cometadata/datacite-titles-descriptions-related-identifiers huggingface dataset:
DOI: 10.4121/uuid:c9000770-1f56-4040-bd19-ba789ad27709
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2010-10

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:cbc76752-cd08-4e26-9ce1-3f2c1b01d856
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2008-01

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:d2689550-f99e-47b2-92f5-ea8fae011352
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2009-12

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:d2cd30fc-ddec-4323-bcb5-519f4ac7b6a0
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2008-08

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:d2e940d9-5b7e-41dd-a957-12063bcc91cf
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2010-07

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:d39b1b7d-b509-40b8-bafc-b70fc1c0547e
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2008-07

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:df0c2403-abd2-4e31-9b62-7cc435d833d1
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2009-11

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:e1433a41-84ba-4861-9ba0-880f9ac0b226
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2008-02

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:e4217dd0-38f0-491c-b8e2-1ea2efcb7f87
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2007-04

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:e479e0f5-de76-4e9a-bb3e-54e0ad250438
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2008-10

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:e4d0fb09-8b79-4d0b-94fc-dd6009d144cb
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2008-03

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:e5fbbf46-988d-469f-890b-559c51c8cae7
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2009-04

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:e79a9d0f-07b9-4921-9e11-6096e3d7809f
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2010-04

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:e819d326-45aa-4857-831c-5bb00f852647
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2010-05

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:e8f5b8b5-961f-4b19-bc19-f71c78c3d680
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2009-11

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:e9130f87-d37e-49bb-ba53-9b082693136e
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2009-01

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:ebffa30d-b034-4c74-a627-0d1749a867d9
title: Maisbich (Luxemburg) river basin discharge data, sensor Q3, month 2007-08

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q3 (Qdown), 2006-11 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:ef9bbc12-cfc9-42e6-9a29-c28f022cd314
title: Maisbich (Luxemburg) river basin discharge data, sensor Q4, month 2010-03

description: Parent item: Maisbich (Luxemburg), data from discharge meter Q4 (Qup), 2006-12 and later Measuring instrument: Maisbich basin: V-notch discharge meter Q4 (Qup) Type: Van Essen TD-Driver, SN 19613 This instrument measures pressure with one absolute sensor. It also measures temperature. Height V-notch above sensor: 0.0018m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)