Analytical Evaluation of BioRad D-100 HPLC Analyzer and Workflow Comparison to BioRad TurboLink Variant II HPLC With Reflex to Boronate Affinity
Authors: Galior Kornelia , Leonard Lisha , Dolan Christopher , Deiter Ryan , Baumann Nikola A , Block Darci R 
Published: September 2019

Abstract

Introduction
Measurement of hemoglobin A1c (HbA1C) is used for the diagnosis and management of patients with diabetes. Methods for measuring HbA1C are classified on the basis of charge differences (cation exchange chromatography) or structural differences (boronate affinity chromatography). Some cation exchange high-performance liquid chromatography (HPLC) analyzers may be prone to interferences from hemoglobin variants. Historically, our lab used two methods to report HbA1C results: cation exchange HPLC (VariantII) with reflex to boronate affinity HPLC (Ultra2) methods. A new analyzer (BioRad D-100) with improved interference detection and thresholds for interference was evaluated. The objectives of this study were (1) assess the comparability of HbA1c results between D-100, VariantII, and Ultra2; (2) evaluate the need for maintaining a reflex method; and (3) calculate cancellation rates before and after implementing D-100.

Methods
HbA1c was measured by cation exchange methods using VariantII Turbolink, D-100 (BioRad), and boronate affinity using Ultra2Affinity (Trinity Biotech) according to the manufacturer’s recommendations with the following analytical measuring ranges (AMRs), 4.0% to 17.6%, 4.0% to 18.0%, and 4.0% to 17.6%, respectively. D-100 was compared to VariantII (n = 26) and Ultra2 (n = 31) using residual patient samples from provider-ordered HbA1c having no chromatography flags and with the following chromatography flags on VariantII (n = 113): variant window (n = 41), HbA1c <4.0% (n = 24), HbA1a + HbA1b >5% (n = 19), HbF >5% (n = 9), P3 >10% (n = 6), HbA1c >17.6% (n = 4), unknown peak (n = 3), labile 5% to 15% (n = 3), labile >15% (n = 3), and HbS >60% (n = 1). The cancelation rates were calculated at baseline (January 1-31, 2018) and postimplementation of D-100 (September 20, 2018, to January 22, 2019).

Results
HbA1c results from D-100 were within ±0.3 or 5% with the following frequencies: Ultra2 in 96.2% (25/26) with VariantII in 96.8% (30/31) using samples with no chromatography flags. A total of 113 samples with abnormal VariantII flags were tested using D-100 with 82% (n = 92) not exceeding interference limits for result reporting. These D-100 results were within ±0.3 or 5% of Ultra2 results with the following frequencies: 38.5% (10/26) of samples below (n = 23) or above (n = 3) the AMR, 85.3% (29/34) of results with Hb variant flags, 68.0% (17/25) with unknown/minor peaks, and 71.4% (5/7) with HbF <15%. The clinical concordance was also assessed according to the following decision limits: normal <5.6%, prediabetic = 5.7% to 6.4%, and diabetic >6.5%. Results were concordant in 88% with variant peaks (12/13 w/HbA1c <5.6%, 8/11 w/HbA1c = 5.7%-6.4%, 10/10 w/HbA1c >6.5%), 84% with minor peaks (5/6 w/HbA1c <5.6%, 0/2 w/HbA1c = 5.7%-6.4%, 16/17 w/HbA1c >6.5%), and 86% with HbF (3/3 w/HbA1c <5.6%, 2/3 w/HbA1c = 5.7%-6.4%, 1/1 w/HbA1c >6.5%). Results above/below AMR had 100% concordance (>16.0% or <4.4%). The frequency of VariantII rule violation was 2.1% in January 2018, with 53 of 117 having results confirmed as <4.0% or >16.0% by Ultra2, 2 of 117 reported by Ultra2, and 62 being cancelled due to HbF >15% (n = 23) or variant Hb >60% (n = 39). The frequency of rule violation using only D-100 since implementation was 2.2%.

Conclusion
Implementation of D-100 has removed the need for boronate affinity reflex testing and maintained similar cancellation rates.