title: ÐÑÐµÐ½ÐºÐ° ÑÑÑÐµÐºÑÐ¸Ð²Ð½Ð¾ÑÑÐ¸ Ð²Ð½ÐµÐ´ÑÐµÐ½Ð¸Ñ Ð°Ð²ÑÐ¾Ð¼Ð°ÑÐ¸Ð·Ð¸ÑÐ¾Ð²Ð°Ð½Ð½Ð¾Ð¹ ÑÐ¸ÑÑÐµÐ¼Ñ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ ÑÐµÐ¼Ð¿ÐµÑÐ°ÑÑÑÐ½Ð¾Ð³Ð¾ ÐºÐ¾ÑÑÑÐ¸ÑÐ¸ÐµÐ½ÑÐ° ÐµÐ¼ÐºÐ¾ÑÑÐ¸
title: Ð²ÑÐ¿ÑÑÐºÐ½Ð°Ñ ÐºÐ²Ð°Ð»Ð¸ÑÐ¸ÐºÐ°ÑÐ¸Ð¾Ð½Ð½Ð°Ñ ÑÐ°Ð±Ð¾ÑÐ° Ð±Ð°ÐºÐ°Ð»Ð°Ð²ÑÐ°
titleType: Subtitle
title: Evaluation of the effectiveness of the implementation of an automated system for measuring the temperature coefficient of capacity
Creators: name: ÐÐ°ÑÐµÑÐ½Ð¸Ðº, ÐÐºÐ°ÑÐµÑÐ¸Ð½Ð°
Publisher: name: Ð¡Ð°Ð½ÐºÑ-ÐÐµÑÐµÑÐ±ÑÑÐ³ÑÐºÐ¸Ð¹ Ð¿Ð¾Ð»Ð¸ÑÐµÑÐ½Ð¸ÑÐµÑÐºÐ¸Ð¹ ÑÐ½Ð¸Ð²ÐµÑÑÐ¸ÑÐµÑ ÐÐµÑÑÐ° ÐÐµÐ»Ð¸ÐºÐ¾Ð³Ð¾
Year: 2022
Subjects: ÐºÐµÑÐ°Ð¼Ð¸ÑÐµÑÐºÐ¸Ð¹ ÐºÐ¾Ð½Ð´ÐµÐ½ÑÐ°ÑÐ¾Ñ
ÐµÐ¼ÐºÐ¾ÑÑÑ
ÑÐµÐ¼Ð¿ÐµÑÐ°ÑÑÑÐ½ÑÐ¹ ÐºÐ¾ÑÑÑÐ¸ÑÐ¸ÐµÐ½Ñ ÐµÐ¼ÐºÐ¾ÑÑÐ¸
ÑÑÐµÐ½Ð´
Ð¾ÑÐµÐ½ÐºÐ° ÑÑÑÐµÐºÑÐ¸Ð²Ð½Ð¾ÑÑÐ¸
Ð¼ÐµÑÐ¾Ð´ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ
ceramic capacitor
capacitance
temperature coefficient of capacitance
stand
efficiency evaluation
measurement method

description: ÐÐ° 43 Ñ., 19 ÑÐ¸ÑÑÐ½ÐºÐ¾Ð². Ð¢ÐµÐ¼Ð° Ð²ÑÐ¿ÑÑÐºÐ½Ð¾Ð¹ ÐºÐ²Ð°Ð»Ð¸ÑÐ¸ÐºÐ°ÑÐ¸Ð¾Ð½Ð½Ð¾Ð¹ ÑÐ°Ð±Ð¾ÑÑ: Â«ÐÑÐµÐ½ÐºÐ° ÑÑÑÐµÐºÑÐ¸Ð²Ð½Ð¾ÑÑÐ¸ Ð²Ð½ÐµÐ´ÑÐµÐ½Ð¸Ñ Ð°Ð²ÑÐ¾Ð¼Ð°ÑÐ¸Ð·Ð¸ÑÐ¾Ð²Ð°Ð½Ð½Ð¾Ð¹ ÑÐ¸ÑÑÐµÐ¼Ñ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ ÑÐµÐ¼Ð¿ÐµÑÐ°ÑÑÑÐ½Ð¾Ð³Ð¾ ÐºÐ¾ÑÑÑÐ¸ÑÐ¸ÐµÐ½ÑÐ° ÐµÐ¼ÐºÐ¾ÑÑÐ¸Â». Ð Ð´Ð°Ð½Ð½Ð¾Ð¹ Ð´Ð¸Ð¿Ð»Ð¾Ð¼Ð½Ð¾Ð¹ ÑÐ°Ð±Ð¾ÑÐµ Ð±ÑÐ»Ð¾ ÑÐ°ÑÑÐ¼Ð¾ÑÑÐµÐ½Ð¾: 1. ÐÑÐ½Ð¾Ð²Ð½ÑÐµ Ð¿Ð°ÑÐ°Ð¼ÐµÑÑÑ ÐºÐ¾Ð½Ð´ÐµÐ½ÑÐ°ÑÐ¾ÑÐ¾Ð² 2. ÐÐµÑÐ¾Ð´ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ ÑÐµÐ¼Ð¿ÐµÑÐ°ÑÑÑÐ½Ð¾Ð³Ð¾ ÐºÐ¾ÑÑÑÐ¸ÑÐ¸ÐµÐ½ÑÐ° ÐµÐ¼ÐºÐ¾ÑÑÐ¸ 3. Ð ÑÑÐ½Ð¾Ðµ Ð¸ Ð°Ð²ÑÐ¾Ð¼Ð°ÑÐ¸Ð·Ð¸ÑÐ¾Ð²Ð°Ð½Ð½Ð¾Ðµ Ð¾Ð±Ð¾ÑÑÐ´Ð¾Ð²Ð°Ð½Ð¸Ðµ Ð´Ð»Ñ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ ÑÐµÐ¼Ð¿ÐµÑÐ°ÑÑÑÐ½Ð¾Ð³Ð¾ ÐºÐ¾ÑÑÑÐ¸ÑÐ¸ÐµÐ½ÑÐ° ÐµÐ¼ÐºÐ¾ÑÑÐ¸ 4. Ð Ð°Ð·ÑÐ°Ð±Ð¾ÑÐºÐ° Ð°Ð»Ð³Ð¾ÑÐ¸ÑÐ¼Ð° ÑÐ°Ð±Ð¾ÑÑ Ð¿ÑÐ¾Ð³ÑÐ°Ð¼Ð¼Ð½Ð¾Ð³Ð¾ Ð¾Ð±ÐµÑÐ¿ÐµÑÐµÐ½Ð¸Ñ 5. ÐÑÐ¾Ð²ÐµÐ´ÐµÐ½Ð¸Ðµ Ð¸ÑÐ¿ÑÑÐ°Ð½Ð¸Ð¹ Ð¸ ÑÑÐ°Ð²Ð½ÐµÐ½Ð¸Ñ ÑÐµÐ·ÑÐ»ÑÑÐ°ÑÐ¾Ð² Ð ÑÐ¾Ð´Ðµ Ð²ÑÐ¿Ð¾Ð»Ð½ÐµÐ½Ð¸Ñ Ð²ÑÐ¿ÑÑÐºÐ½Ð¾Ð¹ ÑÐ°Ð±Ð¾ÑÑ Ð±ÑÐ»Ð¸ Ð¿ÑÐ¾Ð²ÐµÐ´ÐµÐ½Ñ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ ÑÐµÐ¼Ð¿ÐµÑÐ°ÑÑÑÐ½Ð¾Ð³Ð¾ ÐºÐ¾ÑÑÑÐ¸ÑÐ¸ÐµÐ½ÑÐ° ÐµÐ¼ÐºÐ¾ÑÑÐ¸ Ð½Ð° ÑÑÑÐ½Ð¾Ð¼ ÑÑÐµÐ½Ð´Ðµ Ð¢ÐÐ-10Ð Ð¸ Ð°Ð²ÑÐ¾Ð¼Ð°ÑÐ¸Ð·Ð¸ÑÐ¾Ð²Ð°Ð½Ð½Ð¾Ð¼ ÑÑÐµÐ½Ð´Ðµ ÐÐ¡Ð¢ÐÐ. ÐÐ¾ÑÐ»Ðµ ÑÑÐ°Ð²Ð½ÐµÐ½Ð¸Ñ ÑÐµÐ·ÑÐ»ÑÑÐ°ÑÐ¾Ð² Ð±ÑÐ»Ð¾ Ð´Ð¾ÐºÐ°Ð·Ð°Ð½Ð¾, ÑÑÐ¾ Ð´Ð°Ð½Ð½ÑÐµ, Ð¿Ð¾Ð»ÑÑÐµÐ½Ð½ÑÐµ Ð¿ÑÐ¸ Ð¿Ð¾Ð¼Ð¾ÑÐ¸ ÑÑÐµÐ½Ð´Ð° ÐÐ¡Ð¢ÐÐ Ð±Ð¾Ð»ÐµÐµ ÑÐ¾ÑÐ½ÑÐµ, Ð¿Ð¾Ð³ÑÐµÑÐ½Ð¾ÑÑÑ ÑÐ¼ÐµÐ½ÑÑÐ¸Ð»Ð°ÑÑ, Ð¿ÑÐ¾Ð¸Ð·Ð²Ð¾Ð´Ð¸ÑÐµÐ»ÑÐ½Ð¾ÑÑÑ Ð²ÑÑÐ¾ÑÐ»Ð°. Ð¢Ð°ÐºÐ¶Ðµ Ð±ÑÐ»Ð° Ð²ÑÐ¿Ð¾Ð»Ð½ÐµÐ½Ð° Ð¾ÑÐµÐ½ÐºÐ° ÑÑÑÐµÐºÑÐ¸Ð²Ð½Ð¾ÑÑÐ¸ Ð²Ð½ÐµÐ´ÑÐµÐ½Ð¸Ñ ÐÐ¡Ð¢ÐÐ. Ð ÑÐµÐ·ÑÐ»ÑÑÐ°ÑÐµ Ð²Ð½ÐµÐ´ÑÐµÐ½Ð¸Ñ Ð¿ÑÐ¾Ð¸Ð·Ð²Ð¾Ð´Ð¸ÑÐµÐ»ÑÐ½Ð¾ÑÑÑ ÑÐ²ÐµÐ»Ð¸ÑÐ¸Ð»Ð°ÑÑ, ÑÐºÐ¾Ð½Ð¾Ð¼Ð¸ÑÐµÑÐºÐ¸Ðµ Ð·Ð°ÑÑÐ°ÑÑ ÑÐ¼ÐµÐ½ÑÑÐ¸Ð»Ð¸ÑÑ, Ð¿ÑÐ¾ÑÐµÑÑ Ð¸Ð·Ð¼ÐµÑÐµÐ½Ð¸Ñ ÑÑÐ°Ð» Ð¼ÐµÐ½ÐµÐµ ÑÑÑÐ´Ð¾Ð·Ð°ÑÑÐ°ÑÐ½ÑÐ¹. â
description: On 43 p., 19 drawings. Topic of the final qualification work: "Evaluation of the effectiveness of the implementation of an automated system for measuring the temperature coefficient of capacity". In this dissertation, it was considered: 1. Main parameters of capacitors 2. Capacitance temperature coefficient measurement method 3. Manual and automated equipment for measuring the temperature coefficient of the capacitance 4. Development of the software operation algorithm 5. Conducting tests and comparing results In the course of the final work, the temperature coefficient of capacitance was measured on a manual stand TKE-10A and an automated stand ASTKE. After comparing the results, it was proved that the data obtained using the ASTKE stand is more accurate, the error has decreased, and productivity has increased. Also, an assessment was made of the effectiveness of the implementation of ASTKE. As a result of the implementation, productivity has increased, economic costs have decreased, and the measurement process has become less labor-intensive.