title: ÐÑÐ¸Ð±Ð¾Ñ Ð´Ð»Ñ ÑÐ½ÑÑÐ¸Ñ ÑÐµÑÐ¼Ð¾Ð³ÑÐ°ÑÐ¸ÑÐµÑÐºÐ¸Ñ Ð´Ð°Ð½Ð½ÑÑ
title: Ð²ÑÐ¿ÑÑÐºÐ½Ð°Ñ ÐºÐ²Ð°Ð»Ð¸ÑÐ¸ÐºÐ°ÑÐ¸Ð¾Ð½Ð½Ð°Ñ ÑÐ°Ð±Ð¾ÑÐ° Ð±Ð°ÐºÐ°Ð»Ð°Ð²ÑÐ°
titleType: Subtitle
title: Device for taking thermographic data
Creators: name: Ð¢Ð¾Ð»ÑÑÐ¾Ð¹, ÐÐ½Ð´ÑÐµÐ¹
Publisher: name: Ð¡Ð°Ð½ÐºÑ-ÐÐµÑÐµÑÐ±ÑÑÐ³ÑÐºÐ¸Ð¹ Ð¿Ð¾Ð»Ð¸ÑÐµÑÐ½Ð¸ÑÐµÑÐºÐ¸Ð¹ ÑÐ½Ð¸Ð²ÐµÑÑÐ¸ÑÐµÑ ÐÐµÑÑÐ° ÐÐµÐ»Ð¸ÐºÐ¾Ð³Ð¾
Year: 2022
Subjects: Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ðµ ÑÐµÐ¼Ð¿ÐµÑÐ°ÑÑÑÑ
Ð¸Ð½ÑÑÐ°ÐºÑÐ°ÑÐ½ÑÐ¹ Ð´Ð¸Ð°Ð¿Ð°Ð·Ð¾Ð½
Ð¸Ðº ÐºÐ°Ð¼ÐµÑÐ°
ÑÐµÐ¿Ð»Ð¾Ð²Ð¸Ð·Ð¾Ñ
ÑÐµÑÐ¼Ð¾Ð³ÑÐ°ÑÐ¸Ñ
Ð±ÐµÑÐºÐ¾Ð½ÑÐ°ÐºÑÐ½Ð¾Ðµ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ðµ
temperature measurement
infrared range
ir camera
thermal imager
thermography
non-contact measurement

description: Ð¢ÐµÐ¼Ð° Ð²ÑÐ¿ÑÑÐºÐ½Ð¾Ð¹ ÐºÐ²Ð°Ð»Ð¸ÑÐ¸ÐºÐ°ÑÐ¸Ð¾Ð½Ð½Ð¾Ð¹ ÑÐ°Ð±Ð¾ÑÑ: Â«ÐÑÐ¸Ð±Ð¾Ñ Ð´Ð»Ñ ÑÐ½ÑÑÐ¸Ñ ÑÐµÑÐ¼Ð¾Ð³ÑÐ°ÑÐ¸ÑÐµÑÐºÐ¸Ñ Ð´Ð°Ð½Ð½ÑÑÂ». ÐÐ°Ð½Ð½Ð°Ñ ÑÐ°Ð±Ð¾ÑÐ° Ð¿Ð¾ÑÐ²ÑÑÐµÐ½Ð° ÑÐ°Ð·ÑÐ°Ð±Ð¾ÑÐºÐµ Ð¿ÑÐ¸Ð±Ð¾ÑÐ° Ð´Ð»Ñ Ð±ÐµÑÐºÐ¾Ð½ÑÐ°ÐºÑÐ½Ð¾Ð³Ð¾ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ ÑÐµÐ¼Ð¿ÐµÑÐ°ÑÑÑÑ Ð¸ Ð¿Ð¾Ð»ÑÑÐµÐ½Ð¸Ñ ÑÐµÐ¿Ð»Ð¾Ð²Ð¾Ð¹ ÐºÐ°ÑÑÑ Ð¾Ð±ÑÐµÐºÑÐ°. ÐÐ°Ð´Ð°ÑÐ¸, ÐºÐ¾ÑÐ¾ÑÑÐµ ÑÐµÑÐ°Ð»Ð¸ÑÑ Ð² ÑÐ¾Ð´Ðµ ÑÐ°Ð±Ð¾ÑÑ: 1. ÐÐ½Ð°Ð»Ð¸Ð· Ð¾ÑÐ½Ð¾Ð²Ð½ÑÑ Ð¼ÐµÑÐ¾Ð´Ð¾Ð² Ð±ÐµÑÐºÐ¾Ð½ÑÐ°ÐºÑÐ½Ð¾Ð³Ð¾ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ ÑÐµÐ¼Ð¿ÐµÑÐ°ÑÑÑÑ. 2. ÐÐ¾Ð´Ð±Ð¾Ñ ÐºÐ¾Ð¼Ð¿Ð¾Ð½ÐµÐ½ÑÐ¾Ð² ÑÑÑÑÐ¾Ð¹ÑÑÐ²Ð°, ÑÐ°Ð·ÑÐ°Ð±Ð¾ÑÐºÐ° ÑÑÑÑÐºÑÑÑÐ½Ð¾Ð¹ ÑÑÐµÐ¼Ñ. 3. Ð Ð°Ð·ÑÐ°Ð±Ð¾ÑÐºÐ° Ð°Ð»Ð³Ð¾ÑÐ¸ÑÐ¼Ð° ÑÐ°Ð±Ð¾ÑÑ Ð¿ÑÐ¸Ð±Ð¾ÑÐ°. 4. Ð Ð°ÑÑÐµÑ Ð¸ Ð¾ÑÐµÐ½ÐºÐ° Ð¿Ð¾Ð³ÑÐµÑÐ½Ð¾ÑÑÐ¸ Ð¿ÑÐ¸Ð±Ð¾ÑÐ°. Ð Ð°Ð·ÑÐ°Ð±Ð¾ÑÐ°Ð½Ð½Ð¾Ðµ ÑÑÑÑÐ¾Ð¹ÑÑÐ²Ð¾ Ð±Ð°Ð·Ð¸ÑÑÐµÑÑÑ Ð½Ð° ÐºÐ¾Ð½ÑÑÐ¾Ð»Ð»ÐµÑÐµ MSP 432, Ð´Ð¸Ð°Ð¿Ð°Ð·Ð¾Ð½ Ð¸Ð·Ð¼ÐµÑÑÐµÐ¼ÑÑ ÑÐµÐ¼Ð¿ÐµÑÐ°ÑÑÑ Ð¾Ñ -10Â°Ð¡ Ð´Ð¾ +200Â°Ð¡, Ð²ÑÐµÐ¼Ñ Ð°Ð²ÑÐ¾Ð½Ð¾Ð¼Ð½Ð¾Ð¹ ÑÐ°Ð±Ð¾ÑÑ Ð½Ðµ Ð¼ÐµÐ½ÐµÐµ ÑÐµÑÑÑÐµÑ ÑÐ°ÑÐ¾Ð², ÑÐ°Ð·ÑÐµÑÐµÐ½Ð¸Ðµ Ð¸Ðº-Ð¼Ð°ÑÑÐ¸ÑÑ Ð½Ðµ Ð¼ÐµÐ½ÐµÐµ 60x80 Ð¿Ð¸ÐºÑÐµÐ»ÐµÐ¹. ÐÑÐ¾Ð²ÐµÐ´ÐµÐ½ Ð°Ð½Ð°Ð»Ð¸Ð· Ð¿Ð¾Ð³ÑÐµÑÐ½Ð¾ÑÑÐµÐ¹ Ð¸ Ð¾Ð¿Ð¸ÑÐ°Ð½ Ð¼ÐµÑÐ¾Ð´ ÐºÐ°Ð»Ð¸Ð±ÑÐ¾Ð²ÐºÐ¸ Ð½Ðµ Ð² Ð»Ð°Ð±Ð¾ÑÐ°ÑÐ¾ÑÐ½ÑÑ ÑÑÐ»Ð¾Ð²Ð¸ÑÑ, Ð¾Ð¿Ð¸ÑÐ°Ð½ Ð°Ð»Ð³Ð¾ÑÐ¸ÑÐ¼ Ð¿ÐµÑÐµÐ´Ð°ÑÐ¸ Ð¿Ð°ÐºÐµÑÐ¾Ð² Ð´Ð°Ð½Ð½ÑÑ Ñ ÑÐµÐ½ÑÐ¾ÑÐ° Ð¿Ð¾ ÑÐ¸Ð½Ðµ VoSPI Ð¸ Ð¿Ð¾ Ð¿ÑÐ¾ÑÐ¾ÐºÐ¾Ð»Ñ UDP.
description: The topic of the final qualification work: "Device for taking thermographic data".Â  This work is devoted to the development of a device for non-contact temperature measurement and obtaining a heat map of the object. Tasks that were solved in the course of work: 1. Analysis of the main methods of non-contact temperature measurement. 2. Selection of device components, development of a structural scheme. 3. Development of the algorithm of the device. 4. Calculation and evaluation of the error of the device.Â The developed device is based on the controller MSP 432, the range of measured temperatures from -10 Â° C to +200 Â° C, battery life of at least four hours, the resolution of the ir matrix is at least 60x80 pixels. The analysis of errors is carried out and the calibration method is described not in the laboratory, the algorithm for transmitting data packets from the sensor via the VoSPI bus and via the UDP protocol is described.