run System Bias Check and Drift Assessment.
    How do I confirm that each sample I collect is valid? After each 
run, repeat the system bias check or 2-point system calibration 
error check (for dilution systems) to validate the run. Do not make 
adjustments to the measurement system (other than to maintain the 
target sampling rate or dilution ratio) between the end of the run 
and the completion of the post-run system bias or system calibration 
error check. Note that for all post-run system bias or 2-point 
system calibration error checks, you may inject the low-level gas 
first and the upscale gas last, or vice-versa. If conducting a 
relative accuracy test or relative accuracy test audit, consisting 
of nine runs or more, you may risk sampling for up to three runs 
before performing the post-run bias or system calibration error 
check provided you pass this test at the conclusion of the group of 
three runs. A failed post-run bias or system calibration error check 
in this case will invalidate all runs subsequent to the last passed 
check. When conducting a performance or compliance test, you must 
perform a post-run system bias or system calibration error check 
after each individual test run.
* * * * *

0
23. Amend Appendix A-5 to part 60, Method 12 by:
0
a. Revising sections ``7.1.2'', ``8.7.1.6'', ``8.7.3.1'', ``8.7.3.3'', 
``8.7.3.6'', ``12.1'', ``12.3'', ``16.1'' through ``16.5'';
0
b. Adding sections 16.5.1 and 16.5.2; and
0
c. Removing section 16.6.
    The revisions and additions read as follows:

Appendix A-5 to Part 60--Test Methods 11 Through 15A

* * * * *

Method 12--Determination of Inorganic Lead Emissions From Stationary 
Sources

* * * * *
    7.1.2 Silica Gel and Crushed Ice. Same as Method 5, sections 
7.1.2 and 7.1.4, respectively.
* * * * *
    8.7.1.6 Brush and rinse with 0.1 N HNO3 the inside of 
the front half of the filter holder. Brush and rinse each surface 
three times or more, if needed, to remove visible sample matter. 
Make a final rinse of the brush and filter holder. After all 0.1 N 
HNO3 washings and sample matter are collected in the 
sample container, tighten the lid on the sample container so that 
the fluid will not leak out when it is shipped to the laboratory. 
Mark the height of the fluid level to determine whether leakage 
occurs during transport. Label the container to identify its 
contents clearly.
* * * * *
    8.7.3.1 Cap the impinger ball joints.
* * * * *
    8.7.3.3 Treat the impingers as follows: Make a notation of any 
color or film in the liquid catch. Measure the liquid that is in the 
first three impingers by weighing it to within 0.5 g at a minimum by 
using a balance. Record the weight of liquid present. The liquid 
weight is needed, along with the silica gel data, to calculate the 
stack gas moisture content (see Method 5, Figure 5-6).
* * * * *
    8.7.3.6 Rinse the insides of each piece of connecting glassware 
for the impingers twice with 0.1 N HNO3; transfer this 
rinse into Container No. 4. Do not rinse or brush the glass-fritted 
filter support. Mark the height of the fluid level to determine 
whether leakage occurs during transport. Label the container to 
identify its contents clearly.
* * * * *
    12.1 Nomenclature.
    Am = Absorbance of the sample solution.
    An = Cross-sectional area of nozzle, m\2\ (ft\2\).
    At = Absorbance of the spiked sample solution.
    Bws = Water in the gas stream, proportion by volume.
    Ca = Lead concentration in standard solution, 
[micro]g/ml.
    Cm = Lead concentration in sample solution analyzed 
during check for matrix effects, [micro]g/ml.
    Cs = Lead concentration in stack gas, dry basis, 
converted to standard conditions, mg/dscm (gr/dscf).
    I = Percent of isokinetic sampling.

[[Page 63415]]

    L1 = Individual leakage rate observed during the 
leak-check conducted prior to the first component change, m\3\/min 
(ft\3\/min).
    La = Maximum acceptable leakage rate for either a 
pretest leak-check or for a leak-check following a component change; 
equal to 0.00057 m\3\/min (0.020 cfm) or 4 percent of the average 
sampling rate, whichever is less.
    Li = Individual leakage rate observed during the 
leak-check conducted prior to the ``ith'' component change (i = 1, 
2, 3 * * * n), m\3\/min (cfm).
    Lp = Leakage rate observed during the post-test leak-
check, m\3\/min (cfm).
    mt = Total weight of lead collected in the sample, 
[micro]g.
    Mw = Molecular weight of water, 18.0 g/g-mole (18.0 
lb/lb-mole).
    Pbar = Barometric pressure at the sampling site, mm 
Hg (in. Hg).
    Ps = Absolute stack gas pressure, mm Hg (in. Hg).
    Pstd = Standard absolute pressure, 760 mm Hg (29.92 
in. Hg).
    R = Ideal gas constant, 0.06236 [(mm Hg) (m\3\)]/[([deg]K) (g-
mole)] {21.85 [(in. Hg) (ft\3\)]/[([deg]R) (lb-mole)]{time} .
    Tm = Absolute average dry gas meter temperature (see 
Figure 5-3 of Method 5), [deg]K ([deg]R).
    Tstd = Standard absolute temperature, 293 [deg]K (528 
[deg]R).
    vs = Stack gas velocity, m/sec (ft/sec).
    Vm = Volume of gas sample as measured by the dry gas 
meter, dry basis, m\3\ (ft\3\).
    Vm(std) = Volume of gas sample as measured by the dry 
gas meter, corrected to standard conditions, m\3\ (ft\3\).
    Vw(std) = Volume of water vapor collected in the 
sampling train, corrected to standard conditions, m\3\ (ft\3\).
    Y = Dry gas meter calibration factor.
    [Delta]H = Average pressure differential across the orifice 
meter (see Figure 5-3 of Method 5), mm H2O (in. 
H2O).
    [thetas] = Total sampling time, min.
    [thetas]l = Sampling time interval, from the 
beginning of a run until the first component change, min.
    [thetas]i = Sampling time interval, between two 
successive component changes, beginning with the interval between 
the first and second changes, min.
    [thetas]p = Sampling time interval, from the final 
(nth) component change until the end of the sampling run, 
min.
* * * * *
    12.3 Dry Gas Volume, Volume of Water Vapor Condensed, and 
Moisture Content. Using data obtained in this test, calculate 
Vm(std), Vw(std), and Bws according 
to the procedures outlined in Method 5, sections 12.3 through 12.5.
* * * * *
    16.1 Simultaneous Determination of Particulate Matter and Lead 
Emissions. Method 12 may be used to simultaneously determine Pb and 
particulate matter provided:
    (1) A glass fiber filter with a low Pb background is used and 
this filter is checked, desiccated and weighed per section 8.1 of 
Method 5,
    (2) An acetone rinse, as specified by Method 5, sections 7.2 and 
8.7.6.2, is used to remove particulate matter from the probe and 
inside of the filter holder prior to and kept separate from the 0.1 
N HNO3 rinse of the same components,
    (3) The recovered filter, the acetone rinse, and an acetone 
blank (Method 5, section 7.2) are subjected to the gravimetric 
analysis of Method 5, sections 6.3 and 11.0 prior to the analysis 
for Pb as described below, and
    (4) The entire train contents, including the 0.1 N 
HNO3 impingers, filter, acetone and 0.1 N HNO3 
probe rinses are treated and analyzed for Pb as described in 
sections 8.0 and 11.0 of this method.
    16.2 Filter Location. A filter may be used between the third and 
fourth impingers provided the filter is included in the analysis for 
Pb.
    16.3 In-Stack Filter. An in-stack filter may be used provided: 
(1) A glass-lined probe and at least two impingers, each containing 
100 ml of 0.1 N HNO3 after the in-stack filter, are used 
and (2) the probe and impinger contents are recovered and analyzed 
for Pb. Recover sample from the nozzle with acetone if a particulate 
analysis is to be made as described in section 16.1 of this method.
    16.4 Inductively Coupled Plasma-Atomic Emission Spectrometry 
(ICP-AES) Analysis. ICP-AES may be used as an alternative to atomic 
absorption analysis provided the following conditions are met:
    16.4.1 Sample collection/recovery, sample loss check, and sample 
preparation procedures are as defined in sections 8.0, 11.1, and 
11.2, respectively, of this method.
    16.4.2 Analysis shall be conducted following Method 6010D of SW-
846 (incorporated by reference, see Sec.  60.17). The limit of 
detection for the ICP-AES must be demonstrated according to section 
15.0 of Method 301 in appendix A of part 63 of this chapter and must 
be no greater than one-third of the applicable emission limit. 
Perform a check for matrix effects according to section 11.5 of this 
method.
    16.5 Inductively Coupled Plasma-Mass Spectrometry (ICP-MS) 
Analysis. ICP-MS may be used as an alternative to atomic absorption 
analysis provided the following conditions are met:
    16.5.1 Sample collection/recovery, sample loss check, and sample 
preparation procedures are as defined in sections 8.0, 11.1, and 
11.2, respectively of this method.
    16.5.2 Analysis shall be conducted following Method 6020B of SW-
846 (incorporated by reference, see Sec.  60.17). The limit of 
detection for the ICP-MS must be demonstrated according to section 
15.0 of Method 301 in appendix A to part 63 of this chapter and must 
be no greater than one-third of the applicable emission limit. Use 
the multipoint calibration curve option in section 10.4 of Method 
6020B and perform a check for matrix effects according to section 
11.5 of this method.
* * * * *

0
24. Amend Appendix A-6 to part 60 by:
0
a. In Method 16B by:
0
i. Revising sections 2.1, 6.1, 8.2;
0
ii. Removing section 8.3;
0
iii. Redesignating sections 8.4, 8.4.1, and 8.4.2 as 8.3, 8.3.1, and 
8.3.2, respectively;
0
iv. Revising section 11.1; and
0
v. Adding section 11.2; and
0
b. In Method 16C, revising section 13.1.
    The revisions and addition read as follows:

Appendix A-6 to Part 60--Test Methods 16 Through 18

* * * * *

Method 16B--Determination of Total Reduced Sulfur Emissions From 
Stationary Sources

* * * * *
    2.1 A gas sample is extracted from the stack. The SO2 
is removed selectively from the sample using a citrate buffer 
solution. The TRS compounds are then thermally oxidized to 
SO2 and analyzed as SO2 by gas chromatography 
(GC) using flame photometric detection (FPD).
* * * * *
    6.1 Sample Collection. The sampling train is shown in Figure 
16B-1. Modifications to the apparatus are accepted provided the 
system performance check in section 8.3.1 is met.
* * * * *
    8.2 Sample Collection. Before any source sampling is performed, 
conduct a system performance check as detailed in section 8.3.1 to 
validate the sampling train components and procedures. Although this 
test is optional, it would significantly reduce the possibility of 
rejecting tests as a result of failing the post-test performance 
check. At the completion of the pretest system performance check, 
insert the sampling probe into the test port making certain that no 
dilution air enters the stack though the port. Condition the entire 
system with sample for a minimum of 15 minutes before beginning 
analysis. If the sample is diluted, determine the dilution factor as 
in section 10.4 of Method 15.
* * * * *
    11.1 Analysis. Inject aliquots of the sample into the GC/FPD 
analyzer for analysis. Determine the concentration of SO2 
directly from the calibration curves or from the equation for the 
least-squares line.
    11.2 Perform analysis of a minimum of three aliquots or one 
every 15 minutes, whichever is greater, spaced evenly over the test 
period.
* * * * *

Method 16C--Determination of Total Reduced Sulfur Emissions From 
Stationary Sources

* * * * *
    13.1 Analyzer Calibration Error. At each calibration gas level 
(low, mid, and high), the calibration error must either not exceed 
5.0 percent of the calibration span or [bond]CDir-
Cv[bond] must be <=0.5 ppmv.
* * * * *

0
25. Amend Appendix A-7 to part 6 by:
0
a. In Method 24, revising section 6.2.
0
b. In Method 25C, revising sections 8.4.2, 9.1, 12.5, 12.5.1, and 
12.5.2.

[[Page 63416]]

    The revisions read as follows:

Appendix A-7 to Part 60--Test Methods 19 Through 25E

* * * * *

Method 24--Determinaton of Volatile Matter Content, Water Content, 
Density, Volume Solids, and Weight Solids of Surface Coatings

* * * * *
    6.2 ASTM D 2369-81, 87, 90, 92, 93, 95, or 10. Standard Test 
Method for Volatile Content of Coatings.
* * * * *

Method 25C--Determination of Nonmethane Organic Compounds (NMOC) in 
Landfill Gases

* * * * *
    8.4.2 Use Method 3C to determine the percent N2 and 
O2 in each cylinder. The presence of N2 and 
O2 indicate either infiltration of ambient air into the 
landfill gas sample or an inappropriate testing site has been chosen 
where anaerobic decomposition has not begun. The landfill gas sample 
is acceptable if the concentration of N2 is less than 20 
percent. Alternatively, the oxygen content of each cylinder must be 
less than 5 percent. Landfills with 3-year average annual rainfalls 
equal to or less than 20 inches annual rainfalls samples are 
acceptable when the N2 to O2 concentration 
ratio is greater than 3.71.
* * * * *
    9.1 Miscellaneous Quality Control Measures.

----------------------------------------------------------------------------------------------------------------
                  Section                         Quality control measure                    Effect
----------------------------------------------------------------------------------------------------------------
8.4.2......................................  If the 3-year average annual       Ensures that ambient air was not
                                              rainfall is greater than 20        drawn into the landfill gas
                                              inches, verify that landfill gas   sample and gas was sampled from
                                              sample contains less than 20       an appropriate location. If
                                              percent N2 and 5 percent O2.       outside of range, invalidate
                                              Landfills with 3-year average      sample and repeat sample
                                              annual rainfalls equal to or       collection.
                                              less than 20 inches annual
                                              rainfalls samples are acceptable
                                              when the N2 to O2 concentration
                                              ratio is greater than 3.71.
10.1, 10.2.................................  NMOC analyzer initial and daily    Ensures precision of analytical
                                              performance checks.                results.
----------------------------------------------------------------------------------------------------------------

* * * * *
    12.5 You must correct the NMOC Concentration for the 
concentration of nitrogen or oxygen based on which gas or gases 
passes the requirements in section 9.1 or based on the 3-year 
average annual rainfall based on the closest NOAA land-based 
station.
    12.5.1 NMOC Concentration with nitrogen correction. Use Equation 
25C-4 to calculate the concentration of NMOC for each sample tank 
when the nitrogen concentration is less than 20 percent.
[GRAPHIC] [TIFF OMITTED] TR07OC20.009

    12.5.2 NMOC Concentration with oxygen correction. Use Equation 
25C-5 to calculate the concentration of NMOC for each sample tank if 
the landfill gas oxygen is less than 5 percent and the landfill gas 
nitrogen concentration is greater than 20 percent, or 3-year average 
annual rainfall based annual rainfall of less than 20 inches.
[GRAPHIC] [TIFF OMITTED] TR07OC20.010

* * * * *

0
26. Amend Appendix A-8 to part 60 by:
0
a. In Method 26, revising section 8.1.2; and
0
b. In Method 26A, revising sections 6.1.3 and 8.1.5.
    The revisions read as follows:

Appendix A-8 to Part 60--Test Methods 26 Through 30B

* * * * *

Method 26--Determination of Hydrogen Halide and Halogen Emissions From 
Stationary Sources Non-Isokinetic Method

* * * * *
    8.1.2 Adjust the probe temperature and the temperature of the 
filter and the stopcock (i.e., the heated area in Figure 26-1) to a 
temperature sufficient to prevent water condensation. This 
temperature must be maintained between 120 and 134 [deg]C (248 and 
273 [deg]F). The temperature should be monitored throughout a 
sampling run to ensure that the desired temperature is maintained. 
It is important to maintain a temperature around the probe and 
filter in this range since it is extremely difficult to purge acid 
gases off these components. (These components are not quantitatively 
recovered and, hence, any collection of acid gases on these 
components would result in potential under reporting of these 
emissions. The applicable subparts may specify alternative higher 
temperatures.)
* * * * *

Method 26A--Determination of Hydrogen Halide and Halogen Emissions From 
Stationary Sources--Isokinetic Method

* * * * *
    6.1.3 Pitot Tube, Differential Pressure Gauge, Filter Heating 
System, Filter Temperature Sensor with a glass or Teflon encasement, 
Metering System, Barometer, Gas Density Determination Equipment. 
Same as Method 5, sections 6.1.1.3, 6.1.1.4, 6.1.1.6, 6.1.1.7, 
6.1.1.9, 6.1.2, and 6.1.3.
* * * * *
    8.1.5 Sampling Train Operation. Follow the general procedure 
given in Method 5, Section 8.5. It is important to maintain a 
temperature around the probe, filter (and cyclone, if used) between 
120 and 134 [deg]C (248 and 273 [deg]F) since it is extremely 
difficult to purge acid gases off these components. (These 
components are not quantitatively recovered and hence any collection 
of acid gases on these components would result in potential under 
reporting these emissions. The applicable subparts may specify 
alternative higher temperatures.) For each run, record the data 
required on a data sheet such as the one shown in Method 5, Figure 
5-3. If the condensate impinger becomes too full, it may be emptied, 
recharged with 50 ml of 0.1 N H2SO4, and replaced during the sample 
run. The condensate emptied must be saved and included in the 
measurement of the volume of moisture collected and

[[Page 63417]]

included in the sample for analysis. The additional 50 ml of 
absorbing reagent must also be considered in calculating the 
moisture. Before the sampling train integrity is compromised by 
removing the impinger, conduct a leak-check as described in Method 
5, section 8.4.2.
* * * * *

0
27. Amend Appendix B to part 60 by:
0
a. In Performance Specification 4B, revising section 4.5;
0
b. In Performance Specification 5, revising sections 5.0 and 8.1;
0
c. In Performance Specification 6, revising sections 13.1 and 13.2;
0
d. In Performance Specification 8, redesignating sections 8.3, 8.4, and 
8.5 as 8.4, 8.5, and 8.6, respectively;
0
e. Adding new section 8.3;
0
f. In Performance Specification 9, revising sections 7.2, 8.3, 8.4, 
10.1, 10.2, 13.1, and 13.2;
0
g. Adding section 13.4;
0
h. In Performance Specification 18, revising sections 2.3 and 11.9.1.
    The revisions and additions read as follows:

Appendix B to Part 60--Performance Specifications

* * * * *

Performance Specification 4B--Specifications and Test Procedures for 
Carbon Monoxide and Oxygen Continuous Monitoring Systems in Stationary 
Sources

* * * * *
    4.5 Response Time. The response time for the CO or O2 
monitor must not exceed 240 seconds.
* * * * *

Performance Specification 5--Specifications and Test Procedures for TRS 
Continuous Emission Monitoring Systems in Stationary Sources

* * * * *
    5.0 Safety
    This performance specification may involve hazardous materials, 
operations, and equipment. This performance specification may not 
address all of the safety problems associated with its use. It is 
the responsibility of the user to establish appropriate safety and 
health practices and determine the applicable regulatory limitations 
prior to performing this performance specification. The CEMS user's 
manual should be consulted for specific precautions to be taken with 
regard to the analytical procedures.
* * * * *
    8.1 Relative Accuracy Test Procedure. Sampling Strategy for 
reference method (RM) Tests, Number of RM Tests, and Correlation of 
RM and CEMS Data are the same as PS 2, sections 8.4.3, 8.4.4, and 
8.4.5, respectively.
    Note: For Method 16, a sample is made up of at least three 
separate injects equally spaced over time. For Method 16A, a sample 
is collected for at least 1 hour. For Method 16B, you must analyze a 
minimum of three aliquots spaced evenly over the test period.
* * * * *

Performance Specification 6--Specifications and Test Procedures for 
Continuous Emission Rate Monitoring Systems in Stationary Sources

* * * * *
    13.1 Calibration Drift. Since the CERMS includes analyzers for 
several measurements, the CD shall be determined separately for each 
analyzer in terms of its specific measurement. The calibration for 
each analyzer associated with the measurement of flow rate shall not 
drift or deviate from each reference value of flow rate by more than 
3 percent of the respective high-level reference value over the CD 
test period (e.g., seven-day) associated with the pollutant 
analyzer. The CD specification for each analyzer for which other PSs 
have been established (e.g., PS 2 for SO2 and 
NOX), shall be the same as in the applicable PS.
    13.2 CERMS Relative Accuracy. Calculate the CERMS Relative 
Accuracy using Eq. 2-6 of section 12 of Performance Specification 2. 
The RA of the CERMS shall be no greater than 20 percent of the mean 
value of the RM's test data in terms of the units of the emission 
standard, or in cases where the average emissions for the test are 
less than 50 percent of the applicable standard, substitute the 
emission standard value in the denominator of Eq. 2-6 in place of 
the RM.
* * * * *

Performance Specification 8--Performance Specifications for Volatile 
Organic Compound Continuous Emission Monitoring Systems in Stationary 
Sources

* * * * *
    8.3 Calibration Drift Test Procedure. Same as section 8.3 of PS 
2.
    8.4 Reference Method (RM). Use the method specified in the 
applicable regulation or permit, or any approved alternative, as the 
RM.
    8.5 Sampling Strategy for RM Tests, Correlation of RM and CEMS 
Data, and Number of RM Tests. Follow PS 2, sections 8.4.3, 8.4.5, 
and 8.4.4, respectively.
    8.6 Reporting. Same as section 8.5 of PS 2.
* * * * *

Performance Specification 9--Specifications and Test Procedures for Gas 
Chromatographic Continuous Emission Monitoring Systems in Stationary 
Sources

* * * * *
    7.2 Performance Audit Gas. Performance Audit Gas is an 
independent cylinder gas or cylinder gas mixture. A certified EPA 
audit gas shall be used, when possible. A gas mixture containing all 
the target compounds within the calibration range and certified by 
EPA's Traceability Protocol for Assay and Certification of Gaseous 
Calibration Standards may be used when EPA performance audit 
materials are not available. If a certified EPA audit gas or a 
traceability protocol gas is not available, use a gas manufacturer 
standard accurate to 2 percent.
* * * * *
    8.3 Seven (7)-Day Calibration Error (CE) Test Period. At the 
beginning of each 24-hour period, set the initial instrument set 
points by conducting a multi-point calibration for each compound. 
The multi-point calibration shall meet the requirements in sections 
13.1, 13.2, and 13.3. Throughout the 24-hour period, sample and 
analyze the stack gas at the sampling intervals prescribed in the 
regulation or permit. At the end of the 24-hour period, inject the 
calibration gases at three concentrations for each compound in 
triplicate and determine the average instrument response. Determine 
the CE for each pollutant at each concentration using Equation 9-2. 
Each CE shall be <=10 percent. Repeat this procedure six more times 
for a total of 7 consecutive days.
    8.4 Performance Audit Test Periods. Conduct the performance 
audit once during the initial 7-day CE test and quarterly 
thereafter. Performance Audit Tests must be conducted through the 
entire sampling and analyzer system. Sample and analyze the EPA 
audit gas(es) (or the gas mixture) three times. Calculate the 
average instrument response. Results from the performance audit test 
must meet the requirements in sections 13.3 and 13.4.
* * * * *
    10.1 Multi-Point Calibration. After initial startup of the GC, 
after routine maintenance or repair, or at least once per month, 
conduct a multi-point calibration of the GC for each target analyte. 
Calibration is performed at the instrument independent of the sample 
transport system. The multi-point calibration for each analyte shall 
meet the requirements in sections 13.1, 13.2, and 13.3.
* * * * *
    10.2 Daily Calibration. Once every 24 hours, analyze the mid-
level calibration standard for each analyte in triplicate. 
Calibration is performed at the instrument independent of the sample 
transport system. Calculate the average instrument response for each 
analyte. The average instrument response shall not vary by more than 
10 percent from the certified concentration value of the cylinder 
for each analyte. If the difference between the analyzer response 
and the cylinder concentration for any target compound is greater 
than 10 percent, immediately inspect the instrument making any 
necessary adjustments, and conduct an initial multi-point 
calibration as described in section 10.1.
* * * * *
    13.1 Calibration Error (CE). The CEMS must allow the 
determination of CE at all three calibration levels. The average 
CEMS calibration response must not differ by more than 10 percent of 
calibration gas value at each level after each 24-hour period and 
after any triplicate calibration response check.
    13.2 Calibration Precision and Linearity. For each triplicate 
injection at each concentration level for each target analyte, any 
one injection shall not deviate more than 5 percent from the average 
concentration measured at that level. When the CEMS response is 
evaluated over three concentration levels, the linear regression 
curve for each organic compound shall be determined using Equation 
9-1 and must have an r\2\ >=0.995.
* * * * *
    13.4 Performance Audit Test Error. Determine the error for each 
average

[[Page 63418]]

pollutant measurement using the Equation 9-2 in section 12.3. Each 
error shall be less than or equal to 10 percent of the cylinder gas 
certified value. Report the audit results including the average 
measured concentration, the error and the certified cylinder 
concentration of each pollutant as part of the reporting 
requirements in the appropriate regulation or permit.
* * * * *

Performance Specification 18--Performance Specifications and Test 
Procedures for Gaseous Hydrogen Chloride (HCl) Continuous Emission 
Monitoring Systems at Stationary Sources

* * * * *
    2.3 The relative accuracy (RA) must be established against a 
reference method (RM) (e.g., Method 26A, Method 320, ASTM 
International (ASTM) D6348-12, including mandatory annexes, or 
Method 321 for Portland cement plants as specified by the applicable 
regulation or, if not specified, as appropriate for the source 
concentration and category). Method 26 may be approved as a RM by 
the Administrator on a case-by-case basis if not otherwise allowed 
or denied in an applicable regulation.
* * * * *
    11.9.1 Unless otherwise specified in an applicable regulation, 
use Method 26A in 40 CFR part 60, appendix A-8, Method 320 in 40 CFR 
part 63, appendix A, or ASTM D6348-12 including all annexes, as 
applicable, as the RMs for HCl measurement. Obtain and analyze RM 
audit samples, if they are available, concurrently with RM test 
samples according to the same procedure specified for performance 
tests in the general provisions of the applicable part. If Method 26 
is not specified in an applicable subpart of the regulations, you 
may request approval to use Method 26 in appendix A-8 to this part 
as the RM on a site-specific basis under Sec. Sec.  63.7(f) or 
60.8(b). Other RMs for moisture, O2, etc., may be 
necessary. Conduct the RM tests in such a way that they will yield 
results representative of the emissions from the source and can be 
compared to the CEMS data.
* * * * *

0
28. Amend Appendix F to part 60, in Procedure 1, by revising section 
5.2.3(2) to read as follows:

Appendix F to Part 60--Quality Assurance Procedures

Procedure 1--Quality Assurance Requirements for Gas Continuous Emission 
Monitoring Systems Used for Compliance Determination

* * * * *
    5.2.3 * * *
    (2) For the CGA, 15 percent of the average audit value 
or 5 ppm, whichever is greater; for diluent monitors, 
15 percent of the average audit value.
* * * * *

PART 61--NATIONAL EMISSION STANDARDS FOR HAZARDOUS AIR POLLUTANTS

0
29. The authority citation for part 61 continues to read as follows:

    Authority:  42 U.S.C. 7401 et seq.

0
30. Amend Appendix B to part 61 by:
0
a. Adding the entries Method 114--Test Methods for Measuring 
Radionuclide Emissions from Stationary Sources and Method 115--
Monitoring for Radon-222 Emissions at the end of the index for appendix 
B to part 61.
0
b. In Method 107, revising section 12.3, equation 107-3.
    The additions and revisions read as follows:

Appendix B to Part 61--Test Methods

* * * * *

Method 114--Test Methods for Measuring Radionuclide Emissions From 
Stationary Sources

Method 115--Monitoring for Radon-222 Emissions

* * * * *

Method 107--Determination of Vinyl Chloride Content of In-Process 
Wastewater Samples, and Vinyl Chloride Content of Polyvinyl Chloride 
Resin Slurry, Wet Cake, and Latex Samples

* * * * *
    12.3 * * *
    [GRAPHIC] [TIFF OMITTED] TR07OC20.011
    
* * * * *

PART 63--NATIONAL EMISSION STANDARDS FOR HAZARDOUS AIR POLLUTANTS 
FOR SOURCE CATEGORIES

0
31. The authority citation for part 63 continues to read as follows:

    Authority:  42 U.S.C. 7401 et seq.

0
32. Amend Sec.  63.2 by revising the definition of ``Alternative test 
method'' to read as follows:

Sec.  63.2  Definitions.

* * * * *
    Alternative test method means any method of sampling and analyzing 
for an air pollutant that has been demonstrated to the Administrator's 
satisfaction, using Method 301 in appendix A of this part, to produce 
results adequate for the Administrator's determination that it may be 
used in place of a test method specified in this part.
* * * * *

Subpart LLL--National Emission Standards for Hazardous Air 
Pollutants from the Portland Cement Manufacturing Industry

0
33. Amend Sec.  63.1349, by revising paragraphs (b)(7)(viii)(A) and 
(B), (b)(8)(vi), and (b)(8)(vii)(B) and (C) to read as follows:

Sec.  63.1349  Performance testing requirements.

* * * * *
    (b) * * *
    (7) * * *
    (viii) * * *
    (A) Determine the THC CEMS average value in ppmvw, and the average 
of your corresponding three total organic HAP compliance test runs, 
using Equation 12.
[GRAPHIC] [TIFF OMITTED] TR07OC20.012

[[Page 63419]]

    Where:

x = The average THC CEMS value in ppmvw, as propane.
Xi = The THC CEMS data points in ppmvw, as propane, for 
all three test runs.
y = The average organic HAP value in ppmvd, corrected to 7 percent 
oxygen.
Yi = The organic HAP concentrations in ppmvd, corrected 
to 7 percent oxygen, for all three test runs.
n = The number of data points.

    (B) You must use your 3-run average THC CEMS value and your 3-run 
average organic HAP concentration from your Method 18 and/or Method 320 
compliance tests to determine the operating limit. Use equation 13 to 
determine your operating limit in units of ppmvw THC, as propane.
[GRAPHIC] [TIFF OMITTED] TR07OC20.013

    Where:

Tl = The 30-day operating limit for your THC CEMS, ppmvw, 
as propane.
y = The average organic HAP concentration from Eq. 12, ppmvd, 
corrected to 7 percent oxygen.
x = The average THC CEMS concentration from Eq. 12, ppmvw, as 
propane.
9 = 75 percent of the organic HAP emissions limit (12 ppmvd, 
corrected to 7 percent oxygen)

* * * * *
    (8) * * *
    (vi) If your kiln has an inline kiln/raw mill, you must conduct 
separate performance tests while the raw mill is operating (``mill 
on'') and while the raw mill is not operating (``mill off''). Using the 
fraction of time that the raw mill is on and the fraction of time that 
the raw mill is off, calculate this limit as a weighted average of the 
SO2 levels measured during raw mill on and raw mill off 
compliance testing with Equation 17.
[GRAPHIC] [TIFF OMITTED] TR07OC20.014

    Where:

R = Operating limit as SO2, ppmv.
y = Average SO2 CEMS value during mill on operations, 
ppmv.
t = Percentage of operating time with mill on, expressed as a 
decimal.
x = Average SO2 CEMS value during mill off operations, 
ppmv.
1-t = Percentage of operating time with mill off, expressed as a 
decimal.

* * * * *
    (vii) * * *
    (B) Determine your SO2 CEMS instrument average ppmv, and 
the average of your corresponding three HCl compliance test runs, using 
Equation 18.
[GRAPHIC] [TIFF OMITTED] TR07OC20.015

    Where:

x = The average SO2 CEMS value in ppmv.
X1 = The SO2 CEMS data points in ppmv for the 
three runs constituting the performance test.
y = The average HCl value in ppmvd, corrected to 7 percent oxygen.
Y1 = The HCl emission concentration expressed as ppmvd, 
corrected to 7 percent oxygen for the threeruns constituting the 
performance test.
n = The number of data points.

    (C) With your instrument zero expressed in ppmv, your 
SO2 CEMS three run average expressed in ppmv, and your 3-run 
HCl compliance test average in ppmvd, corrected to 7 percent 
O2, determine a relationship of ppmvd HCl corrected to 7 
percent O2 per ppmv SO2 with Equation 19.
[GRAPHIC] [TIFF OMITTED] TR07OC20.016

    Where:

R = The relative HCl ppmvd, corrected to 7 percent 
oxygen, per ppmv SO2 for your SO2 
CEMS.
y = The average HCl concentration from Eq. 18 in ppmvd, corrected to 
7 percent oxygen.
x = The average SO2 CEMS value from Eq. 18 in ppmv.
z = The instrument zero output ppmv value.

* * * * *

0
34. Amend Appendix A to part 63 by:
0
a. In Method 301, revising section 11.1.3;
0
b. In Method 308, revising section 12.4, equation 308-3 and section 
12.5, equation 308-5;
0
c. In Method 311, revising sections 1.1 and 17;
0
d. In Method 315, revising Figure 315-1;
0
e. In Method 316, revising section 1.0; and
0
f. In Method 323, revising the method heading and section 2.0.
    The revisions read as follows:

Appendix A to Part 63--Test Methods Pollutant Measurement Methods From 
Various Waste Media

* * * * *

Method 301--Field Validation of Pollutant Measurement Methods From 
Various Waste Media

* * * * *
    11.1.3 T Test. Calculate the t-statistic using Equation 301-13.
    [GRAPHIC] [TIFF OMITTED] TR07OC20.017
    
    * * *
* * * * *

Method 308--Procedure for Determination of Methanol Emission From 
Stationary Sources

* * * * *
    12.4 * * *

[[Page 63420]]

[GRAPHIC] [TIFF OMITTED] TR07OC20.018

    12.5 * * *
    [GRAPHIC] [TIFF OMITTED] TR07OC20.019
    
* * * * *

Method 311--Analysis of Hazardous Air Pollutant Compounds in Paints and 
Coatings by Direct Injection Into a Gas Chromatograph

* * * * *
    1.1 Applicability. This method is applicable for determination 
of most compounds designated by the U.S. Environmental Protection 
Agency as volatile hazardous air pollutants (HAP's) (See Reference 
1) that are contained in paints and coatings. Styrene, ethyl 
acrylate, and methyl methacrylate can be measured by ASTM D 4827-03. 
Formaldehyde can be measured by ASTM D 5910-05 or ASTM D 1979-91. 
Toluene diisocyanate can be measured in urethane prepolymers by ASTM 
D 3432-89. Method 311 applies only to those volatile HAP's which are 
added to the coating when it is manufactured, not to those that may 
form as the coating cures (reaction products or cure volatiles). A 
separate or modified test procedure must be used to measure these 
reaction products or cure volatiles in order to determine the total 
volatile HAP emissions from a coating. Cure volatiles are a 
significant component of the total HAP content of some coatings. The 
term ``coating'' used in this method shall be understood to mean 
paints and coatings.
* * * * *
    17. * * *
    4. Standard Test Method for Determination of Dichloromethane and 
1,1,1-Trichloroethane in Paints and Coatings by Direct Injection 
into a Gas Chromatograph. ASTM Designation D4457-02.
    5. Standard Test Method for Determining the Unreacted Monomer 
Content of Latexes Using Capillary Column Gas Chromatography. ASTM 
Designation D4827-03.
    6. Standard Test Method for Determining Unreacted Monomer 
Content of Latexes Using Gas-Liquid Chromatography, ASTM Designation 
D4747-02.
* * * * *

Method 315--Determination of Particulate and Methylene Chloride 
Extractable Matter (MCEM) From Selected Sources at Primary Aluminum 
Production Facilities

* * * * *
BILLING CODE 6560-50-P

[[Page 63421]]

[GRAPHIC] [TIFF OMITTED] TR07OC20.020

[[Page 63422]]

[GRAPHIC] [TIFF OMITTED] TR07OC20.021

Method 316--Sampling and Analysis for Formaldehyde Emissions From 
Stationary Sources in the Mineral Wool and Wool Fiberglass Industries

    1.0 Scope and Application
    This method is applicable to the determination of formaldehyde, 
CAS Registry number 50-00-0, from stationary sources in the mineral 
wool and wool fiber glass industries. High purity water is used to 
collect the formaldehyde. The formaldehyde concentrations in the 
stack samples are determined using the modified pararosaniline 
method. Formaldehyde can be detected as low as 8.8 x 
10-10 lbs/cu ft (11.3 ppbv) or as high as 1.8 x 
10-3 lbs/cu ft (23,000,000 ppbv), at standard conditions 
over a 1-hour sampling period, sampling approximately 30 cu ft.
* * * * *

Method 323--Measurement of Formaldehyde Emissions From Natural Gas-
Fired Stationary Sources--Acetyl Acetone Derivatization Method

* * * * *
    2.0 Summary of Method. An emission sample from the combustion 
exhaust is drawn through a midget impinger train containing chilled 
reagent water to absorb formaldehyde. The formaldehyde concentration 
in the impinger is determined by reaction with acetyl acetone to 
form a colored derivative which is measured colorimetrically.
* * * * *
[FR Doc. 2020-18824 Filed 10-6-20; 8:45 am]
 BILLING CODE 6560-50-C