Ablation Index-guided 50W ablation for left atrial posterior wall isolation compared with lower powers: feasibility and lesion level analysis
Authors: Tovmassian L , Bierme C , Kozhuharov N , Ding WY , Obeidat M , Chu G , O"brien J , Snowdon RL , Gupta D 
Published: May 2021

Abstract

Funding Acknowledgements
Type of funding sources: Public grant(s) – National budget only. Main funding source(s): Groupe de Rythmologie de la Société Française de Rythmologie
Background
Posterior Wall Isolation (PWI) is increasingly performed for Atrial Fibrillation (AF). The use of Ablation Index (AI)-guided 50W ablation for PWI has not been described, nor the interplay between ablation parameters at this power when compared to lower powers.
Methods
40 consecutive AF patients (26 males, 65.5 ± 10.0 years. 95% non-paroxysmal AF) underwent PWI following pulmonary vein isolation. A roof line and floor line were created with point-by-point ablation, targeting a contact force (CF) of 10-30g, AI 550-600 on the roof and 400-450 on the floor, and inter-tag distance of &lt;6mm. 35-40W powers were used for the first 20 patients, and 50W used for the next 20. Generator impedance was monitored in real time for each lesion. Ablation inside the box was delivered in case of failure of first pass isolation (FPI). All VisiTags (n = 959) were analyzed retrospectively.
Results
PWI was successful in 19(95%) of the 35-40W group and in all 20 patients in the 50W group, with FPI seen in 8(40%) and 10(50%) respectively, p = 0.53. The mean CF and number of RF applications on the roof. floor and inside the box were similar between the two groups. Ablation time per lesion (10.4 [8.8-12.5]sec) and total ablation time per patient (3.84[3.34-4.66] min) were shorter in the 50W group as compared to 35-40W (13.0 [11.6-16.2] sec and 5.86 [4.23-7.73] min respectively), p &lt; 0.005. The mean AI and Impedance Drop were larger in the 50W group (Table). There was no steam pop observed in any of the 959 radiofrequency applications.
Conclusion
Ablation Index guided 50W ablation has a very high success rate for posterior wall isolation with shorter ablation times and higher impedance drop compared to conventional powers. Steam pops may be avoidable by targeting CF &lt; 30g, and by monitoring impedance in real-time. 50W Group(N = 458) 35-40W Group(N = 501) p-value Number of lesions (s, IQR)Roof lineFloor lineAdditional ablation inside box 21.5 [19.5-26.3]7.0 [5.8-9.0]13.0 [10.8-14.3]6.0 [6.0-6.8] 24.0 [20.8-29.5]8.0 [6.0-10.0]12.5 [10.8-14.0]5.5 [2.8-9.0] 0.330.180.850.59 Ablation Time per lesion (s, IQR)Roof lineFloor lineAdditional ablation inside box 10.4 [8.8-12.5]13.0 [10.9-16.0]9.9 [8.7-11.4]8.1 [6.9-9.0] 13.0 [11.6-16.2]14.5 [12.4-19.0]12.7 [11.4-15.9]11.8 [10.6-14.0] &lt;0.005&lt;0.005&lt;0.005&lt;0.005 Total RF Time (min, IQR)Roof lineFloor lineAdditional ablation inside box 3.84 [3.34-4.66]1.54 [1.15-1.90]2.06 [1.68-2.54]0.79 [0.65-1.07] 5.86 [4.23-7.73]1.98 [1.62-2.59]2.78 [2.28-3.25]1.07 [0.59-1.42] &lt; 0.0050.0190.0090.50 Impedance Drop (ohms, IQR)Roof lineFloor lineAdditional ablation inside box 7.4 [5.2-10.3]8.7 [6.1-11.3]6.9 [5.0-10.1]7.1 [5.4-9.8] 6.9 [4.8-9.7]7.5 [5.1-10.0]6.0 [4.2-8.3]8.3 [5.8-10.9] 0.0070.04&lt; 0.0050.17 Contact Force (g, IQR)Roof lineFloor lineAdditional ablation inside box 21.1 [14.5-30.3]23.9 [17.8-32.7]19.2 [13.2-25.3]25.5 [18.5-36.9] 21.2 [14.9-28.1]24.3 [17.2-30.3]19.0 [14.1-25.0]23.1 [16.9-31.7] 0.560.450.870.21 Ablation Index (IQR)Roof lineFloor lineAdditional ablation inside box 471 [441-519]560 [509-571]453 [436-475]461 [430-488] 461 [434-493]502 [466-541]446 [426-464]455 [434-478] &lt; 0.005&lt; 0.005&lt; 0.0050.59 Lesion level analysis for Posterior Wall Isolation