bluuebunny/crossref_metadata_2025_split huggingface dataset:
An optimized continuous-monitoring procedure for semiautomated determination of serum acid phosphatase activity.
Authors: Bais R , Edwards J B 
Published: January 2020

Abstract
A continuous-monitoring method for measuring acid phosphatase activity with alpha-naphthyl phosphate as the substrate was critically evaluated and modified. Using partially purified prostatic acid phosphatase, we show that certain conditions for the assay must be satisfied to ensure linearity. These conditions include maintaining the pH between 5.6 and 5.9 and the addition of detergent to sustain linearity. The results obtained with alpha-naphthyl phosphate have been compared with those obtained by using p-nitrophenyl phosphate as substrate. When used with an automatic rate analyzer, the modified method is as sensitive but more reproducible.
Stabilization of blood glucose by cooling with ice: an effective procedure for preservation of samples from adults and newborns.
Authors: Lin Y L , Smith C H , Dietzler D N 
Published: January 2020

Abstract
Glycolysis causes a considerable decrease in blood glucose when whole blood is kept at room temperature without preservative. The most commonly used preservative, NaF, makes analysis of other serum constituents such as sodium and calcium and urea difficult or impossible, an especially serious limitation when sample size must be restricted. In samples at room temperature without preservative, plasma glucose decreased 36 mg/liter per hour in blood from adults and 60 mg/liter per hour in blood from newborns. Cooling on ice slowed these rates to 3.9 and 11, respectively. Plasma potassium increased 0.3 mmol/liter per hour in cooled specimens from both adults and newborns. Sodium, calcium, chloride and urea values were unaffected. We conclude that cooling effectively stabilizes plasma glucose for 4 h in samples from both adults and newborns and that potassium may be measured with negligible change for as long as 1 h and other constituents for the entire period.
Automated Enzymatic Determination of L-Lactate in Serum, with Use of a Miniature Centrifugal Anlyzer
Authors: Hadjiioannou T P , Hadjiioannou S I , Brunk S D , Malmstadt H V 
Published: January 2020

Abstract
We describe an automated enzymatic reaction-rate method for spectrophotometric determination of lactate in serum with a miniature centrifugal analyzer. The L(+)-lactate is selectively oxidized in the presence of lactate dehydrogenase (EC 1.1.1.27) and NAD+ to from NADH, whitch is measured from its absorption. Reaction rates are determined automatically, and unknown concentrations are calculated from a computer0generated calibration curve with aqueous lithium lactate standards. Lactte concentrations in the range 0.32-1.6 µg/4 µl (80-400 mg/liter) of sample were determined with relative errors and coefficient of variation of 4.8%. Analytical recovery of lactate added to pooled serum was 89-112% (average, 101%). Comparison with a kit ("Rapid Lactate") method gave a correlation coefficient squared of 0.979 over a concentration range of 39-779 mg/liter.
Enzymatic rate method for measuring cholesterol in serum.
Authors: Pesce M A , Bodourian S H 
Published: January 2020

Abstract
An enzymatic rate assay is described for measuring cholesterol in serum. Cholesterol is analyzed by mixing 5 mul of sample with a reagent consisting of cholesterol esterase, cholesterol oxidase, catalase, acetylacetone, methanol, and hydroxypolyethoxydodecane in a ammonium phosphate buffer at pH 7.0. The rate of increase in absorbance of the dihydrolutidine product is measured at 37 degrees C and 405 nm. The change in absorbance between 4 and 10 min is used to calculate the cholesterol concentrations by using simultaneously determined free cholesterol standards. The change is linearly related to cholesterol concentration up to 4 g/liter. Samples containing bilirubin up to 200 mg/liter, uric acid up to 200 mg/liter, and hemoglobin up to 1 g/liter, or certain drugs (clofibrate, phenobarbital, nicotinic acid, salicylate, Ketochol, and Ovral) gave no interference. Ascorbic acid added to serum caused a positive interference. Lipemic samples gave values that were slightly lower than did the method of Abell et al., used for comparison. Our kinetic assay, compared with the method of Abell et al., the enzymatic assay used with Abbott's Bichromatic Analyzer, and the Technicon SMA 12/60 enzymatic procedure gave correlation coefficients of 0.992, 0.985, and 0.986, respectively.
Determination of the molar absorptivity of NADH.
Authors: McComb R B , Bond L W , Burnett R W , Keech R C , Bowers G N 
Published: January 2020

Abstract
The molar absorptivity of NADH at 340 nm has been determined by an indirect procedure in which high-purity glucose is phosphorylated by ATP in the presence of hexokinase, coupled to oxidation of the glucose-6-phosphate by NAD+ in the presence of glucose-6-phosphate dehydrogenase. The average value from 85 independent determinations is 6317 liter mol-1 cm-1 at 25 degrees C and pH 7.8. The overall uncertainty is -4.0 to +5.5 ppt (6292 to 6352 liter mol-1 cm-1), based on a standard error of the mean of 0.48 ppt and an estimate of systematic error of -2.6 to +4.1 ppt. Effects of pH, buffer, and temperature on the molar absorptivity are also reported.
Double-antibody enzyme immunoassay applied to human alpha 1-fetoprotein.
Authors: Belanger L , Hamel D , Dufour D , Pouliot M 
Published: January 2020

Abstract
We report the development of a double-antibody system for enzyme immunoassay of human alpha 1-fetoprotein. Equilibrium and sequential saturation procedures were evaluated and compared with a similar double-antibody radioimmunoassay. With our method, 3 mug of alpha 1-fetoprotein per liter can be detected. The dose-response curve covers a 100-fold range of analyte concentrations. Within-run and between-run coefficients of variation are respectively 6.9% and 10.0%. Analytical recovery of added antigen is 99.5 +/- 10.6%. Results are routinely obtained in 24 h but can be obtained in less than 8 h. Alpha 1-fetoprotein was measured in a variety of clinical samples, including sera from 488 pregnant women. The advantages of the present method over radioimmunoassay and other enzyme immunoassay procedures are discussed.
Complement C3 in serum and plasma, as measured by radial immunodiffusion with four commercial kits.
Authors: Vladutiu A O , Winiarski B M 
Published: January 2020

Abstract
Quantitation of the C3 component of complement by single radial immunodiffusion is subject to error because the C3 molecule has several antigenic determinants and anti-C3 sera differ in their specificity to these determinants. C3 was measured in plasma (ethylenediaminetetraacetate anticoagulant) and serum, in five normal individuals, by use of four such commercial kits. The effects of ethylenediaminetetraacetate and of incubation at -20, 4, and 37 degrees C for 1, 2, and 7 days were investigated. We found wide variations in C3 in the same sample, as measured with different kits. Incubation for longer than 48 h at 4 degrees C changed C3 concentrations as compared to those in fresh samples. Reference sera for C3 assay and standardized procedures for sample processing are both essential to valid results.
Changes in electrophoretic mobilities of alpha-and beta-lipoproteins as a result of plasma delipidation.
Authors: Cham B E , Knowles B R 
Published: January 2020

Abstract
A two-phase system containing the ternary mixture butanol/disopropyl ether/plasma in different proportions yields ordered delipidation of alpha-lipoproteins, pre-beta-lipoproteins- and beta-lipoproteins in plasma, as quantitated by densitometry after electrophoresis. As a consequence of delipidation the electrophoretic mobilities of pre-beta-lipoproteins and beta-lipoprotein increased, that of alpha-lipoprotein decreased.
Simultaneous quantitation of 4-hydroxy-3-methoxymandelic (vanilmandelic) and 4-hydroxy-3-methoxyphenylacetic (homovanillic) acids in human urine.
Authors: Addanki A , Hinnenkamp E R , Sotos J F 
Published: January 2020

Abstract
We describe a simultaneous analysis for vanilmandelic and homovanillic acids in urine. An aliquot of a final extract, containing phenylpropriopionic acid as external standard, is dried, an internal standard (resorcinol) is added, and the mixture is silylated and chromatographed. The two compounds are identified from their characteristic retention times and from their elution behhavior in relation to four different marker compounds (the external and internal standards and two other naturally occurring compounds, p-hydroxyphenylacetic and hippuric acids) and are quantitated from their relative response factors. Within-run precision (CV) varied from 2 to 14% for vanilmandelic acid and 1 to 18% for homovanillic acid. The ratio of the amount of phenylpropionic acid extracted to the amount of added resorcinol found is a useful quality-control index. Values for vanilmandelic acid obtained by this method and by the technique of Pisano et al. [Clin. Chim. Acta 7, 285 (1962)] correlated well (r = 0.844).
Use of enzyme-loaded erythrocytes in in-vitro correction of arginase-deficient erythrocytes in familial hyperargininemia.
Authors: Adriaenssens K , Karcher D , Lowenthal A , Terheggen H G 
Published: January 2020

Abstract
The capacity of arginase-deficient erythrocytes of patients with familial hyperargininemia to produce urea and to catabolize arginine can be increased in vitro by introducing human liver arginase into their erythrocytes. The results of this study on a specific human model show that it is possible to change the metabolic function of a genetically defective erythrocyte by incorporating exogenous human enzyme. The in vivo application of enzyme-loaded erythrocytes for enzyme replacement therapy of inborn metabolic errors in humans must await in vivo studies on animal models.
Gas-chromatographic analysis for succinimide anticonvulsants in serum: macro- and micro-scale methods.
Authors: Bonitati J 
Published: January 2020

Abstract
A gas-chromatographic analysis for the succinimide anticonvulsant drugs--ethosuximide, methsuximide, and phensuximide--in 1.0 ml of serum was modified to improve its reliability, speed, and precision. A separate procedure for 10-100 mul of serum was also developed. Neither method requires an initial preparation of derivatives. The working range of each method is about 10-100 mg/liter for the macro-method, 2-100 mg/liter for the micro-method. The methsuximide metabolite, N-desmethylmethsuximide, is included in both methods. Concentrations of N-desmethylmethsuximide in the blood of two patients with petit mal epilepsy are reported.
Nonchromatographic radioimmunoassay of unconjugated estriol in plasma, with polyethylene glycol as precipitant.
Authors: Schiller H S , Brammall M A 
Published: January 2020

Abstract
We describe a rapid, reliable radioimmunoassay for unconjugated estriol in plasma. Polyethylene glycol (Carbowax 6000) is used to separate antibody-bound and free steroid. The assay is sensitive (25 pg for standards), precise, and accurate. At high and low concentrations of estriol, intra-assay coefficients of variation were 7.1% and 7.6%, respectively, and inter-assay coefficients of variation were 7.2% and 10.0 %, respectively. Free [3H] estriol is not precipitated by polyethylene glycol. This radioimmunoassay of estriol, with a highly specific antiserum and with polyethylene glycol as the antibody precipitant, is a reliable one-day assay that is practical both for the clinical laboratory and the obstetrician.
Measurement of 25-hydroxyvitamin D3 in serum.
Authors: Bouillon R , Kerkhove P V , De Moor P 
Published: January 2020

Abstract
We describe a method for measuring 25-hydroxyvitamin D3 in serum. Extraction with dichloromethane/methanol (2/1 by vol), followed by chromatography on a column of Sephadex LH-20, resulted in an overall analytical recovery of 82% +/- 3.5% (SD). Diluted normal rat serum was used as binding protein because it contains a transport protein that has both a high affinity (Ka = 2 X 10(10) liter/mol) and a high capacity (3 X 10(-6) mol/liter) for 25-hydroxyvitamin D3. There is no advantage in using more complex binding proteins derived either from rachitic animals or from cytosol preparations. Concentrations of 25-hydroxyvitamin D3 (13.4 +/- 4 mug/liter) in the serum of apparently normal Belgian subjects are lower than those reported for North Americans, but resemble those reported for the United Kingdom.
Gamma-glutamyltransferase: Substrate inhibition, kinetic mechanism, and assay conditions.
Authors: Stromme J H , Theodorsen L 
Published: January 2020

Abstract
Gamma-glutamyltransferase activity in serum is shown to be competitively inhibited by the two substrates gamma-glutamyl-4-nitroanilide and glycylglycine. Awareness of this is of importance when one is choosing final reaction conditions for the assay of the enzyme. Gamma-glutamyltransferase probably acts by a "ping-pong bi-bi" kinetic mechanism, which fits with the double competitive substrate inhibition demonstrated. The product, 4-nitro-aniline, appears to be an uncompetitive dead-end inhibitor of both substrates. Various amino acids, particularly glycine and L-alanine, inhibit the enzyme. Their inhibition patterns are uncompetitive with glycylglycine and competitive with gamma-glutamyl-4-nitroanilide. On the basis of the present and other studies, the Scandinavian Society for Clinical Chemistry and Clinical Physiology is going to recommend for routine use a gamma-glutamyltransferase method in which the final concentrations of gamma-glutamyl-4-nitroanilide and glycylglycine are 4 and 75 mmol/liter, respectively.
Clinical laboratory evaluation of the Perkin-Elmer KA-150 Enzyme Analyzer.
Authors: Hardin E , Passey R B , Gillum R L , Fuller J B , Lawrence D 
Published: January 2020

Abstract
We evaluated the Perkin-Elmer KA-150 Enzyme Analyzer in a clinical laboratory situation according to a rigid evaluation protocol. The factors evaluated for five commonly assayed enzymes were as follows: (a) rate of analysis of patient samples, (b) analytical linearity, (c) overall precision (within-run and between-day), (d) carry-over from a sample with activity near 1000 U/liter, (e) error detection at low and high concentrations, (f) correlation of values with those obtained by methods used in our laboratory and (g) normal-value ranges. We find the KA-150 to be an effective and rapid enzyme analyzer.
Electrophoretic pattern of amylase isoenzymes in serum and urine of normal persons.
Authors: Otsuki M , Saeki S , Yuu H , Maeda M , Baba S 
Published: January 2020

Abstract
We separated and measured amylase isoenzymes in the serum and urine of 3036 normal persons by electrophoresis on a thin layer of polyacrylamide gel. We wished to establish the normal pattern of these isoenzymes and to evaluate the usefulness of this method of electrophoresis in clinical diagnosis. Results for patients with hyper- or hypofunctioning pancreas and salivary glands suggested that essentially all the isoamylases in human serum and urine are derived from the salivary glands and the pancreas, and revealed that isoamylases of more than 98% of normal persons consisted of two major isoenzymes and two to three minor ones. Although these observations indicate that data on changes in the proportion of amylase activity of each isoenzyme can be useful in clinical medicine, the following points should be remembered: (a) quantitative differences in the isoenzyme pattern were observed, depending upon the condition of the samples; (b) because the proportion of isoamylase activity in serum of different normal persons differs, seriatim determination of amylase isoenzymes is necessary; and (c) because five different genetically controlled types of isoamylases were observed in normal persons, genetic investigations are also necessary.
Comparison of three procedures for isolating human ferritin, for use as a standard in an immunoradiometric assay.
Authors: Gonyea L M , Lamb C M , Sundberg R D , Deinard A S 
Published: January 2020

Abstract
We evaluated three methods for isolating ferritin for use as a standard, with respect to purity of the products, ease of preparation, and yield. Examination of the respective products by gel filitration on Sephadex G-200 and Sepharose 6B suggested that the preparations isolated by ammonium sulfate and cadmium sulfate precipitation (Method 1) and by ultracentrifugation (Method 2) were homogeneous, while the product of a procedure including precipitation with ammonium sulfate (Method 3) contained significant amounts of nonferritin protein. The ratios of ferritin as measured by immunoradiometric assay to the amount of protein in the product indicated the ferritin prepared by Method 1 to be the most highly purified. Methods 1 and 2 were both comparatively simple. Although the yield from Method 1 was lowest, it is probably the method of choice, on the basis of the ease of obtaining a highly purified product. The most appropriate method for estimating protein in the isolated preparations appears to be that of Lowry et al.
Theoretical approaches to estimation of plasma renin activity: a review and some original observations.
Authors: Oparil S 
Published: January 2020

Abstract
Performance of accurrate, reproducible, and interpretable assays for plasma renin activity and other components of the renin/angiotensin system in the clinical setting requires a clear understanding of the various reactions in the renin/angiotensin cascade and the nature of their interactions. Plasma renin activity, the rate of angiotensin generations from plasma incubated in vitro, is the most commonly used clinical index of function in the renin/angiotensin system. Renin activity is measured by radioimmunoassay of angiotensin I generated in vitro under carefully controlled conditions. The value obtained for plasma renin activity depends on pH and duration of incubation and on the method used to protect the angiotensin I generated. We recommend incubation at neutral pH in buffered plasma for three hours in the presence of either ethylenediaminetetraacetate + 8-hydroxyquinolone + dimercaprol or ethylenediaminetetraacetate + phenylmethylsulfonylfluoride. Addition of a standard preparation of human renin to the plasma incubation step of the renin activity assay serves the dual purpose of permitting measurement of renin activity and furnishing an internal standard for comparison of assay procedures. The many variables among renin assay methods can be cancelled by referring to a common internal renin standard.
Gas-liquid chromatographic profile of neutral and acidic metabolites in cerebrospinal fluid from newborns and infants.
Authors: Malcolm R D , Leonards R 
Published: January 2020

Abstract
A profile (chromatographic pattern) of the neutral and acidic metabolites present in cerebrospinal fluid from newborns and infants was obtained by gas-liquid chromatography. The metabolites are those extracted with ethyl acetate and diethyl ether. They are converted to trimethylsilyl ether derivatives, and chromatographed on a column containing 5% silicone OV-101 (methyl) packed on Chromosorb W. Several substances were tentatively identified from their methylene unit values. We established a control profile for infants, and compared profiles for infants and adults. Noteworthy qualitative and quantitative differences from the control were seen for cerebrospinal fluid from subjects with neurological disorders of infections. The technique may be of use in neurological diagnosis.
Improved determination of prostatic acid phosphatase (sodium thymolphthalein monophosphate substrate).
Authors: Ewen L M , Spitzer R W 
Published: January 2020

Abstract
We have modified a previously described method for determining acid phosphatase, with thymolphthalein monophosphate as substrate, to increase its sensitivity. We assessed effects of serum on variables influencing acid phosphatase activity as measured by this method. The method is shown to be not completely specific for prostatic acid phosphatase. The importance of standardizing methodology in measurement of enzyme activities is demonstrated.
Creatine kinase in serum: 1. Determination of optimum reaction conditions.
Authors: Szasz G , Gruber W , Bernt E 
Published: January 2020

Abstract
To establish optimum conditions for creatine kinase (EC 2.7.3.2) activity measurement with the creatine phosphate in equilibrium creatine reaction, we re-examined all kinetics factors relevant to an optimal and standardized enzyme assay at 30 and 25 degrees C. We determined the pH optimum in vaious buffers, considering the effect of the type and concentration of the buffer, as well as the influence of various buffer anions on the activity. The relation between activity and substrate concentration was shown and the apparent Michaelis constants of creatine kinase for creatine phosphate and ADP were evaluated. We tested the effect on creatine kinase measurement of the concentration of substrates (glucose and NADP+) in the auxillary and indicator reactions, especially the influence of the added auxiliary (hexokinase) and indicator (glucose-6-phosphate dehydrogenase) enzymes on the lag phase, at different temperatures. The NADP+ concentration proved to be the factor limiting the duration of constant reaction rate. We studied the inhibition of creatine kinase and adenylate kinase by AMP and established a convenient AMP concentration. For reactivation of creatine kinase, N-acetyl cysteine as sulfhydryl compound was introduced. Finally, we examined the relationship between activity and temperature.
Interference of fatty acids in the competitive protein-binding assay for serum thyroxine.
Authors: Shaw W , Hubert I L , Spierto F W 
Published: January 2020

Abstract
An increase in apparent thyroxine values obtained by competitive protein-binding assay on storage of sera is well documented. We find that the major source of this positive bias is probably the unsaturated nonesterified fatty acids. Nonesterified fatty acids cause a positive basis in the competitive protein-binding assay for serum thyroxine because they inhibit the binding of radioactive thyroxine by the serum-binding reagent (probably thyroxine-binding globulin). This inhibition by fatty acids may be due to the formation of a fatty acid/thyroxine complex. The degree of inhibition caused by the fatty acids depends on the length of the carbon chain of the fatty acid and the degree of saturation. Short-chain fatty acids are more potent inhibitors of thyroxine binding than those with a longer chain, and unsaturated fatty acids are more potent inhibitors of thyroxine binding than are saturated fatty acids. The polyunsaturated fatty acid, arachidonic acid, was the most potent inhibitor of all the fatty acids tested. Triglycerides (triacylglycerols) insignificantly inhibit thyroxine binding in this assay.
Four commercial urease reagents and a laboratory-prepared reagent compared for analysis of blood urea nitrogen with the Beckman analyzer.
Authors: Norkus N S , Kubasik N P , Sine H E 
Published: January 2020

Abstract
Three laboratory-prepared urease reagents were compared with a commercial preparation supplied for routine use on the Beckman Blood Urea Nitrogen Analyzer. There were discrepancies in results for urea nitrogen among the four urease reagents when matching serum and the corresponding oxalate/fluoride treated plasma were compared as measured with the Beckman Analyzer and continuous-flow (AutoAnalyzer) method. All four urease preparations were affected by fluoride, but to different extents. We believe that an effective laboratory reagent can be prepared in the laboratory at significantly lower cost.