Evaluation of total ozone measurements from Geostationary Environmental Monitoring Spectrometer (GEMS)
Authors: Baek Kanghyun , Kim Jae Hwan , Bak Juseon , Haffner David P. , Kang Mina , Hong Hyunkee 
Published: November 2023

Abstract. The continued interest in air pollution and stratospheric ozone variability has motivated the development of a Geostationary Environmental Monitoring Spectrometer (GEMS) for hourly ozone monitoring. This paper provides the atmospheric science community with the world's first assessment of GEMS total column ozone (TCO) retrieval performance and diurnal ozone variation. The algorithm used for GEMS is a more advanced version of its predecessor, the Total Ozone Mapping Spectrometer (TOMS) V8, that incorporates several improvements, including a new lookup table, a simple Lambertian-equivalent reflectivity model, and a spectral dependence correction. The GEMS algorithm also uses the optimal estimation method (OEM) to make error analysis more accessible and robust. The estimated retrieval errors range from 1.5 to 2 DU in September and 2 DU in December, with a constant degree of freedom of the signal (DFS) of 1 in September and a variable DFS of 1.25 to 1.4 in December throughout the day, depending on solar zenith angle (SZA). To assess the performance of the GEMS algorithm, the hourly GEMS total ozone was compared with ground-based measurements from Pandora instruments and other satellite platforms from TROPOMI (TROPOspheric Monitoring Instrument) and OMPS (Ozone Mapping and Profiler Suite Nadir Mapper). GEMS has a high correlation of 0.97 and small RMSE values compared to Pandora TCO at Busan and Seoul in South Korea. It is notable that despite exhibiting seasonal dependence in the mean bias of GEMS with Pandora, GEMS is capable of observing daily variations in ozone that are highly consistent with Pandora measurements, with a bias of approximately 1 %. The comparison of GEMS TCO data with TROPOMI and OMPS TCO data shows a high correlation of 0.99 and low RMSE compared to TROPOMI and OMPS TCO data, but the data have a negative bias of −2.38 % and −2.17 %, with standard deviations of 1.33 % and 1.57 %, respectively. Similar to OMPS, the influence of SO2 from volcanic eruptions is not properly removed in some regions, leading to GEMS overestimating TCO in those areas. The mean biases of GEMS TCO data with TROPOMI and OMPS TCO are within ±1 % at low latitudes but become negative at midlatitudes, with an increasingly negative dependence on latitude. Furthermore, this dependence becomes more prominent from summer to winter. The empirical correction applied to the GEMS irradiance data improves the dependence of the mean bias on season and latitude, but a consistent bias still remains, and a marginal positive trend was observed in December. Therefore, further investigation into correction methods is needed. The results are a meaningful scientific advance by providing the first validated, hourly UV ozone retrievals from a satellite in geostationary orbit. This experience can be used to advance research with future geostationary environmental satellite missions, including the incoming TEMPO (Tropospheric Emissions: Monitoring of Pollution) and Sentinel-4.