Comparison of geometric design, metallurgical features and mechanical behavior of ProTaper Gold SX and two replica-like instruments
Authors:   Department of Endodontics, Faculdade de Medicina Dentária, Universidade de Lisboa, Lisboa, Portugal, Martins Jorge ,   Unidade de Investigação em Ciências Orais e Biomédicas (UICOB), Faculdade de Medicina Dentária, Universidade de Lisboa, Portugal,   Centro de Estudo de Medicina Dentária Baseada na Evidência (CEMDBE), Faculdade de Medicina Dentária, Universidade de Lisboa, Portugal, Silva Emmanuel ,   Department of Endodontics, School of Dentistry, Grande Rio University (UNIGRANRIO), Rio de Janeiro, Rio de Janeiro, Brazil,   Department of Endodontics, Fluminense Federal University, Niteroi, Rio de Janeiro, Brazil, Marques Duarte ,   Centro de Estudo de Medicina Dentária Baseada na Evidência (CEMDBE), Faculdade de Medicina Dentária, Universidade de Lisboa, Portugal,   LIBPhys-FCT UID/FIS/04559/2013, Lisboa, Portugal,   Prosthodontics Postgraduation Department, Faculdade de Medicina Dentária, Universidade de Lisboa, Lisboa, Portugal, Oliveira Sofia ,   LIBPhys-FCT UID/FIS/04559/2013, Lisboa, Portugal,   Faculdade de Medicina Dentária, Universidade de Lisboa, Lisboa, Portugal, Caramês João ,   LIBPhys-FCT UID/FIS/04559/2013, Lisboa, Portugal,   Department of Oral Implantology, Faculdade de Medicina Dentária, Universidade de Lisboa, Lisboa, Portugal, Versiani Marco ,   Dental Specialty Center, Brazilian Military Police, Minas Gerais, Brazil
Published: January 2021

Objectives: To assess the characteristics, performance, and safety of ProTaper Gold SX (PTG SX) replica-like instruments. Methods: A multimethod research involving the assessment of geometric design (macro and through scanning electron microscope), Ti/Ni proportions (energy-dispersive X-ray spectroscopy), phase transformation temperatures (differential scanning calorimetry), torsional resistance, and flexibility was conducted to compare two SX replica-like instruments (Premium Taper Gold and Go-Taper Flex) with the original PTG SX. Results were compared using one-way ANOVA with post hoc Tukey or Kruskal-Wallis tests according to gaussian or non-gaussian distribution (Shapiro-Wilk test). The significance level was set at 0.05. Results: The three SX instruments have nine cutting blades with helix angles of approximately 21º, symmetrical blade geometry without radial lands, a convex triangular cross-section design, and near equiatomic Ti/Ni atomic proportions. Design differences were observed at the tip geometry and surface finishing, which was smoother in the Premium Taper Gold and irregular in the Go-Taper Flex. Distinct R-phase start (Rs) and finish (Rf) phase transformation temperatures were noted between PTG (Rs ~48ºC and Rf ~30ºC), Go-Taper Flex (Rs ~43ºC and Rf ~25ºC), and Premium Taper Gold (Rs ~30ºC and Rf ~15ºC). In the mechanical tests, Go-Taper Flex presented lower maximum torque (mean 0.5 N.cm) and higher bending resistance (mean 582.2 gf) (less flexibility) than PTG (means 0.8 N.cm and 447.1 gf) (P<0.05). No significant differences were observed in the mechanical tests between Premium Taper Gold and PTG (P>0.05). Conclusions: Overall, the tested replica-like systems showed different features compared to the original PTG SX instrument.