Development and implementation of EPID‐based quality assurance tests for the small animal radiation research platform (SARRP)
Authors: Anvari Akbar , Poirier Yannick , Sawant Amit 
Published: April 2018

PurposeAlthough small animal image‐guided radiotherapy (SA‐IGRT) systems are used increasingly in preclinical research, tools for performing routine quality assurance (QA) have not been optimized and are not readily available. Robust, efficient, and reliable QA tools are needed to ensure the accuracy and reproducibility of SA‐IGRT systems. Several investigators have reported custom‐made phantoms and protocols for SA‐IGRT systems QA. These are typically time and resource intensive and are therefore not well suited to the preclinical radiotherapy environment, in which physics support is limited and routine QA is performed by technical staff. We investigated the use of the inbuilt electronic portal imaging device (EPID) to develop and validate routine QA tests and procedures. In this work, we focus on the Xstrahl Small Animal Radiation Research Platform (SARRP) EPID. However, the methodology and tests developed here are applicable to any SA‐IGRT system that incorporates an EPID.MethodsWe performed a comprehensive characterization of the dosimetric properties of the camera‐based EPID at kilovoltage energies over a 11‐month period, including detector warm‐up time, radiation dose history effect, stability and short‐ and long‐term reproducibility, gantry angle dependency, output factor, and linearity of the EPID response. We developed a test to measure the constancy of beam quality in terms of half‐value layer and tube peak potential using the EPID. We verified the SARRP daily output and beam profile constancy using the imager. We investigated the use of the imager to monitor beam‐targeting accuracy at various gantry and couch angles.ResultsThe EPID response was stable and reproducible, exhibiting maximum variations of ≤0.3% and ≤1.9% for short and long terms, respectively. The detector showed no dependence on response at different gantry angles, with a maximum variation ≤0.5%. We found close agreement in output factor measurement between the portal imager and reference dosimeters, with maximum differences ≤3% for ionization chamber and ≤1.7% for Gafchromic EBT3 dosimetry film, respectively. We have shown that the EPID response is linear with tube current (mA) for the entire range of tube kilovoltage peak. Notably, a close relationship was seen between the detector response vs mA slope, and the kilovoltage peak, allowing an independent verification of kilovoltage peak stability based solely on EPID response. In addition to dosimetry tests, according to the beam‐targeting measurement using portal images, maximum displacement of the central axis of the x‐ray beam (due to sag) was 0.76 ± 0.09 mm at gantry 135°/couch 0° and 0.89 ± 0.06 mm at gantry 0°/couch −135°.ConclusionWe performed the first comprehensive analysis on the dosimetric properties of an EPID operating at kilovoltage x‐ray energies. We characterized the detector performance over a 11‐month period. Our results indicate that the imager is a stable and convenient tool for SARRP routine QA tests. We then developed EPID‐based tests to perform routine SA‐IGRT systems QA tasks, such as verifying constancy of beam quality, energy, output, and profile measurements, relative output factors, and beam targeting.