technological advances in three - dimensional ( 3d ) imaging such as cone - beam computed tomography ( cbct ) appear to offer significant advantages in both quality and quantity of data representing true anatomy 
 , 




 . the increased use of computed tomography in dentistry has also spurred the improvement of existing software designed primarily for dental use , as well as the development of new software 
 . however , it is necessary to select protocols and to assess the reliability and accuracy of landmark identification in 3d images 
 . development of 3d imaging into a practical and usable cephalometric method was hindered for many years by technologic limitations , high costs , and high radiation doses 
 . advances in cbct technology make its use feasible for specific indications , such as impacted teeth , craniofacial anomalies , airway and temporomandibular joint disorders . in these cases , cbct images could be used for routine cephalometric purposes beyond the diagnostic purposes for which they were primarily taken . this has been extensively done for cephalometric measurements 


 , but to a much lesser extent for maxillofacial computed tomography imaging 


 . whether cbct should be used routinely in all orthodontic patients the voxel is the smallest unit of ct images and its size has an influence on image spatial resolution 
 . by decreasing the voxel size the image spatial resolution increases as there is a decrease in partial volume averaging 


 . on the other hand , by decreasing the voxel size , the scan time increases , as does the probability of patient movement 
 . this is especially of concern in the case of younger individuals , who constitute the overwhelming majority of orthodontic patients . this issue of increased radiation dose to children , and specifically for orthodontic purposes , has caused widespread concern and has been the source of much discussion even in the lay literature 
 . in order to verify the clinical reliability of measurements on multiplanar sections and 3d reconstructions derived from cbct , it is also necessary to evaluate the measurements using anatomic landmarks without the use of metallic markers . the most recent studies involving cbct scans have shown reproducibility and accuracy of cephalometric measurements performed on lateral cephalograms reformatted from cbct images 
  or in 3d - cbct images 


 . however , most of the studies used radiopaque markers 


  and these might have an influence on the accuracy and reproducibility of the measurements . the presence of metallic markers eliminates errors from landmark identification because metallic landmarks can be easily identified and located with a high degree of accuracy and precision . the aim of this study was to verify the influence of voxel size on the accuracy and reproducibility of linear measurements of the mandible performed without metallic markers on 3d cbct images , comparing multiplanar sections and 3d rendering images . approval for this study was received from the ethics committee at the university of so paulo , bauru , brazil . ten dry adult human mandibles , selected from the collection of dry skulls of the department of anatomy from the bauru school of dentistry were scanned . to fit the inclusion criteria , the mandibles had to be well preserved with no large metallic restorations on the teeth , in order to avoid scattering and artifacts . the sample size of each group was calculated based on an alpha significance level of 0.05 and beta of 0.2 to achieve 80% of power 
 . to provide soft - tissue equivalent attenuation , the mandibles were positioned with the mandibular plane parallel and the midsagittal plane perpendicular to the floor , in accordance with the manufacturer s instructions for performing the scans in the clinical setting . the cbct images were acquired using the i - cat classic 3d dental imaging system ( i - cat classic , imaging sciences international , hatfield , pa , usa ) . each mandible was scanned twice : once with a 0.20 voxel resolution ( 36.12 mas , 40 sec , fov of 8 cm , 120 kv ) and once with a 0.40 voxel resolution ( 18.45 mas , 20 sec , fov of 8 cm , 120 kv ) . cbct data were exported in dicom ( digital imaging and communication in medicine ) multi - file format and imported into dolphin11.5 software ( dolphin imaging & management solutions , patterson technology , chatsworth , ca , usa ) . a computer a505-s6975 ( satellite , toshiba , tokyo , japan ) with a dedicated 512-mb video card on a 17-in crystalbrite lcd flat - panel color screen with a maximum resolution of 1440x900 pixels measurements were made separately on the multiplanar sections ( 2d ) and on 3d rendering images display window of dolphin 3d . the multiplanar window displays the sagittal , coronal and axial slices , as well as the 3d rendered image . it was possible to visualize the 3 projections and the 3d rendered image together or choose one of the four images in full screen . linear measurements of 3d coordinates were obtained using several craniometric anatomic landmarks ( figure 1 ) . for multiplanar groups , each landmark ( figure 2 ) was identified and marked on the three orthogonal slices simultaneously ( axial , sagittal and coronal ) . the corresponding linear measurements ( figure 3 ) were determined electronically with the  measurement  tool on the sectional images . for performing the linear measurements which landmarks were not located in the same sectional image , as c  mf , the first mark of a line was performed in the sectional image showing the first landmark and the second mark was performed on a different section showing the second landmark . as the volume rendering was reoriented , the positions of the landmarks were verified in the axial , sagittal and coronal slices and , if necessary , relocated . 
 figure 1frontal ( left ) and sagittal ( right ) views of mandible showing the points 
 
 figure 2points . reference in the anatomical view and in the cone - beam computed tomography ( cbct ) images 
 
 figure 3definitions of mandibular linear measurements 
  for 3d rendering images , the landmarks were identified directly on the surface for 3d images . the measurements were performed twice by 2 blinded independent examiners ( tmff and ja ) . the physical measurements of the dry mandibles were obtained with a high - precision digital caliper ( mitutoyo , mitutoyo sul americana , suzano , sp , brazil ) , identical to those used in similar studies 
 . in this initial study , we selected landmarks that are relatively easy to identify linear measurements were identified on two occasions , one month apart , by one operator ( mlp ) . the mean of the two measurements of each distance was calculated and used as the gold standard to evaluate the accuracy of the cbct 3d images . the measurements were grouped into five groups : physical measurements ( g1 ) ; measurements performed on multiplanar slices with 0.20 mm voxel size ( g2 ) ; measurements performed on multiplanar slices with 0.40 mm voxel size ( g3 ) ; measurements performed on 3d images with voxel size of 0.20 mm ( g4 ) and measurements performed on 3d images with voxel size of 0.40 mm ( g5 ) . intraclass correlation and bland - altman test was used to evaluate the intra- and interobserver reliability . analysis of variance for repeated measures ( anova ) and dunnett s test were used to compare the groups on cbct images and the physical measurements . all tests were performed with statistica software 6.0 ( statistica for windows ; statsoft , tulsa , ok , usa ) and spss version 10 ( spss , spss inc . measurements were made separately on the multiplanar sections ( 2d ) and on 3d rendering images display window of dolphin 3d . the multiplanar window displays the sagittal , coronal and axial slices , as well as the 3d rendered image . it was possible to visualize the 3 projections and the 3d rendered image together or choose one of the four images in full screen . linear measurements of 3d coordinates were obtained using several craniometric anatomic landmarks ( figure 1 ) . for multiplanar groups , each landmark ( figure 2 ) was identified and marked on the three orthogonal slices simultaneously ( axial , sagittal and coronal ) . the corresponding linear measurements ( figure 3 ) were determined electronically with the  measurement  tool on the sectional images . for performing the linear measurements which landmarks were not located in the same sectional image , as c  mf , the first mark of a line was performed in the sectional image showing the first landmark and the second mark was performed on a different section showing the second landmark . as the volume rendering was reoriented , the positions of the landmarks were verified in the axial , sagittal and coronal slices and , if necessary , relocated . figure 1frontal ( left ) and sagittal ( right ) views of mandible showing the points 
 
 figure 2points . reference in the anatomical view and in the cone - beam computed tomography ( cbct ) images 
 
 figure 3definitions of mandibular linear measurements 
  for 3d rendering images , the landmarks were identified directly on the surface for 3d images . the measurements were performed twice by 2 blinded independent examiners ( tmff and ja ) . the physical measurements of the dry mandibles were obtained with a high - precision digital caliper ( mitutoyo , mitutoyo sul americana , suzano , sp , brazil ) , identical to those used in similar studies 
 . in this initial study , we selected landmarks that are relatively easy to identify . linear measurements were identified on two occasions , one month apart , by one operator ( mlp ) . the mean of the two measurements of each distance was calculated and used as the gold standard to evaluate the accuracy of the cbct 3d images . the measurements were grouped into five groups : physical measurements ( g1 ) ; measurements performed on multiplanar slices with 0.20 mm voxel size ( g2 ) ; measurements performed on multiplanar slices with 0.40 mm voxel size ( g3 ) ; measurements performed on 3d images with voxel size of 0.20 mm ( g4 ) and measurements performed on 3d images with voxel size of 0.40 mm ( g5 ) . intraclass correlation and bland - altman test was used to evaluate the intra- and interobserver reliability . analysis of variance for repeated measures ( anova ) and dunnett s test were used to compare the groups on cbct images and the physical measurements . all tests were performed with statistica software 6.0 ( statistica for windows ; statsoft , tulsa , ok , usa ) and spss version 10 ( spss , spss inc . , chicago , il , usa ) . the measurements for investigator 1 showed excellent reliability with intraclass correlation coefficients ( icc ) ranging from 0.93 to 1 ( icc average g2:0.97 ; g3:0.97 ; g4:0.98 and g5:0.98 ) . additionally , the mean absolute differences between the first and second measurements were less than 0.40 mm ( mean average difference g2:0.11 mm and g3:0.02 mm ) for multiplanar images and less than 0.64 mm ( mean average difference g4:0.14 mm and g5:0.03 mm ) for 3d model images . for investigator 2 , the results also showed excellent reliability , with icc coefficients ranging from 0.88 to 0.99 ( icc average - g2:0.96 ; g3:0.95 ; g4:0.97 and g5:0.97 ) . table 1intraobserver and interobserver variation for each group and analysis methods [ intraclass correlation coefficient ( icc ) and bland - altman ]  multiplanar images3d model imagesvariable  g2g3g4g5  0.2 voxel ( n=10)0.4 voxel ( n=10)0.2 voxel ( n=10)0.4 voxel ( n=10)intraobserver  obser veradsd95% ciiccadsd95% ciiccadsd95% ciiccadsd95% ciiccc  c 
 10.010.36 - 0.69 ; 0.711 - 0.010.15 - 0.31 ; 0.2910.120.67 - 1.20 ; 1.440.990.180.77 - 1.34 ; 1.700.98  2 - 0.410.75 - 1.89 ; 1.070.98 - 0.680.75 - 2.16 ; 0.800.970.170.87 - 1.54 ; 1.880.980.460.73 - 0.98 ; 1.900.98mf  mf 
 1 - 0.130.46 - 1.03 ; 0.770.99 - 0.190.49 - 1.14 ; 0.760.980.170.72 - 1.25 ; 1.590.96 - 0.210.49 - 1.17 ; 0.750.98  2 - 0.301.18 - 2.16 ; 2.010.93 - 0.080.68 - 1.41 ; 1.250.97 - 0.210.82 - 1.82 ; 1.400.96 - 0.031.33 - 2.63 ; 2.570.92c  mf 
 1 - 0.041.09 - 2.18 ; 2.100.970.310.67 - 0.99 ; 1.610.980.640.78 - 0.90 ; 2.180.980.260.81 - 1.32 ; 1.840.98  2 - 0.581.21 - 2.95 ; 1.790.95 - 0.920.97 - 2.81 ; 0.970.960.480.92 - 1.33 ; 2.290.970.420.75 - 1.04 ; 1.880.98c  mf 
 10.190.84 - 1.46 ; 1.840.990.240.90 - 1.53 ; 2.010.970.390.67 - 0.93 ; 1.710.980.100.70 - 1.28 ; 1.480.99  2 - 0.660.81 - 2.24 ; 0.920.97 - 0.811.27 - 3.30 ; 1.680.940.100.73 - 1.34 ; 1.540.990.630.85 - 1.03 ; 2.290.97c  mf 
 10.401.22 - 2.0 ; 2.800.950.051.40 - 2.68 ; 2.780.930.041.00 - 1.93 ; 2.010.960.261.06 - 1.83 ; 2.350.95  2 - 1.050.90 - 2.82 ; 0.720.91 - 0.291.01 - 2.27 ; 1.690.95 - 0.090.61 - 1.28 ; 1.100.990.410.68 - 0.92 ; 1.740.97c  mf 
 10.121.40 - 2.62 ; 2.860.94 - 0.041.13 - 2.27 ; 2.190.950.090.90 - 1.67 ; 1.850.970.090.73 - 1.35 ; 1.530.98  2 - 0.650.88 - 2.36 ; 1.060.96 - 0.690.72 - 2.10 ; 0.720.96 - 0.020.99 - 1.96 ; 1.920.970.061.01 - 1.93 ; 2.050.97mco  lco ( right)1 - 0.030.39 - 0.80 ; 0.740.98 - 0.070.40 - 0.84 ; 0.700.980.140.28 - 0.42 ; 0.700.99 - 0.030.45 - 0.91 ; 0.850.98  20.260.47 - 0.66 ; 1.180.970.180.71 - 1.22 ; 1.580.950.770.54 - 0.29 ; 1.830.900.680.55 - 0.40 ; 1.760.92mco  lco ( left)10.390.44 - 0.47 ; 1.250.94 - 0.380.53 - 1.42 ; 0.660.930.010.35 - 0.67 ; 0.690.98 - 0.050.53 - 1.10 ; 1.000.95  2 - 0.050.53 - 1.08 ; 0.990.950.300.81 - 1.29 ; 1.890.88 - 0.080.43 - 0.93 ; 0.770.980.140.33 - 0.50 ; 0.780.98lco  lco 
 10.300.38 - 0.34 ; 0.940.99 - 0.220.50 - 1.22 ; 0.780.99 - 0.030.31 - 0.65 ; 0.591 - 0.150.70 - 1.52 ; 1.220.99  2 - 0.300.70 - 1.09 ; 1.690.980.150.67 - 1.15 ; 1.450.990.770.53 - 0.28 ; 1.820.970.780.53 - 0.27 ; 1.830.97mco  mco 
 1 - 0.070.35 - 0.76 ; 0.620.990.080.54 - 0.98 ; 1.140.98 - 0.220.35 - 0.91 ; 0.470.99 - 0.120.50 - 1.11 ; 0.870.99  2 - 0.330.51 - 1.34 ; 0.680.98 - 0.620.35 - 1.31 ; 0.070.97 - 0.080.35 - 0.78 ; 0.620.99 - 0.051.29 - 0.62 ; 0.520.99total10.11  0.970.02  0.970.14  0.980.03  0.98  2 - 0.41  0.96 - 0.35  0.950.18  0.970.35  0.97interobserverc  c 
 1 x 2 - 0.540.98 - 2.46 ; 1.380.960.000.49 - 0.95 ; 0.960.99 - 0.110.88 - 1.84 ; 1.630.98 - 0.150.56 - 1.25 ; 0.950.99mf  mf 
 1 x 21.000.61 - 0.20 ; 2.190.910.940.54 - 0.12 ; 1.990.923.591.490.66 ; 6.510.263.251.450.38 ; 6.090.45c  mf 
 1 x 2 - 1.201.18 - 3.51 ; 1.150.93 - 1.491.09 - 3.63 ; 0.640.90 - 2.420.80 - 3.99 ; -0.840.84 - 2.560.66 - 3.86 ; -1.260.82c  mf 
 1 x 20.631.05 - 1.45 ; 2.700.970.060.80 - 1.51 ; 1.640.980.071.20 - 2.30 ; 2.430.96 - 0.340.57 - 1.46 ; 0.790.99c  mf 
 1 x 2 - 1.091.34 - 3.72 ; 1.530.89 - 1.070.94 - 2.91 ; 0.790.92 - 2.421.38 - 5.13 ; 0.300.70 - 2.471.06 - 4.56 ; -0.370.69c  mf 
 1 x 20.400.60 - 0.78 ; 1.590.98 - 0.790.81 - 0.79 ; 2.380.950.070.68 - 1.28 ; 1.410.980.100.58 - 1.05 ; 1.240.99mco  lco ( right)1 x 2 - 1.770.54 - 2.83 ; -0.710.69 - 1.840.50 - 2.83 ; -0.860.67 - 0.180.84 - 1.83 ; 1.460.92 - 0.490.46 - 1.40 ; 0.410.95mco  lco ( left)1 x 2 - 1.430.54 - 2.49 ; -0.360.63 - 1.390.63 - 2.63 ; -0.140.63 - 1.130.84 - 2.79 ; 0.520.72 - 0.730.56 - 1.85 ; 0.380.86lco  lco 
 1 x 2 - 1.420.55 - 2.50 ; -0.340.92 - 1.450.45 - 2.34 ; -0.550.93 - 0.091.16 - 2.35 ; 2.190.96 - 0.150.93 - 1.96 ; 1.670.97mco mco 
 1 x 21.990.461.08 ; 2.890.762.120.431.27 ; 2.960.761.340.610.14 ; 2.530.861.170.480.21 ; 2.130.90total  -0.34  0.86 - 0.49  0.87 - 0.13  0.82 - 0.24  0.86ad : average difference ; sd = standard deviation ; ci : confidence interval ; icc : intraclass correlation coefficient 
  ad : average difference ; sd = standard deviation ; ci : confidence interval ; icc : intraclass correlation coefficient cbct multiplanar reconstruction measurements showed good to high interobserver reliability with icc ranging from 0.63 to 0.99 ( mean average g2=0.86 and g3=0.87 ) with the absolute differences between the first and second measurements ranging for 0 to 2.12 mm ( mean average difference g2:-0.34 mm and g3:-0.49 mm , table 1 ) . for 3d reconstruction measurements , independently of the voxel size ( g4 or g5 ) , most of the variables presented good to high interobserver reliability , with the exception of inter - mental foramen width ( mf  mf ) that had fair reliability on 0.2 mm voxel images ( g4=0.26 ) and moderate reliability for 0.4 voxel size ( g5=0.45 ) . the mean values and standard deviations of the physical measurements and of the cbct measurements for all groups are presented in table 2 . two variables ( mf  mf and c  mf ) at groups g2 and g3 were significantly smaller than physical measurements ( g1 ) . for 3d rendering images ( g4 and g5 ) , only two variables out of 10 were not significantly smaller than the physical measurements . no significant difference was found between the voxel size both for 3d rendering and multiplanar sections . table 2mean ( mm ) and standard deviation ( sd ) of linear measurements for dry mandibles compared to cone - beam computed tomography ( cbct ) images with different voxel sizes [ analysis of variance ( anova ) for repeated measures and dunnett s tests ]  physical measurementmultiplanar images3d model images  variableg1g2g3g4g5p  0.2 voxel0.4 voxel0.2 voxel0.4 voxel  ( n=10)(n=10)(n=10)(n=10)(n=10 )  meansdmeansdmdmeansdmdmeansdmdmeansdmd  c  c 
 101.033.98101.154.150.11100.974.01 - 0,06101.124.060,09101.184.260.140.697mf  mf 
 49.17 
 3.0347.91 
 2.98 - 1.2647.91 
 2.81 - 1,2647.16 
 2.64 - 2,0147.76 
 3.01 - 1.420.000*c  mf 76.43 
 4.7176.50 
 4.420.0776.24 
 4.17 - 0,1875.59 
 4.60 - 0,8475.51 
 4.49 - 0.920.000*c  mf 
 104.41 
 4.63103.82 
 4.79 - 0.58103.56 
 4.26 - 0,85102.76 
 4.38 - 1,64103.10 
 4.47 - 1.310.000*c  mf 
 75.16 
 3.9975.66 
 3.640.5075.69 
 3.580,5374.48 
 3.46 - 0,6874.25 
 3.30 - 0.910.000*c  mf 
 102.65 
 3.92102.44 
 4.02 - 0.21102.33 
 3.76 - 0,32101.29 
 3.90 - 1,36101.38 
 3.81 - 1.270.000*mco  lco 
 21.42 
 2.4321.68 
 2.280.2621.69 
 2.280,2619.61 
 2.08 - 1,8120.07 
 2.19 - 1.350.000*mco  lco 
 21.37 
 2.0521.28 
 1.67 - 0.0921.02 
 1.68 - 0,3419.31 
 1.82 - 2,0619.14 
 1.74 - 2.230.000*lco  lco 
 124.25 
 4.05124.00 
 4.03 - 0.25123.78 
 4.10 - 0,47122.73 
 4.21 - 1,53122.74 
 4.04 - 1.510.000*mco  mco 
 83.873.0984.072.860.2084.132.970,2684.072.810,283.762.98 - 0.110.251sd , standard deviation ; md , mean difference * statistically significantdifferent letters represent statistically significant differences ( dunnett s test ) 
  sd , standard deviation ; md , mean difference   statistically significant different letters represent statistically significant differences ( dunnett s test ) the availability of this technology is undoubtedly expanding the use and application of 3d imaging in the field of orthodontics 
 . however , while cbct provides many advantages , patient radiation dose is still higher than conventional cephalometric and panoramic digital imaging modalities 


 . this study was performed to evaluate the reliability and accuracy of linear measurements between common landmarks of the mandible taken from cbct multiplanar sections and 3d volumetric renderings with different voxel sizes . in addition , larger voxel sizes are consistent with the alara principle , keeping the dose as low as reasonably achievable for the intended purpose of the scan 
 . most of the previous studies 






  performed to validate cbct for 3d cephalometrics measured linear distances between anatomical landmarks in dry skulls in combination with radiopaque markers . cephalometric analyses are subject to the influence of the examiner , landmark identification and type of image analyzed ( 2d sections or 3d image ) . the major question of this study was if the difficulty in identifying the landmarks in 3d images can influence the reliability and accuracy of the measurements , and for this reason , metallic markers were not used . this method is close to the real clinical procedures and is more clinically relevant than making measurements between well - defined metallic landmarks . there was excellent intraobserver reliability for both examiners with the mean icc for all groups above 0.95 and mean average differences less than 0.41 mm , independently of the protocol analyzed ( table 1 ) . these differences are not clinically significant for orthodontic / surgical diagnosis and treatment planning , and acceptable for skeletal measurements 


 . additionally , good to high interobserver precision was identified for all groups , with exception of measurement mf the main explanations for the good reliability found in this study are the use of landmarks which are easily identifiable , the calibration between observers and the high spatial definition of cbct images . the differences between multiplanar sections and 3d volume rendering images were probably because landmarks on multiplanar images were identified on the orthogonal slices simultaneously instead of being identified directly on the surface of 3d reformatted images . our study also evaluated the accuracy of measurements performed in cbct images ( table 2 ) . two variables at g2 and g3 and 8 variables at g4 and g5 were significantly smaller than physical measurements ( g1 ) . ( 2009 ) , who showed a tendency for the cbct measurements to slightly underestimate the gold standard . the consistency of the differences we found suggests a systematic error in the 3d model measurement method . one possible explanation is that this error might have been introduced by the measurement software , mainly building line . although few software systems currently have a 3d cephalometric module , most of them have not been tested or validated 
 . the results of the validation measurements for a specific machine or software can not be extrapolated to the entire cbct machine and software 
 . most of them are designed differently and software updates are constant and frequent . in addition , the difference ranged from 0.06 to 2.23 mm . in agreement with previous reports , the average difference in our study is below clinical significance . according to grauer , et al . ( 2009 ) , generating measurements in 3d volumetric images rather than simultaneously in sets of 2d multiplanar images introduces error because of the difficulty involved in locating landmarks in 3d space and the inaccuracies of the user entered threshold used for the construction of 3d virtual surface models . rendered for this reason , greater caution is suggested for linear measurements performed on 3d models 
 . the spatial resolution of the image has an influence in the accuracy of the measurements 
 . most studies on cbct with a large field of view used 0.4 mm voxel size 




 , whereas others with a small field of view used a 0.2 mm or smaller voxel sizes 


 . none of the measurements showed significant difference between the voxel sizes , independently of the method of measurement ( table 2 ) . ( 2010 ) , who have suggested that 0.4 mm voxel resolution is adequate for performing measurements of craniofacial structures . while small voxel sizes increase the image resolution , they also may increase the image noise 
 . the benefits of a shorter scanning time , with its reduced likelihood of patient movement and especially the lower exposure dose compensates for the poorer resolution . besides , large measurements are less influenced by small differences in spatial resolution 
 . on the other hand , the diagnostic ability of cbct images for evaluating small structures appears to be influenced by voxel size . the two most common voxel sizes used in orthodontics  0.3 and 0.4 mm  provide lower spatial resolution than smaller voxel sizes . orthodontic scans with resolutions of 0.3 and 0.4 voxels are ideal for general treatment planning but should be used with caution if the goal is to assess small variations in bone thickness 
 . ( 2009 ) investigated the usefulness of cbct for identifying artificial external root resorption in images with voxel sizes of 0.4 , 0.3 , and 0.2 mm . they concluded that , even though the results were the same for the different voxel sizes , diagnosis was easier at smaller voxel sizes of 0.3 and 0.2 mm . although the benefits of a shorter scanning time satisfy the   as low as reasonably achievable principle , the risks of misdiagnosis and treatment complications must also be weighed . consequently , a scanning protocol with a 0.40 mm voxel size might not be suitable for every patient . voxel size may vary according to the size of the structures to be analyzed as well as to the level of detail desired . in this study , we measured distances that are in excess of the resolutions 0.2 and 0.4 mm of the scan , so that voxel size would not very likely influence the measurements . unlike in conventional cephalometrics , where all the landmarks are identified in one image , namely , the lateral cephalogram , with cbct each landmark must be identified in three different images ( axial , coronal and sagittal ) , making the process of performing the measurements in cbct images more time - consuming for the orthodontist . further , cbct demands a higher radiation dose than traditional cephalometric images . for these reasons , its use should be limited to specific indications , for example , patients with impacted teeth , or those with facial asymmetries or craniofacial anomalies where cbct is better able to quantify the differences between the right and the left side of craniofacial structures . evidence - based will tell whether future advances , especially in terms of dose reduction , will make cbct appropriate for routine use in all orthodontic patients . linear measurements obtained on multiplanar 2d cbct images with 0.2 and 0.4 voxel sizes using i - cat scanner and dolphin software are reliable and accurate for clinical diagnosis and treatment planning . caution should be taken in linear measurements on 3d rendering images , because the measurements were reliable , but not accurate . reducing the voxel size from 0.4 to 0.2 mm does not influence the accuracy and reliability of measurements of large craniofacial structures on cbct-3d images .
objective the purpose of this study was to determine the accuracy and reliability of two methods of measurements of linear distances ( multiplanar 2d and tridimensional reconstruction 3d ) obtained from cone - beam computed tomography ( cbct ) with different voxel sizes.material and methods ten dry human mandibles were scanned at voxel sizes of 0.2 and 0.4 mm . 
 craniometric anatomical landmarks were identified twice by two independent operators on the multiplanar reconstructed and on volume rendering images that were generated by the software dolphin. subsequently , physical measurements were performed using a digital caliper . 
 analysis of variance ( anova ) , intraclass correlation coefficient ( icc ) and bland - altman were used for evaluating accuracy and reliability ( p<0.05).results excellent intraobserver reliability and good to high precision interobserver reliability values were found for linear measurements from cbct 3d and multiplanar images . 
 measurements performed on multiplanar reconstructed images were more accurate than measurements in volume rendering compared with the gold standard . 
 no statistically significant difference was found between voxel protocols , independently of the measurement method.conclusions linear measurements on multiplanar images of 0.2 and 0.4 voxel are reliable and accurate when compared with direct caliper measurements . 
 caution should be taken in the volume rendering measurements , because the measurements were reliable , but not accurate for all variables . 
 an increased voxel resolution did not result in greater accuracy of mandible measurements and would potentially provide increased patient radiation exposure .