cometadata/datacite-titles-descriptions-related-identifiers huggingface dataset:
DOI: 10.4121/uuid:cd707304-32fb-435c-b9ee-55fd0ac1fe08
title: Measured data from sensor Q3 at Maisbich for month 2004-09

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:cf18ed61-f40d-4e07-b17e-8ac7ba5a5052
title: Measured data from sensor Dis at Attert for month 2004-11

description: measured water pressure Parent item: All data measured by discharge meter in Attert basin Measuring instrument: Useldange, Attert basin: Discharge meter ISCO 4120 ultrasonic sensor (ceramic pressure membrane). Measures water pressure (?). The sensor is installed under a bridge (?).
DOI: 10.4121/uuid:cf630965-9afb-43cf-af8c-874f6b9ce909
title: Measured data from sensor TB at Maisbich for month 2005-02

description: Parent item: All data measured by tipping bucket discharge meter in Maisbich Measuring instrument: Maisbich basin: Tipping bucket rainfall meter Type: Casella Tipping Bucket 100000 E This instrument measures rainfall by counting the times a bucket tips over due to filling with rainfall. Aperture 400 cm3 (?) Bucket size 0.2mm (?)
DOI: 10.4121/uuid:d003fcf9-b96d-4209-a123-1f9803c49fd2
title: Measured data from sensor IMD at Westerbork for month 2005-09

description: Parent item: All data measured by interception measurement device in Westerbork Measuring instrument: Westerbork: Interception measurment device Evaporation measurement. Area basin 0.36 m2, 2x3 strain gauges: type FLA-6, resistance 120 Ohm. Valve (&lt; 18 jan 2006): electronic magnetic valve Type: MBMG2Z-1263008-12V. (KOBOLD Instrumentatie BV, Arnhem) Valve (&gt; 18 jan 2006): electronic magnetic valve (12VDC) Type: MBMG2Z-1264020. Messing 3/4 "(?) (KOBOLD Instrumentatie BV, Arnhem) Temperature sensor: LM335, output 10 mV/K (Farnell).
DOI: 10.4121/uuid:d0e1b27f-ce99-46ea-b481-c20c4c4a4da8
title: Measured data from sensor Dis at Attert for month 2004-08

description: measured water pressure Parent item: All data measured by discharge meter in Attert basin Measuring instrument: Useldange, Attert basin: Discharge meter ISCO 4120 ultrasonic sensor (ceramic pressure membrane). Measures water pressure (?). The sensor is installed under a bridge (?).
DOI: 10.4121/uuid:d0ffacf0-2e01-41f9-ae7f-1858b116a3f2
title: Measured data from sensor METEO at Huewelerbach2 for month 2005-06

description: Parent item: All data measured by meteorological station in Huewerbach basin Measuring instrument: Huewelerbach basin: Meteorological station
DOI: 10.4121/uuid:d23c720d-0c83-41a8-8456-7d9e89f80ad2
title: Calculated data from sensor Q2 at Maisbich for month 2004-11

description: Height notch above sensor0.017m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:d366f5df-6533-4dd6-b8ab-5900d65f02e0
title: Measured data from sensor GRW21 at Maisbich for month 2004-12

description: Height piezometer above sensor1.25m Parent item: All data measured by groundwater level piezometer GRW21 at Maisbich Measuring instrument: Maisbich basin: Groundwater level piezometer Code: GRW 21 Type: Temperature and Pressure sensor Make: Keller DCX-22 AA This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of piezometer above sensor: 1.18m.
DOI: 10.4121/uuid:d36d180a-c235-4e25-aadf-ab8e31dd9976
title: Measured data from sensor Dis at Michelau for month 2005-09

description: measured distance between water level and sensor level Parent item: All data measured by discharge meter in Michelau, Sure basin Measuring instrument: Michelau, Sure basin: Discharge meter ISCO 4110 ultrasonic sensor. The sensor is installed under a bridge. Measured distance between water level and sensor level (?)
DOI: 10.4121/uuid:d37f4954-4ef9-4bdd-847a-482c7bb5139a
title: Measured data from sensor TB at Maisbich for month 2006-11

description: Parent item: All data measured by tipping bucket discharge meter in Maisbich Measuring instrument: Maisbich basin: Tipping bucket rainfall meter Type: Casella Tipping Bucket 100000 E This instrument measures rainfall by counting the times a bucket tips over due to filling with rainfall. Aperture 400 cm3 (?) Bucket size 0.2mm (?)
DOI: 10.4121/uuid:d4551f73-4840-48a1-a379-af6e541719db
title: Calculated data from sensor Q2 at Maisbich for month 2006-02

description: Height V-notch above sensor-0.017m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:d4d077b8-76f8-4a22-8424-12c5efaee97c
title: Calculated data from sensor Q2 at Maisbich for month 2004-07

description: Height notch above sensor unknown Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:d4de03f8-3816-40dd-9382-0300534bca3a
title: Measured data from sensor GRW21 at Maisbich for month 2006-08

description: Height piezometer above sensor1.25m Parent item: All data measured by groundwater level piezometer GRW21 at Maisbich Measuring instrument: Maisbich basin: Groundwater level piezometer Code: GRW 21 Type: Temperature and Pressure sensor Make: Keller DCX-22 AA This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of piezometer above sensor: 1.18m.
DOI: 10.4121/uuid:d547f434-6531-4bd1-9e6c-0b7245f3bc2d
title: Measured data from sensor GRW21 at Maisbich for month 2005-12

description: Height piezometer above sensor1.25m Parent item: All data measured by groundwater level piezometer GRW21 at Maisbich Measuring instrument: Maisbich basin: Groundwater level piezometer Code: GRW 21 Type: Temperature and Pressure sensor Make: Keller DCX-22 AA This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of piezometer above sensor: 1.18m.
DOI: 10.4121/uuid:d556d067-31ec-47c0-98c6-eac3caf5d720
title: Calculated data from sensor Q1 at Maisbich for month 2005-10

description: Height V-notch above sensor 0.145 m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff). Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:d6723018-b57c-4230-a689-7d385f8b2c6a
title: Calculated data from sensor Q1 at Maisbich for month 2006-07

description: Height V-notch above sensor 0.145 m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff). Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:d683bb5f-d75a-497c-b0a7-b3a4ae251791
title: Measured data from sensor Dis at Michelau for month 2004-06

description: measured distance between water level and sensor level Parent item: All data measured by discharge meter in Michelau, Sure basin Measuring instrument: Michelau, Sure basin: Discharge meter ISCO 4110 ultrasonic sensor. The sensor is installed under a bridge. Measured distance between water level and sensor level (?)
DOI: 10.4121/uuid:d8607908-5df3-4299-8d9d-d84d0e38dc73
title: Measured data from sensor Q2 at Maisbich for month 2006-06

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m.
DOI: 10.4121/uuid:d8ea722c-687c-4344-ba39-56a7802adb6d
title: Measured data from sensor Q1 at Maisbich for month 2005-05

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:d9391c58-7d35-4524-b218-ff876b52c630
title: Measured data from sensor TB at Maisbich for month 2006-02

description: Parent item: All data measured by tipping bucket discharge meter in Maisbich Measuring instrument: Maisbich basin: Tipping bucket rainfall meter Type: Casella Tipping Bucket 100000 E This instrument measures rainfall by counting the times a bucket tips over due to filling with rainfall. Aperture 400 cm3 (?) Bucket size 0.2mm (?)
DOI: 10.4121/uuid:d99c78f5-c016-4a14-88c1-76d114214251
title: Measured data from sensor Q1 at Maisbich for month 2006-06

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:dab41128-9159-42df-b6d0-270a17e9e798
title: Measured data from sensor Dis at Attert for month 2005-12

description: measured water pressure Parent item: All data measured by discharge meter in Attert basin Measuring instrument: Useldange, Attert basin: Discharge meter ISCO 4120 ultrasonic sensor (ceramic pressure membrane). Measures water pressure (?). The sensor is installed under a bridge (?).
DOI: 10.4121/uuid:daf55258-3eb3-42bb-9d65-70b86ceb016f
title: Calculated data from sensor Q1 at Maisbich for month 2005-06

description: Height V-notch above sensor 0.145 m Missing values till 06-21 Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff). Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:db1413ed-9581-40de-858e-f75803335035
title: Measured data from sensor Q2 at Maisbich for month 2006-09

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m.
DOI: 10.4121/uuid:db7df3d1-b079-4ec2-a5f5-6906bab2edd9
title: Measured data from sensor Dis at Attert for month 2004-07

description: measured water pressure Parent item: All data measured by discharge meter in Attert basin Measuring instrument: Useldange, Attert basin: Discharge meter ISCO 4120 ultrasonic sensor (ceramic pressure membrane). Measures water pressure (?). The sensor is installed under a bridge (?).
DOI: 10.4121/uuid:dc73d74d-de8c-4d8e-a012-1d106f1062fc
title: Measured data from sensor Q3 at Maisbich for month 2006-09

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:dcf6b361-8a34-4837-82ae-edb7b9280d73
title: Calculated data from sensor Q1 at Maisbich for month 2006-11

description: Height V-notch above sensor 0.145 m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff). Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:dd2ed20d-c94e-4e2c-b190-3c6d79053433
title: Measured data from sensor Q1 at Maisbich for month 2004-08

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:def0cf48-6979-4210-8357-0784ebc7f110
title: Calculated data from sensor Q1 at Maisbich for month 2006-05

description: Height V-notch above sensor 0.145 m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff). Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:df8493b8-263e-4b4a-8c9b-79512759806e
title: Measured data from sensor METEO at Huewelerbach2 for month 2005-01

description: Parent item: All data measured by meteorological station in Huewerbach basin Measuring instrument: Huewelerbach basin: Meteorological station
DOI: 10.4121/uuid:e05578fd-09b1-4758-b8af-ee1ef168500c
title: Calculated data from sensor Q3 at Maisbich for month 2004-06

description: Height notch above sensor 0.0015m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4) Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:e1a10a64-e527-4f9f-be64-dde0821647f4
title: Calculated data from sensor Q2 at Maisbich for month 2004-09

description: Height notch above sensor 0.017m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:e2e566a4-8069-4e18-b76e-3a8fb8d7bdf8
title: Measured data from sensor Q1 at Maisbich for month 2004-12

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:e31b1543-a077-4cb2-8d41-17fa56096474
title: Calculated data from sensor Q2 at Maisbich for month 2004-03

description: Height notch above sensor 0.017m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:e4d9a8c5-9e84-4837-9cd4-a60d5d9ae749
title: Calculated data from sensor Q3 at Maisbich for month 2004-07

description: Height notch above sensor unknown Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4) Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:e52d03ed-35af-4dca-9ce5-c38d50492810
title: Measured data from sensor METEO at Huewelerbach2 for month 2005-05

description: Parent item: All data measured by meteorological station in Huewerbach basin Measuring instrument: Huewelerbach basin: Meteorological station
DOI: 10.4121/uuid:e5caace8-2d6f-4272-87f2-e1738e243144
title: Measured data from sensor Q1 at Maisbich for month 2006-10

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).