cometadata/datacite-titles-descriptions-related-identifiers huggingface dataset:
DOI: 10.4121/uuid:0b395d1b-ea70-4dfa-b988-60316cf867aa
title: Measured data from sensor Q3 at Maisbich for month 2004-03

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:0ce6081a-1da9-4928-aa99-9e1af924b972
title: Measured data from sensor Q3 at Maisbich for month 2004-12

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:0d05074a-de09-4049-9acd-23bed9939b34
title: Measured data from sensor IMD at Westerbork for month 2004-11

description: new data set (RG121004_01) 2004-10-12-13.55.58 Parent item: All data measured by interception measurement device in Westerbork Measuring instrument: Westerbork: Interception measurment device Evaporation measurement. Area basin 0.36 m2, 2x3 strain gauges: type FLA-6, resistance 120 Ohm. Valve (&lt; 18 jan 2006): electronic magnetic valve Type: MBMG2Z-1263008-12V. (KOBOLD Instrumentatie BV, Arnhem) Valve (&gt; 18 jan 2006): electronic magnetic valve (12VDC) Type: MBMG2Z-1264020. Messing 3/4 "(?) (KOBOLD Instrumentatie BV, Arnhem) Temperature sensor: LM335, output 10 mV/K (Farnell).
DOI: 10.4121/uuid:0d3af2d7-76e8-4618-a4be-b5ea1784c0f6
title: Measured data from sensor Q1 at Maisbich for month 2007-01

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:0dd39e7c-a4c3-4d04-b32a-6d14feec8084
title: Measured data from sensor TB at Maisbich for month 2006-10

description: Parent item: All data measured by tipping bucket discharge meter in Maisbich Measuring instrument: Maisbich basin: Tipping bucket rainfall meter Type: Casella Tipping Bucket 100000 E This instrument measures rainfall by counting the times a bucket tips over due to filling with rainfall. Aperture 400 cm3 (?) Bucket size 0.2mm (?)
DOI: 10.4121/uuid:0e18a88e-9ebc-403f-9c4a-64d5da48fe03
title: Calculated data from sensor Q1 at Maisbich for month 2005-01

description: Height V-notch above sensor0.145m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff). Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:0e93a0d8-04f5-4dc5-b7c7-6e8594351893
title: Calculated data from sensor Q2 at Maisbich for month 2006-10

description: Height V-notch above sensor -0.017 m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:105c1c0a-e94e-4279-97e0-001a4214eab2
title: Calculated data from sensor Q2 at Maisbich for month 2006-11

description: Height V-notch above sensor -0.017 m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:10b8df33-ad49-41e9-8784-b790512a00c2
title: Calculated data from sensor Q2 at Maisbich for month 2004-01

description: Height notch above sensor 0.017m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:10dde364-ce87-4d12-a8ac-52a5a54072b7
title: Measured data from sensor Q3 at Maisbich for month 2006-05

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:11bfb7e8-6de5-4dcb-b160-bb4a95a00044
title: Calculated data from sensor Q3 at Maisbich for month 2006-09

description: Height V-notch above sensor -0.0015 m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4) Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:11f36ec3-5411-4ab1-b648-27ff9169b5e1
title: Measured data from sensor Q1 at Maisbich for month 2006-01

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:1289b045-a601-4e75-99e9-94d6fbcef419
title: Measured data from sensor Q1 at Maisbich for month 2005-09

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:129f7301-bcac-46e1-a162-c1479f6ffe1b
title: Measured data from sensor IMD at Westerbork for month 2006-02

description: Parent item: All data measured by interception measurement device in Westerbork Measuring instrument: Westerbork: Interception measurment device Evaporation measurement. Area basin 0.36 m2, 2x3 strain gauges: type FLA-6, resistance 120 Ohm. Valve (&lt; 18 jan 2006): electronic magnetic valve Type: MBMG2Z-1263008-12V. (KOBOLD Instrumentatie BV, Arnhem) Valve (&gt; 18 jan 2006): electronic magnetic valve (12VDC) Type: MBMG2Z-1264020. Messing 3/4 "(?) (KOBOLD Instrumentatie BV, Arnhem) Temperature sensor: LM335, output 10 mV/K (Farnell).
DOI: 10.4121/uuid:12d22662-061c-4bf8-93e2-c327aea57f63
title: Measured data from sensor Q3 at Maisbich for month 2004-02

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:12dd178c-a7c5-42a4-a7b1-77b4ab39951a
title: Measured data from sensor Q2 at Maisbich for month 2004-09

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m.
DOI: 10.4121/uuid:12dd96c4-0f66-4f59-a9c4-583f4ce5110d
title: Measured data from sensor Q1 at Maisbich for month 2006-08

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:1457efcf-cc2b-4196-8578-22c7fdae1f50
title: Calculated data from sensor Q2 at Maisbich for month 2005-12

description: Height V-notch above sensor-0.017m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:5a0764f4-6900-48d4-adef-a0f79373e4f6
title: Measured data from sensor Q2 at Maisbich for month 2003-12

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m.
DOI: 10.4121/uuid:1536b9b6-3b2c-445e-ab36-f9a34f86c8ad
title: Measured data from sensor RR1 at Maisbich for month 2004-07

description: Measuring instrument: Maisbich basin: Road runoff discharge V-notch Type: Minitroll ssp-100. This instrument measures pressure with one absolute sensor. It also measures temperature.
DOI: 10.4121/uuid:153c9633-ce72-4cef-ae6d-256c35ee02a3
title: Measured data from sensor Q2 at Maisbich for month 2006-07

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m.
DOI: 10.4121/uuid:15a86163-dfc5-42f7-a36b-30ceb99cd22d
title: Measured data from sensor Dis at Michelau for month 2005-01

description: measured distance between water level and sensor level Parent item: All data measured by discharge meter in Michelau, Sure basin Measuring instrument: Michelau, Sure basin: Discharge meter ISCO 4110 ultrasonic sensor. The sensor is installed under a bridge. Measured distance between water level and sensor level (?)
DOI: 10.4121/uuid:15b63b0c-8f69-4a2d-a48d-cf4703f31bad
title: Measured data from sensor IMD at Huewelerbach2 for month 2005-01

description: measurements missing between 2005-01-10 16:19:34 and 2005-01-10 17:12:12 and after 2005-01-26 01:08:55 Parent item: All data measured by interception measurement device in Huewerbach basin Measuring instrument: Huewelerbach basin: Interception measurement device According to original data the location is Westerbork NL but this is probably wrong Evaporation measument. Area basin 1.00 m2, 2x3 strain gauges: type FLA-6, resistance 120 Ohm. Valve (&lt; 18 jan 2006): electronic magnetic valve Type: MBMG2Z-1263008-12V. (KOBOLD Instrumentatie BV, Arnhem) Valve (&gt;=18 jan 2006): electronic magnetic valve (12VDC) Type: MBMG2Z-1264020. Messing 3/4 " (KOBOLD Instrumentatie BV, Arnhem) Temperature sensor: LM335, output 10 mV/K (Farnell). Throughfall: tipping bucket & pluvio meters.
DOI: 10.4121/uuid:15fc1834-3155-4b0d-ab11-6e2f24cc708c
title: Measured data from sensor Dis at Michelau for month 2006-04

description: measured distance between water level and sensor level Parent item: All data measured by discharge meter in Michelau, Sure basin Measuring instrument: Michelau, Sure basin: Discharge meter ISCO 4110 ultrasonic sensor. The sensor is installed under a bridge. Measured distance between water level and sensor level (?)
DOI: 10.4121/uuid:1638c2a7-8657-45f3-a6ec-0a35f516086f
title: Calculated data from sensor Q2 at Maisbich for month 2006-12

description: Height V-notch above sensor -0.017 m Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:16a7982e-8446-485e-8c86-4fad4016513b
title: Measured data from sensor IMD at Westerbork for month 2005-06

description: Parent item: All data measured by interception measurement device in Westerbork Measuring instrument: Westerbork: Interception measurment device Evaporation measurement. Area basin 0.36 m2, 2x3 strain gauges: type FLA-6, resistance 120 Ohm. Valve (&lt; 18 jan 2006): electronic magnetic valve Type: MBMG2Z-1263008-12V. (KOBOLD Instrumentatie BV, Arnhem) Valve (&gt; 18 jan 2006): electronic magnetic valve (12VDC) Type: MBMG2Z-1264020. Messing 3/4 "(?) (KOBOLD Instrumentatie BV, Arnhem) Temperature sensor: LM335, output 10 mV/K (Farnell).
DOI: 10.4121/uuid:1786c445-c7cf-4873-b637-e876ce655ad5
title: Measured data from sensor Dis at Attert for month 2005-07

description: measured water pressure Parent item: All data measured by discharge meter in Attert basin Measuring instrument: Useldange, Attert basin: Discharge meter ISCO 4120 ultrasonic sensor (ceramic pressure membrane). Measures water pressure (?). The sensor is installed under a bridge (?).
DOI: 10.4121/uuid:179121f4-2604-4329-8363-e57e834c1237
title: Calculated data from sensor Q2 at Maisbich for month 2006-05

description: Height V-notch above sensor -0.017 m ; For 2006-05-02 at 10:56 measurement values are missing. Parent item: Maisbich (Luxemburg), calculated data from discharge meter Q2 Measuring instrument: Maisbich basin: V-notch discharge meter Q2 Type: Pressure Sensor Keller DCX-4.0, SN 1466 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.017m. Contributor: Durr, E.H.(Eugene)
DOI: 10.4121/uuid:17c65024-ec69-489f-821e-6ea9a34ba99e
title: Measured data from sensor Q1 at Maisbich for month 2003-12

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:1a5f4e80-e860-4b3d-98fd-9ccdf071db10
title: Measured data from sensor Q1 at Maisbich for month 2005-08

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:1b558437-ba9d-41b6-9c71-92cafd9f21b8
title: Measured data from sensor Q1 at Maisbich for month 2005-06

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q1, 2003-12 to 2008-05 Measuring instrument: Maisbich basin: V-notch discharge meter Q1 Type: Pressure Sensor Keller DCX-4.0, SN 1454 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: hV=0.145m. The discharge is calculated with the rating curve Q=C*(h-hV)^2.5 where C is a calibration constant and h results from the pressure difference (Willem Luxemburg, Martijn Westhoff).
DOI: 10.4121/uuid:1b81d75d-3b21-42c9-b117-31f9566615bc
title: Measured data from sensor Q3 at Maisbich for month 2004-04

description: Parent item: Maisbich (Luxemburg), measured data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)
DOI: 10.4121/uuid:1b899663-0045-4ca3-9710-c1677895f558
title: Measured data from sensor METEO at Huewelerbach2 for month 2005-04

description: Parent item: All data measured by meteorological station in Huewerbach basin Measuring instrument: Huewelerbach basin: Meteorological station