DOI: 10.16904/envidat.157
title: Aerosol Data Davos Wolfgang

description: Aerosol properties were measured between February 8 and March 31 2019 at the measurement site Davos Wolfgang (LON: 9.853594, LAT: 46.835577). Optical and aerodynamic particle counters, as well as a scanning mobility particle size spectrometer and an ice nuclei counter were deployed to report particle concentrations and size distributions in fine (10-1000 nm) and coarse mode (- 1000 nm), cloud condensation nuclei concentrations (CCNCs) and ice nuclei particle concentrations (ICNCs). The ambient particles were transported via a heated inlet to be distributed to the particle detecting devices inside the setup room.
Optical Particle Counter (OPC): Light scattering of a diode laser beam caused by travelling particles is used in the both, the OPC-N3 (0.41 - 38.5 &956;m) and GT-526S (0.3 – 5 &956;m), to determine their size and number concentration. For the OPC-N3, particle size spectra and concentration data are used afterwards to calculate PM&8321;, PM&8322;,&8325; and PM&8321;&8320; (assumptions: particle density: 1.65 g cm&713;&179;, refractive index: 1.5+i0).
Aerodynamic Particle Sizer (APS): The APS (3321, TSI Inc.) measured the particle size distribution for aerodynamic diameters between 0.5 &956;m and ~20 &956;m by the particle’s time-of-flight and light-scattering intensity (assumptions: particle density 1 g cm&713;&179;).
Scanning Mobility Particle Size Spectrometer (SMPS): Particle number concentrations in a size range between 12 and 460 nm (electrical mobility diameter) were measured at Davos Wolfgang, using a Scanning Mobility Particle Sizer Spectrometer (3938, TSI Inc.). The classifier (3082, TSI Inc.) was equipped with a neutralizer (3088, TSI Inc.) and a differential mobility analyzer working with negative polarity (3081, TSI Inc.). The size selected particles were counted by a water-based condensation particle counter (3788 , TSI Inc.). The TSI AIM software was used to provide particle size distributions by applying multiple charge and diffusion loss corrections (assumptions: particle density 1 g cm&713;&179;). 
Coriolis &956; and DRINCZ: A microbial air sampler (Coriolis &956;, bertin Instruments) was used to collect airborne particles for investigating their ice nucleating ability with a droplet freezing device. Particles larger than 0.5 &956;m were drawn with an air flow rate of up to 300 l min&713;&185; into the cone and centrifuged into the wall of the cone due to the forming vortex. The liquid sample was transferred into the DRoplet Ice Nuclei Counter Zurich (DRINCZ, ETH Zurich) to study heterogeneous ice formation (immersion freezing mode) of ambient airborne particles.
description: Resource Title: OPC-N3 Dataset Davos Wolfgang
 Size: 30756674 bytes
 Format: text/csv
 URL: https://www.envidat.ch/dataset/03a2914e-d0c6-438a-b4c0-5f576db641da/resource/2632179e-3c62-45f8-9454-c3d946ccf430/download/opc-n3_wolfgang.csv
 Description: Particle concentrations in a size range of 0.35 μm to 40 μm were measured with an OPC-N3 (Alphasense, Inc.) in a resolution of 20 seconds (time in UTC). PM₁, PM₂,₅ and PM₁₀ were directly derived.
description: Resource Title: OPC GT-526S Dataset Davos Wolfgang
 Size: 14718206 bytes
 Format: text/csv
 URL: https://www.envidat.ch/dataset/03a2914e-d0c6-438a-b4c0-5f576db641da/resource/a0a29389-e885-4e4d-bc5f-a65913a0ba46/download/opc-gt526s_wolfgang.csv
 Description: Particle counts were recorded in six size channels (0.3, 0.5, 0.8, 1.5, 2.5 and 5 μm) in a resolution of 20 seconds (time in UTC) by a particle counter (GT-526S, Met One Instruments, Inc.).
description: Resource Title: APS Dataset Davos Wolfgang
 Size: 100296188 bytes
 Format: application/x-netcdf
 URL: https://www.envidat.ch/dataset/03a2914e-d0c6-438a-b4c0-5f576db641da/resource/1b62d21e-674c-4fa4-884b-cce7f18801e6/download/aps_wolfgang.nc
 Description: The APS file contains aerosol data (total number concentration, median, mean, geometric mean, mode and geometric standard deviation of the size distribution) for aerodynamic diameters between 0.5 μm and ~20 μm as well as instrument characteristics, in a temporal resolution of 20 seconds, with time in UTC, measured at Davos Wolfgang.
description: Resource Title: SMPS Dataset Davos Wolfgang
 Size: 32383567 bytes
 Format: application/x-netcdf
 URL: https://www.envidat.ch/dataset/03a2914e-d0c6-438a-b4c0-5f576db641da/resource/1135c794-6c1c-49cb-9189-4d0a67f15b0d/download/smps_wolfgang.nc
 Description: The SMPS file contains aerosol data (particle concentration, total number concentration, median, mean, geometric mean, mode and geometric standard deviation of the size distribution) in a size range between 12 and 460 nm (electrical mobility diameter) as well as instrument characteristics (sample temperature, pressure, relative humidity, mean free path and gas viscosity), in a temporal resolution of 2 minutes, with time in UTC, measured at Davos Wolfgang.
description: Resource Title: DRINCZ Dataset Davos Wolfgang
 Size: 181636 bytes
 Format: application/x-netcdf
 URL: https://www.envidat.ch/dataset/03a2914e-d0c6-438a-b4c0-5f576db641da/resource/6c39beac-0605-4915-9f93-ef22ff5b29f4/download/raclets_inp_wop.nc
 Description: The DRINCZ file contains information on temperature (in °C) dependent frozen fractions, number of Ice Nuleating Particles (INP, per standard liter) and differential INP concentrations (per standard liter and Kelvin).