Diagnostic Accuracy of CareStart™ Malaria HRP2 and SD Bioline Pf/PAN for Malaria in Febrile Outpatients in Varying Malaria Transmission Settings in Cameroon
Authors: Ali Innocent Mbulli , Nji Akindeh Mbuh , Bonkum Jacob Chefor , Moyeh Marcel Nyuylam , Carole Guenang Kenfack , Efon Agni , Dabou Solange , Tchuenkam Valery Pacome Kom , Tah Calvino , Kengne Jean-Paul Chedjou , Achu Dorothy Fosah , Bigoga Jude Daiga , Mbacham Wilfred Fon 
Published: August 2021

Background: There was an increase in the number of malaria cases in Cameroon in 2018 that could reflect changes in provider practice, despite effective interventions. In this study, we assessed the diagnostic performance of two malaria rapid diagnostic tests (mRDTs) for diagnostic confirmation of suspected cases of malaria in public and private health facilities in two malaria transmission settings in Cameroon. Methods: We evaluated the diagnostic performance of CareStart pf and SD Bioline Pf/PAN mRDT and compared these parameters by RDT type and transmission setting. Nested PCR and blood film microscopy were used as references. The chi square test was used for independent sample comparisons, while the McNemar’s test was used to test for the dependence of categorical data in paired sample testing. A p < 0.05 was considered significant in all comparisons. The R (v.4.0.2) software was used for analyses. Results: A total of 1126 participants consented for the study in the four sites. The diagnostic accuracy of the CareStart Pf mRDT was 0.93.6% (0.911–0.961) in Yaoundé, 0.930% (0.90–0.960) in Ngounso, 0.84% (0.794–0.891) in St Vincent Catholic Hospital Dschang and 0.407 (0.345–0.468) in Dschang district hospital. For SD Bioline Pf/PAN the accuracy was 0.759 (0.738–0.846) for St Vincent Catholic Hospital Dschang and 0.426 (0.372–0.496) for the Dschang district hospital. The accuracy was slightly lower in each case but not statistically different when PCR was considered as the reference. The likelihood ratios of the positive and negative tests were high in the high transmission settings of Yaoundé (10.99 (6.24–19.35)) and Ngounso (14.40 (7.89–26.28)) compared to the low transmission settings of Dschang (0.71 (0.37–1.37)) and St Vincent Catholic hospital (7.37 (4.32−12.59)). There was a high degree of agreement between the tests in Yaoundé (Cohen’s Kappa: 0.85 ± 0.05 (0.7–0.95)) and Ngounso (Cohen’s Kappa: 0.86 ± 0.05 (0.74, 0.97)) and moderate agreement in St Vincent hospital Dschang (k: 0.58 ± 0.06 (0.44–0.71)) and poor agreement in the District Hospital Dschang (Cohen’s Kappa: −0.11 ± 0.05 (−0.21–0.01)). The diagnostic indicators of the SD Bioline Pf/PAN were slightly better than for CareStart Pf mRDT in St Vincent Catholic hospital Dschang, irrespective of the reference test. Conclusions: Publicly procured malaria rapid diagnostic tests in Cameroon have maintained high accuracy (91–94%) in the clinical diagnosis of malaria in high malaria transmission regions of Cameroon, although they failed to reach WHO standards. We observed an exception in the low transmission region of Dschang, West region, where the accuracy tended to be lower and variable between facilities located in this town. These results underscore the importance of the routine monitoring of the quality and performance of malaria RDTs in diverse settings in malaria endemic areas.