title: ÐÐ·Ð¼ÐµÑÐ¸ÑÐµÐ»Ñ ÑÐ°Ð·Ð¾Ð²ÑÑ ÑÑÐ¼Ð¾Ð² ÑÑÐ¸Ð»Ð¸ÑÐµÐ»Ñ Ð¼Ð¸Ð»Ð»Ð¸Ð¼ÐµÑÑÐ¾Ð²Ð¾Ð³Ð¾ Ð´Ð¸Ð°Ð¿Ð°Ð·Ð¾Ð½Ð° Ð´Ð»Ð¸Ð½ Ð²Ð¾Ð»Ð½, ÑÐ°Ð±Ð¾ÑÐ°ÑÑÐµÐ³Ð¾ Ð² Ð¸Ð¼Ð¿ÑÐ»ÑÑÐ½Ð¾Ð¼ ÑÐµÐ¶Ð¸Ð¼Ðµ
title: Ð²ÑÐ¿ÑÑÐºÐ½Ð°Ñ ÐºÐ²Ð°Ð»Ð¸ÑÐ¸ÐºÐ°ÑÐ¸Ð¾Ð½Ð½Ð°Ñ ÑÐ°Ð±Ð¾ÑÐ° Ð¼Ð°Ð³Ð¸ÑÑÑÐ°
titleType: Subtitle
title: Phase noise meter for the millimeter-wave amplifier operating in pulsed mode
Creators: name: ÐÐ½Ð´ÑÐµÐµÐ², ÐÐ½Ð´ÑÐµÐ¹
Publisher: name: Ð¡Ð°Ð½ÐºÑ-ÐÐµÑÐµÑÐ±ÑÑÐ³ÑÐºÐ¸Ð¹ Ð¿Ð¾Ð»Ð¸ÑÐµÑÐ½Ð¸ÑÐµÑÐºÐ¸Ð¹ ÑÐ½Ð¸Ð²ÐµÑÑÐ¸ÑÐµÑ ÐÐµÑÑÐ° ÐÐµÐ»Ð¸ÐºÐ¾Ð³Ð¾
Year: 2019
Subjects: Ð¸Ð·Ð¼ÐµÑÐ¸ÑÐµÐ»Ñ ÑÐ°Ð·Ð¾Ð²ÑÑ ÑÑÐ¼Ð¾Ð²
ÑÑÐ¼Ñ
Ð²Ð¾Ð»Ð½Ð¾Ð²Ð¾Ð´Ð½Ð¾-ÑÐµÐ»ÐµÐ²Ð¾Ð¹ Ð¼Ð¾ÑÑ
Ð´ÐµÑÐµÐºÑÐ¾ÑÐ½Ð°Ñ ÑÐµÐºÑÐ¸Ñ
ÑÑÐ²ÑÑÐ²Ð¸ÑÐµÐ»ÑÐ½Ð¾ÑÑÑ
ÑÐ°Ð·Ð¾Ð²ÑÐµ ÑÐ»ÑÐºÑÑÐ°ÑÐ¸Ð¸
Ð°Ð¼Ð¿Ð»Ð¸ÑÑÐ´Ð½ÑÐµ ÑÐ»ÑÐºÑÑÐ°ÑÐ¸Ð¸
ÑÐ²ÐµÑÑ Ð²ÑÑÐ¾ÐºÐ¸Ðµ ÑÐ°ÑÑÐ¾ÑÑ
phase noise meter
noise
waveguide-slot bridge
detector section
sensitivity
phase fluctuations
amplitude fluctuations
ultrahigh frequencies

description: Ð Ð°Ð±Ð¾ÑÐ° ÑÐ¾Ð´ÐµÑÐ¶Ð¸Ñ 40 ÑÑÑÐ°Ð½Ð¸Ñ, 30 ÑÐ¸ÑÑÐ½ÐºÐ¾Ð², 1 ÑÐ°Ð±Ð»Ð¸ÑÑ, 26 Ð±Ð¸Ð±Ð»Ð¸Ð¾Ð³ÑÐ°ÑÐ¸ÑÐµÑÐºÐ¸Ñ Ð½Ð°Ð·Ð²Ð°Ð½Ð¸Ð¹, 3 Ð¿ÑÐ¸Ð»Ð¾Ð¶ÐµÐ½Ð¸Ñ. Ð Ð´Ð°Ð½Ð½Ð¾Ð¹ ÑÐ°Ð±Ð¾ÑÐµ Ð¿ÑÐµÐ´ÑÑÐ°Ð²Ð»ÐµÐ½Ð° ÑÐ°Ð·ÑÐ°Ð±Ð¾ÑÐºÐ° ÐÐ¤Ð¨ Ð¸Ð¼Ð¿ÑÐ»ÑÑÐ½Ð¾Ð³Ð¾ ÑÑÐ¸Ð»Ð¸ÑÐµÐ»Ñ Ð´Ð¸Ð°Ð¿Ð°Ð·Ð¾Ð½Ð° ÑÐ°Ð±Ð¾ÑÐ¸Ñ ÑÐ°ÑÑÐ¾Ñ 33,5-34,5 ÐÐÑ, Ð¾Ð±ÐµÑÐ¿ÐµÑÐ¸Ð²Ð°ÑÑÐµÐ³Ð¾ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ Ð¤Ð¨ Ð½Ð° ÑÐ°ÑÑÐ¾ÑÐ°Ñ Ð°Ð½Ð°Ð»Ð¸Ð·Ð° F Ð¾Ñ 1 Ð´Ð¾ 10 ÐºÐÑ, Ñ ÑÑÐ²ÑÑÐ²Ð¸ÑÐµÐ»ÑÐ½Ð¾ÑÑÑÑ Ð½Ðµ Ð¼ÐµÐ½ÐµÐµ Ð¼Ð¸Ð½ÑÑ 105 Ð´Ð/ÐÑ |F=1ÐºÐÑ Ð¸ Ð¼Ð¸Ð½ÑÑ 115 Ð´Ð/ÐÑ |F=10ÐºÐÑ Ñ ÑÐ¾ÑÐ½Ð¾ÑÑÑÑ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ Â± 1 Ð´Ð. Ð Ð»Ð¸ÑÐµÑÐ°ÑÑÑÐ½Ð¾Ð¼ Ð¾Ð±Ð·Ð¾ÑÐµ Ð±ÑÐ»Ð¸ ÑÐ°ÑÑÐ¼Ð¾ÑÑÐµÐ½Ñ ÑÐ°ÐºÐ¸Ðµ Ð²Ð¾Ð¿ÑÐ¾ÑÑ ÐºÐ°Ðº ÑÐ°ÑÐ°ÐºÑÐµÑÐ¸ÑÑÐ¸ÐºÐ¸ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ð¹ ÑÐ°Ð·Ð¾Ð²Ð¾Ð³Ð¾ ÑÑÐ¼Ð°, ÑÐ½ÐµÑÐ³ÐµÑÐ¸ÑÐµÑÐºÐ¸Ð¹ ÑÐ¿ÐµÐºÑÑ ÑÐ»ÑÑÐ°Ð¹Ð½Ð¾Ð³Ð¾ Ð¿ÑÐ¾ÑÐµÑÑÐ°, Ð¼ÐµÑÐ¾Ð´Ñ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ ÑÐ°Ð·Ð¾Ð²Ð¾Ð³Ð¾ ÑÑÐ¼Ð° ÑÑÐ¸Ð»Ð¸ÑÐµÐ»Ñ, ÑÐ¿ÐµÑÐ¸ÑÐ¸ÐºÐ° Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ Ð¤Ð¨ Ð² Ð¸Ð¼Ð¿ÑÐ»ÑÑÐ½Ð¾Ð¼ ÑÐµÐ¶Ð¸Ð¼Ðµ. Ð Ð¿ÑÐ°ÐºÑÐ¸ÑÐµÑÐºÐ¾Ð¹ ÑÐ°ÑÑÐ¸ ÑÐ°Ð±Ð¾ÑÑ Ð±ÑÐ»Ð¸ ÑÐµÑÐµÐ½Ñ ÑÐ»ÐµÐ´ÑÑÑÐ¸Ðµ Ð·Ð°Ð´Ð°ÑÐ¸: Ð²ÑÐ±Ð¾Ñ Ð¸ Ð¾Ð±Ð¾ÑÐ½Ð¾Ð²Ð°Ð½Ð¸Ðµ ÑÑÑÑÐºÑÑÑÐ½Ð¾Ð¹ ÑÑÐµÐ¼Ñ ÐÐ¤Ð¨, Ð²ÑÐ±Ð¾Ñ Ð¸ Ð¾Ð±Ð¾ÑÐ½Ð¾Ð²Ð°Ð½Ð¸Ðµ ÑÑÐµÐ±Ð¾Ð²Ð°Ð½Ð¸Ð¹ Ðº ÑÐ»ÐµÐ¼ÐµÐ½ÑÐ°Ð¼ ÑÑÑÑÐºÑÑÑÐ½Ð¾Ð¹ ÑÑÐµÐ¼Ñ ÐÐ¤Ð¨, ÑÐ°Ð·ÑÐ°Ð±Ð¾ÑÐºÐ° Ð¿ÑÐ¸Ð½ÑÐ¸Ð¿Ð¸Ð°Ð»ÑÐ½Ð¾Ð¹ ÑÑÐµÐ¼Ñ ÐÐ¤Ð¨ Ð¸ Ð¿ÑÐ¾Ð²ÐµÑÐºÐ° ÐµÑ Ð½Ð° Ð¾ÑÐ½Ð¾Ð²Ð°Ð½Ð¸Ð¸ Ð¼Ð¾Ð´ÐµÐ»Ð¸ÑÐ¾Ð²Ð°Ð½Ð¸Ñ, Ð° ÑÐ°ÐºÐ¶Ðµ Ð¾ÑÐµÐ½ÐºÐ° ÑÑÐ²ÑÑÐ²Ð¸ÑÐµÐ»ÑÐ½Ð¾ÑÑÐ¸ ÑÑÐµÐ¼Ñ ÐÐ¤Ð¨.
description: The work contains 40 pages, 30 figures, 1 table, 26 bibliographic names, 3 applications. This work presents the development of phase noise meter for a pulsed amplifier operating frequency range of 33.5-34.5 GHz, providing measurements of phase noise at analysis frequencies from 1 to 10 kHz, with a sensitivity of at least minus 105 dB / Hz | F = 1 kHz and minus 115 dB / Hz | F = 10 kHz with a measurement accuracy of Â± 1 dB. In the literature review, such issues as the characteristics of measurements of phase noise, the energy spectrum of a random process, methods for measuring the phase noise of an amplifier, the specifics of measuring phase noise in a pulsed mode were considered. In the practical part of the work, the following tasks were solved: selection and justification of the phase noise meter structural scheme, selection and justification of the requirements for the elements of the phase noise meter structural scheme, development of the phase noise meter schematic diagram and verification based on modeling, as well as assessment of the sensitivity of the phase noise meter scheme.