title: ÐÐ´Ð°Ð¿ÑÐ°ÑÐ¸Ñ Ð°Ð»Ð³Ð¾ÑÐ¸ÑÐ¼Ð¾Ð² Ð´Ð¾Ð· Ð² ÑÐ¸ÑÑÐµÐ¼Ñ Ð´Ð¾Ð·Ð¸Ð¼ÐµÑÑÐ¸ÑÐµÑÐºÐ¾Ð³Ð¾ Ð¿Ð»Ð°Ð½Ð¸ÑÐ¾Ð²Ð°Ð½Ð¸Ñ Ð½Ð° Ð¿ÑÐ¸Ð¼ÐµÑÐµ TPS â Monaco 5.11
title: Ð²ÑÐ¿ÑÑÐºÐ½Ð°Ñ ÐºÐ²Ð°Ð»Ð¸ÑÐ¸ÐºÐ°ÑÐ¸Ð¾Ð½Ð½Ð°Ñ ÑÐ°Ð±Ð¾ÑÐ° Ð¼Ð°Ð³Ð¸ÑÑÑÐ°
titleType: Subtitle
title: Adaptation of dose algorithms to the dosimetric planning system using the example of TPS - Monaco 5.11
Creators: name: Ð§Ð¸Ð¶Ð¾Ð², ÐÐ¸ÐºÐ¸ÑÐ°
Publisher: name: Ð¡Ð°Ð½ÐºÑ-ÐÐµÑÐµÑÐ±ÑÑÐ³ÑÐºÐ¸Ð¹ Ð¿Ð¾Ð»Ð¸ÑÐµÑÐ½Ð¸ÑÐµÑÐºÐ¸Ð¹ ÑÐ½Ð¸Ð²ÐµÑÑÐ¸ÑÐµÑ ÐÐµÑÑÐ° ÐÐµÐ»Ð¸ÐºÐ¾Ð³Ð¾
Year: 2020
Subjects: ÐºÐ»Ð¸Ð½Ð¸ÑÐµÑÐºÐ°Ñ Ð´Ð¾Ð·Ð¸Ð¼ÐµÑÑÐ¸Ñ
Ð»Ð¸Ð½ÐµÐ¹Ð½ÑÐ¹ ÑÑÐºÐ¾ÑÐ¸ÑÐµÐ»Ñ
Ð»ÑÑÐµÐ²Ð°Ñ ÑÐµÑÐ°Ð¿Ð¸Ñ
ÑÐ¸ÑÑÐµÐ¼Ð° Ð¿Ð»Ð°Ð½Ð¸ÑÐ¾Ð²Ð°Ð½Ð¸Ñ
Ð²ÐµÑÐ¸ÑÐ¸ÐºÐ°ÑÐ¸Ñ
Ð¼ÐµÐ´Ð¸ÑÐ¸Ð½ÑÐºÐ°Ñ ÑÐ¸Ð·Ð¸ÐºÐ°
Ð¼Ð¾Ð½ÑÐµ ÐºÐ°ÑÐ»Ð¾
clinical dosimetry
linear accelerator
radiation therapy
planning system
verification
medical physics
collapsed cone algorithm
monte carlo algorithm

description: Ð¢ÐµÐ¼Ð° Ð²ÑÐ¿ÑÑÐºÐ½Ð¾Ð¹ ÐºÐ²Ð°Ð»Ð¸ÑÐ¸ÐºÐ°ÑÐ¸Ð¾Ð½Ð½Ð¾Ð¹ ÑÐ°Ð±Ð¾ÑÑ: Â«ÐÐ´Ð°Ð¿ÑÐ°ÑÐ¸Ñ Ð°Ð»Ð³Ð¾ÑÐ¸ÑÐ¼Ð¾Ð² Ð´Ð¾Ð· Ð² ÑÐ¸ÑÑÐµÐ¼Ñ Ð´Ð¾Ð·Ð¸Ð¼ÐµÑÑÐ¸ÑÐµÑÐºÐ¾Ð³Ð¾ Ð¿Ð»Ð°Ð½Ð¸ÑÐ¾Ð²Ð°Ð½Ð¸Ñ Ð½Ð° Ð¿ÑÐ¸Ð¼ÐµÑÐµ TPS â Monaco 5.11Â». ÐÐ°Ð½Ð½Ð°Ñ ÑÐ°Ð±Ð¾ÑÐ° Ð¿Ð¾ÑÐ²ÑÑÐµÐ½Ð° Ð¿ÑÐ¾Ð²ÐµÐ´ÐµÐ½Ð¸Ñ Ð°Ð´Ð°Ð¿ÑÐ°ÑÐ¸Ð¸ Ð¼Ð¾Ð´ÐµÐ»ÐµÐ¹ ÑÐ¾ÑÐ¾Ð½Ð½ÑÑ Ð¿ÑÑÐºÐ¾Ð² Ð¼ÐµÐ´Ð¸ÑÐ¸Ð½ÑÐºÐ¾Ð³Ð¾ Ð»Ð¸Ð½ÐµÐ¹Ð½Ð¾Ð³Ð¾ ÑÑÐºÐ¾ÑÐ¸ÑÐµÐ»Ñ Â«SIEMENS PRIMUSÂ» Ð² ÑÐ¸ÑÑÐµÐ¼Ðµ Ð¿Ð»Ð°Ð½Ð¸ÑÐ¾Ð²Ð°Ð½Ð¸Ñ (TPS) Ð»ÑÑÐµÐ²Ð¾Ð¹ ÑÐµÑÐ°Ð¿Ð¸Ð¸ (ÐÐ¢) Â«Monaco 5.11Â» Ñ ÑÐ½ÐµÑÐ³Ð¸ÐµÐ¹ 6ÐÑÐ. ÐÑÑÐ»ÐµÐ´Ð¾Ð²Ð°Ð½Ð¸Ñ Ð¿ÑÐ¾Ð²Ð¾Ð´Ð¸Ð»Ð¸ÑÑ Ð½Ð° Ð±Ð°Ð·Ðµ ÐÐÐ£Ð Â«Ð¡Ð°Ð½ÐºÑ-ÐÐµÑÐµÑÐ±ÑÑÐ³ÑÐºÐ¸Ð¹ ÐºÐ»Ð¸Ð½Ð¸ÑÐµÑÐºÐ¸Ð¹ Ð½Ð°ÑÑÐ½Ð¾-Ð¿ÑÐ°ÐºÑÐ¸ÑÐµÑÐºÐ¸Ð¹ ÑÐµÐ½ÑÑ ÑÐ¿ÐµÑÐ¸Ð°Ð»Ð¸Ð·Ð¸ÑÐ¾Ð²Ð°Ð½Ð½ÑÑ Ð²Ð¸Ð´Ð¾Ð² Ð¼ÐµÐ´Ð¸ÑÐ¸Ð½ÑÐºÐ¾Ð¹ Ð¿Ð¾Ð¼Ð¾ÑÐ¸ (Ð¾Ð½ÐºÐ¾Ð»Ð¾Ð³Ð¸ÑÐµÑÐºÐ¸Ð¹)Â», Ð³Ð´Ðµ Ð¿ÑÐ¾ÑÐ¾Ð´Ð¸Ð»Ð° Ð¾ÑÐ½Ð¾Ð²Ð½Ð°Ñ ÑÐ°Ð±Ð¾ÑÐ° Ð¿Ð¾ ÑÐ±Ð¾ÑÑ Ð¸ÑÑÐ»ÐµÐ´ÑÐµÐ¼ÑÑ Ð¼Ð°ÑÐµÑÐ¸Ð°Ð»Ð¾Ð². Ð ÑÐ¾Ð´Ðµ ÑÐ°Ð±Ð¾ÑÑ Ð±ÑÐ»Ð¸ Ð¿ÑÐ¾Ð²ÐµÐ´ÐµÐ½Ð° Ð¿ÑÐ¾Ð²ÐµÑÐºÐ° Ð¼Ð¾Ð´ÐµÐ»ÐµÐ¹ Ð¿ÑÑÐºÐ¾Ð² Ð¼ÐµÐ´Ð¸ÑÐ¸Ð½ÑÐºÐ¾Ð³Ð¾ Ð»Ð¸Ð½ÐµÐ¹Ð½Ð¾Ð³Ð¾ ÑÑÐºÐ¾ÑÐ¸ÑÐµÐ»Ñ Â«Siemens PrimusÂ» Ð² Ð¿Ð»Ð°Ð½Ð¸ÑÑÑÑÐµÐ¹ ÑÐ¸ÑÑÐµÐ¼Ðµ Ð»ÑÑÐµÐ²Ð¾Ð¹ ÑÐµÑÐ°Ð¿Ð¸Ð¸ Â«Elekta Monaco 5.11Â» Ñ Ð¸ÑÐ¿Ð¾Ð»ÑÐ·Ð¾Ð²Ð°Ð½Ð¸ÐµÐ¼ Ð°Ð»Ð³Ð¾ÑÐ¸ÑÐ¼Ð° ÑÐ°ÑÑÐµÑÐ° Ð´Ð¾Ð·Ñ Â«Monte CarloÂ», Ð¿ÑÐ¾Ð°Ð½Ð°Ð»Ð¸Ð·Ð¸ÑÐ¾Ð²Ð°Ð½Ñ Ð´Ð¾Ð·Ð¸Ð¼ÐµÑÑÐ¸ÑÐµÑÐºÐ¸Ðµ ÑÐ°ÑÐ°ÐºÑÐµÑÐ¸ÑÑÐ¸ÐºÐ¸ Ð¼ÐµÐ´Ð¸ÑÐ¸Ð½ÑÐºÐ¾Ð³Ð¾ Ð»Ð¸Ð½ÐµÐ¹Ð½Ð¾Ð³Ð¾ ÑÑÐºÐ¾ÑÐ¸ÑÐµÐ»Ñ Â«SIEMENS PRIMUSÂ», Ð° ÑÐ°ÐºÐ¶Ðµ Ð¿ÑÐ¸Ð¼ÐµÐ½ÐµÐ½Ð° ÐºÐ¾Ð¼Ð¼ÐµÑÑÐµÑÐºÐ°Ñ ÑÐ¸ÑÑÐµÐ¼Ð° Ð¿Ð»Ð°Ð½Ð¸ÑÐ¾Ð²Ð°Ð½Ð¸Ñ Ð»ÑÑÐµÐ²Ð¾Ð¹ ÑÐµÑÐ°Ð¿Ð¸Ð¸ Â«Monaco 5.11Â» Ð´Ð»Ñ ÑÐ°ÑÑÐµÑÐ° Ð¿ÑÐ¾Ð²ÐµÑÐ¾ÑÐ½ÑÑ Ð¿Ð»Ð°Ð½Ð¾Ð². ÐÐ¾ Ð´Ð¾Ð·Ð¸Ð¼ÐµÑÑÐ¸ÑÐµÑÐºÐ¸Ð¼ Ð´Ð°Ð½Ð½ÑÐµ Ð¸ Ð¾ÑÐ½Ð¾Ð²Ð½ÑÐ¼ ÑÐ°ÑÐ°ÐºÑÐµÑÐ¸ÑÑÐ¸ÐºÐ°Ð¼ ÑÐµÑÐ°Ð¿ÐµÐ²ÑÐ¸ÑÐµÑÐºÐ¾Ð¹ ÑÑÑÐ°Ð½Ð¾Ð²ÐºÐ¸ Ð±ÑÐ»Ð¸ ÑÐ¾Ð·Ð´Ð°Ð½Ñ Ð¼Ð¾Ð´ÐµÐ»Ð¸ Ð¿ÑÑÐºÐ¾Ð² ÑÑÐºÐ¾ÑÐ¸ÑÐµÐ»Ñ Ð¸ Ð²Ð²ÐµÐ´ÐµÐ½Ñ Ð² TPS - Â«Monaco 5.11Â», Ð¿ÑÐ¾Ð²ÐµÑÐµÐ½Ñ Ð½Ð° ÑÑÐ´Ðµ ÑÐ¾Ð·Ð´Ð°Ð½Ð½ÑÑ Ð¿Ð»Ð°Ð½Ð¾Ð². ÐÐ¾ Ð¸ÑÐ¾Ð³Ñ Ð¿ÑÐ¾Ð´ÐµÐ»Ð°Ð½Ð½Ð¾Ð³Ð¾ Ð¸ÑÑÐ»ÐµÐ´Ð¾Ð²Ð°Ð½Ð¸Ñ Ð±ÑÐ»Ð¸ ÑÐ´ÐµÐ»Ð°Ð½Ñ Ð²ÑÐ²Ð¾Ð´Ñ, ÑÑÐ¾ Ð¿Ð¾Ð»ÑÑÐµÐ½Ð½ÑÐµ Ð² ÑÐ¾Ð´Ðµ ÑÐ°Ð±Ð¾ÑÑ Ð¼Ð¾Ð´ÐµÐ»Ð¸ Ð¿ÑÑÐºÐ¾Ð² Ð¼ÐµÐ´Ð¸ÑÐ¸Ð½ÑÐºÐ¾Ð³Ð¾ Ð»Ð¸Ð½ÐµÐ¹Ð½Ð¾Ð³Ð¾ ÑÑÐºÐ¾ÑÐ¸ÑÐµÐ»Ñ Â«SIEMENS PRIMUSÂ» Ð¼Ð¾Ð³ÑÑ Ð±ÑÑÑ Ð¸ÑÐ¿Ð¾Ð»ÑÐ·Ð¾Ð²Ð°Ð½Ñ Ð² ÐºÐ»Ð¸Ð½Ð¸ÑÐµÑÐºÐ¾Ð¹ Ð¿ÑÐ°ÐºÑÐ¸ÐºÐµ Ð´Ð»Ñ ÑÐ°ÑÑÐµÑÐ¾Ð² Ð¿Ð»Ð°Ð½Ð¾Ð² Ð¾Ð±Ð»ÑÑÐµÐ½Ð¸Ñ Ð¿Ð°ÑÐ¸ÐµÐ½ÑÐ¾Ð², Ð±Ð¾Ð»ÑÐ½ÑÑ Ð¾Ð½ÐºÐ¾Ð»Ð¾Ð³Ð¸ÑÐµÑÐºÐ¸Ð¼Ð¸ Ð·Ð°Ð±Ð¾Ð»ÐµÐ²Ð°Ð½Ð¸ÑÐ¼Ð¸.
description: Theme of final qualification work: "Adaptation of dose algorithms to the dosimetric planning system using the example of TPS - Monaco 5.11." This work is devoted to the adaptation of the photon beam models of the SIEMENS PRIMUS linear medical accelerator in the 6MeV Monaco 5.11 radiation therapy planning (TPS) system. The studies were carried out on the basis of the "Saint-Petersburg Clinical Scientific and Practical Center for Specialized Types of Medical Care (Oncology) ", where the main work on collecting the studied materials took place. During the work, the beam models of the "Siemens Primus" medical linear accelerator were tested in the planning radiation therapy system "Elekta Monaco 5.11" using the "Monte Carlo" dose calculation algorithm, the dosimetric characteristics of the "SIEMENS PRIMUS" medical linear accelerator were analyzed, and the commercial "Monaco 5.11" radiation therapy planning system for calculating test plans. Based on the dosimetric data and the main characteristics of the therapeutic unit, accelerator beam models were created and introduced into TPS - âMonaco 5.11â, tested on a number of created plans. Based on the results of the study, it was concluded that the "SIEMENS PRIMUS" linear medical accelerator beam models obtained during the work can be used in clinical practice to calculate radiation plans for patients with cancer.