Cone beam CT dosimetry: A unified and self‐consistent approach including all scan modalities—With or without phantom motion
Authors: Dixon Robert L. , Boone John M. 
Published: June 2010

PurposeThis article describes a common methodology and measurement technique, encompassing both conventional (helical and axial) CT scanningusing phantom translation and cone beam (or narrow fan beam) CT scans about a stationary phantom. Cone beam CT systems having beam widths along the ‐axis wide enough to cover a significant anatomical length (50–160 mm) in a single axial rotation (e.g., in cardiac CT) are rapidly proliferating in the clinic, referred to herein as stationary cone beam CT (SCBCT). The integral format of the CTDI paradigm is not appropriate for a stationary phantom, and is not useful for predicting the dose in SCBCT, nor for perfusion studies or CT fluoroscopy. Likewise, the pencil chamber has limited utility in this domain (even one of extended length).MethodsBy demonstrating, both experimentally and theoretically, the match between the dose distribution for a wide cone beam and that due to an axial scan series , it is shown that the dose on the central ray of the cone beam is both spatially colocated and numerically equal to the dose predicted by CTDI for the axial series; and thus is the logical (and unique) choice for a SCBCT dose‐descriptor consistent with the CTDI‐based dose of conventional CT. This dose can be readily measured using a conventional (short) ionization chamber. Additionally, Monte Carlo simulations of Boone [J. M. Boone, “Dose spread functions in computed tomography: A Monte Carlo study,” Med. Phys. 36, 4547–4554 (2009)], expressed as a scatter LSF (or DSF), allow the application of a convolution‐based model [R. L. Dixon, M. T. Munley, and E. Bayram, “An improved analytical model for CT dose simulation with a new look at the theory of CT dose,” Med. Phys. 32, 3712–3728 (2005)] of the axial dose profile for any primary beam width (any ), fan beam and cone beam alike, from a single LSF kernel; its simple form allows the results to be expressed as simple analytical equations. The experimental data of Mori et al. [S. Mori, M. Endo, K. Nishizawa, T. Tsunoo, T. Aoyama, H. Fujiwara, and K. Murase, “Enlarged longitudinal dose profiles in cone‐beam CT and the need for modified dosimetry,” Med. Phys. 32, 1061–1069 (2005)] from a 256 channel cone beam scanner for a variety of beam widths (28–138 mm) are used to corroborate the theory.ResultsUseful commonalities between SCBCT and conventional CT dose are revealed, including a common equilibrium dose parameter , which is independent of ‐collimator aperture (or ), and a common analytical (exponential growth) function describing the relative approach to scatter equilibrium at for both modalities (with or length ). This function exhibits good agreement with the above‐mentioned cone beam data of Mori et al. for as well as with data obtained from conventional CT scanning [R. L. Dixon and A. C. Ballard, “Experimental validation of a versatile system of CT dosimetry using a conventional ion chamber: Beyond ,” Med. Phys. 34(8), 3399–3413 (2007)] for the same directly irradiated, phantom length .ConclusionsThis methodology and associated mathematical theory provide a physically self‐consistent description of dose between stationary phantom CT and conventional CT, and has predictive capabilities which can be used to effect a substantial reduction in data collection; provide a bridge between modalities; and predict the relevant peak doses for perfusion studies.