Experts , however , have commented that there may be under-utilization of OAT due to patient factors , physician factors , or regional practice variations and that sub-optimal patient management may also occur .
These models are not suitable for all patients and require the identification and education of suitable c and i date s. Potential advantages of POC devices include improved convenience to patients , better treatment compliance and satisfaction , more frequent monitoring and fewer thromboembolic and hemorrhagic complications .
Potential disadvantages of the device include the tendency to underestimate high INR values and overestimate low INR values , low thromboplastin sensitivity , inability to calculate a mean normal PT , and errors in INR determination in patients with antiphospholipid antibodies with certain instruments .
Although treatment satisfaction and quality of life ( QoL ) may improve with POC INR monitoring , some patients may experience increased anxiety or preoccupation with their disease with these strategies .
The majority of studies reported favourable impacts of POC INR monitoring on QoL and found better treatment satisfaction with POC monitoring .
UNLABELLED SUBJECT OF THE EVIDENCE -BASED ANALYSIS : The purpose of this evidence based analysis report is to examine the safety and effectiveness of point-of-care ( POC ) international normalized ratio ( INR ) monitoring devices for patients on long-term oral anticoagulation therapy ( OAT ) .
CLINICAL NEED TARGET POPULATION AND CONDITION Long-term OAT is typically required by patients with mechanical heart valves , chronic atrial fibrillation , venous thromboembolism , myocardial infa rct ion , stroke , and /or peripheral arterial occlusion .
It is estimated that approximately 1 % of the population receives anticoagulation treatment and , by applying this value to Ontario , there are an estimated 132,000 patients on OAT in the province , a figure that is expected to increase with the aging population .
Patients on OAT  are regularly monitored and their medications adjusted to ensure that their INR scores remain in the therapeutic range .
This can be challenging due to the narrow therapeutic window of warfarin and variation in individual responses .
Optimal INR scores depend on the underlying indication for treatment and patient level characteristics , but for most patients the therapeutic range is an INR score of between 2.0 and 3.0 .
There is currently no population -based Ontario data to permit the assessment of patient care , but recent systematic review s have estimated that less that 50 % of patients receive OAT on a routine basis and that patients are in the therapeutic range only 64 % of the time .
OVERVIEW OF POC INR DEVICES POC INR devices offer an alternative to laboratory-based testing and venipuncture , enabling INR determination from a fingerstick sample of whole blood .
Independent evaluations have shown POC devices to have an acceptable level of precision .
They permit INR results to be determined immediately , allowing for more rapid medication adjustments .
POC devices can be used in a variety of setting s including physician offices , ACCs , long-term care facilities , pharmacies , or by the patients themselves through self-testing ( PST ) or self-management ( PSM ) techniques .
With PST , patients measure their INR values and then contact their physician for instructions on dose adjustment , whereas with PSM , patients adjust the medication themselves based on pre-set algorithms .
EffectivenessDoes POC INR monitoring improve clinical outcomes in various setting s compared to st and ard laboratory-based testing?Does POC INR monitoring impact patient satisfaction , QoL , compliance , acceptability , convenience compared to st and ard laboratory-based INR determination ?
Cost-effectivenessWhat is the cost-effectiveness of POC INR monitoring devices in various setting s compared to st and ard laboratory-based INR determination ?
ANTICOAGULATION CONTROL Anticoagulation control is measured by the percentage of time INR is within the therapeutic range or by the percentage of INR values in the therapeutic range .
Patient self‐testing is a reliable and acceptable alternative to laboratory INR monitoring
An ageing population and the continuing expansion of clinical indications for coumadin therapy have increased pressure on hospital anticoagulant clinics . One solution is patient self‐testing ( PST ) of the international normalized ratio ( INR ) using capillary blood sample s on point‐of‐care coagulation monitors at home . We conducted a prospect i ve study to determine whether patients can achieve accurate INR values through PST , using the CoaguChek S ( Roche Diagnostics , Lewes , UK ) . The main outcome measurements were : comparability of INR values obtained by PST and the hospital laboratory , patient acceptability as assessed by a question naire and anticoagulant control . Eighty‐four patients [ 53 men , 31 women ; median age 59 years ( range 26–83 ) ] , receiving long‐term oral anticoagulation ( warfarin ) , were recruited from our Anticoagulation Clinic . Patients were r and omized to weekly self‐testing or continuing 4‐weekly hospital laboratory monitoring of INR . Comparison of INRs ( n = 234 ) showed no significant differences between the CoaguChek ( median INR 3·02 ) and laboratory testing ( median INR 3·07 ) . There was excellent correlation between the two methods ( r = 0·95 ) , with 85 % of CoaguChek results within 0·5 INR units of the laboratory method . On four occasions , differences of > 1 unit INR were obtained , but in each case the patient 's anticoagulation was unstable ( INR > 4·5 by both methods ) and the differences in INR would not have altered patient management . 87 % of patients found self‐testing straightforward , 87 % were confident in the result they obtained and 77 % preferred self‐testing . We conclude that PST is a reliable alternative to hospital clinic attendance and is acceptable to the majority of suitably trained patients
Experts , however , have commented that there may be under-utilization of OAT due to patient factors , physician factors , or regional practice variations and that sub-optimal patient management may also occur .
These models are not suitable for all patients and require the identification and education of suitable c and i date s. Potential advantages of POC devices include improved convenience to patients , better treatment compliance and satisfaction , more frequent monitoring and fewer thromboembolic and hemorrhagic complications .
Potential disadvantages of the device include the tendency to underestimate high INR values and overestimate low INR values , low thromboplastin sensitivity , inability to calculate a mean normal PT , and errors in INR determination in patients with antiphospholipid antibodies with certain instruments .
Although treatment satisfaction and quality of life ( QoL ) may improve with POC INR monitoring , some patients may experience increased anxiety or preoccupation with their disease with these strategies .
The majority of studies reported favourable impacts of POC INR monitoring on QoL and found better treatment satisfaction with POC monitoring .
UNLABELLED SUBJECT OF THE EVIDENCE -BASED ANALYSIS : The purpose of this evidence based analysis report is to examine the safety and effectiveness of point-of-care ( POC ) international normalized ratio ( INR ) monitoring devices for patients on long-term oral anticoagulation therapy ( OAT ) .
CLINICAL NEED TARGET POPULATION AND CONDITION Long-term OAT is typically required by patients with mechanical heart valves , chronic atrial fibrillation , venous thromboembolism , myocardial infa rct ion , stroke , and /or peripheral arterial occlusion .
It is estimated that approximately 1 % of the population receives anticoagulation treatment and , by applying this value to Ontario , there are an estimated 132,000 patients on OAT in the province , a figure that is expected to increase with the aging population .
Patients on OAT  are regularly monitored and their medications adjusted to ensure that their INR scores remain in the therapeutic range .
This can be challenging due to the narrow therapeutic window of warfarin and variation in individual responses .
Optimal INR scores depend on the underlying indication for treatment and patient level characteristics , but for most patients the therapeutic range is an INR score of between 2.0 and 3.0 .
There is currently no population -based Ontario data to permit the assessment of patient care , but recent systematic review s have estimated that less that 50 % of patients receive OAT on a routine basis and that patients are in the therapeutic range only 64 % of the time .
OVERVIEW OF POC INR DEVICES POC INR devices offer an alternative to laboratory-based testing and venipuncture , enabling INR determination from a fingerstick sample of whole blood .
Independent evaluations have shown POC devices to have an acceptable level of precision .
They permit INR results to be determined immediately , allowing for more rapid medication adjustments .
POC devices can be used in a variety of setting s including physician offices , ACCs , long-term care facilities , pharmacies , or by the patients themselves through self-testing ( PST ) or self-management ( PSM ) techniques .
With PST , patients measure their INR values and then contact their physician for instructions on dose adjustment , whereas with PSM , patients adjust the medication themselves based on pre-set algorithms .
EffectivenessDoes POC INR monitoring improve clinical outcomes in various setting s compared to st and ard laboratory-based testing?Does POC INR monitoring impact patient satisfaction , QoL , compliance , acceptability , convenience compared to st and ard laboratory-based INR determination ?
Cost-effectivenessWhat is the cost-effectiveness of POC INR monitoring devices in various setting s compared to st and ard laboratory-based INR determination ?
ANTICOAGULATION CONTROL Anticoagulation control is measured by the percentage of time INR is within the therapeutic range or by the percentage of INR values in the therapeutic range .