Biomechanical comparison of the therapeutic effect of a novel proximal femoral bionic intramedullary nail and traditional inverted triangle hollow screw on femoral neck fracture
Authors: Zhang Yi-Fan , Ren Chuan , Yao Meng-Xuan , Zhao Shu-Guang , Ding Kai , Wang Hai-Cheng , Chen Wei , Zhang Qi , Zhang Ying-ze 
Published: June 2024

Objective

A novel Proximal Femoral Bionic Nail (PFBN) has been developed by a research team for the treatment of femoral neck fractures. This study aims to compare the biomechanical properties of the innovative PFBN with those of the conventional Inverted Triangular Cannulated Screw (ITCS) fixation method through biomechanical testing.

Methods

Sixteen male femoral specimens preserved in formalin were selected, with the donors’ age at death averaging 56.1 ± 6.3 years (range 47–64 years), and a mean age of 51.4 years. The femurs showed no visible damage and were examined by X-rays to exclude diseases affecting bone quality such as tumors, severe osteoporosis, and deformities. The 16 femoral specimens were randomly divided into an experimental group (n = 8) and a control group (n = 8). All femurs were prepared with Pauwels type III femoral neck fractures, fixed with PFBN in the experimental group and ITCS in the control group. Displacement and stress limits of each specimen were measured through cyclic compression tests and failure experiments, and vertical displacement and strain values under a 600 N vertical load were measured in all specimens through vertical compression tests.

Results

In the vertical compression test, the average displacement at the anterior head region of the femur was 0.362 mm for the PFBN group, significantly less than the 0.480 mm for the ITCS group (p < 0.001). At the fracture line area, the average displacement for the PFBN group was also lower than that of the ITCS group (0.196 mm vs. 0.324 mm, p < 0.001). The difference in displacement in the shaft area was smaller, but the average displacement for the PFBN group (0.049 mm) was still significantly less than that for the ITCS group (0.062 mm, p = 0.016). The situation was similar on the posterior side of the femur. The average displacements in the head area, fracture line area, and shaft area for the PFBN group were 0.300 mm, 0.168 mm, and 0.081 mm, respectively, while those for the ITCS group were 0.558 mm, 0.274 mm, and 0.041 mm, with significant differences in all areas (p < 0.001). The average strain in the anterior head area for the PFBN group was 4947 μm/m, significantly less than the 1540 μm/m for the ITCS group (p < 0.001). Likewise, in the fracture line and shaft areas, the average strains for the PFBN group were significantly less than those for the ITCS group (p < 0.05). In the posterior head area, the average strain for the PFBN group was 4861 μm/m, significantly less than the 1442 μm/m for the ITCS group (p < 0.001). The strain conditions in the fracture line and shaft areas also showed the PFBN group was superior to the ITCS group (p < 0.001). In cyclic loading experiments, the PFBN fixation showed smaller maximum displacement (1.269 mm vs. 1.808 mm, p < 0.001), indicating better stability. In the failure experiments, the maximum failure load that the PFBN-fixated fracture block could withstand was significantly higher than that for the ITCS fixation (1817 N vs. 1116 N, p < 0.001).

Conclusion

The PFBN can meet the biomechanical requirements for internal fixation of femoral neck fractures. PFBN is superior in biomechanical stability compared to ITCS, particularly showing less displacement and higher failure resistance in cyclic load and failure experiments. While there are differences in strain performance in different regions between the two fixation methods, overall, PFBN provides superior stability.