cometadata/datacite-titles-descriptions-related-identifiers huggingface dataset:
DOI: 10.5439/1973057
title: kasacrspeccmaskcross (a0)

description: The Ka-Band Scanning ARM Cloud Radar (KASACR) records cloud properties. ARM's scanning cloud radars are dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (1.7 kW peak power) or an X-band (20 kW peak power). Beamwidths for Ka-bands paired with W-bands are roughly matched at 0.3 degrees. The X-band beamwidth is approximately 1 degree. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the KASACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the KASACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ratio. KASACR data from the 2018–2019 Cloud, Aerosol, and Complex Terrain Interactions (CACTI) field campaign in Argentina are now available as b1-level products. Building on the original CACTI operational data, the b1-level products feature improved data quality resulting from extensive analyses and corrections. The data are cross-calibrated to a common point, datastreams are corrected for operational issues that occurred during the campaign, and several data quality masks and basic derived products are incorporated. For more information, read the CACTI radar b1-level processing report.
DOI: 10.5439/1973061
title: aosaeth (01)

description: The aethalometer provides a real-time readout of the concentration of “black” or “elemental” carbon aerosol particles (BC or E) in an air stream. It measures the rate of change of optical transmission through a spot on a filter where aerosol is continuously collected and calculates the concentration of optically absorbing material in the sample. With a built-in reference check of the optical source and a mass flowmeter to monitor air flow rate, the aethalometer calculates the mean BC content of the sampled air stream and provides on-the-spot readout.
DOI: 10.5439/1973062
title: aosuhsas (01)

description: The ultra-high-sensitivity aerosol spectrometer (UHSAS) is an optical-scattering, laser-based, aerosol particle spectrometer system for sizing particles in the 60 to 1000 nanometer (nm) range [1–3]. The instrument counts particles in up to 100 user-specified sizing bins. The instrument’s laser illuminates particles, which scatter light. The system captures the peak light signals that are generated. These signals are used for particle sizing, since the amount of light scattered correlates strongly with particle size. The UHSAS is part of the aerosol observing system (AOS).
DOI: 10.5439/1973371
title: aosccn100 (01)

description: The cloud condensation nuclei counter measures the concentration of aerosol particles by drawing an air sample through a column with thermodynamically unstable supersaturated water vapor that can condense onto aerosol particles. Particles that are activated, i.e., grown larger in the process, are counted and sized by an optical particle counter (OPC). In this way the CCN measures activated ambient aerosol particle number concentration as a function of supersaturation. ARM operates both one- and two-column versions of the CCN.
DOI: 10.5439/1973569
title: mwacrspeccmaskcopol (a0)

description: The Marine W-Band (95 GHz) ARM Cloud Radar measures cloud properties, including radar Doppler and radar reflectivity. This instrument is a zenith-pointing W-Band ARM Cloud Radar (WACR) mounted on a stabilized platform for use aboard ships.
DOI: 10.5439/1973572
title: xsapr (i0)

description: The X-band Scanning ARM Precipitation Radar (X-SAPR) is an X-band dual-polarization Doppler weather radar. The X-SAPR operates in a simultaneous transmit and receive (STAR) mode, meaning that the transmit signal is split so that power is transmitted on both horizontal and vertical polarizations at the same time. The X-SAPR transmitter is a 200 kW (peak power) magnetron. The receiver is based upon the Vaisala RVP-900 and runs Vaisala’s IRIS software. In addition to the first three Doppler moments (reflectivity, radial velocity, and spectra width), the X-SAPR also provides differential reflectivity, correlation coefficient, and specific differential phases. The dual-polarization variables enable estimates of rainfall rates and identification of precipitation types. At the Southern Great Plains site, three X-SAPRs surround the Central Facility, allowing the use of multi-Doppler velocity retrievals to estimate wind fields.
DOI: 10.5439/1973910
title: mwacraux (a0)

description: The Marine W-Band (95 GHz) ARM Cloud Radar measures cloud properties, including radar Doppler and radar reflectivity. This instrument is a zenith-pointing W-Band ARM Cloud Radar (WACR) mounted on a stabilized platform for use aboard ships.
DOI: 10.5439/1973915
title: kazrspec (00)

description: The Ka-band ARM Zenith Radar (KAZR) remotely probes the extent and composition of clouds at millimeter wavelengths. The KAZR is a zenith-pointing Doppler radar that operates at a frequency of approximately 35 GHz. The main purpose of this radar is to determine the first three Doppler moments (reflectivity, vertical velocity, and spectral width) at a range resolution of approximately 30 meters from near-ground to nearly 20 kilometers in altitude. The KAZR replaces the millimeter-wavelength cloud radar (MMCR) and uses a new digital receiver that provides higher spatial and temporal resolution than the MMCR. In addition, spectral artifacts in the data are significantly reduced in the KAZR, allowing researchers to study cloud dynamics much more closely than with the MMCR. KAZR data from the 2018–2019 Cloud, Aerosol, and Complex Terrain Interactions (CACTI) field campaign in Argentina are now available as b1-level products. Building on the original CACTI operational data, the b1-level products feature improved data quality resulting from extensive analyses and corrections. The data are cross-calibrated to a common point, datastreams are corrected for operational issues that occurred during the campaign, and several data quality masks and basic derived products are incorporated. For more information, read the CACTI radar b1-level processing report.
DOI: 10.5439/1974513
title: wsacrcrcal (b1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1974676
title: wsacrslr (b1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1974678
title: kasacrcrraster (a1)

description: The Ka-Band Scanning ARM Cloud Radar (KASACR) records cloud properties. ARM's scanning cloud radars are dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (1.7 kW peak power) or an X-band (20 kW peak power). Beamwidths for Ka-bands paired with W-bands are roughly matched at 0.3 degrees. The X-band beamwidth is approximately 1 degree. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the KASACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the KASACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ratio. KASACR data from the 2018–2019 Cloud, Aerosol, and Complex Terrain Interactions (CACTI) field campaign in Argentina are now available as b1-level products. Building on the original CACTI operational data, the b1-level products feature improved data quality resulting from extensive analyses and corrections. The data are cross-calibrated to a common point, datastreams are corrected for operational issues that occurred during the campaign, and several data quality masks and basic derived products are incorporated. For more information, read the CACTI radar b1-level processing report.
DOI: 10.5439/1974855
title: aosnephwet (c1)

description: The nephelometer measures the total scattering and hemispheric backscattering of aerosol. Nephelometers are deployed in pairs, with one measuring the ambient conditions and the other measuring the scattering as a function of slowly increasing or decreasing relative humidity (RH). The combination can then be used to derive the hygroscopic growth factor as a function of RH: f(RH). The nephelometer is part of the aerosol observing system (AOS).
DOI: 10.5439/1974857
title: aospsap3w (c1)

description: The particle soot absorption photometer collects aerosol particles on a substrate and measures the change in light transmission relative to a reference filter. Bulk particle absorption is derived after correcting for scattering effects. The PSAP is part of the aerosol observing system (AOS).
DOI: 10.5439/1974858
title: aosnephdry (c1)

description: The nephelometer measures the total scattering and hemispheric backscattering of aerosol. Nephelometers are deployed in pairs, with one measuring the ambient conditions and the other measuring the scattering as a function of slowly increasing or decreasing relative humidity (RH). The combination can then be used to derive the hygroscopic growth factor as a function of RH: f(RH). The nephelometer is part of the aerosol observing system (AOS).
DOI: 10.5439/1974860
title: aosaeth (a1)

description: The aethalometer provides a real-time readout of the concentration of “black” or “elemental” carbon aerosol particles (BC or E) in an air stream. It measures the rate of change of optical transmission through a spot on a filter where aerosol is continuously collected and calculates the concentration of optically absorbing material in the sample. With a built-in reference check of the optical source and a mass flowmeter to monitor air flow rate, the aethalometer calculates the mean BC content of the sampled air stream and provides on-the-spot readout.
DOI: 10.5439/1974861
title: aoscpcf (01)

description: The condensation particle counter (CPC) measures sub-micron aerosol particle number concentration by condensing fluid onto particles and causing them to grow to sizes that are easily detectable by optical scattering. The condensing liquid is typically an alcohol (e.g., butanol) or water. Depending on the CPC type, typical CPC instruments can count particles with sizes that range from 3 to 3000 nanometers (ultrafine CPC) or from 10 to 3000 nanometers (fine CPC). The CPC is part of the aerosol observing system (AOS).
DOI: 10.5439/1974862
title: wsacrcrraster (a1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1974863
title: wsacrblrhi (b1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1974864
title: wsacrawrhi (b1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1975386
title: wsacrrhi (b1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1975387
title: wsacrcwrhi (b1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1975388
title: wsacrhsrhi (b1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1975389
title: wsacrppi (b1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1975390
title: wsacrvpt (b1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1975394
title: kasacrcrcal (a1)

description: The Ka-Band Scanning ARM Cloud Radar (KASACR) records cloud properties. ARM's scanning cloud radars are dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (1.7 kW peak power) or an X-band (20 kW peak power). Beamwidths for Ka-bands paired with W-bands are roughly matched at 0.3 degrees. The X-band beamwidth is approximately 1 degree. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the KASACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the KASACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ratio. KASACR data from the 2018–2019 Cloud, Aerosol, and Complex Terrain Interactions (CACTI) field campaign in Argentina are now available as b1-level products. Building on the original CACTI operational data, the b1-level products feature improved data quality resulting from extensive analyses and corrections. The data are cross-calibrated to a common point, datastreams are corrected for operational issues that occurred during the campaign, and several data quality masks and basic derived products are incorporated. For more information, read the CACTI radar b1-level processing report.
DOI: 10.5439/1975467
title: wsacrcrcal (a1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.
DOI: 10.5439/1975468
title: wsacrcrrhi (a1)

description: ARM's scanning cloud radars are fully coherent dual-frequency, dual-polarization Doppler radars mounted on a common scanning pedestal. Each pedestal includes a Ka-band radar (2kW peak power) and the deployment location determines whether the second radar is a W-band (WSACR; 1.7 kW peak power) or X-band (XSACR; 20 kW peak power). Beamwidths at Ka and W bands are roughly matched at 0.3 degrees. Due to the narrow antenna beamwidth, ARM’s scanning cloud radars use scanning strategies that are unlike typical weather radars. Rather than focusing on plan position indicator, or PPI, scans, the Ka-SACR uses range height indicator, or RHI, scans at numerous azimuths to obtain cloud volume data. Measurements collected with the W-SACR are copolar and cross-polar radar reflectivity, Doppler velocity, spectra width and spectra when not scanning, and linear depolarization ration.