Verification of the validity of air quality measurements related to the Directive 2004/107/EC in 2019-2020 (DIRME2019)
Authors:   Finnish Meteorological Institute, Kyllönen Katriina , Saarnio Karri , Makkonen Ulla , Hellén Heidi 
Published: August 2021

This project summarizes the results from 2000–2020and evaluates the trueness andthequality control (QC) procedures of the ongoing polycyclic aromatic hydrocarbon (PAH)and trace element measurements in Finlandrelating to Air Quality (AQ) Directive 2004/107/EC. The evaluation was focused on benzo(a)pyrene and other PAH compounds as well as arsenic, cadmium and nickel in PM10and deposition. Additionally, it included lead and other metals in PM10and deposition, gaseous mercury and mercury deposition, andbriefly other specificAQ measurements such as volatile organic compounds (VOC)and PM2.5chemical composition. This project was conducted by the National Reference Laboratory on air quality and thiswas the first time these measurements were assessed. A major part of the project was field and laboratory audits of the ongoing PAH and metal measurements. Other measurements were briefly evaluated through interviews and available literature. In addition, the national AQ database, the expertise of local measurement networks and related publications were utilised. In total, all theseven measurement networks performing PAH and metal measurements in 2019–2020took part in the audits. Eleven stations were audited while these measurements are performed at 22 AQ stations in Finland. For the large networks, one station was chosen to represent the performance of the network. The audits included also six laboratories performing the analysis of the collected samples. The audits revealed the compliance of the measurements with the AQ Decree 113/2017, Directive 2004/107/EC and Standards of the European Committee for Standardization(CEN). In addition, general information of the measurements, instruments and quality control procedures were gained. The results of the laboratory audits were confidential,but this report includes general findings, and the measurement networks were informed on the audit results with the permission of the participating laboratories. As a conclusion, the measurementmethodsusedwere mainly reference methods. Currently, all sampling methods were reference methods; however, before 2018 three networks used other methods that may have underestimated concentrations. Regarding these measurements, it should be noted the results are notcomparable with the reference method. Laboratory methods were reference methods excluding two cases, where the first was considered an acceptable equivalent method. For the other, a change to a reference method was strongly recommended and this realized in 2020. For some new measurements, the ongoing QC procedures were not yet fully established, and advice were given. Some networks used consultant for calibration and maintenance, and thus theywere not fully aware of the QC procedures. EN Standards were mostly followed. Main concerns were related to the checks of flow and calculation of measurement uncertainty, and suggestions for improvement were given. When the measurement networks implement the recommendations given inthe audits, it can be concluded that the EN Standards are adequately followed in the networks. In the ongoing sampling, clear factors risking the trueness of the result were not found. This applies also for the laboratory analyses in 2020. One network had concentrations above the target value, and theindicative measurementsshould be updated to fixed measurements.