bluuebunny/crossref_metadata_2025_split huggingface dataset:
Automatic Quantitative Radiometric Assay of Bacterial Metabolism
Authors: Buddemeyer Edward , Hutchinson Robert , Cooper Malcolm 
Published: January 2020

Abstract
In a two-compartment scintillation vial, suspensions of bacteria were cultured with 1 µCi of [U-14C] glucose and the released 14CO2 was measured continuosly, cumulatively, and automatically in a liquid-scintillation counter modified to maintain sample temprature at 37 °C. We could follow the metabolism of bacterial populations through their early phase of exponential growth with good precision. The data were obtained conveniently, with use of conventional regents, glassware, and counting equipment. From analysis of the exponential portion of the curves for cumulative activity vs. time, we could measure cell replication rate precisely in units of time. The resulting values were demonstrably independent of some common experimental variables, including the number of bacteria in the inoculum and counting system sensitivity. Sensitivity of the bacteria to antibiotics was measured to within a few percent by noting the relative prolongation of replication time in the presence of those inhibitors. The digital data from the scintillation counter are susceptible to on- or off-line computer analysis, thus providing the prospect for a totaly-automated analytical system. The method shows promise for the mechanized quantitative analysis of bacterial growth, and its inhibition.
Use of laser nephelometry in the measurement of serum proteins.
Authors: Deaton C D , Maxwell K W , Smith R S , Creveling R L 
Published: January 2020

Abstract
Measurement of immunoglobulins (Ig) and of complement component C3 in human serum by automated and manual nephelometric techniques is tedious, and the effective linear range is too narrow. We describe a laser nephelometer/reagent system for measuring serum proteins by the use of forward light scatter (which enhances the ratio of reaction/blank), electronic blank subtraction, laser light (632 nm), and electronic signal selection. We report data establishing the range of linearity for immunoglobulins IgG, IgA, IgM, and complement C3 with this system, and correlations with results by a radial immunodiffusion. We also compared an electroimmunodiffusion system for quantitation of albumin and IgG in cerebrospinal fluid to our technique. The nephelometric system described provides a rapid, accurate, precise, and objective way to measure immunoglobulins and C3.
A minicomputer-automated array spectrometer for liquid-chromatographic detection of metabolites.
Authors: Dessy R E , Reynolds W D , Nunn W G , Titus C A , Moler G F 
Published: January 2020

Abstract
A third-generation multiwavelength array spectrometer was developed as a detector for the high-resolution liquid-chromatographic characterization of metabolites. Components include a PDP-8/e minicomputer, matched pair of linear photodiode arrays, holographically-ruled gratings, fiber optics, flow cells, and high intensity xenon light source. The wavelength range is 256 nm differential with 1-nm resolution and can be adjusted from 200 to 800 nm. The system is capable of storing 20 spectra per second (200-456 nm) in a dual-beam mode. Special features include minicomputer-driven signal enhancement via integration as a function of signal strength. The display output includes presentation of the total absorption chromatogram vs. elution time in both real and post-run time as well as selectable single absorption band vs. elution time (post-run time). Application of this dedicated system is illustrated by the separation and charcterization of the metabolites of a carcinogen, 4-ethylsulfonyl-1 napthalenesulfonamide.
Reverse-phase chromatography of polar biological substances: separation of catechol compounds by high-performance liquid chromatography.
Authors: Molnár I , Horváth C 
Published: January 2020

Abstract
Catecholamines and their metabolites have been separated isocratically by reverse-phase chromatography with aqueous (no organic solvent admixed) eluents. Unlike ion-exchange or ion-pair chromatography, mixtures of both acidic and basic substances can be separated in a single chromatographic run, because the retention is governed by hydrophobic interactions between the nonpolar moiety of the solute molecules and the octadecyl-silica stationary phase. The relative retention values strongly depend on the pH of the eluent, which governs the degree of dissociation of ionogenic solutes. The reproducibility of the results and the stability and efficiency of the chromatographic systems make this approach particularly attractive for use in clinical analysis.
Anticonvulsants in serum, determined with a fully mechanized enzyme analyzer.
Authors: McClean S W , Young D S , Yonekawa W 
Published: January 2020

Abstract
The "EMIT" enzyme immunoassay system (Syva) that involves use of glucose-6-phosphate dehydrogenase (EC 1.1.1.49) as the enzyme label has been adapted to a fully mechanized kinetic enzyme analyzer for analysis of phenobarbital, phenytoin, and primidone. This procedure, compared with the gas-chromatographic procedure of Kupferberg [Clin. Chim. Acta 29, 283 (1970)], gave a weighted regression line--forced through the origin--of y=(1.01+/-0.04) x for phenobarbital, y=(0.95+/-0.04) x for primidone. Within-run coefficients of variation based on single determinations were 9,11, and 22% for primidone, phenobarbital, and phenytoin, respectively. Run-to-run assay CV was 6% for primidone and phenobarbital and 13% for phenytoin, based on the means of triplicate determinations of a sample with a mid-range concentration.
Automated turbidimetry of amylase activity by use of a discrete kinetic analyzer.
Authors: Malkus H , Ibanez J , Castro A , DiCesare J L 
Published: January 2020

Abstract
We adopted an automated turbidimetric rate method for determining amylase activity to the KA-150 Kinetic Analyzer. In the method, an insoluble amylopectin substrate is used with activity determined by the rate of decrease in turbidity. Run-to-run CV for 59 samples with activities up to 400 units (arbitrary amylase units per 100 ml of sample) was 2.8%. A comparison with a similar method, performed by nephelometry, for 104 sera, showed a correlation coefficient (r) of 0.992, with a slope of 1.02. In an additional comparison with an amyloclastic method, for 52 sera r was 0.997, with a slope of 1.01. Day-to-day precision for control sera with activities near the upper limit of normal (279 and 216 units) averaged 2.5% during two months. Measured and calculated activity were linearly related to well above the upper limit of normal (normal range, 60-200 arbitrary units), showing a deviation from linearity of about 10% at 450 units. Commercial reagents available for the Perkin-Elmer Model 91 Amylase Lipase Analyzer can be used with the KA-150.
Quantitative alkaline phosphatase isoenzyme determination by electrophoresis on cellulose acetate membranes.
Authors: Siede W H , Seiffert U B 
Published: January 2020

Abstract
We present a new method for quantitative determination of alkaline phosphatase isoenzymes. This method consists of electrophoretic separation on cellulose acetate membranes, special fixation technique to avoid elution and diffusion of enzyme protein during incubation, specific staining, and quantitative evaluation by densitometric measurement. We highly recommend the precedure for routine clinical laboratory use. In all normal individuals we observe two isoenzymes of hepatic origin and one isoenzyme each of osseous, intestinal, and biliary origin. Quantitative normal values are presented. Precision of the method is calculated, the CV being less than 10%. The exactness of densitometric quantification is proved by comparison with kinetic assay of alkaline phosphatase isoenzymes by use of an elution method. Clinical implications of alkaline phosphatase isoenzymograms are reported and discussed in detail.
Fetal lung maturity, as assessed by gas-liquid chromatographic determination of phospholipid palmitic acid in amniotic fluid.
Authors: Ip M P , Draisey T F , Thibert R J , Gagneja G L , Jasey G M 
Published: January 2020

Abstract
We describe a new and specific method for measurement of lecithin palmitic acid in amniotic fluid. Dipalmitoyl lecithin, the major alveolar surfactant, has previously been estimated by measuring the lecithin-sphingomyelin ratio, total lecithin, total phospholipid phosphorus, and (or) total palmitic acid. Our method is more specific for estimation of dipalmitoyl lecithin, because nonphospholipid sources of palmitic acid are removed by solvent extraction. Using a hexane/2-propanol/sulfuric acid system, we obviated the major interferences from triglycerides and free fatty acids. The palmitic acid derived from the phospholipid fraction is measured by gas-liquid chromatography of its methyl ester. No contribution appears to be made by sphingomyelin palmitic acid--probably owing to the mild hydrolysis conditions. The measured palmitic acid therefore appears to be derived from lecithins, principally dipalmitoyl lecithin. The value for palmitic acid determined by this method correlates well with the lecithin-sphingomyelin ratio and total phospholipid phosphorus. Infants are unlikely to develop respiratory distress syndrome when the measured palmitic acid in amniotic fluid exceeds 8.0 mg/liter, which corresponds to an lecithin-sphingomyelin ratio of 2.0.
Determination of plasma and erythrocyte lithium concentrations by atomic absorption spectrophotometry.
Authors: Hisayasu G H , Cohen J L , Nelson R W 
Published: January 2020

Abstract
We describe a method for determining plasma and erythrocyte lithium concentrations by atomic absorption spectrophotometry. Plasma and hemolyzed whole-blood are diluted and analyzed, with use of a lithium hollow-cathode lamp, at 670.8 nm. Erythrocyte lithium concentrations are calculated indirectly from the hematocrit. The standard deviation for a 0.43 mmol/liter pool of whole blood, run daily over 11 months, was +/-20 mumol/liter (CV=5.1). The lithium concentration of a lyophilized pool assayed periodically over the same period (n=127) was 1.84+/-0.05 mmol/liter (CV=2.7%). The relatively low erythrocyte/plasma lithium ratio and the microhematocrit centrifugation force (9600 to 13 600 X g) make corrections for trapped plasma insignificant. Problems with matrix matching and viscosity are overcome by using a plasma pool standard for calculations. Values for erythrocyte lithium concentrations were unchanged in samples stored at room temperature up to 24 h. Hemolysis appears to be of possible clinical significance. This method is useful as a routine clinical laboratory procedure for monitoring patients with affective disorders, who are undergoing therapy with lithium.
Gas-chromatographic/mass-spectrometric determination of theophylline in whole blood.
Authors: Sheehan M , Hertel R H , Kelly C T 
Published: January 2020

Abstract
We describe a combined gas-chromatographic mass-spectrometric method for determination of theophylline in whole blood. With use of a probability-based matching approach, the extensive gas-chromatographic interference did not affect mass-spectrometric quantitation of theophylline. Concentrations expected and found were linearly related from 1 to 60 mg/liter; the run-to-run coefficient of variation was less than 8%, within-run less than 6%. Some other xan-thines and drugs frequently co-administered with theophylline did not affect results. Only 0.2 ml of blood is required, which makes the procedure useful for monitoring theophylline in newborns and pediatric populations.
Estimation of maprotiline in serum by gas-chromatography, with use of a nitrogen-specific detector.
Authors: Gupta R N , Molnar G , Gupta M L 
Published: January 2020

Abstract
We describe a gas-chromatographic procedure for estimating therapeutic concentrations of maprotiline, a new tetracyclic antidepressant, in serum by use of a nitrogen-specific detector. Desmethyldoxepin, a secondary amine similar in structure to maprotiline, is added as a mass internal standard to the specimen before extraction, to obviate the need for accurate measurements of volumes during extraction and analysis. Both maprotiline and the internal standard are converted to acetyl derivatives, to avoid the adsorption of secondary amines on the column. A highly selective liquid phase is used, which allows a very good separation of desmethyldoxepin, maprotiline, desmethylmaprotiline, and the interferences from serum and reagents.
Calorimetry of cholinesterase activity in serum.
Authors: O'Farrell H K , Chattopadhyay S K , Brown H D 
Published: January 2020

Abstract
Cholinesterase activity in human serum was measured calorimetrically and compared to values obtained with the spectrophotometric method of Rappaport et al. [Clin. Chim. Acta 4, 227 (1959)]. The response of the calorimeter upon mixing serum with acetylcholine corresponded to the cholinesterase activity of the serum, the thermogram peak height being linearly related to enzyme activity at serum dilution volumes at which interference heat was negligible. The interference heat contributed by the dilution mixing characteristics of serum, studied separately, was found to be similar for heat-denatured, aged, or lyophilized samples. In all cases, heat of mixing was negligible when 1 ml of serum in buffer (2/3 by vol) was mixed with 1 ml of buffer. When this amount of serum was assayed with acetylcholine, cholinesterase activities of 2 and 3 kU/liter were recorded as thermograms with peak heights of 2.3 and 3.5 cm, respectively.
Protective effect of glycerol against the increase in alkaline phosphatase activity of lyophilized quality-control serum.
Authors: Tanishima K , Minamikawa Y , Yokogawa N , Takeshita M 
Published: January 2020

Abstract
We studied the effectiveness of glycerol or ethylene glycol in preventing the increase in alkaline phosphatase activity of lyophilized or refrigerated quality-control serum after reconstitution or repeated freezing and thawing. Control serum reconstituted completely from the lyophilized state with subsequent storage at -20 degrees C showed a considerable decrease in alkaline phosphatase activity immediately after thawing, and a gradual increase in activity on allowing it to stand at room temperature. Adding glycerol or ethylene glycol to the reconstituted serum obviated these changes in activity, glycerol being more effective than ethylene glycol. Reconstituted serum with added glycerol maintained maximum activity before refrigeration during either storage for 30 days or on repeated freezing and thawing. Practical applications of this glycerol-supplemented control serum are discussed.
Creatine kinase in serum: 3. Further study of adenylate kinase inhibitors.
Authors: Szasz G , Gerhardt W , Gruber W 
Published: January 2020

Abstract
In search of an appropriate inhibitor to suppress the interference of adenylate kinase with the creatine kinase assay, we found that the combination diadenosine pentaphosphate (10 mumol/liter) and AMP (5 mmol/liter) is a better inhibitor than is fluoride (25 mmol/liter). The latter inhibits adenylate kinase uncompetitively and weakly (Ki = 2.5 mmol/liter), and must be incorporated in the starting reagent, and at 30 degrees C it becoms fully effective only after a lag phase of 6 min. In this concentration, fluoride inhibits adenylate kinase from erythrocytes, muscle, liver or platelets by 94, 92, 88, and 87%, respectively, and creatine kinase by 8%. Bromide and chloride also inhibit creatine kinase. Attempts to replace AMP by a specific inhibitor of liver adenylate kinase failed. Homologs of diadenosine pentaphosphate with either fewer or more phosphoryl groups in the polyphosphate bridge inhibited even more weakly than did the pentaphosphate. Platelets can significantly contribute to adenylate kinase activity in serum. The inhibitor combination inhibited adenylate kinase from platelets by 90%.
Direct potentiometric measurement of sodium and potassium in whole blood.
Authors: Ladenson J H 
Published: January 2020

Abstract
I compared results for sodium and potassium in whole blood and plasma as measured with a newly available potentiometric analyzer, the "Orion SS-30". No significant difference was found for either sodium or potassium in 207 such comparisons. With use of the flowing, high mixing-velocity liquid junction of the Orion SS-30, the residual liquid junction potential due to blood cells was found to be less than 0.1 mV and to be independent of the hematocrit. This is in contrast to the hematocrit-dependent residual liquid junction potential of approximately 0.6 mV noted by others at normal hematocrit values with the open capillary liquid junctions now commonly used in pH instruments. I also found that the potassium concentration can increase significantly during the mixing of whole blood, and such samples should be mixed gently, if at all. Evidently sodium and potassium can be accutately and easily measured directly in heparinized blood.
Design of quality-control specimens for use with a small multi-channel analyzer.
Authors: Caragher T E , Grannis G F 
Published: January 2020

Abstract
Specimens having linear relationships between concentration and instrument response for seven analytes were prepared for use with a small multi-channel analyzer. The specimens had inter-specimen and inter-constituent relationships that facilitated the performance evaluation of all channels simultaneously with each individual specimen. Techniques are described for use of the specimens in various aspects of quality control.
Specific serum quinidine assay by high-performance liquid chromatography.
Authors: Crouthamel W G , Kowarski B , Narang P K 
Published: January 2020

Abstract
The cardioactive drug quinidine has a narrow therapeutic index; consequently, determination of serum quinidine concentrations can be important. We describe a relatively rapid and specific assay for quinidine in serum by high-performance liquid chromatography. It is suitable for use in patient monitoring or pharmacodynamic studies. Alkalinized serum is extracted with benzene, which is evaporated under nitrogen and reconstituted with methanol; an aliquot is chromatographed. Quinidine is separated from its metabolites and dihydroquinidine (a contaminant in quinidine raw materials). The retention time for quinidine is 4 min 10 s, for dihydroquinidine it is 5.5 min. Results for patients' sera by this assay method and the double-extraction method of Cramer and Isaksson [Scand. J. Clin. Lab. Invest. 15, 553 (1963] correlate well (r = .975).
Binding proteins from fish sera and intrinsic factor compared in vitamin B12 radioassay.
Authors: Ithakissios D S , Kubiatowicz D O , Windorski D C , Wicks J H 
Published: January 2020

Abstract
We compare serum proteins from rainbow trout, chinook salmon, coho salmon, and oyster toadfish with intrinsic factor as binding proteins in a simplified radioassay for B12. Regression analysis of B12 values, determined in 21 serum samples, shows good correlation (r greater than .975) between results for the fish sera and intrinsic factor. The accuracy of the five assays, as evaluated by analytical recovery of B12 added to pooled human serum, ranges from 90 to 110%. Intra-assay precision ranges from 2.6% for coho salmon serum to 5.5% for intrinsic factor, Ionic strength and variations in pH influence binding of [57Co]vit B12 to the fish sera. Maximum binding occurs from pH 6 to 10 at an ionic strength of 0.1 for all sera. The sera are stable for longer than two years when stored at -20 degrees C. Important advantages of fish sera are their high binding capacity (typical assay dilutions range from 1500-fold for trout serum to more than 50 000-fold for chinook salmon); high affinity for B12 (K greater than 10(12) liter/mol); their relative constant binding characteristics as compared to commercial intrinsic factor preparations; and the finding that the accuracy of radioassays with use of fish sera is not significantly affected by the amount of B12 or human serum proteins present.
Electrophoretic assay of specific estrogen receptors: a contribution to methodology.
Authors: van Netten J P , Algard F T , Montessori G , Weare B 
Published: January 2020

Abstract
Experimental evidence is presented that supports the use of the cold agar-gel electrophoretic method for the clinical quantitation of specific estrogen-binding protein present in some human mammary carcinomas. It is necessary to dilute tumor extracts to avoid interference by serum-borne, non-relevant hormone-binding proteins such as albumin, which migrates to the same anodal region as does the binding protein. Dilution to 2.5 mg or less of total protein per milliliter circumvents such interference while still permitting reliable quantitation of the binding protein. Seventy-two mammary carcinomas were compared for binding-protein content by both the cold agar-gel electrophoresis and a single-point dextran-coated charcoal assay. The correlation coefficient (0.96) indicated excellent agreement between results by the two methods. In addition results are presented which indicate that the preparation of tumor extracts for electrophoresis does not require the use of an ultracentrifuge.
Methotrexate assay by enzymatic inhibition, with use of centrifugal analyzer.
Authors: Finley P R , Williams R J 
Published: January 2020

Abstract
We present a method for measuring methotrexate by enzymatic inhibition, with use of the centrifugal analyzer. In the presence of dihydrofolate reductase (EC 1.5.1.3), dihydrofolic acid is converted to tetrahydrofolate. NADPH acts as a coenzyme. Methotrexate inhibits the action of dihydrofolate reductase. Increasing concentrations of methotrexate depress the conversion of NADPH to NADP+, and the reaction may be followed at 340 nm. Precision, accuracy, and sensitivity are satisfactory. A comparison to the radioimmunoassay method (x) showed the following least-squares regression: y = 0.9813x + 5.6 microgram/liter; r = .9812; Sy = 6.3 microgram/liter. Some interference was noted from bilirubin, lipermia, and hemoglobin. Up to 60 patients' samples per hour may be estimated by this technique.
Method for measuring increased plasma hemoglobin in the presence of erythrocytes.
Authors: Kruszyna R , Smith R P , Ou L 
Published: January 2020

Abstract
Despite its many drawbacks the classic benzidine technique is the method most widely used for determining plasma hemoglobin. An additional disadvantage, reported here, is that the method measures hemoglobin in erythrocytes as well as in plasma. Thus, if plasma specimens are accidentally contaminated with erythrocytes, apparent free hemoglobin in plasma will be artifactually high. Although the present method lacks the sensitivity of the benzidine technique, it self-corrects for the presence of erythrocytes. We have found it particularly useful for specimens of rodent (rat mouse) blood, where some cells appear to resist centrifugation, where the small sample sizes frequently result in accidental contamination of plasma with cells, and where normal values may be higher than those for human plasma.