A Portable Smartphone-Based Laryngoscope System for High-Speed Vocal Cord Imaging of Patients With Throat Disorders: Instrument Validation Study
Authors: Kim Youngkyu , Oh Jeongmin , Choi Seung-Ho , Jung Ahra , Lee June-Goo , Lee Yoon Se , Kim Jun Ki 
Published: May 2021

Background

Currently, high-speed digital imaging (HSDI), especially endoscopic HSDI, is routinely used for the diagnosis of vocal cord disorders. However, endoscopic HSDI devices are usually large and costly, which limits access to patients in underdeveloped countries and in regions with inadequate medical infrastructure. Modern smartphones have sufficient functionality to process the complex calculations that are required for processing high-resolution images and videos with a high frame rate. Recently, several attempts have been made to integrate medical endoscopes with smartphones to make them more accessible to people in underdeveloped countries.

Objective

This study aims to develop a smartphone adaptor for endoscopes, which enables smartphone-based vocal cord imaging, to demonstrate the feasibility of performing high-speed vocal cord imaging via the high-speed imaging functions of a high-performance smartphone camera, and to determine the acceptability of the smartphone-based high-speed vocal cord imaging system for clinical applications in developing countries.

Methods

A customized smartphone adaptor optical relay was designed for clinical endoscopy using selective laser melting–based 3D printing. A standard laryngoscope was attached to the smartphone adaptor to acquire high-speed vocal cord endoscopic images. Only existing basic functions of the smartphone camera were used for HSDI of the vocal cords. Extracted still frames were observed for qualitative glottal volume and shape. For image processing, segmented glottal and vocal cord areas were calculated from whole HSDI frames to characterize the amplitude of the vibrations on each side of the glottis, including the frequency, edge length, glottal areas, base cord, and lateral phase differences over the acquisition time. The device was incorporated into a preclinical videokymography diagnosis routine to compare functionality.

Results

Smartphone-based HSDI with the smartphone-endoscope adaptor could achieve 940 frames per second and a resolution of 1280 by 720 frames, which corresponds to the detection of 3 to 8 frames per vocal cycle at double the spatial resolution of existing devices. The device was used to image the vocal cords of 4 volunteers: 1 healthy individual and 3 patients with vocal cord paralysis, chronic laryngitis, or vocal cord polyps. The resultant image stacks were sufficient for most diagnostic purposes. The cost of the device including the smartphone was lower than that of existing HSDI devices. The image processing and analytics demonstrated the successful calculation of relevant diagnostic variables from the acquired images. Patients with vocal pathologies were easily differentiable in the quantitative data.

Conclusions

A smartphone-based HSDI endoscope system can function as a point-of-care clinical diagnostic device. The resulting analysis is of higher quality than that accessible by videostroboscopy and promises comparable quality and greater accessibility than HSDI. In particular, this system is suitable for use as an accessible diagnostic tool in underdeveloped areas with inadequate medical service infrastructure.