Electronics of Time-of-flight Measurement for Back-n at CSNS
Creators: name: Yu, T.
name: Cao, P.
name: Ji, X. Y.
name: Xie, L. K.
name: Huang, X. R.
name: An, Q.
name: Bai, H. Y.
name: Bao, J.
name: Chen, Y. H.
name: Cheng, P. J.
name: Cui, Z. Q.
name: Fan, R. R.
name: Feng, C. Q.
name: Gu, M. H.
name: Han, Z. J.
name: He, G. Z.
name: He, Y. C.
name: He, Y. F.
name: Huang, H. X.
name: Huang, W. L.
name: Ji, X. L.
name: Jiang, H. Y.
name: Jiang, W.
name: Jing, H. Y.
name: Kang, L.
name: Li, B.
name: Li, L.
name: Li, Q.
name: Li, X.
name: Li, Y.
name: Liu, R.
name: Liu, S. B.
name: Liu, X. Y.
name: Luan, G. Y.
name: Ma, Y. L.
name: Ning, C. J.
name: Ren, J.
name: Ruan, X. C.
name: Song, Z. H.
name: Sun, H.
name: Sun, X. Y.
name: Sun, Z. J.
name: Tan, Z. X.
name: Tang, J. Y.
name: Tang, H. Q.
name: Wang, P. C.
name: Wang, Q.
name: Wang, T. F.
name: Wang, Y. F.
name: Wang, Z. H.
name: Wang, Z.
name: Wen, J.
name: Wen, Z. W.
name: Wu, Q. B.
name: Wu, X. G.
name: Wu, X.
name: Yang, Y. W.
name: Yi, H.
name: Yu, L.
name: Yu, Y. J.
name: Zhang, G. H.
name: Zhang, L. Y.
name: Zhang, J.
name: Zhang, Q. M.
name: Zhang, Q. W.
name: Zhang, X. P.
name: Zhao, Y. T.
name: Zhong, Q. P.
name: Zhou, L.
name: Zhou, Z. Y.
name: Zhu, K. J.
Publisher: name: arXiv
Year: 2018
Subjects: subject: Instrumentation and Detectors (physics.ins-det)
subject: Signal Processing (eess.SP)
subject: FOS: Physical sciences
subject: FOS: Physical sciences
subject: FOS: Electrical engineering, electronic engineering, information engineering
subject: FOS: Electrical engineering, electronic engineering, information engineering

description: Back-n is a white neutron experimental facility at China Spallation Neutron Source (CSNS). The time structure of the primary proton beam make it fully applicable to use TOF (time-of-flight) method for neutron energy measuring. We implement the electronics of TOF measurement on the general-purpose readout electronics designed for all of the seven detectors in Back-n. The electronics is based on PXIe (Peripheral Component Interconnect Express eXtensions for Instrumentation) platform, which is composed of FDM (Field Digitizer Modules), TCM (Trigger and Clock Module), and SCM (Signal Conditioning Module). T0 signal synchronous to the CSNS accelerator represents the neutron emission from the target. It is the start of time stamp. The trigger and clock module (TCM) receives, synchronizes and distributes the T0 signal to each FDM based on the PXIe backplane bus. Meantime, detector signals after being conditioned are fed into FDMs for waveform digitizing. First sample point of the signal is the stop of time stamp. According to the start, stop time stamp and the time of signal over threshold, the total TOF can be obtained. FPGA-based (Field Programmable Gate Array) TDC is implemented on TCM to accurately acquire the time interval between the asynchronous T0 signal and the global synchronous clock phase. There is also an FPGA-based TDC on FDM to accurately acquire the time interval between T0 arriving at FDM and the first sample point of the detector signal, the over threshold time of signal is obtained offline. This method for TOF measurement is efficient and not needed for additional modules. Test result shows the accuracy of TOF is sub-nanosecond and can meet the requirement for Back-n at CSNS.
description: 4 pages, 13 figures, 21st IEEE Real Time Conference