bluuebunny/crossref_metadata_2025 huggingface dataset:
A practical application of computer pattern recognition research: the Abbott ADC-500 differential classifier.
Authors: Green J E
Published: April 2011

The ADC-500 is a new blood cell differential classifier manufactured by Abbott Laboratories. It performs 500-cell leukocyte differentials on both normal and abnormal cells, evaluates red cell morphology and estimates platelet sufficiency at a rate of 40 to 50 samples per hour in stand-alone operation. The ADC-500 system consists of a spinner which prepares a uniform blood monolayer on a slide, a stainer which reproducibly stains the slide with Wright's stain, an encoder which attaches an instrument and human readable identification to the slide and an analyzer which accepts a stack of up to 50 slides, evaluates these slides and prints the results and the slide identification on report forms. The system's analysis rate, which represents a 5- to 10-fold increase over other commercially available differential counters, requires a number of specialized techniques for its realization. One key to this performance is the development of a high speed X-Y slide positioning stage which can move to a new cell and settle in 50 msec. Another is the high degree of parallelism used in the system structure and the pipelining of the data processing. A third is the development of uniform and repeatable sample preparation modules. Within the analyzer module, the autofocus, white cell acquisition and high resolution cell analysis systems are independent and operate in parallel. At the same time within the high resolution cell analysis system, one cell is acquired; the digitized image of a second processed; and a third is classified using pattern recognition techniques. All of these tasks, except focus, are under the control of a minicomputer system. Tests of the system reveal good accuracy and an improvement in precision due to the increase in the number of counted cells.
Rapid analysis of hematology image data: the ADC-500 preprocessor.
Authors: Green J E
Published: April 2011

A sequential, pipeline processor (that we have named the ADC-500 preprocessor) has been developed which scene segments the three color image data from the ADC-500 optics one image element at a time, groups together image elements from each object in the scene and extracts features from each object. The processing occurs at television frame rates, requiring 16.7 msec to process the entire image. This speed was instrumental in allowing the ADC-500 automated differential analyzer to perform routine 500-cell differentials. The preprocessor also contains hardware which simplifies compilation of the three color histograms. The segmentation algorithms implemented in the preprocessor are multicolor extensions of the classical monochrome density histogram threshold method. For most cell image analysis tasks, a sequential pipeline processor of this type should be more economical and as fast or faster than a parallel processor.
Correlation between nuclear morphology and rate of deoxyribonucleic acid synthesis in a normal cell line.
Authors: Dörmer P , Abmayr W
Published: April 2011

Chinese hamster fibroblasts were investigated for the existence of correlations between proliferative activity and nuclear morphology. As a proliferative parameter, the rate of DNA synthesis of individual cells was determined by quantitative 14C-autoradiography. In a second step the images of the Feulgen-stained nuclei were digitized for extraction of features of morphology and texture. These features were correlated with the corresponding DNA synthesis rate values. The following relationships were found: Round nuclei have higher rates of DNA synthesis than flat ones. The more chromatin is packed at the nuclear rim, possibly representing heterochromatin, the lower the rate of DNA synthesis. The DNA synthesis rate also correlates with the graininess of chromatin. Larger areas of condensed chromatin are associated with lower rate values. A fine and irregular network of chromatin, as is typical of immature cell types, is associated with a high rate of DNA synthesis. Although these results are presently confined to the cell line investigated, parallels seem to exist to other cell types, such as erythropoietic cells, which await further investigation.
Texture analysis of cervical cell nuclei by segmentation of chromatin patterns.
Authors: Smeulders A W , Leyte-Veldstra L , Ploem J S , Cornelisse C J
Published: April 2011

Texture parameters of the nuclear chromatin pattern can contribute to the automated classification of specimens on the basis of single cell analysis in cervical cytology. Current texture parameters are abstract and therefore hamper understanding. In this paper texture parameters are described that can be derived from the chromatin pattern after segmentation of the nuclear image. These texture parameters are more directly related to the visual properties of the chromatin pattern. The image segmentation procedure is based on a region grow algorithm which specifically isolates high chromatin density. The texture analysis method has been tested on a data set of images of 112 cervical nuclei on photographic negatives digitized with a step size of 0.125 micron. The preliminary results of a classification trial indicate that these visually interpretable parameters have promising discriminatory power for the distinction between negative and positive specimens.
Automated quantitative analysis of single and double label autoradiographs.
Authors: Rüter A , Aus H M , Harms H , Haucke M , ter Meulen V , Maurer-Schultze B , Korr H , Kellerer A
Published: April 2011

A method for the analysis of silver grain content in both single and double label autoradiographs is presented. The total grain area is calculated by counting the number of pixels at which the recorded light intensity in transmission dark field illumination exceeds a selected threshold. The calibration tests included autoradiographs with low (3H-thymidin) and high (3H-desoxyuridin) silver grain density. The results are proportional to the customary visual grain count. For the range of visibly countable grain densities in single labeled specimens, the correlation coefficient between the computed values and the visual grain counts is better than 0.96. In the first emulsion of the two emulsion layer autoradiographs of double labeled specimens (3H-14C-thymidin) the correlation coefficient is 0.919 and 0.906. The method provides a statistical correction for the background grains not due to the isotope. The possibility to record 14C tracks by shifting the focus through the second emulsion of the double labeled specimens is also demonstrated. The reported technique is essentially independent of size, shape and density of the grains.
The automated multiparameter analyzer for cells (AMAC) IIA, a true bridge circuit Coulter-type electronic cell volume transducer.
Authors: Leif R C , Guarino V , Lefkove N
Published: April 2011

A true bridge Coulter effect (electronic cell volume) transducer has been developed. All resistances of this bridge are now the result of current flow through saline channels. Contamination by electrode products including gas bubbles has been completely eliminated since both power electrodes are now remote from the flow chamber. Since the orifice is in series with an approximately 10 K ohm resistance generated by a gel-filled capillary and a displacement rheostat, it floats electrically, at virtual ground. The other side of the bridge consists of a fluid side-wire. Removing the power electrode from the orifice outlet makes possible downward flow and the use of a single outer sheath, and eliminates noise generated by gas bubbles which could possibly be trapped. It should now be possible to combine this design with that of the AMAC III square orifice, to produce an electro-optical sorter where all parameters are measured simultaneously. This true bridge circuit possesses the further advantage that noise due both to the power supply and to overvoltage at the power electrodes is common-mode rejected, and any drift due to changes in electrode polarization is eliminated. Preliminary experiments confirm results with the AMAC II that hemoglobinopathies can be recognized by the increased coefficient of variation (CV) of the erythrocyte spectra.
Flow-system measurement of cell impedance properties.
Authors: Hoffman R A , Britt W B
Published: April 2011

A flow-system technique has been developed that detects the high-frequency resistance and capacitance changes in a sensing orifice due to the passage of cells through the orifice. The resistance and capacitance changes are related to cell properties such as size, plasma membrane capacitance, and electrical resistivity of the cell interior. The relationship between measured impedances and cellular properties is discussed, and the prototype instrument for making such measurements is described. The instrument can simultaneously detect the dc Coulter volume and two ac parameters related to the complex ac impedance change. Some initial tests of the instrument using plastic microspheres and Chinese hamster ovary cells are described.
A comparison of mercury arc lamp and laser illumination for flow cytometers.
Authors: Peters D C
Published: April 2011

Optical differences between a mercury arc lamp and a laser-illuminated flow cytometer are compared. The distributions of spectral intensities of the two light sources are shown in relation to the excitation characteristics of the fluorescent dyes acriflavine, chromomycin A3, mithramycin, ethidium bromide, Hoechst 33258, and 4,6-diamidino-2-phenylindole (DAPI). Fluorescence intensities of microspheres and Hoechst 33258-stained mouse sperm are compared in the two cytometers. The optical efficiencies are similar and depend on the match of the excitation characteristics of the stain with the emission spectra of the light source.
Dual-laser flow cytometry of single mammalian cells.
Authors: Steinkamp J A , Orlicky D A , Crissman H A
Published: April 2011

An improved dual-laser flow cytometric system for quantitative analysis and sorting of mammalian cells has been developed using a low-power argon and high-power krypton laser as illumination sources, thus permitting the excitation of fluorescent dyes having absorption regions ranging from the ultraviolet to infrared. Cells stained in liquid suspension with fluorescent dyes enter a flow chamber where they intersect two spatially separated laser beams. Separate pairs of quartz beam-shaping optics focus each beam onto the cell stream. Electro-optical sensors measure fluorescence and light scatter signals from cells that are processed electronically and displayed as frequency distribution histograms. Cells also can be electronically separated and microscopically identified. The ease and versatility of operation designed into this system represent a marked technological improvement for dual-laser excited flow systems. Details of this instrument are described along with illustrative examples of cells stained with mithramycin and rhodamine and analyzed for DNA content, total protein, and nuclear and cytoplasmic diameter.
Investigations in high-precision sorting.
Authors: Merrill J T , Dean P N , Gray J W
Published: April 2011

We have investigated the accuracy with which droplets containing cells can be sorted individually onto known and thus relocatable positions. The presence and random arrival of cells and particles in the sorter jet disturbs the orderly production and deflection of droplets, causing a dispersion of sorted droplet trajectories. The magnitude of this dispersion is a function of the phase relationship between the arrival of a cell at the end of the jet and the droplet formation. Using a modified Becton Dickinson Fluorescence-Activated Cell Sorter, we selected for sorting only those droplets that formed with a cell near the center of the droplet. We used this technique to sort Lewis lung tumor cells. The dispersion of droplet positions was reduced from over 3% to about 1% of an average deflection of typically 15 mm for a nozzle with a 50-micron diameter orifice. Sorting onto a surface such as magnetic tape or a microscope slide introduces another uncertainty in position because the cell may be located anywhere within the wetted radius of the droplet on the slide. Sorting onto less-wettable surfaces reduces the wetted radius and thus the variation in cell position.
Individual cell sorting.
Authors: Stovel R T , Sweet R G
Published: April 2011

Current cell sorting machines do not preserve the individual identity of processed cells; after analysis, the cells are assigned to a subpopulation where they are pooled with other similar cells. This paper reports progress on a system that sorts cells individually to precise locations on a microscope slide and preserves them for further observation with a light microscope while recording flow measurement data for each cell. Various electronic and mechanical modifications to an existing sorting machine are described that increase drop placement accuracy and permit individual cell sorting.
Selection of optimal model for the DNA histogram by analysis of error of estimated parameters.
Authors: Lindmo T , Aarnaes E
Published: April 2011

The ability of four different mathematical models of the DNA histogram to give accurate estimates for the fractions of cells in G1, S, and G2 + M has been investigated. The models studied differ in the form and number of parameters of the function used to represent cells in S-phase. Results obtained from simulated DNA histograms suggest that the standard deviations of the model parameters increase exponentially with the width of the G1 and G2 + M peaks of the histogram. Error analysis is presented as a method to select a model of optimal complexity in relation to the resolution provided by the data in a given set of DNA histograms. Introduction of additional parameters improves the agreement between model and data but may result in a less well-posed model. A model with an optimal number of parameters can therefore be found that will yield parameter estimates with the smallest possible standard deviations.
A template method for decomposing flow cytometry histograms of human chromosomes.
Authors: Moore D H
Published: April 2011

A new method for decomposing flow cytometry histograms of isolated human metaphase chromosomes is described and tested. The method is based on fitting a template, composed of the means of all chromosomes of a normal karyotype to the flow histogram. The utility of the method is demonstrated by application to flow measurements of chromosomes from a normal person and comparing the results with those obtained by conventional cytophotometry. The power of the method for detecting gross chromosomal abnormalities, such as trisomy 21, as well as more subtle variations such as a single translocation, is determined for simulated data.
Predicted performance of single- versus multiple-slit flow systems.
Authors: Cambier J L , Wheeless L L
Published: April 2011

Flow systems utilizing multiple orthogonal excitation slits have been proposed as a means of reducing some types of false alarms in prescreening systems for gynecologic cytology. Such false alarms include those caused by orientation-dependent events, such as passage of binucleate or overlapping cells through the measurement region with both nuclei entering the excitation slit simultaneously. This paper presents distributions of optimal projection angles for randomly oriented nuclei passing through one, two, and three slit excitation regions. The results are used to compute observed nuclear spacing of binucleate cells and to compare performance of one, two, and three slit systems in recognition of binucleate and overlapping cells.
Imaging in flow.
Authors: Kay D B , Cambier J L , Wheeless L L
Published: April 2011

Imaging in flow has been valuable in investigating discrepancies in flow cell measurements due to cell orientation and flow dynamics. This paper discusses optical consideration in flow imaging, slit and full field imaging systems and various cell motion arresting techniques from the standpoint of image plane exposure and suitable detector choices. It concludes with an explanation of the slit-imaging techniques employed in a multidimensional slit-scan flow system and slit-scan correlation system.
Detection and discrimination of individual viruses by flow cytometry.
Authors: Hercher M , Mueller W , Shapiro H M
Published: April 2011

A new flow cytometer with a very small observation volume has been developed to detect individual viruses with good resolution, and has been used to discriminate between two types of viral particles based on differences in their light scattering. Measurements of light scattering and fluorescence made with such an instrument can provide a basis for quantitative analysis and sorting of viruses and other particles in the micron and submicron size range.
Multiangle light-scattering analysis of murine teratocarcinoma cells.
Authors: Swartzendruber D E , Price B J , Rall L B
Published: April 2011

Stem cells of the mouse testicular teratocarcinoma are capable of giving rise in vivo and in vitro to a wide variety of cell and tissue types representative of each embryonic germ layer. Multiangle light-scattering measurements in a flow system have been made on these stem cells and on a variety of their differentiated derivatives. This technique is capable of distinguishing the stem cells from parietal yolk sac cells, visceral yolk sac cells, neuronal cells and squamous cells. However, multipotential stem cells cannot be distinguished from stem cells that are restricted in their development to a single pathway.
Cell differentiation based on absorption and scattering.
Authors: Eisert W G
Published: April 2011

Improvements of the flow system allow calibrated cell length measurements down to less than 2 micron at a very high rate. An optical index match to plane viewing windows perpendicular to the optical axis in the flow system keeps the axial symmetry for forward scattered light. Cell size, axial light loss and scattering intensity within different angles were found to be powerful tools to differentiate cell populations. Red cells were analyzed according to various cell surface structures. Lymphocyte populations isolated from different parts of the lymphatic system in rats have been distinguished. Experimental tumor cells showed typical data pattern after different chemical treatments.
Cytofluorometric analysis of R-Thy-1. antigens in various rat lymphocytes with different electrophoretic mobility and organ distribution.
Authors: Voet L , Hannig K , Zeiller K
Published: April 2011

Small bone marrow lymphocytes, which had been previously enriched by velocity sedimentation, thymocytes, lymph node cells and spleen cells were electrophoretically separated, stained with fluorescein conjugated rabbit a-rat-Thy-1. globulin and their fluorescence intensities analyzed with a flow cytophotometer. Thy-1. antigens were found in 80% of the bone marrow small lymphocytes showing low electrophoretic mobility (EPM), in all thymocytes, about 80% of which show low and the rest medium to high EPM, and in a few lymph node cells of high EPM. Thy-1. positive cells were not observed in the spleen. All fluorescence intensity histograms obtained were modal and could be properly fitted with normal curves showing coefficients of variation (C.V.) in the range of 20% to 30%. It was observed that the thymocytes of low EPM had an antibody binding affinity significantly different from that of the other stained lymphocytes. Moreover the surface antigen density decreased in the sequence: thymocytes of low EPM, bone marrow lymphocytes of low EPM and thymocytes of high EPM. The fluorescence intensity of stained lymph node cells of high EPM appeared similar to that of thymocytes of high EPM but was not evaluated precisely. Thus the two dimensional cell analysis provided by a combination of EPM and surface fluorescence of Thy-1.+ cells, allows the characterization of different lymphocyte populations which cannot be clearly identified with normal one dimensional techniques. The biological significance of the results is discussed briefly.
An immunocytochemical approach to cell kinetics automation.
Authors: Gratzner H G , Young I T , Sher S E
Published: April 2011

Antibodies specific for 5-bromodeoxyuridine (BrdUrd) can be used to measure labeling indices in an automated system by image analysis. The antibody, used with an indirect immunoperoxidase technique, will detect de novo DNA synthesis subsequent to growing the cells for various time intervals in 5-bromodeoxyuridine-containing medium. Asynchronously growing CHO cells were pulsed with 3H-5-bromodeoxyuridine, fixed, denatured and then stained with anti-bromouridine antiserum. Peroxidase-coupled goat anti-rabbit IgG was used as the secondary antibody, and slides were stained with diaminobenzidine. Cells which are positive display a reticular pattern indicative of replicating chromatin. "Labeling indices" were generated by scanning the nuclei by TV image analysis. The percentage of labeled cells by the immunocytochemical technique correlates well with that found by autoradiography. Some of the applications of this automated method include cell kinetics and analysis of S-phase by pattern recognition technique.
Flow cytometric analysis of DNA content for ploidy determination in human solid tumors.
Authors: Barlogie B , Göhde W , Drewinko B
Published: April 2011

Flow cytometric studies of cellular DNA content were conducted in 26 patients with a variety of neoplasms. Cell dispersal was achieved with pepsin treatment, and a combination of ethidium bromide and mithramycin was used as DNA specific staining procedure. All measurements were conducted with a new sheath flow chamber in a PHYWE ICP 11 pulse cytophotometer. All but one patient with multiple myeloma had unimodal tumor cell DNA distributions. With human granulocytes as reference standard, 24 of 26 tumors were aneuploid; and of these, 23 showed varying degrees of hyperdiploidy. Except for one patient, ploidy abnormalities were stable on repeat examination.
Automated cervical cytology screening: performance requirements and instrument evaluation.
Authors: Miller R E , Herman C J
Published: April 2011

Continuing progress towards automation of cervical cancer screening requires that criteria for clinical acceptability of automated systems be defined, and that methods be devised for effectively evaluating new technology. The potential roles of automation in cervical cancer detection, performance requirements, instrument evaluation and useful contributions from tuberculosis screening, automated white blood cell differential counting, signal detection theory and decision theory are discussed.
The diagnostic significance of nuclear deoxyribonucleic acid measurement in automated cytology.
Authors: Sprenger E , Witte S
Published: April 2011

Flow cytometric analysis of cytologic samples from four different organs shows that nuclear DNA content of malignant cell populations depends to a large extent on organ of origin of the tumor. This fact must be considered in planning screening systems.
Proximity of lectin receptors on the cell surface measured by fluorescence energy transfer in a flow system.
Authors: Chan S S , Arndt-Jovin D J , Jovin T M
Published: April 2011

Molecules of the lectin concanavalin A have been labeled separately with the fluorescein and rhodamine chromophores and jointly bound to the surface of transformed Friend erythroleukemia cells. The two dyes constitute an ideal donor-acceptor pair for fluorescence resonance energy transfer thereby permitting the determination of the proximity relationships between bound ligand molecules and the corresponding surface receptors. The transfer efficiency at saturation (about 57%) was measured in a multiparameter flow system using laser excitation at 488 nm and detection of fluorescein and rhodamine emission intensities as well as the emission anisotropy of the rhodamine fluorescence for each cell. The degree of energy transfer was estimated from the quenching of donor emission, the sensitization of acceptor emission, and the depolarization of acceptor fluorescence. The system has been modeled according to a formalism developed by Gennis and Cantor (Biochemistry 11: 2509, 1972). We estimate the separation between the surfaces of bound lectin molecules at saturation to be 0-40 A, a range possibly characteristic for micropatches induced by ligand binding.
Visualization of multidimensional spectra in flow cytometry.
Authors: Stöhr M , Futterman G
Published: April 2011

Flow data from a cell sorter have been processed by hardwired circuits which include amplification, discrimination, coincidence requirements, peak sensing and holding, A-D conversion, and a computerized pulse height analysis with storage of the spectra obtained. Two dimensional spectra can be stored directly in memory, on tape and disk. Three and four parametric cellular events can be recorded on line during the flow measurement in a sequential mode on tape for subsequent recall. Simple processing of these data can be performed for displaying of two dimensional projections from these multidimensional spaces based on threshold conditions for the remaining parameters. Interfaced transmission of the stored data to a large scale computer enables more sophisticated data analysis. Data reduction by means of a multidimensional probability analysis has been carried out in order to transfer the spectra to a computerized picture system for display. This system creates perspective two-dimensional images from a three-dimensional data space. Frequency can be converted into grey levels. Hard copy in color (color as the third dimension and color intensity as frequency) simplifies the visualization of multiparametric flow data sets.
Flow cytometry of human gynecologic specimens using log chromomycin A3 fluorescence and log 90 degrees light scatter.
Authors: Barrett D L , Jensen R H , King E B , Dean P N , Mayall B H
Published: April 2011

Flow cytometry and electronic cell sorting are being investigated to screen gynecologic specimens for cervical neoplasia. Cellular DNA content is quantitated by Chromomycin A3 fluorescence and cell size is quantitated by 90 degrees light scatter; the logarithms of the measured intensities are used to produce a two parameter histogram. To determine the cell types responsible for signals in various histogram regions, systematic electronic cell sorting is performed. The sorted fractions are sedimented into microscope slides and stained by the Papanicolaou technique. The cells in each fraction are identified by conventional cytomorphologic criteria. Morphologic analysis of sorted cells reveals histogram regions corresponding to specific cell types. One very important region contains the highest concentration of signals from abnormal cells and is therefore the best region to analyze for specimen abnormality. However, because a significant number of signals in this region are from normal cells, specimens cannot be diagnosed by their analysis. Another important histogram region is composed primarily of signals from endocervical columnar and metaplastic cells. The presence of such cells is a good criterion for specimen adequacy, therefore analysis of signals in this region is essential to assess specimen adequacy for automatic screening.
False alarms in a slit-scan flow system: causes and occurrence rates. Implications and potential solutions.
Authors: Wheeless L L , Cambier J L , Cambier M A , Kay D B , Wightman L L , Patten S F
Published: April 2011

A slit-scan technique was developed as a basis for an automated prescreening system for gynecologic cytology. A flow system based on this technique was fabricated and tested and results indicated that false alarms (misclassification of objects or events from normal specimens as abnormal) are the greatest remaining obstacle to development of an automated prescreening instrument. A dual view correlation system was fabricated to provide exact image-contour correlation in flow and permit precise determination of causes and occurrence rates of false alarms. This paper presents data from correlation analyses of 23 normal cytologic specimens. Major causes of false alarms and their implications to automated prescreening are discussed. A technique that would eliminate the majority of false alarms in flow is presented.
Preselection of alarms in a hybrid system for screening of cervical cancer.
Authors: von Sengbusch G , Achatz M , Beck R , Betz J W , Härtel W , Mann R , Bader P , Schlüter G , Seger G
Published: April 2011

In this report, a preselection of alarms in a system for automated screening of cervical cancer based on depositing the cell sample linearly as a "cell trace" on a tape and analyzing it at different decision levels with increasing complexity, and preliminary results on analyzing cervical material with this system are discussed. The "cell trace" is analyzed with the slit-scan technique. Six parameters are computed: 1) cellular diameter; 2) nuclear diameter; 3) nuclear fluorescence (acriflavin-Feulgen) as nuclear DNA; 4) cellular fluorescence; 5) nuclear to cytoplasm ratio (N/C ratio); and 6) nuclear density. At present, only nuclear fluorescence is used to define a decision boundary between normal and potentially atypical cells. Under this criteria the slit-scan analysis leaves 5% of the events in a sample that must be rechecked at a second decision level in normal cell samples. A further reduction is expected when several slit-scan parameters are used at the first decision step. All events declared suspicious will be investigated in more detail by a two dimensional image analyzing system where the fluorescence image is generated by a laser scanning system. Results obtained in preliminary experiments are discussed in this paper.
High resolution segmentation of cervical cells.
Authors: Bengtsson E , Eriksson O , Holmquist J , Nordin B , Stenkvist B
Published: April 2011

A major problem in the automation of cervical cytology screening is the segmentation of cell images. This paper presents the present status of the work on that problem at the University of Uppsala. A dual resolution system is used. Suspect malignant cells are located at 4 mu resolution. Each such cell is rescanned at 0.5 mu resolution at two different wavelengths, 530 and 570 nm. The nucleus and the cytoplasm are isolated each by two independent methods. For the nucleus adaptive thresholding in the histogram of the 570 nm image and a contouring in a radially transformed version of that image is used. For the cytoplasm a two dimensional thresholding in the 2D histogram and a contouring in a radially transformed version of the 530 nm image is used. If the two nuclear masks agree the surrounding area is checked for disturbing objects. If also the cytoplasm masks agree and are without disturbing objects the whole cell is accepted. The result of the cytoplasm masks agree and are without disturbing objects the whole cell is accepted. The result of the segmentation is thus three categories; free cells, free nuclei and rejected objects. The shape of the objects belonging to the former two categories is checked and irregularly shaped ones are rejected as probably consisting of several overlapping nuclei. Cells passing also this test are classified as normal or malignant. The experience from using this algorithm is discussed and areas for further research are pointed out.
Iterative image transformations for an automatic screening of cervical smears.
Authors: Meyer F
Published: April 2011

The new generation of image analysis systems permits the use of iterative image transformations. It is now possible to construct algorithms where the elementary steps are not arithmetic operations but image transformations. This will be illustrated by two examples. In the first, the absorption image of Feulgen Stained nuclei is processed by contrast algorithms in order to detect suspect cells. In the second, free lying cells are separated from overlapping cells and other artefacts by the use of skeletonization procedures.
Transformation of nuclear morphology during cellular maturation.
Authors: Teich J M , Young I T , Sher S E , Lee J S
Published: April 2011

Using in vivo labeling of mouse leukocytes with tritiated thymidine, cells in the neutrophilic series were studied to determine the change in their nuclear shape as a function of maturation level. Several morphologic shape parameters including perimeter and bending energy were used to quantify the distribution of the nuclear morphology in a given age cohort. The change in these distributions as a function of calender age level was determined. The two parameters named above were used to test the possibility of inferring the age from the quantitative morphology.
Cell volume decrease during Friend leukemia cell differentiation.
Authors: Zucker R M , Wu N C , Mitrani A , Silverman M
Published: April 2011

Friend leukemia cells (GM86, clone 745) were induced to differentiate with dimethyl sulfoxide or butyric acid. The kinetics of induction were measured by cell growth, cell volume distributions, and [3H]thymidine incorporation. From the volume distributions, it was found that the rate of induction was both agent sensitive and concentration dependent. The changes in volume distributions occurred approximately 4 hr earlier with dimethyl sulfoxide induction relative to butyric acid induction. However, the changes with the butyric acid induction were more dependent on concentration. A decrease in labeling indices during the 12- to 20-hr time period was correlated to a decrease in mean cell volume and an increase in the proportion of G1 cells. After the 20-hr time period of induction, an increase in labeling indices and in the percentage of large cells was observed. The data suggest that a transient block of cells in G1 occurred between 12 to 20 hr, and that the early differentiation involved a volume decrease which was related to a redistribution of cell cycle stages. The study was also shown that the changes of cell volume are a rapid monitor to determine the early events of the differentiation process.
Cellular autofluorescence--is it due to flavins?
Authors: Benson R C , Meyer R A , Zaruba M E , McKhann G M
Published: April 2011

For many cell types, cellular autofluorescence in the 500-600 nm spectral region can result in a significant background signal for measurements of weakly fluorescent probes. Measurements of fluorescence intensity and spectra of a variety of cell types and assays of neuronal homogenates are presented to demonstrate that this autofluorescence is most likely due to endogenous flavoproteins.
Polarization of fluorescein fluorescence in single cells.
Authors: Udkoff R , Norman A
Published: April 2011

Measurement of fluorescence polarization (P) gives information about the immediate environment of the fluorescent molecule. We used a flow polarimeter to investigate the factors influencing P of fluorescein in mammalian cells to determine whether such measurements are useful for characterizing heterogeneous cell populations. Fluorescein was introduced into cells by incubation with FDA. Measurements of the intensity of fluorescence (TI) and polarization (P) revealed an unexpected dependence: P decreased with increasing intensity of fluorescence. This may be accounted for by the classical model of the binding of small molecules to protein in which P is dependent on the ratio bound to unbound molecules. We have been able to estimate the quenching due to binding and construct a Scatchard plot. We estimated a wavelength shift from in vitro data consistent with the dependence of P on wavelength seen in our cell work. Generally, the distributions of P are symmetrical. Photon statistics broadens the P distribution of dim cells. However, structure does develop in the P distribution when the cells are deprived of calcium or incubated in the cold. This appears as a shoulder on the P distribution or resolves into two peaks. Calcium deprivation may differentially affect a subpopulation of cells whose significance remains to be explored in various cell types.
Centrifugal separation of cells in sputum.
Authors: Frost J K , Pressman N J , Albright C D , Gill G W , Vansickel M H
Published: April 2011

A centrifugation technique was developed and used to separate fixed cells from the sputa of patients with bronchogenic squamous cell carcinoma and ones with no evidence of cancer. This article presents the relative frequencies of occurrence of five cell types (i.e., leukocytes, macrophages, squamous, columnar and atypical/cancer) in specimen fractions separated from a discontinuous aqueous Ficoll density gradient. These differential counts show that individual cell types may be selectively collected. Atypical and cancer cells are found at high-density gradient fractions (p congruent to 1.138-1.155 g/ml) with a 10-fold enrichment over unprocessed samples.
Model studies on the acriflavine-Feulgen reaction.
Authors: van Ingen E M , Tanke H J , Ploem J S
Published: April 2011

The specificity and quantitative reliability of the Feulgen-acriflavine-SO2 procedure was tested on polyacrylamide model films containing DNA. Noncovalent binding of acriflavine to DNA was observed when the washing procedure, as used in the classical way, was applied. The noncovalently bound acriflavine could be removed with an extra wash in acid-ethanol. The presence of SO2 in the staining solution has been found to enhance covalent binding significantly. The absorbance of films stained by our Feulgen-acriflavine-SO2 procedure is directly proportional to that obtained by the classical Feulgen-pararosanilline-SO2 procedure. The acriflavine-Feulgen procedure has also been tested using a commercial and a purified dye. The use of purified acriflavine, compared to a commercial sample did not result in a significant difference in the maximum absorbance value of stained DNA nor in the absorption or the fluorescence emission spectra of acriflavin covalently bound to DNA.
Cytomorphologic results of preparation experiments for monolayer deposition of cervical material.
Authors: Höffken H , Otto K , Soost H J
Published: April 2011

For automated prescreening methods by high resolution analysis serving as a detecting method in gynecologic mass screening programs a new monolayer deposition method of cervical material has been used. This method will be outlines briefly and the mode of evaluation as well as current cytomorphological findings will be presented. With regard to measurability of the slides prepared according to the new method a number of cytologic criteria were thought to be of particular importance. These criteria are delineated and compiled in a table. A form in which these criteria were listed was filled in by cytopathologists for each slide evaluated. When performing isolation and centrifugation procedures several new morphologic questions arose to the cytopathologist which can only partly be answered by now. If taking into account all criteria of evaluation it may be followed from the present experiences that slides of cervical material are much more suited for automated prescreening methods by high resolution analysis if prepared after isolation and centrifugation in macromolecular liquids than are conventional Papanicolaou smears or slides from suspensions with isolated cells that were not subjected to centrifugation procedures.
Cytofluorometry of electromagnetically controlled cell dedifferentiation.
Authors: Chiabrera A , Hinsenkamp M , Pilla A A , Ryaby J , Ponta D , Belmont A , Beltrame F , Grattarola M , Nicolini C
Published: April 2011

Cellular morphology changes, which appear related to dedifferentiation (despecialization), have been produced in vitro in the nucleated red blood cell of the frog. This has been achieved by controlled alterations in the electrochemical environment of these living cells, both by a selective modification of the ionic concentrations of an isotonic amphibian Ringer solution, and by the electromagnetic induction of pulsating current having specific waveform parameters. Laser flow microfluorometry shows that the modified Ringer solution is able, per se, to partially trigger the process in the same time interval that certain induced current waveforms can significantly affect the number of cells in the so-called dedifferentiated state. It has also been found that, for a given waveform, the repetition rate appears to have a significant effect on the rate of cell change. Preliminary automated image analysis of cell smears suggests that dedifferentiated and normal cells have the same integrated optical density but different nuclear areas. In conclusion, it appears that, after the initial electrochemical trigger, the early stage of the process, when the cells move from a state of specialized function to one of less specific activity, is the unfolding of their chromatin supercoil, not involving DNA synthesis. Then cytofluorometry allowed us to identify, for the first time, fundamental modifications which occur in the cell nucleus under electromagnetic exposure.
Fluorescence flow analysis of lymphocyte activation using Hoechst 33342 dye.
Authors: Lalande M E , Miller R G
Published: April 2011

The in vitro response of murine lymphocytes to allogeneic and mitogenic stimulation has been studied by using the nontoxic fluorescent DNA probe, Hoechst 33342, and a fluorescence flow cytometer-cell sorter. Under appropriate conditions, two peaks of fluorescent intensity, not related to cellular DNA content, can be seen. As early as 12 hr after culture set up, lymphocyte activation can be identified. It appears that Hoechst-labeled lymphocytes of higher fluorescence intensity represent those cells activated by allogeneic stimulation whereas cells obtained from the lower intensity peak are nonresponding lymphocytes.
Fluorescence polarization and pulse width analysis of chromosomes by a flow system.
Authors: Cram L S , Arndt-Jovin D J , Grimwade B G , Jovin T M
Published: April 2011

Isolated Chinese hamster chromosomes have been analyzed using a multiparameter computer-controlled cell sorter to obtain information about unique properties of individual chromosomes. Parameters other than DNA content were sought that would further aid in distinguishing among chromosomes. The polarized emission of the DNA-specific bis-benzimidazole dye Hoechst 33342 was measured for each class of chromosomes identified by a distinct peak, i.e., differeing in DNA content. The emission anisotropy values for all chromosome classes was constant (emission anisotropy = 0.30), and the same value was obtained for purified DNA in solution. Pulse width was found to be a good parameter for resolving chromosomes as a function of total emission in the case of the smaller chromosomes and orientation (i.e., arm length) for large chromosomes. A simple theoretical model for predicting the pulse shapes generated by arbitrarily oriented, thin, rigid rods was developed and applied to the evaluation of the experimental data.
Automatic cell identification and enrichment in lung cancer. II. Acridine orange for cell sorting of sputum.
Authors: Tyrer H W , Golden J F , Vansickel M H , Echols C K , Frost J K , West S S , Pressman N J , Albright C D , Adams L A , Gill G W
Published: April 2011

Fluorescence spectra were obtained from cells from sputum and pleural effusions stained with different fluorescent dyes and fixed by alternate methods. The spectra were referenced to a standard allowing for fluorescence comparisons of unstained and stained cells under various conditions. The metachromasia of acridine orange-stained cells offers nuclear/cytoplasmic differentiation in a single stain; mithramycin and propidium iodide do not. Unstained cells have an appreciable amount of green (546 nm) fluorescence, as does Carbowax in Saccomanno's preservative. Cytoplasm stained with acidine orange also has appreciable green fluorescence. Consequently, cells with much cytoplasm have high total fluorescence. Cytoplasmic fluorescence is negligible with mithramycin or propidium iodide. The metachromasia of acridine orange-stained cells is altered by alcohol and Carbowax levels in fixatives, keeping other factors constant.
The Heidelberg flow analyzer and sorter (HEIFAS) approach on the prescreening of uterine cancer.
Authors: Stöhr M , Goerttler K
Published: April 2011

A dual laser flow system has been proved and used with a novel staining method for simultaneous quantitative DNA and protein analysis. A diamidinophenylindole compound (DAPI) for DNA has been employed in combination with sulforhodamine (SR 101) for protein. With this dye mixture various cell types of the cervical smear could be identified. Despite some overlapping, clusters of leukocytes, endocervicals, (para)basals, intermediates, superficials and dys/neoplastic cells together with artificial events could be discriminated. Up to now the problem of negative cells which create positive signals (false alarms) in the dys/neoplastic region could not be sufficiently solved.
Detection of suspicious cells and rejection of artefacts in cervical cytology using the Leyden Television Analysis System.
Authors: Al I , Ploem J S
Published: April 2011

In slide based automation of cervical cytology the first stage of analysis involves finding possibly suspicious cells, or areas on the slide with these types of cells. By using a television based system such as the Leyden Television Analysis System (LEYTAS), a number of detection methods can be applied to rapidly screen a large number of fields automatically for suspicious cells. In this paper, results using a parameter based on increased nuclear DNA content of cells are given and a second detection method based on a chromatin pattern feature, called chromatin contrast, is discussed. Two blind trials on 41 positive and 22 negative cervical slides, using the Leyden Television Analysis System to detect suspicious cells with an increased nuclear DNA content, were promising. In 1 of the 41 positive cases no suspicious cells were found. In the negative specimens, suspicious cells were detected in 1 of 9 cases and 1 of 13 cases, with the two detection parameters investigated. These findings are discussed and some automatic artefact rejection procedures with preliminary results are given.
Pathology of the endocrine pancreas.
Authors: Heitz P U , Kasper M , Polak J M , Klöppel G
Published: April 2011

An immunocytochemical analysis of 94 pancreatic endocrine tumors revealed that 73 tumors were multicellular. Significant amounts of somatostatin and human pancreatic polypeptide were found by radioimmunoassay in extracts of 19 and 17 tumors resp., in addition to the hormone causing the clinical syndrome. Numerous tumors contained ductular structures. In the surrounding pancreatic parenchyma a proliferation of small ducts and budding-off from the ductular epithelium of endocrine cells was often observed. These features are hallmarks of nesidioblastosis of the endocrine pancreas which is a hyperplasia. In multiple endocrine neoplasia I hyperplasia of the endocrine pancreas is combined with larger nodules, currently labeled tumors. On the basis of these findings it is conceivable that pancreatic endocrine tumors are not primarily neoplastic and autonomous but that they are rather of hyperplastic origin.
Peroxisomes in disease.
Authors: Goldfischer S
Published: April 2011

Cytochemical, biochemical and morphological changes in peroxisomes have been described in human metabolic disorders, in experimental models of disease and in response to drugs and toxins. These include the cerebrohepatorenal syndromes, in which peroxisomes can not be detected and mitochondrial respiration is inhibited, atherosclerosis, alcoholic cardiomyopathy, and tolerance to oxygen toxicity. Although information on the role of peroxisomes in disease is limited, increased awareness of their widespread distribution and the availability of an improved cytochemical procedure for staining peroxisomes in human specimens should provide new insights into their function.