Nonwoven Textile for Use in a Nanoparticle Respiratory Deposition Sampler
Creators: name: Vosburgh, Donna J.H.
name: Park, Jae Hong
name: Mines, Levi W. D.
name: Imali A. Mudunkotuwa
name: T. Renée Anthony
name: Peters, Thomas M.
Publisher: name: Taylor & Francis
Year: 2016
Subjects: subject: Biophysics
subject: Space Science
subject: 29999 Physical Sciences not elsewhere classified
subject: FOS: Physical sciences
subject: FOS: Physical sciences
subject: Medicine
subject: Cell Biology
subject: Biotechnology
subject: 59999 Environmental Sciences not elsewhere classified
subject: FOS: Earth and related environmental sciences
subject: FOS: Earth and related environmental sciences
subject: 39999 Chemical Sciences not elsewhere classified
subject: FOS: Chemical sciences
subject: FOS: Chemical sciences
subject: Ecology
subject: FOS: Biological sciences
subject: FOS: Biological sciences
subject: 20199 Astronomical and Space Sciences not elsewhere classified
subject: 69999 Biological Sciences not elsewhere classified
subject: Developmental Biology
subject: Cancer
subject: Science Policy

description: The nanoparticle respiratory deposition (NRD) sampler is a personal sampler that combines a cyclone, impactor, and a nylon mesh diffusion stage to measure a worker's exposure to nanoparticles. The concentration of titanium in the nylon mesh of the diffusion stage complicates the application of the NRD sampler for assessing exposures to titanium dioxide nanoparticles. This study evaluated commercially available nonwoven textiles for use as an alternative media in the diffusion stage of the NRD sampler. Three textiles were selected as containing little titanium from an initial screening of 11 textiles by field portable x-ray fluorescence (FPXRF). Further evaluation on these three textiles was conducted to determine the concentration of titanium and other metals by inductively coupled plasma – optical emission spectroscopy (ICP-OES), the number of layers required to achieve desired collection characteristics for use as the diffusion stage in the NRD sampler (i.e., the nanoparticulate matter, NPM, criterion), and the pressure drop associated with that number of layers. Only three (two composed of cotton fibers, C1 and C2; and one of viscose bamboo and cotton fibers, BC) of 11 textiles screened had titanium concentrations below the limit of detection the XRF device (0.15 µg/cm<sup>2</sup>). Multiple metals, including small amounts of titanium, were found in each of the three nonwoven textiles using ICP-OES. The number of 25-mm-diameter layers required to achieve the collection efficiency by size required for the NRD sampler was three for C1 (R<sup>2</sup> = 0.95 with reference to the NPM criterion), two for C2 (R<sup>2</sup> = 0.79), and three for BC (R<sup>2</sup> = 0.87). All measured pressure drops were less than the theoretical and even the greatest pressure drop of 65.4 Pa indicated that a typical personal sampling pump could accommodate any of the three nonwoven textiles in the NRD sampler. The titanium concentration, collection efficiency, and measured pressure drops show there is a potential for nonwoven textiles to be used as the diffusion stage of the NRD sampler.