cometadata/datacite-titles-descriptions-related-identifiers huggingface dataset:
DOI: 10.48322/64q8-vn69
title: MMS 1 Fast Plasma Investigation, Dual Ion Spectrometer (FPI, DIS) Partial Distribution Moments, Level 2 (L2), Burst Mode, 0.15 s Data
title: MMS1_FPI_BRST_L2_DIS-PARTMOMS

description: FPI usually operates in Fast Survey (FS) Mode in the MMS Region Of Interest (ROI) for the current Mission Phase. Data are taken at Burst (30/150 ms for DES/DIS) Resolution in this Mode. Data are also made available at Survey (4.5 s, etc.) Resolution. Per Mission Design, not all Burst Resolution Data are downlinked, but all Survey Data are downlinked. Planning around Calibration Activities, avoidance of Earth Radiation Belts, etc., when possible, FPI usually operates in Slow Survey (SS) Mode (60 s Resolution) outside of ROI, and then only the 60 s Resolution Survey Data are available. This Product contains Partial Moments that come from performing the Standard Moment Integrals over a limited Portion of Velocity Space. The resulting Quantities are named similarly to their corresponding Standard Moments, but are decorated with 'part' to differentiate. For example, density_part is the Density Moment integrated from a particular Energy Step to Infinity. These Partial Moments are formed from the indicated Data Type (DES/DIS Burst, FS or SS). For convenience, some additional Parameters are included to augment those most commonly found in a Moments Product of this sort, plus Time Stamps and other Annotation characterizing the State of the Instrument System at the indicated Time.
DOI: 10.48322/bqvd-yf57
title: MMS 1 Fast Plasma Investigation, Dual Ion Spectrometer (FPI, DIS) Instrument Distributions, Level 2 (L2), Fast Mode, 4.5 s Data
title: MMS1_FPI_FAST_L2_DIS-DIST

description: The Fast Plasma Instrument (FPI) usually Operates in Fast Survey (FS) Mode in the MMS Region Of Interest (ROI) for the current Mission Phase. Data are taken at Burst (30/150 ms for DES/DIS) Resolution are aggregated onboard and made available at Survey (4.5 s) Resolution in this Mode. This Product contains Phase Space Distribution Maps of Results from surveying the High Resolution Observations during each 4.5 s Period. In particular, the (highest possible Quality at the Time of Release) corrected/converted "Fast Survey Sky Map" Distributions are reported with Time Stamps and other Annotation characterizing the State of the Instrument System at the indicated Time.
DOI: 10.48322/kgr0-yb53
title: MMS 1 Fast Plasma Investigation, Dual Ion Spectrometer (FPI, DIS) Distribution Moments, Level 2 (L2), Fast Mode, 4.5 s Data
title: MMS1_FPI_FAST_L2_DIS-MOMS

description: The Fast Plasma Instrument (FPI) usually Operates in Fast Survey (FS) Mode in the MMS Region Of Interest (ROI) for the current Mission Phase. Data are taken at Burst (30/150 ms for DES/DIS) Resolution are aggregated onboard and made available at Survey (4.5 s) Resolution in this Mode. Planning around Calibration Activities, avoidance of Earth Radiation Belts, etc., when possible, FPI usually Operates in Slow Survey (SS) Mode outside of ROI, and then only the 60 s Resolution Survey Data are available. This Product contains Results from integrating the standard Moments of Phase Space Distributions formed from the indicated Data Type (DES/DIS Burst, FS or SS). For Convenience, some additional Parameters are included to augment those most commonly found in a Moments Product of this sort, plus Time Stamps and other Annotation characterizing the State of the Instrument System at the indicated Time.
DOI: 10.48322/0bbk-4s61
title: MMS 1 Fast Plasma Investigation, Dual Ion Spectrometer (FPI, DIS) Partial Distribution Moments, Level 2 (L2), Fast Mode, 4.5 s Data
title: MMS1_FPI_FAST_L2_DIS-PARTMOMS

description: FPI usually operates in Fast Survey (FS) Mode in the MMS Region Of Interest (ROI) for the current Mission Phase. Data are taken at Burst (30/150 ms for DES/DIS) Resolution in this Mode. Data are also made available at Survey (4.5 s, etc.) Resolution. Per Mission Design, not all Burst Resolution Data are downlinked, but all Survey Data are downlinked. Planning around Calibration Activities, avoidance of Earth Radiation Belts, etc., when possible, FPI usually operates in Slow Survey (SS) Mode (60 s Resolution) outside of ROI, and then only the 60 s Resolution Survey Data are available. This Product contains Partial Moments that come from performing the Standard Moment Integrals over a limited Portion of Velocity Space. The resulting Quantities are named similarly to their corresponding Standard Moments, but are decorated with 'part' to differentiate. For example, density_part is the Density Moment integrated from a particular Energy Step to Infinity. These Partial Moments are formed from the indicated Data Type (DES/DIS Burst, FS or SS). For convenience, some additional Parameters are included to augment those most commonly found in a Moments Product of this sort, plus Time Stamps and other Annotation characterizing the State of the Instrument System at the indicated Time.
DOI: 10.48322/jt8z-7x15
title: MMS 1 Digital Signal Processor (DSP) Search Coil Magnetometer (SCM), Magnetic Field Power Spectral Density, Level 2 (L2), Fast Mode, 2 s Data
title: MMS1_DSP_FAST_L2_BPSD

description: The MMS magnetic field power spectral density (BPSD) is computed onboard by the Digital Signal Processor (DSP). The fast Fourier transform (FFT) calculation is performed on a digitized version of analog signals from the Search Coil Magnetometer (SCM) in the SCM123 coordinate system, see SCM data product guide for details, https://lasp.colorado.edu/mms/sdc/public/datasets/fields/. This data product is computed in space from individual components that are not synchronized to the 1 second pulse. Therefore, the timing between channels can be inaccurate by a fraction of a second. The samples times are interval start times taken from the x component. The spectra are calculated via a 1024-point FFT algorithm on piecewise continuous sets of waveform data. Nine signals can be processed simultaneously. Six of the twelve DC-coupled E, DC-coupled V, or SCM signals (16384 samples/s) are selected for spectral processing at 100% duty cycle. In addition, the three AC-coupled signals (262,144 kS/s) each can be processed at 6.25% duty cycle. Each of the nine signals has 16, 1024-point FFT operations every second; the field-programmable gate array (FPGA) performs 144 FFTs per second. The FFT is performed by an arithmetic logic unit (ALU), which is controlled by a state machine. Both are hard-coded into the FPGA. The operation starts by applying a 1024-point Hanning window onto a waveform. Next, an FFT is implemented. The FFT is broken into a series of "butterfly" operations performed by the ALU. The result has real and imaginary data. Power spectra are calculated by taking the sum of squares of real and imaginary values (the ALU includes a multiplier), which produces a power spectrum with 512 frequency bins. The frequency bins are then combined to give pseudo-logarithmic frequency spacing (del f)/f. The spectra are reduced to 88 frequency bins with (del f)/f between 6% and 12% when possible. Narrow-band emissions can be fit to an accuracy of (del f)/f ~3%, allowing for an accurate determination of plasma density. The spectra can be averaged in time. The fastest reporting rate of any signal is 16 spectra per second. Reporting rates can be as slow a one spectra every 16 s (averaging 256 spectra). The DSP and SCM instrument papers can be found at https://link.springer.com/article/10.1007/s11214-014-0115-x and https://link.springer.com/article/10.1007/s11214-014-0096-9, respectively. The DSP and SCM data product guides can be found at https://lasp.colorado.edu/mms/sdc/public/datasets/fields/.
DOI: 10.48322/n5ky-bm34
title: MMS 1 Digital Signal Processor (DSP) Double Probe (ADP, SDP), Electric Field Power Spectral Density, Level 2 (L2), Fast Mode, 2 s Data
title: MMS1_DSP_FAST_L2_EPSD

description: The MMS electric field power spectral density (EPSD) is computed onboard by the Digital Signal Processor (DSP). The fast Fourier transform (FFT) calculation is performed on a digitized version of analog signals from the Axial Double Probe (ADP) and Spin-Plane Double Probe (SDP). This data product is computed in space from individual components that are not synchronized to the 1 second pulse. Therefore, the timing between channels can be inaccurate by a fraction of a second. The samples times are interval start times taken from the x component. The spectra are calculated via a 1024-point FFT algorithm on piecewise continuous sets of waveform data. Nine signals can be processed simultaneously. Six of the twelve DC-coupled E, DC-coupled V, or SCM signals (16384 samples/s) are selected for spectral processing at 100% duty cycle. In addition, the three AC-coupled signals (262,144 kS/s) each can be processed at 6.25% duty cycle. Each of the nine signals has 16, 1024-point FFT operations every second; the field-programmable gate array (FPGA) performs 144 FFTs per second. The FFT is performed by an arithmetic logic unit (ALU), which is controlled by a state machine. Both are hard-coded into the FPGA. The operation starts by applying a 1024-point Hanning window onto a waveform. Next, an FFT is implemented. The FFT is broken into a series of "butterfly" operations performed by the ALU. The result has real and imaginary data. Power spectra are calculated by taking the sum of squares of real and imaginary values (the ALU includes a multiplier), which produces a power spectrum with 512 frequency bins. The frequency bins are then combined to give pseudo-logarithmic frequency spacing (del f)/f. The spectra are reduced to 56 frequency bins with (del f)/f between 6% and 12% when possible. Narrow-band emissions can be fit to an accuracy of (del f)/f ~3%, allowing for an accurate determination of plasma density. The spectra can be averaged in time. The fastest reporting rate of any signal is 16 spectra per second. Reporting rates can be as slow a one spectra every 16 s (averaging 256 spectra). The averaging process has 48-bit accuracy to maximize the dynamic range. The amplitudes undergo a pseudo-logarithmic compression to an 8-bit number representing over 120 dB of dynamic range at ~5% precision.
DOI: 10.48322/k8g5-s923
title: MMS 1 Digital Signal Processor (DSP) Double Probe (ADP, SDP), Electric Field Power Spectral Density, Level 2 (L2), Slow Mode, 16 s Data
title: MMS1_DSP_SLOW_L2_EPSD

description: The MMS electric field power spectral density (EPSD) is computed onboard by the Digital Signal Processor (DSP). The fast Fourier transform (FFT) calculation is performed on a digitized version of analog signals from the Axial Double Probe (ADP) and Spin-Plane Double Probe (SDP). This data product is computed in space from individual components that are not synchronized to the 1 second pulse. Therefore, the timing between channels can be inaccurate by a fraction of a second. The samples times are interval start times taken from the x component. The spectra are calculated via a 1024-point FFT algorithm on piecewise continuous sets of waveform data. Nine signals can be processed simultaneously. Six of the twelve DC-coupled E, DC-coupled V, or SCM signals (16384 samples/s) are selected for spectral processing at 100% duty cycle. In addition, the three AC-coupled signals (262,144 kS/s) each can be processed at 6.25% duty cycle. Each of the nine signals has 16, 1024-point FFT operations every second; the field-programmable gate array (FPGA) performs 144 FFTs per second. The FFT is performed by an arithmetic logic unit (ALU), which is controlled by a state machine. Both are hard-coded into the FPGA. The operation starts by applying a 1024-point Hanning window onto a waveform. Next, an FFT is implemented. The FFT is broken into a series of "butterfly" operations performed by the ALU. The result has real and imaginary data. Power spectra are calculated by taking the sum of squares of real and imaginary values (the ALU includes a multiplier), which produces a power spectrum with 512 frequency bins. The frequency bins are then combined to give pseudo-logarithmic frequency spacing (del f)/f. The spectra are reduced to 56 frequency bins with (del f)/f between 6% and 12% when possible. Narrow-band emissions can be fit to an accuracy of (del f)/f ~3%, allowing for an accurate determination of plasma density. The spectra can be averaged in time. The fastest reporting rate of any signal is 16 spectra per second. Reporting rates can be as slow a one spectra every 16 s (averaging 256 spectra). The averaging process has 48-bit accuracy to maximize the dynamic range. The amplitudes undergo a pseudo-logarithmic compression to an 8-bit number representing over 120 dB of dynamic range at ~5% precision.
DOI: 10.48322/1cbf-hb73
title: MMS 1 Digital Signal Processor (DSP) Search Coil Magnetometer (SCM), Magnetic Field Power Spectral Density, Level 2 (L2), Slow Mode, 16 s Data
title: MMS1_DSP_SLOW_L2_BPSD

description: The MMS magnetic field power spectral density (BPSD) is computed onboard by the Digital Signal Processor (DSP). The fast Fourier transform (FFT) calculation is performed on a digitized version of analog signals from the Search Coil Magnetometer (SCM) in the SCM123 coordinate system (scm1 = - x sensor; scm2 = -z sensor; scm3 = -y sensor). This data product is computed in space from individual components that are not synchronized to the 1 second pulse. Therefore, the timing between channels can be inaccurate by a fraction of a second. The samples times are interval start times taken from the x component. The spectra are calculated via a 1024-point FFT algorithm on piecewise continuous sets of waveform data. Nine signals can be processed simultaneously. Six of the twelve DC-coupled E, DC-coupled V, or SCM signals (16384 samples/s) are selected for spectral processing at 100% duty cycle. In addition, the three AC-coupled signals (262,144 kS/s) each can be processed at 6.25% duty cycle. Each of the nine signals has 16, 1024-point FFT operations every second; the field-programmable gate array (FPGA) performs 144 FFTs per second. The FFT is performed by an arithmetic logic unit (ALU), which is controlled by a state machine. Both are hard-coded into the FPGA. The operation starts by applying a 1024-point Hanning window onto a waveform. Next, an FFT is implemented. The FFT is broken into a series of "butterfly" operations performed by the ALU. The result has real and imaginary data. Power spectra are calculated by taking the sum of squares of real and imaginary values (the ALU includes a multiplier), which produces a power spectrum with 512 frequency bins. The frequency bins are then combined to give pseudo-logarithmic frequency spacing (del f)/f. The spectra are reduced to 88 frequency bins with (del f)/f between 6% and 12% when possible. Narrow-band emissions can be fit to an accuracy of (del f)/f ~3%, allowing for an accurate determination of plasma density. The spectra can be averaged in time. The fastest reporting rate of any signal is 16 spectra per second. Reporting rates can be as slow a one spectra every 16 s (averaging 256 spectra). The averaging process has 48-bit accuracy to maximize the dynamic range. The amplitudes undergo a pseudo-logarithmic compression to an 8-bit number representing over 120 dB of dynamic range at ~5% precision.
DOI: 10.48322/7t5y-eq71
title: MMS 1 Electric Double Probe (EDP) Spacecraft Potential, Level 2 (L2), Burst Mode, 0.1220703125 ms Data
title: MMS1_EDP_BRST_L2_SCPOT

description: Electric Double Probe, Dual Probe Spacecraft Potential, Level 2, Burst Mode Data. Spacecraft potential is etstimatrd by averaging measurements of the probe-to-spacecraft potential from several probes.
DOI: 10.48322/rxbn-r719
title: MMS 1 Electric Double Probe (EDP) Three-Dimensional Electric Field, Level 2 (L2), Burst Mode, 0.1220703125 ms Data
title: MMS1_EDP_BRST_L2_DCE

description: Electric Double Probe, Three-Dimensional Electric Field, Level 2, Burst Mode Data
DOI: 10.48322/9ssk-hm65
title: MMS 1 Electric Double Probe (EDP) Axial Double Probe, Spin Plane Double Probe (ADP-SDP) Three-Dimensional HMFE Electric Field, Level 2 (L2), Burst Mode, 0.01525878906 ms Data
title: MMS1_EDP_BRST_L2_HMFE

description: Electric Double Probe, Three-Dimensional HMFE Electric Field, Level 2, Burst Mode, Level 1B AC Electric Field Data
DOI: 10.48322/1tgb-hx06
title: MMS 1 Electric Double Probe (EDP) Three-Dimensional Electric Field, Level 2 (L2), Fast Mode, 31.25 ms Data
title: MMS1_EDP_FAST_L2_DCE

description: Electric Double Probe, Quick-Look Three-Dimensional Electric Field, Level 2, Fast Survey Data
DOI: 10.48322/jjsp-6g51
title: MMS 1 Electric Double Probe (EDP) Spacecraft Potential, Level 2 (L2), Fast Mode, 31.25 ms Data
title: MMS1_EDP_FAST_L2_SCPOT

description: Electric Double Probe, Dual Probe Spacecraft Potential, Level 2, Fast Survey Data. Spacecraft potential is etstimatrd by averaging measurements of the probe-to-spacecraft potential from several probes.
DOI: 10.48322/3s8b-0j08
title: MMS 1 Electric Double Probe (EDP) Three-Dimensional Electric Field, Level 2 (L2), Slow Mode, 0.125 s Data
title: MMS1_EDP_SLOW_L2_DCE

description: Electric Double Probe, Quick-Look Three-Dimensional Electric Field, Level 2, Slow Survey Data
DOI: 10.48322/5t77-ka57
title: MMS 1 Electric Double Probe (EDP) Spacecraft Potential, Level 2 (L2), Slow Mode, 0.125 s Data
title: MMS1_EDP_SLOW_L2_SCPOT

description: Electric Double Probe, Dual Probe Spacecraft Potential, Level 2, Slow Survey Data. Spacecraft potential is etstimatrd by averaging measurements of the probe-to-spacecraft potential from several probes.
DOI: 10.48322/81bq-5z12
title: MMS 1 Electric Double Probe (EDP) Axial Double Probe, Spin Plane Double Probe (ADP-SDP) Electric Double Probe, High-Frequency Electric Field Spectra, Level 2 (L2), Survey Mode, 16 s Data
title: MMS1_EDP_SRVY_L2_HFESP

description: Electric Double Probe, High-Frequency AC Electric Field Spectra, Level 2, Survey Data, Level 1B AC Electric Field Data
DOI: 10.48322/pj0n-m695
title: MMS 1 Flux Gate Magnetometer (FGM) DC Magnetic Field, Level 2 (L2), Burst Mode, 128 Sample/s, v4/5 Data
title: MMS1_FGM_BRST_L2

description: The Fluxgate Magnetometers (FGM) on Magnetospheric Multiscale consist of a traditional Analog Fluxgate Magnetometer (AFG) and a Digital Fluxgate magnetometer (DFG). The dual magnetometers are operated as a single instrument providing a single intercalibrated data product. Range changes occur at different times on the two instruments so the gains checked each periapsis can be carried out unambiguously to apoapsis. Cross correlation of calibration parameters can separate causes of the any apparent calibration changes. Use of Electron Drift Instrument (EDI) to determine the field along the rotation axis allows accurate monitoring of the zero levels along the rotation axis. Prior to launch the magnetometers were calibrated at the Technical University, Braunschweig, except for the AFG magnetometers on MMS3 and MMS4, which were calibrated at UCLA. Both sets of sensors are operated for the entire MMS orbit, with slow survey (8 samples per second) outside of the Region of Interest (ROI), and fast survey (16 samples per second) inside the ROI. Within the ROI, burst mode data (128 samples per second) are also acquired. A detailed description of the MMS fluxgate magnetometers, including science objectives, instrument description, calibration, magnetic cleanliness program, and data flow can be found at http://link.springer.com/article/10.1007%2Fs11214-014-0057-3 (DOI 10.1007/s11214-014-0057-3). Additional information can also be found at http://www-spc.igpp.ucla.edu/ssc/mms (UCLA), and http://www.iwf.oeaw.ac.at/de/forschung/erdnaher-weltraum/mms/dfg (IWF, Graz). For the purpose of creating a unified FGM Level 2 data product, burst mode data is taken from DFG and survey mode data is taken from AFG. Because AFG and DFG are cross-calibrated on an orbit-averaged basis, small differences in offset may be observed between Level 2 burst and survey mode data. Consequently, any differences are within the error of the measurement. Based on preliminary analysis of the data, the absolute error within the Region of Interest (ROI) is estimated to be no more than 0.1 nT in the spin-plane, 0.15 nT along the spin-axis and 0.2 nT in total magnitude.
DOI: 10.48322/5dsj-t876
title: MMS 1 Search Coil Magnetometer (SCM) AC Magnetic Field Level 2 (L2), High Speed Burst Mode, 16384 Sample/s Data
title: MMS1_SCM_BRST_L2_SCHB

description: Search Coil Magnetometer (SCM) AC Magnetic Field (16384 samples/s), Level 2, High Speed Burst Mode Data. The tri-axial Search-Coil Magnetometer with its associated preamplifier measures three-dimensional magnetic field fluctuations. The analog magnetic waveforms measured by the SCM are digitized and processed inside the Digital Signal Processor (DSP), collected and stored by the Central Instrument Data Processor (CIDP) via the Fields Central Electronics Box (CEB). Prior to launch, all SCM Flight models were calibrated by LPP team members at the National Magnetic Observatory, Chambon-la-Foret (Orleans). Once per orbit, each SCM transfer function is checked thanks to the onboard calibration signal provided by the DSP. The SCM is operated for the entire MMS orbit in survey mode. Within scientific Regions Of Interest (ROI), burst mode data are also acquired as well as high speed burst mode data. This SCM data set corresponds to the AC magnetic field waveforms in nanoTesla and in the GSE frame. The SCM instrument paper for SCM can be found at http://link.springer.com/article/10.1007/s11214-014-0096-9 and the SCM data product guide at https://lasp.colorado.edu/mms/sdc/public/datasets/fields/.
DOI: 10.48322/mxbx-r466
title: MMS 1 Flux Gate Magnetometer (FGM) DC Magnetic Field, Level 2 (L2), Survey Mode, 8 or 16 Sample/s, v4/5 Data
title: MMS1_FGM_SRVY_L2

description: The Fluxgate Magnetometers (FGM) on Magnetospheric Multiscale consist of a traditional Analog Fluxgate Magnetometer (AFG) and a Digital Fluxgate magnetometer (DFG). The dual magnetometers are operated as a single instrument providing a single intercalibrated data product. Range changes occur at different times on the two instruments so the gains checked each periapsis can be carried out unambiguously to apoapsis. Cross correlation of calibration parameters can separate causes of the any apparent calibration changes. Use of Electron Drift Instrument (EDI) to determine the field along the rotation axis allows accurate monitoring of the zero levels along the rotation axis. Prior to launch the magnetometers were calibrated at the Technical University, Braunschweig, except for the AFG magnetometers on MMS3 and MMS4, which were calibrated at UCLA. Both sets of sensors are operated for the entire MMS orbit, with slow survey (8 samples per second) outside of the Region of Interest (ROI), and fast survey (16 samples per second) inside the ROI. Within the ROI, burst mode data (128 samples per second) are also acquired. A detailed description of the MMS fluxgate magnetometers, including science objectives, instrument description, calibration, magnetic cleanliness program, and data flow can be found at http://link.springer.com/article/10.1007%2Fs11214-014-0057-3 (DOI 10.1007/s11214-014-0057-3). Additional information can also be found at http://www-spc.igpp.ucla.edu/ssc/mms (UCLA), and http://www.iwf.oeaw.ac.at/de/forschung/erdnaher-weltraum/mms/dfg (IWF, Graz). For the purpose of creating a unified FGM Level 2 data product, burst mode data is taken from DFG and survey mode data is taken from AFG. Because AFG and DFG are cross-calibrated on an orbit-averaged basis, small differences in offset may be observed between Level 2 burst and survey mode data. Consequently, any differences are within the error of the measurement. Based on preliminary analysis of the data, the absolute error within the Region of Interest (ROI) is estimated to be no more than 0.1 nT in the spin-plane, 0.15 nT along the spin-axis and 0.2 nT in total magnitude.