cometadata/datacite-titles-descriptions-related-identifiers huggingface dataset:
DOI: 10.4121/uuid:a061bd75-6175-4666-8e21-dc8a57ab0719
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2007-09

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:a1b967f8-a979-4382-a625-5084578e508e
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2010-06

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:a4c21f52-cc60-418b-9ab4-ff42035066ef
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2010-02

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:a7315e4b-2a27-4acc-b5a7-ac7bb45d87bc
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2008-04

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:a93a1fb8-48bd-4c07-b54c-bea0b9b4fc08
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2007-07

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:ac326663-18ca-44ee-9cba-4e35872d8faf
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2009-08

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:ad5bcd9e-1674-45ad-ac61-9d1f6690ad33
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2008-02

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:b35f5df1-5d7f-48b1-be57-2000b740616b
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2007-02

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:b67370fb-4f8e-4577-8c70-c9020361822b
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2007-10

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:b6dce0c2-26d3-42be-90a2-179ac0d4e940
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2007-12

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:b985dbc2-591b-4241-9308-e0e1a2aac911
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2010-09

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:c0bb7587-6fe6-48e6-97ba-980a38149dbd
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2007-02

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:c12078da-7766-436c-87f7-596de2b55736
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2008-12

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:c294581e-7ec0-4a39-916e-f2596cba5b4a
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2007-12

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:c2c609b5-7159-4a98-aa99-7e66d8422dc2
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2009-06

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:c3302c1b-d949-45b8-8080-517bd504b0c0
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2007-11

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:c79d33f6-bcda-4983-a6be-3b52b34aedec
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2009-05

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:c86779bf-1b53-4f9d-b169-0511591b7dc6
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2007-01

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)