DOI: 10.5281/zenodo.5573872
title: An Audibility Model of the Bone Conduction Device during Headband Trial in Single-sided Deaf Subjects.

description: <strong>Data of the study, including primary data, calculated data, analysis results and graphs.</strong> <strong>Related software</strong><br> zenodo.7295482 <strong>Abstract<br> Objective</strong><br> Modelling the head-shadow-effect compensation and speech intelligibility outcomes, we studied the benefits of fitting a bone conduction device (BCD) during the headband trial in single-sided deafened (SSD) subjects. <strong>Design</strong><br> The participants’ BCD settings were retrospectively used for measurements on the skull simulator. The sensation levels of the Bone-Conduction and Air-Conduction sound paths were compared, modelling three spatial conditions with the speech in quiet. When the difference between sensation levels was equivalent or greater than zero, this was scored as full head-shadow-effect compensation. We calculated the phoneme score using the Speech Intelligibility Index for the three conditions in quiet and seven in noise. <strong>Study sample</strong><br> Data from eighty-five SSD adults fitted with a BCD during the headband trial. <strong>Results</strong><br> According to our model, most subjects did not achieve a full head-shadow-effect compensation with the signal at the BCD side and in front. The modelled speech intelligibility in the quiet condition did not improve with the transcutaneous BCD compared with the unaided condition. In noise, we found a slight improvement in some specific conditions and minimal worsening in others. <strong>Conclusions</strong><br> Based on an audibility model, this study challenges the fundamentals of a trial period with a transcutaneous BCD in SSD subjects.
description: Related code at https://github.com/guidocattani/Audibility_Model_BCD_headband_trial_SSD.git This project is related to: Peters, J. P., van Zon, A., Smit, A. L., van Zanten, G. A., de Wit, G. A., Stegeman, I., Grolman, W. (2015). CINGLE-trial: cochlear implantation for siNGLE-sided deafness, a randomised controlled trial and economic evaluation. BMC Ear, Nose and Throat Disorders, 15(1), 3. https://doi.org/10.1186/s12901-015-0016-y
description: {"references": ["Bentler, R. A., Pavlovic, C. V, & others. (1989). Transfer functions and correction factors used in hearing aid evaluation and research. Ear Hear, 10(1), 58\u201363.", "Bosman, A. J., Kruyt, I. J., Mylanus, E. A. M., Hol, M. K. S., & Snik, A. F. M. (2018). On the evaluation of a superpower sound processor for bone-anchored hearing. Clinical Otolaryngology, 43(2), 450\u2013455. https://doi.org/10.1111/coa.12989", "Bosman, A.J. and G.F. Smoorenburg. (1992). Word lists for speech audiometry. [CD-ROM]. Dutch Association of Audiology (NVA).", "Bosman, A. J., & Smoorenburg, G. F. (1995). Intelligibility of Dutch CVC syllables and sentences for listeners with normal hearing and with three types of hearing impairment. Audiology\u202f: Official Organ of the International Society of Audiology, 34(5), 260\u2013284. Retrieved from http://www.ncbi.nlm.nih.gov/pubmed/8837785", "Carhart, R., & Jerger, J. F. (1959). Preferred Method For Clinical Determination Of Pure-Tone Thresholds. Journal of Speech and Hearing Disorders, 24(4), 330. https://doi.org/10.1044/jshd.2404.330", "Carlsson, P., H\u00e5kansson, B., & Ringdahl, A. (1995). Force threshold for hearing by direct bone conduction. The Journal of the Acoustical Society of America, 97(2), 1124\u20131129. https://doi.org/10.1121/1.412225", "Dillon, H. (2012). Hearing aids. Sydney\u202f: New York: Boomerang Press\u202f; Thieme.", "Faber, H. T., de Wolf, M. J. F., de Rooy, J. W. J., Hol, M. K. S., Cremers, C. W. R. J., & Mylanus, E. A. M. (2009). Bone-Anchored Hearing Aid Implant Location in Relation to Skin Reactions. Archives of Otolaryngology-Head & Neck Surgery, 135(8), 742. https://doi.org/10.1001/archoto.2009.99", "Flynn, M. C., & Hillbratt, M. (2012). Improving the Accuracy of Baha\u00ae Fittings through Measures of Direct Bone Conduction. Clinical and Experimental Otorhinolaryngology, 5(Suppl 1), S43. https://doi.org/10.3342/ceo.2012.5.S1.S43", "Gr\u00fcnder, I., Seidl, R. O., Ernst, A., & Todt, I. (2008). Wertigkeit der BAHA-Testung f\u00fcr das postoperative H\u00f6rergebnis. HNO, 56(10), 1020\u20131024. https://doi.org/10.1007/s00106-007-1608-7", "Hol, M. K. S., Kunst, S. J. W., Snik, A. F. M., & Cremers, C. W. R. J. (2010). Pilot study on the effectiveness of the conventional CROS, the transcranial CROS and the BAHA transcranial CROS in adults with unilateral inner ear deafness. European Archives of Oto-Rhino-Laryngology, 267(6), 889\u2013896. https://doi.org/10.1007/s00405-009-1147-9", "Holube, I., Fredelake, S., Vlaming, M., & Kollmeier, B. (2010) Development and analysis of an International Speech Test Signal (ISTS). International Journal of Audiology, 49:12, 891-903. https://doi.org/10.3109/14992027.2010.506889", "Kitterick, P. T., Smith, S. N., & Lucas, L. (2016). Hearing Instruments for Unilateral Severe-to-Profound Sensorineural Hearing Loss in Adults. Ear and Hearing, 37(5), 495\u2013507. https://doi.org/10.1097/AUD.0000000000000313", "Kompis, M., Pfiffner, F., Krebs, M., & Caversaccio, M.-D. (2011). Factors influencing the decision for Baha in unilateral deafness: the Bern benefit in single-sided deafness questionnaire. Advances in Oto-Rhino-Laryngology, 71, 103\u2013111. https://doi.org/10.1159/000323591", "Kurz, A., Flynn, M., Caversaccio, M., & Kompis, M. (2014). Speech Understanding with a New Implant Technology: A Comparative Study with a New Nonskin Penetrating Baha System. BioMed Research International, 2014, 1\u20139. https://doi.org/10.1155/2014/416205", "Lucas, L., Katiri, R., & Kitterick, P. T. (2018). The psychological and social consequences of single-sided deafness in adulthood. International Journal of Audiology, 57(1), 21\u201330. https://doi.org/10.1080/14992027.2017.1398420", "Mattingly, J. K., Greene, N. T., Jenkins, H. A., Tollin, D. J., Easter, J. R., & Cass, S. P. (2015). Effects of Skin Thickness on Cochlear Input Signal Using Transcutaneous Bone Conduction Implants. Otology & Neurotology, 36(8), 1403\u20131411. https://doi.org/10.1097/MAO.0000000000000814", "Monini, S., Musy, I., Filippi, C., Atturo, F., & Barbara, M. (2015). Bone conductive implants in single-sided deafness. Acta Oto-Laryngologica, 135(4), 381\u2013388. https://doi.org/10.3109/00016489.2014.990057", "Pennings, R. J. E., Gulliver, M., & Morris, D. P. (2011). The importance of an extended preoperative trial of BAHA in unilateral sensorineural hearing loss: a prospective cohort study. Clinical Otolaryngology\u202f: Official Journal of ENT-UK\u202f; Official Journal of Netherlands Society for Oto-Rhino-Laryngology & Cervico-Facial Surgery, 36(5), 442\u2013449. https://doi.org/10.1111/j.1749-4486.2011.02388.x", "Peters, J. P., van Zon, A., Smit, A. L., van Zanten, G. A., de Wit, G. A., Stegeman, I., & Grolman, W. (2015). CINGLE-trial: cochlear implantation for siNGLE-sided deafness, a randomised controlled trial and economic evaluation. BMC Ear, Nose and Throat Disorders, 15(1), 3. https://doi.org/10.1186/s12901-015-0016-y", "Peters, J. P. M., van Heteren, J. A. A., Wendrich, A. W., van Zanten, G. A., Grolman, W., Stokroos, R. J., & Smit, A. L. (2020). Short-term Outcomes of Cochlear Implantation for Single-Sided Deafness compared to Bone Conduction Devices and Contralateral Routing of Sound hearing aids \u2013 Results of a Randomised Controlled Trial (CINGLE-trial). Manuscript submitted for publication. https://dspace.library.uu.nl/handle/1874/400352", "Pfiffner, F., Kompis, M., Flynn, M., Asnes, K., Arnold, A., & Stieger, C. (2011). Benefits of low-frequency attenuation of baha\u00ae in single-sided sensorineural deafness. Ear and Hearing, 32(1), 40\u201345. https://doi.org/10.1097/AUD.0b013e3181ecd002", "Snapp, H. A., Fabry, D. A., Telischi, F. F., Arheart, K. L., & Angeli, S. I. (2010). A clinical protocol for predicting outcomes with an implantable prosthetic device (baha) in patients with single-sided deafness. Journal of the American Academy of Audiology, 21(10), 654\u2013662. https://doi.org/10.3766/jaaa.21.10.5", "Snapp, H., Angeli, S., Telischi, F. F., & Fabry, D. (2012). Postoperative Validation of Bone-Anchored Implants in the Single-Sided Deafness Population. Otology & Neurotology, 34(4), 777\u2013778. https://doi.org/10.1097/MAO.0b013e31828bb1cc", "Snapp, H. A., Morgenstein, K. E., & Kuzbyt, B. (2019). Speech Perception Outcomes in Transcutaneous Versus Percutaneous Bone Conduction Stimulation in Individuals With Single-sided Deafness. Otology & Neurotology, 40(8), 1068-1075. https://doi.org /10.1097/MAO.0000000000002362", "Snik, F. M., Mylanus, E. a M., Proops, D. W., Wolfaardt, J. F., Hodgetts, W. E., Somers, T., \u2026 Tjellstr\u00f6m, A. (2005). Consensus Statements on the BAHA System: Where Do We Stand at Present? Annals of Otology, Rhinology & Laryngology, 114(12_suppl), 2\u201312. https://doi.org/10.1177/0003489405114S1201", "Stenfelt, S. (2005). Bilateral fitting of BAHAs and BAHA\u00ae fitted in unilateral deaf persons: Acoustical aspects. International Journal of Audiology, 44(3), 178\u2013189. https://doi.org/10.1080/14992020500031561", "Stenfelt, S. (2012). Transcranial attenuation of bone-conducted sound when stimulation is at the mastoid and at the bone conduction hearing aid position. Otology & Neurotology\u202f: Official Publication of the American Otological Society, American Neurotology Society [and] European Academy of Otology and Neurotology, 33(2), 105\u2013114. https://doi.org/10.1097/MAO.0b013e31823e28ab", "Verstraeten, N., Zarowski, A. J., Somers, T., Riff, D., & Offeciers, E. F. (2009). Comparison of the Audiologic Results Obtained With the Bone-Anchored Hearing Aid Attached to the Headband, the Testband, and to the \"Snap\" Abutment. Otology & Neurotology, 30(1), 70\u201375. https://doi.org/10.1097/MAO.0b013e31818be97a", "Wendrich, A. W., Kroese, T. E., Peters, J. P. M., Cattani, G., & Grolman, W. (2017). Systematic Review on the Trial Period for Bone Conduction Devices in Single-Sided Deafness: Rates and Reasons for Rejection. Otology & Neurotology\u202f: Official Publication of the American Otological Society, American Neurotology Society [and] European Academy of Otology and Neurotology, 38(5), 632\u2013641. https://doi.org/10.1097/MAO.0000000000001405", "Zwartenkot, J. W., Snik, A. F. M., Mylanus, E. A. M., & Mulder, J. J. S. (2014). Amplification Options for Patients With Mixed Hearing Loss. Otology & Neurotology, 35(2), 221\u2013226. https://doi.org/10.1097/MAO.0000000000000258", "American National Standards Institute. (1997). Methods For Calculation Of The Speech Intelligibility Index (ANSI/ASA Standard No. 3.5).", "International Organization for Standardization. (1998). Acoustics \u2014 Reference zero for the calibration of audiometric equipment \u2014 Part 1: Reference equivalent threshold sound pressure levels for pure tones and supra-aural earphones (ISO Standard No. 389-1)."]}