Blood Pressure Measurement Based on the Camera and Inertial Measurement Unit of a Smartphone: Instrument Validation Study
Authors: Yoon Yong-Hoon , Kim Jongin , Lee Kwang Jin , Cho Dongrae , Oh Jin Kyung , Kim Minsu , Roh Jae-Hyung , Park Hyun Woong , Lee Jae-Hwan 
Published: July 2023

Abstract

Background
Even though several mobile apps that can measure blood pressure have been developed, the data about the accuracy of these apps are limited.

Objective
We assessed the accuracy of AlwaysBP (test) in blood pressure measurement compared with the standard, cuff-based, manual method of brachial blood pressure measurement (reference).

Methods
AlwaysBP is a smartphone software that estimates systolic blood pressure (SBP) and diastolic blood pressure (DBP) based on pulse transit time (PTT). PTT was calculated with a finger photoplethysmogram and seismocardiogram using, respectively, the camera and inertial measurement unit sensor of a commercially available smartphone. After calculating PTT, SBP and DBP were estimated via the Bramwell-Hill and Moens-Korteweg equations. A calibration process was carried out 3 times for each participant to determine the input parameters of the equations. This study was conducted from March to August 2021 at Chungnam National University Sejong Hospital with 87 participants aged between 19 and 70 years who met specific conditions. The primary analysis aimed to evaluate the accuracy of the test method compared with the reference method for the entire study population. The secondary analysis was performed to confirm the stability of the test method for up to 4 weeks in 15 participants. At enrollment, gender, arm circumference, and blood pressure distribution were considered according to current guidelines.

Results
Among the 87 study participants, 45 (52%) individuals were male, and the average age was 35.6 (SD 10.4) years. Hypertension was diagnosed in 14 (16%) participants before this study. The mean test and reference SBPs were 120.0 (SD 18.8) and 118.7 (SD 20.2) mm Hg, respectively (difference: mean 1.2, SD 7.1 mm Hg). The absolute differences between the test and reference SBPs were <5, <10, and <15 mm Hg in 57.5% (150/261), 84.3% (220/261 ), and 94.6% (247/261) of measurements. The mean test and reference DBPs were 80.1 (SD 12.6) and 81.1 (SD 14.4) mm Hg, respectively (difference: mean −1.0, SD 6.0 mm Hg). The absolute differences between the test and reference DBPs were <5, <10, and <15 mm Hg in 75.5% (197/261), 93.9% (245/261), and 97.3% (254/261) of measurements, respectively. The secondary analysis showed that after 4 weeks, the differences between SBP and DBP were 0.1 (SD 8.8) and −2.4 (SD 7.6) mm Hg, respectively.

Conclusions
AlwaysBP exhibited acceptable accuracy in SBP and DBP measurement compared with the standard measurement method, according to the Association for the Advancement of Medical Instrumentation/European Society of Hypertension/International Organization for Standardization protocol criteria. However, further validation studies with a specific validation protocol designed for cuffless blood pressure measuring devices are required to assess clinical accuracy. This technology can be easily applied in everyday life and may improve the general population’s awareness of hypertension, thus helping to control it.