Evaluation of the accuracy of the CyberKnife Synchrony™ Respiratory Tracking System using a plastic scintillator
Authors: Akino Yuichi , Sumida Iori , Shiomi Hiroya , Higashinaka Naokazu , Murashima Yoshiichi , Hayashida Miori , Mabuchi Nobuhisa , Ogawa Kazuhiko 
Published: June 2018

PurposeThe Synchrony™ Respiratory Tracking System of the CyberKnife® Robotic Radiosurgery System (Accuray, Inc., Sunnyvale, CA, USA) enables real‐time tracking of moving targets such as lung and liver tumors during radiotherapy. Although film measurements have been used for quality assurance of the tracking system, they cannot evaluate the temporal tracking accuracy. We have developed a verification system using a plastic scintillator that can evaluate the temporal accuracy of the CyberKnife Synchrony.MethodsA phantom consisting of a U‐shaped plastic frame with three fiducial markers was used. The phantom was moved on a plastic scintillator plate. To identify the phantom position on the recording video in darkness, four pieces of fluorescent tape representing the corners of a 10 cm × 10 cm square around an 8 cm × 8 cm window were attached to the phantom. For a stable respiration model, the phantom was moved with the fourth power of a sinusoidal wave with breathing cycles of 4, 3, and 2 s and an amplitude of 1 cm. To simulate irregular breathing, the respiratory cycle was varied with Gaussian random numbers. A virtual target was generated at the center of the fluorescent markers using the MultiPlan™ treatment planning system. Photon beams were irradiated using a fiducial tracking technique. In a dark room, the fluorescent light of the markers and the scintillation light of the beam position were recorded using a camera. For each video frame, a homography matrix was calculated from the four fluorescent marker positions, and the beam position derived from the scintillation light was corrected. To correct the displacement of the beam position due to oblique irradiation angles and other systematic measurement errors, offset values were derived from measurements with the phantom held stationary.ResultsThe average SDs of beam position measured without phantom motion were 0.16 and 0.20 mm for lateral and longitudinal directions, respectively. For the stable respiration model, the tracking errors (mean ± SD) were 0.40 ± 0.64 mm, −0.07 ± 0.79 mm, and 0.45 ± 1.14 mm for breathing cycles of 4, 3, and 2 s, respectively. The tracking errors showed significant linear correlation with the phantom velocity. The correlation coefficients were 0.897, 0.913, and 0.957 for breathing cycles of 4, 3, and 2 s, respectively. The unstable respiration model also showed linear correlation between tracking errors and phantom velocity. The probability of tracking error incidents increased with decreasing length of the respiratory cycles. Although the tracking error incidents increased with larger variations in respiratory cycle, the effect on the cumulative probability was insignificant. For a respiratory cycle of 4 s, the maximum tracking error was 1.10 and 1.43 mm at the probability of 10% and 5%, respectively. Large tracking errors were observed when there was phase shift between the tumor and the LED marker.ConclusionThis technique allows evaluation of the motion tracking accuracy of the Synchrony™ system over time by measurement of the photon beam. The velocity of the target and phase shift have significant effects on accuracy.