Multicenter early United States feasibility study and periprocedural safety of LVIS EVO for the treatment of unruptured intracranial aneurysms
Authors: Kayan Yasha , Delgado Almandoz Josser E , Copelan Alexander , Matouk Charles , Chaudry M Imran , Altschul David , Essibayi Muhammed Amir , Goren Oded , Yim Benjamin , Tsappidi Sam , Zhang Yi Jonathan , Hui Ferdinand K , Samaniego Edgar A , Gudino Andres , Siddiqui Adnan , Jaikumar Vinay , Puri Ajit S , Kühn Anna Luisa , Singh Jasmeet , Ringer Andrew , Hanel Ricardo A , De Toledo Otavio Frederico , Dabus Guilherme , Gooch M Reid , Sizdahkhani Saman , Field Nicholas C , Paul Alexandra R 
Published: June 2024

Background

Stent development has focused recently on low-profile, self-expandable stents compatible with 0.0165 inch microcatheters. The LVIS EVO is the second-generation version of the Low-Profile Visualized Intraluminal Support (LVIS) with improved visibility and resheathability. The LVIS EVO underwent a limited premarket release (PMR) in December 2023. This study aims to report the early safety and feasibility experience with the LVIS EVO stent for the treatment of intracranial aneurysms in the United States (US).

Methods

This was a multicenter, retrospective, observational study evaluating patients who underwent treatment of an intracranial aneurysm with an LVIS EVO stent after the limited PMR. All physicians who had placed an LVIS EVO stent were asked to input their cases after institutional review board approval was obtained. The data were then sent to a single center for analysis. Any patient aged 18 years or older who underwent treatment of an intracranial aneurysm with a LVIS EVO stent in the US was included from the initial PMR in December 2023 until April 2024. Patient age (or ≤90 years old), sex, preoperative modified Rankin Scale (mRS), aneurysm location, aneurysm measurements, and information about preoperative antiplatelet management were all collected. Data on periprocedural complications, 30-day mortality, discharge mRS, and length of stay were also collected.

Results

Some 53 patients with 55 aneurysms underwent treatment with the LVIS EVO stent at 15 institutions. All aneurysms were unruptured. The most common location was the anterior communicating artery (35%) followed by the middle cerebral artery bifurcation (31%). All patients were on dual antiplatelet therapy. The average aneurysm size was 5.2 mm with a neck size of 3.7 mm. The smallest distal parent vessel size was 1.2 mm and 36% of stents were deployed in distal parent vessels <2 mm. All (100%) cases had successful deployment and the stent was repositioned in 10% of cases. A single stent was utilized in 91% of cases. Coils were placed in 48 cases (87.2%) and a microcatheter was jailed in 98% of those cases. Immediate Raymond Roy (RR) Class I occlusion was obtained in 33%, Class II in 22%, Class IIIa in 37%, and Class IIIb in 8% of cases. There were no delayed thromboembolic or hemorrhagic complications.

Conclusions

The LVIS EVO is a braided, self-expanding, retrievable stent with enhanced visibility and smaller cell size. The drawn filled tube (DFT) technology results in improved visibility of the stent, allowing for more controlled stent positioning and visualization of vessel wall apposition. All cases in our series had complete neck coverage and good wall apposition. There were no thromboembolic or hemorrhagic complications.