Experts , however , have commented that there may be under-utilization of OAT due to patient factors , physician factors , or regional practice variations and that sub-optimal patient management may also occur .
These models are not suitable for all patients and require the identification and education of suitable c and i date s. Potential advantages of POC devices include improved convenience to patients , better treatment compliance and satisfaction , more frequent monitoring and fewer thromboembolic and hemorrhagic complications .
Potential disadvantages of the device include the tendency to underestimate high INR values and overestimate low INR values , low thromboplastin sensitivity , inability to calculate a mean normal PT , and errors in INR determination in patients with antiphospholipid antibodies with certain instruments .
Although treatment satisfaction and quality of life ( QoL ) may improve with POC INR monitoring , some patients may experience increased anxiety or preoccupation with their disease with these strategies .
The majority of studies reported favourable impacts of POC INR monitoring on QoL and found better treatment satisfaction with POC monitoring .
UNLABELLED SUBJECT OF THE EVIDENCE -BASED ANALYSIS : The purpose of this evidence based analysis report is to examine the safety and effectiveness of point-of-care ( POC ) international normalized ratio ( INR ) monitoring devices for patients on long-term oral anticoagulation therapy ( OAT ) .
CLINICAL NEED TARGET POPULATION AND CONDITION Long-term OAT is typically required by patients with mechanical heart valves , chronic atrial fibrillation , venous thromboembolism , myocardial infa rct ion , stroke , and /or peripheral arterial occlusion .
It is estimated that approximately 1 % of the population receives anticoagulation treatment and , by applying this value to Ontario , there are an estimated 132,000 patients on OAT in the province , a figure that is expected to increase with the aging population .
Patients on OAT  are regularly monitored and their medications adjusted to ensure that their INR scores remain in the therapeutic range .
This can be challenging due to the narrow therapeutic window of warfarin and variation in individual responses .
Optimal INR scores depend on the underlying indication for treatment and patient level characteristics , but for most patients the therapeutic range is an INR score of between 2.0 and 3.0 .
There is currently no population -based Ontario data to permit the assessment of patient care , but recent systematic review s have estimated that less that 50 % of patients receive OAT on a routine basis and that patients are in the therapeutic range only 64 % of the time .
OVERVIEW OF POC INR DEVICES POC INR devices offer an alternative to laboratory-based testing and venipuncture , enabling INR determination from a fingerstick sample of whole blood .
Independent evaluations have shown POC devices to have an acceptable level of precision .
They permit INR results to be determined immediately , allowing for more rapid medication adjustments .
POC devices can be used in a variety of setting s including physician offices , ACCs , long-term care facilities , pharmacies , or by the patients themselves through self-testing ( PST ) or self-management ( PSM ) techniques .
With PST , patients measure their INR values and then contact their physician for instructions on dose adjustment , whereas with PSM , patients adjust the medication themselves based on pre-set algorithms .
EffectivenessDoes POC INR monitoring improve clinical outcomes in various setting s compared to st and ard laboratory-based testing?Does POC INR monitoring impact patient satisfaction , QoL , compliance , acceptability , convenience compared to st and ard laboratory-based INR determination ?
Cost-effectivenessWhat is the cost-effectiveness of POC INR monitoring devices in various setting s compared to st and ard laboratory-based INR determination ?
ANTICOAGULATION CONTROL Anticoagulation control is measured by the percentage of time INR is within the therapeutic range or by the percentage of INR values in the therapeutic range .
Patient preferences for capillary vs. venous INR determination in an anticoagulation clinic: a randomized controlled trial.
BACKGROUND Patients who are receiving warfarin therapy require frequent blood testing to monitor the intensity of anticoagulation . Although previous studies suggest that capillary blood monitoring of the international normalize ratio ( INR ) is rapid and reliable , patient preferences for the method of blood drawing have not been investigated . METHODS We performed a r and omized controlled trial of patients attending an anticoagulation clinic in which patients were r and omly allocated to undergo capillary or venous INR monitoring .  Patient satisfaction with the outpatient visit , pain associated with blood drawing , and time spent in the clinic were assessed for each patient . RESULTS Sixty patients were studied . Using a 10-point visual analogue scale to quantify patient satisfaction ( 0-very satisfied ; 10-very dissatisfied ) , patients expressed a strong preference for capillary INR monitoring over venous INR monitoring ( 1.64 vs. 4.45 ; P < 0.001 ) . Using a 10-point visual analogue scale to quantify pain with blood sampling ( 0-no pain ; 10-very painful ) , patients who underwent capillary INR testing had less pain than venous INR testing ( 0.83 vs. 2.23 ; P < or = 0.004 ) . Patients spent , on average , 33 fewer minutes in the clinic with capillary INR testing than venous INR testing ( P < 0.001 ) . DISCUSSION Our findings support the routine use of capillary blood testing , using a portable monitor , for the management of patients in outpatient anticoagulation clinics
Experts , however , have commented that there may be under-utilization of OAT due to patient factors , physician factors , or regional practice variations and that sub-optimal patient management may also occur .
These models are not suitable for all patients and require the identification and education of suitable c and i date s. Potential advantages of POC devices include improved convenience to patients , better treatment compliance and satisfaction , more frequent monitoring and fewer thromboembolic and hemorrhagic complications .
Potential disadvantages of the device include the tendency to underestimate high INR values and overestimate low INR values , low thromboplastin sensitivity , inability to calculate a mean normal PT , and errors in INR determination in patients with antiphospholipid antibodies with certain instruments .
Although treatment satisfaction and quality of life ( QoL ) may improve with POC INR monitoring , some patients may experience increased anxiety or preoccupation with their disease with these strategies .
The majority of studies reported favourable impacts of POC INR monitoring on QoL and found better treatment satisfaction with POC monitoring .
UNLABELLED SUBJECT OF THE EVIDENCE -BASED ANALYSIS : The purpose of this evidence based analysis report is to examine the safety and effectiveness of point-of-care ( POC ) international normalized ratio ( INR ) monitoring devices for patients on long-term oral anticoagulation therapy ( OAT ) .
CLINICAL NEED TARGET POPULATION AND CONDITION Long-term OAT is typically required by patients with mechanical heart valves , chronic atrial fibrillation , venous thromboembolism , myocardial infa rct ion , stroke , and /or peripheral arterial occlusion .
It is estimated that approximately 1 % of the population receives anticoagulation treatment and , by applying this value to Ontario , there are an estimated 132,000 patients on OAT in the province , a figure that is expected to increase with the aging population .
Patients on OAT  are regularly monitored and their medications adjusted to ensure that their INR scores remain in the therapeutic range .
This can be challenging due to the narrow therapeutic window of warfarin and variation in individual responses .
Optimal INR scores depend on the underlying indication for treatment and patient level characteristics , but for most patients the therapeutic range is an INR score of between 2.0 and 3.0 .
There is currently no population -based Ontario data to permit the assessment of patient care , but recent systematic review s have estimated that less that 50 % of patients receive OAT on a routine basis and that patients are in the therapeutic range only 64 % of the time .
OVERVIEW OF POC INR DEVICES POC INR devices offer an alternative to laboratory-based testing and venipuncture , enabling INR determination from a fingerstick sample of whole blood .
Independent evaluations have shown POC devices to have an acceptable level of precision .
They permit INR results to be determined immediately , allowing for more rapid medication adjustments .
POC devices can be used in a variety of setting s including physician offices , ACCs , long-term care facilities , pharmacies , or by the patients themselves through self-testing ( PST ) or self-management ( PSM ) techniques .
With PST , patients measure their INR values and then contact their physician for instructions on dose adjustment , whereas with PSM , patients adjust the medication themselves based on pre-set algorithms .
EffectivenessDoes POC INR monitoring improve clinical outcomes in various setting s compared to st and ard laboratory-based testing?Does POC INR monitoring impact patient satisfaction , QoL , compliance , acceptability , convenience compared to st and ard laboratory-based INR determination ?
Cost-effectivenessWhat is the cost-effectiveness of POC INR monitoring devices in various setting s compared to st and ard laboratory-based INR determination ?
ANTICOAGULATION CONTROL Anticoagulation control is measured by the percentage of time INR is within the therapeutic range or by the percentage of INR values in the therapeutic range .