DOI: 10.6084/m9.figshare.7791152.v1
title: Supplementary Material for: A Pivotal Study to Validate the Performance of a Novel Wearable Sensor and System for Biometric Monitoring in Clinical and Remote Environments

description: <b><i>Background:</i></b> Increasingly, drug and device clinical trials are tracking activity levels and other quality of life indices as endpoints for therapeutic efficacy. Trials have traditionally required intermittent subject visits to the clinic that are artificial, activity-intensive, and infrequent, making trend and event detection between visits difficult. Thus, there is an unmet need for wearable sensors that produce clinical quality and medical grade physiological data from subjects in the home. The current study was designed to validate the BioStamp nPoint® system (MC10 Inc., Lexington, MA, USA), a new technology designed to meet this need. <b><i>Objective:</i></b> To evaluate the accuracy, performance, and ease of use of an end-to-end system called the BioStamp nPoint. The system consists of an investigator portal for design of trials and data review, conformal, low-profile, wearable biosensors that adhere to the skin, a companion technology for wireless data transfer to a proprietary cloud, and algorithms for analyzing physiological, biometric, and contextual data for clinical research. <b><i>Methods:</i></b> A prospective, nonrandomized clinical trial was conducted on 30 healthy adult volunteers over the course of two continuous days and nights. Supervised and unsupervised study activities enabled performance validation in clinical and remote (simulated “at home”) environments. System outputs for heart rate (HR), heart rate variability (HRV) (including root mean square of successive differences [RMSSD] and low frequency/high frequency ratio), activity classification during prescribed activities (lying, sitting, standing, walking, stationary biking, and sleep), step count during walking, posture characterization, and sleep metrics including onset/wake times, sleep duration, and respiration rate (RR) during sleep were evaluated. Outputs were compared to FDA-cleared comparator devices for HR, HRV, and RR and to ground truth investigator observations for activity and posture classifications, step count, and sleep events. <b><i>Results:</i></b> Thirty participants (77% male, 23% female; mean age 35.9 ± 10.1 years; mean BMI 28.1 ± 3.6) were enrolled in the study. The BioStamp nPoint system accurately measured HR and HRV (correlations: HR = 0.957, HRV RMSSD = 0.965, HRV ratio = 0.861) when compared to Actiheart<sup>TM</sup>. The system accurately monitored RR (mean absolute error [MAE] = 1.3 breaths/min) during sleep when compared to a Capnostream35<sup>TM</sup> end-tidal CO<sub>2</sub> monitor. When compared with investigator observations, the system correctly classified activities and posture (agreement = 98.7 and 92.9%, respectively), step count (MAE = 14.7, &lt; 3% of actual steps during a 6-min walk), and sleep events (MAE: sleep onset = 6.8 min, wake = 11.5 min, sleep duration = 13.7 min) with high accuracy. Participants indicated “good” to “excellent” usability (average System Usability Scale score of 81.3) and preferred the BioStamp nPoint system over both the Actiheart (86%) and Capnostream (97%) devices. <b><i>Conclusions:</i></b> The present study validated the BioStamp nPoint system’s performance and ease of use compared to FDA-cleared comparator devices in both the clinic and remote (home) environments.