title: ÐÐµÑÐ¾Ð´Ð¸ÐºÐ° Ð¸ÑÐ¿ÑÑÐ°Ð½Ð¸Ð¹ Ð³Ð°Ð·Ð¾Ð°Ð½Ð°Ð»Ð¸Ð·Ð°ÑÐ¾ÑÐ° Ð¾Ð·Ð¾Ð½Ð°
title: Ð²ÑÐ¿ÑÑÐºÐ½Ð°Ñ ÐºÐ²Ð°Ð»Ð¸ÑÐ¸ÐºÐ°ÑÐ¸Ð¾Ð½Ð½Ð°Ñ ÑÐ°Ð±Ð¾ÑÐ° Ð±Ð°ÐºÐ°Ð»Ð°Ð²ÑÐ°
titleType: Subtitle
title: Method of testing the ozone gas analyzer
Creators: name: ÐÐ°Ð²ÑÐ¸Ð»Ð¾Ð², ÐÐ°Ð²ÐµÐ»
Publisher: name: Ð¡Ð°Ð½ÐºÑ-ÐÐµÑÐµÑÐ±ÑÑÐ³ÑÐºÐ¸Ð¹ Ð¿Ð¾Ð»Ð¸ÑÐµÑÐ½Ð¸ÑÐµÑÐºÐ¸Ð¹ ÑÐ½Ð¸Ð²ÐµÑÑÐ¸ÑÐµÑ ÐÐµÑÑÐ° ÐÐµÐ»Ð¸ÐºÐ¾Ð³Ð¾
Year: 2023
Subjects: Ð³Ð°Ð·Ð¾Ð°Ð½Ð°Ð»Ð¸Ð·Ð°ÑÐ¾Ñ Ð¾Ð·Ð¾Ð½Ð° Ð342Ðµ
Ð³ÐµÐ½ÐµÑÐ°ÑÐ¾Ñ Ð³Ð°Ð·Ð¾Ð²ÑÑ ÑÐ¼ÐµÑÐµÐ¹ T700U
Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ
Ð¼ÐµÑÐ¾Ð´Ð¸ÐºÐ° Ð¸ÑÐ¿ÑÑÐ°Ð½Ð¸Ð¹
Ð·Ð°ÐºÐ¾Ð½ ÐÑÐ³ÐµÑÐ°-ÐÐ°Ð¼Ð±ÐµÑÑÐ°-ÐÐµÑÐ°
ozone gas analyzer Ð342Ðµ
gas mixture generator T700U
measurements
test methods
Booger-Lambert-Beer law

description: Ð¢ÐµÐ¼Ð¾Ð¹ Ð²ÑÐ¿ÑÑÐºÐ½Ð¾Ð¹ ÐºÐ²Ð°Ð»Ð¸ÑÐ¸ÐºÐ°ÑÐ¸Ð¾Ð½Ð½Ð¾Ð¹ ÑÐ°Ð±Ð¾ÑÑ ÑÐ²Ð»ÑÐµÑÑÑ Â«ÐÐµÑÐ¾Ð´Ð¸ÐºÐ° Ð¸ÑÐ¿ÑÑÐ°Ð½Ð¸Ð¹ Ð³Ð°Ð·Ð¾Ð°Ð½Ð°Ð»Ð¸Ð·Ð°ÑÐ¾ÑÐ° Ð¾Ð·Ð¾Ð½Ð°Â».Ð¦ÐµÐ»ÑÐ¼Ð¸ Ð´Ð°Ð½Ð½Ð¾Ð¹ Ð²ÑÐ¿ÑÑÐºÐ½Ð¾Ð¹ ÐºÐ²Ð°Ð»Ð¸ÑÐ¸ÐºÐ°ÑÐ¸Ð¾Ð½Ð½Ð¾Ð¹ ÑÐ°Ð±Ð¾ÑÑ ÑÐ²Ð»ÑÑÑÑÑ ÑÐ°Ð·ÑÐ°Ð±Ð¾ÑÐºÐ°Â Ð¿ÑÐ¾ÐµÐºÑÐ° Ð¼ÐµÑÐ¾Ð´Ð¸ÐºÐ¸ Ð¸ÑÐ¿ÑÑÐ°Ð½Ð¸Ð¹ Ð³Ð°Ð·Ð¾Ð°Ð½Ð°Ð»Ð¸Ð·Ð°ÑÐ¾ÑÐ° Ð¾Ð·Ð¾Ð½Ð° Â«Ð342ÐµÂ», Ð¿ÑÐ¾Ð²ÐµÐ´ÐµÐ½Ð¸Ðµ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ð¹ Ð¸ Ð¾Ð¿ÑÐ¾Ð±Ð¾Ð²Ð°Ð½Ð¸Ðµ ÑÑÐ¾Ð¹ Ð¼ÐµÑÐ¾Ð´Ð¸ÐºÐ¸.ÐÐµÑÐ¾Ð´ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ð¹ Ð¾ÑÐ½Ð¾Ð²Ð°Ð½ Ð½Ð° Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ð¸ Ð¿Ð¾Ð³Ð»Ð¾ÑÐµÐ½Ð¸Ñ Ð¾Ð·Ð¾Ð½Ð¾Ð¼ ÑÐ»ÑÑÑÐ°ÑÐ¸Ð¾Ð»ÐµÑÐ¾Ð²Ð¾Ð³Ð¾ Ð¸Ð·Ð»ÑÑÐµÐ½Ð¸Ñ Ð½Ð° Ð´Ð»Ð¸Ð½Ðµ Ð²Ð¾Ð»Ð½Ñ 255 Ð½Ð¼.ÐÑÐ»Ð¸ Ð¿ÑÐ¾Ð²ÐµÐ´ÐµÐ½Ñ Ð¸ÑÑÐ»ÐµÐ´Ð¾Ð²Ð°Ð½Ð¸Ñ, Ð² ÑÐµÐ·ÑÐ»ÑÑÐ°ÑÐµ ÐºÐ¾ÑÐ¾ÑÑÑ Ð±ÑÐ»Ð° ÑÐ°Ð·ÑÐ°Ð±Ð¾ÑÐ°Ð½Ð° Ð¼ÐµÑÐ¾Ð´Ð¸ÐºÐ° Ð¸ÑÐ¿ÑÑÐ°Ð½Ð¸Ð¹ Ð³Ð°Ð·Ð¾Ð°Ð½Ð°Ð»Ð¸Ð·Ð°ÑÐ¾ÑÐ° Ð¾Ð·Ð¾Ð½Ð° Â«Ð342ÐµÂ», Ð² ÐºÐ¾ÑÐ¾ÑÐ¾Ð¹ Ð±ÑÐ»Ð¸ Ð¾Ð¿ÑÐµÐ´ÐµÐ»ÐµÐ½Ñ Ð¾ÑÐ½Ð¾Ð²Ð½ÑÐµ Ð¼ÐµÑÑÐ¾Ð»Ð¾Ð³Ð¸ÑÐµÑÐºÐ¸Ðµ ÑÐ°ÑÐ°ÐºÑÐµÑÐ¸ÑÑÐ¸ÐºÐ¸ Ð³Ð°Ð·Ð¾Ð°Ð½Ð°Ð»Ð¸Ð·Ð°ÑÐ¾ÑÐ°: Ð´Ð¸Ð°Ð¿Ð°Ð·Ð¾Ð½ Ð¿Ð¾ÐºÐ°Ð·Ð°Ð½Ð¸Ð¹ Ð¾Ð±ÑÐµÐ¼Ð½Ð¾Ð¹ Ð´Ð¾Ð»Ð¸ (Ð¼Ð°ÑÑÐ¾Ð²Ð¾Ð¹ ÐºÐ¾Ð½ÑÐµÐ½ÑÑÐ°ÑÐ¸Ð¸) Ð¾Ð·Ð¾Ð½Ð°, Ð´Ð¸Ð°Ð¿Ð°Ð·Ð¾Ð½ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ð¹ Ð¾Ð±ÑÐµÐ¼Ð½Ð¾Ð¹ Ð´Ð¾Ð»Ð¸ (Ð¼Ð°ÑÑÐ¾Ð²Ð¾Ð¹ ÐºÐ¾Ð½ÑÐµÐ½ÑÑÐ°ÑÐ¸Ð¸) Ð¾Ð·Ð¾Ð½Ð°, Ð¿ÑÐµÐ´ÐµÐ»Ñ Ð´Ð¾Ð¿ÑÑÐºÐ°ÐµÐ¼Ð¾Ð¹ Ð¾ÑÐ½Ð¾Ð²Ð½Ð¾Ð¹ Ð¿ÑÐ¸Ð²ÐµÐ´ÐµÐ½Ð½Ð¾Ð¹ Ð¸ Ð¾ÑÐ½Ð¾ÑÐ¸ÑÐµÐ»ÑÐ½Ð¾Ð¹ Ð¿Ð¾Ð³ÑÐµÑÐ½Ð¾ÑÑÐ¸ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ð¹ Ð¾Ð±ÑÐµÐ¼Ð½Ð¾Ð¹ Ð´Ð¾Ð»Ð¸ (Ð¼Ð°ÑÑÐ¾Ð²Ð¾Ð¹ ÐºÐ¾Ð½ÑÐµÐ½ÑÑÐ°ÑÐ¸Ð¸) Ð¾Ð·Ð¾Ð½Ð°, Ð¿ÑÐµÐ´ÐµÐ» Ð´Ð¾Ð¿ÑÑÐºÐ°ÐµÐ¼Ð¾Ð¹ Ð²Ð°ÑÐ¸Ð°ÑÐ¸Ð¸ Ð¿Ð¾ÐºÐ°Ð·Ð°Ð½Ð¸Ð¹, Ð² Ð´Ð¾Ð»ÑÑ Ð¾Ñ Ð¿ÑÐµÐ´ÐµÐ»Ð° Ð´Ð¾Ð¿ÑÑÐºÐ°ÐµÐ¼Ð¾Ð¹ Ð¾ÑÐ½Ð¾Ð²Ð½Ð¾Ð¹ Ð¿Ð¾Ð³ÑÐµÑÐ½Ð¾ÑÑÐ¸.ÐÐ° Ð¾ÑÐ½Ð¾Ð²Ðµ ÑÐµÐ·ÑÐ»ÑÑÐ°ÑÐ¾Ð² Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ð¹ Ð¸ ÑÐ°Ð·ÑÐ°Ð±Ð¾ÑÐ°Ð½Ð½Ð¾Ð¹ Ð¼ÐµÑÐ¾Ð´Ð¸ÐºÐ¸ Ð¸ÑÐ¿ÑÑÐ°Ð½Ð¸Ð¹ Ð² Ð´Ð°Ð»ÑÐ½ÐµÐ¹ÑÐµÐ¼ Ð±ÑÐ´ÐµÑ ÑÐ°Ð·ÑÐ°Ð±Ð¾ÑÐ°Ð½Ð° Ð¸ Ð¾Ð¿ÑÐ¾Ð±Ð¾Ð²Ð°Ð½Ð° Ð¼ÐµÑÐ¾Ð´Ð¸ÐºÐ¸ Ð¿Ð¾Ð²ÐµÑÐºÐ¸ Ð³Ð°Ð·Ð¾Ð°Ð½Ð°Ð»Ð¸Ð·Ð°ÑÐ¾ÑÐ° Ð¾Ð·Ð¾Ð½Ð° Â«Ð342ÐµÂ» Ð´Ð»Ñ ÑÑÐ²ÐµÑÐ¶Ð´ÐµÐ½Ð¸Ñ ÑÐ¸Ð¿Ð° Ð¸ Ð¸ÑÐ¿Ð¾Ð»ÑÐ·Ð¾Ð²Ð°Ð½Ð¸Ñ Ð³Ð°Ð·Ð¾Ð°Ð½Ð°Ð»Ð¸Ð·Ð°ÑÐ¾ÑÐ° Ð² ÑÑÐµÑÐµ Ð³Ð¾ÑÑÐ´Ð°ÑÑÑÐ²ÐµÐ½Ð½Ð¾Ð³Ð¾ ÑÐµÐ³ÑÐ»Ð¸ÑÐ¾Ð²Ð°Ð½Ð¸Ñ.ÐÐ¾ ÑÐµÐ·ÑÐ»ÑÑÐ°ÑÐ°Ð¼ Ð¿ÑÐ¾Ð²ÐµÐ´ÐµÐ½Ð½ÑÑ Ð¸ÑÐ¿ÑÑÐ°Ð½Ð¸Ð¹ Ð±ÑÐ»Ð¾ ÑÑÑÐ°Ð½Ð¾Ð²Ð»ÐµÐ½Ð¾, ÑÑÐ¾ ÑÐ°ÑÐ°ÐºÑÐµÑÐ¸ÑÑÐ¸ÐºÐ¸ Ð³Ð°Ð·Ð¾Ð°Ð½Ð°Ð»Ð¸Ð·Ð°ÑÐ¾ÑÐ° Ð¾Ð·Ð¾Ð½Ð° Â«Ð342ÐµÂ» ÑÐ¾Ð¾ÑÐ²ÐµÑÑÑÐ²ÑÑÑ ÑÐ°ÑÐ°ÐºÑÐµÑÐ¸ÑÑÐ¸ÐºÐ°Ð¼ Ð·Ð°ÑÐ²Ð»ÐµÐ½Ð½ÑÐ¼ Ð¸Ð·Ð³Ð¾ÑÐ¾Ð²Ð¸ÑÐµÐ»ÐµÐ¼, Ð° Ð¿Ð¾Ð³ÑÐµÑÐ½Ð¾ÑÑÐ¸ Ð½Ðµ Ð¿ÑÐµÐ²ÑÑÐ°ÑÑ Ð´Ð¾Ð¿ÑÑÑÐ¸Ð¼ÑÑ Ð·Ð½Ð°ÑÐµÐ½Ð¸Ð¹ Ð¾ÑÐºÐ»Ð¾Ð½ÐµÐ½Ð¸Ð¹.
description: The theme of the final qualifying work is "Method of testing the ozone gas analyzer".The purposesÂ of this final qualification work are the development of a draft methodology for testing the O342e ozone gas analyzer, measurements and refinement of this methodology.The measurement method is based on measuring the absorption of ultraviolet radiation by ozone at a wavelength of 255 nm.Studies were carried out, as a result of which a test procedure for the O342e ozone gas analyzer was developed, in which the main metrological characteristics of the gas analyzer were determined: the range of indications of the volume fraction (mass concentration) of ozone, the range of measurements of the volume fraction (mass concentration) of ozone, the limits of the permissible basic reduced and relative measurement error of the volume fraction (mass concentration) of ozone, the limit of allowable variation of readings, in fractions of the limit of the allowable basic error.Based on the results of measurements and the developed test methodology, in the future, methods for verifying the O342e ozone gas analyzer will be developed and tested to approve the type and use of the gas analyzer in the field of state regulation.According to the results of the tests, it was found that the characteristics of the O342e ozone gas analyzer correspond to the characteristics declared by the manufacturer, and the errors do not exceed the permissible deviations.