Experts , however , have commented that there may be under-utilization of OAT due to patient factors , physician factors , or regional practice variations and that sub-optimal patient management may also occur .
These models are not suitable for all patients and require the identification and education of suitable c and i date s. Potential advantages of POC devices include improved convenience to patients , better treatment compliance and satisfaction , more frequent monitoring and fewer thromboembolic and hemorrhagic complications .
Potential disadvantages of the device include the tendency to underestimate high INR values and overestimate low INR values , low thromboplastin sensitivity , inability to calculate a mean normal PT , and errors in INR determination in patients with antiphospholipid antibodies with certain instruments .
Although treatment satisfaction and quality of life ( QoL ) may improve with POC INR monitoring , some patients may experience increased anxiety or preoccupation with their disease with these strategies .
The majority of studies reported favourable impacts of POC INR monitoring on QoL and found better treatment satisfaction with POC monitoring .
UNLABELLED SUBJECT OF THE EVIDENCE -BASED ANALYSIS : The purpose of this evidence based analysis report is to examine the safety and effectiveness of point-of-care ( POC ) international normalized ratio ( INR ) monitoring devices for patients on long-term oral anticoagulation therapy ( OAT ) .
CLINICAL NEED TARGET POPULATION AND CONDITION Long-term OAT is typically required by patients with mechanical heart valves , chronic atrial fibrillation , venous thromboembolism , myocardial infa rct ion , stroke , and /or peripheral arterial occlusion .
It is estimated that approximately 1 % of the population receives anticoagulation treatment and , by applying this value to Ontario , there are an estimated 132,000 patients on OAT in the province , a figure that is expected to increase with the aging population .
Patients on OAT  are regularly monitored and their medications adjusted to ensure that their INR scores remain in the therapeutic range .
This can be challenging due to the narrow therapeutic window of warfarin and variation in individual responses .
Optimal INR scores depend on the underlying indication for treatment and patient level characteristics , but for most patients the therapeutic range is an INR score of between 2.0 and 3.0 .
There is currently no population -based Ontario data to permit the assessment of patient care , but recent systematic review s have estimated that less that 50 % of patients receive OAT on a routine basis and that patients are in the therapeutic range only 64 % of the time .
OVERVIEW OF POC INR DEVICES POC INR devices offer an alternative to laboratory-based testing and venipuncture , enabling INR determination from a fingerstick sample of whole blood .
Independent evaluations have shown POC devices to have an acceptable level of precision .
They permit INR results to be determined immediately , allowing for more rapid medication adjustments .
POC devices can be used in a variety of setting s including physician offices , ACCs , long-term care facilities , pharmacies , or by the patients themselves through self-testing ( PST ) or self-management ( PSM ) techniques .
With PST , patients measure their INR values and then contact their physician for instructions on dose adjustment , whereas with PSM , patients adjust the medication themselves based on pre-set algorithms .
EffectivenessDoes POC INR monitoring improve clinical outcomes in various setting s compared to st and ard laboratory-based testing?Does POC INR monitoring impact patient satisfaction , QoL , compliance , acceptability , convenience compared to st and ard laboratory-based INR determination ?
Cost-effectivenessWhat is the cost-effectiveness of POC INR monitoring devices in various setting s compared to st and ard laboratory-based INR determination ?
ANTICOAGULATION CONTROL Anticoagulation control is measured by the percentage of time INR is within the therapeutic range or by the percentage of INR values in the therapeutic range .
Comparison of the quality of oral anticoagulant therapy through patient self-management and management by specialized anticoagulation clinics in the Netherlands: a randomized clinical trial.
BACKGROUND Several studies have demonstrated that patient self-management of oral anticoagulant therapy ( OAT )  can improve treatment quality . However , most of these studies were not conducted within a specialized anticoagulation care system . The objective of the present study was to determine whether patient self-management of OAT improves the quality of care delivered by anticoagulation clinics . METHODS In this r and omized study by 2 Dutch anticoagulation clinics 341 patients aged between 18 and 75 years and receiving long-term OAT were divided into 4 groups : an existing routine care group of patients untrained in self-management ; a routine care group of trained patients ; a group managed weekly at an anticoagulation clinic where international normalized ratios were measured by trained patients ; and weekly patient self-management . A 2-step r and omization procedure was followed : first , a Zelen- design r and omization was performed to distribute patients ( without informing them ) to the existing care group or to receive training in self-management ; second , trained patients were r and omized to the 3 other study groups . RESULTS Only 25.6 % of invited patients agreed to participate in the training program . Patients who remained in the existing care group were within the international normalized ratio target range 63.5 % of the time . The type of coumarin taken was a major predicting factor of OAT quality . In all study groups phenprocoumon outperformed acenocoumarol by 11.6 % ( 95 % confidence interval [ CI ] , 6.6%-16.5 % ) . Weekly management with phenprocoumon led to a 6.5 % improvement ( 95 % CI , 0.0%-13.1 % ) in time in the international normalized ratio target range when patients were managed at an anticoagulation clinic and to an 8.7 % improvement ( 95 % CI , 1.6%-15.9 % ) when patients were self-managed . Weekly management with acenocoumarol did not improve the quality of OAT . CONCLUSION With selected patients , the quality of OAT obtained through patient self-management is at least as high as that delivered by specialized physicians at anticoagulation clinics . Weekly management of OAT with long-acting phenprocoumon has to be preferred at anticoagulation clinics or , where possible , through patient self-management
Experts , however , have commented that there may be under-utilization of OAT due to patient factors , physician factors , or regional practice variations and that sub-optimal patient management may also occur .
These models are not suitable for all patients and require the identification and education of suitable c and i date s. Potential advantages of POC devices include improved convenience to patients , better treatment compliance and satisfaction , more frequent monitoring and fewer thromboembolic and hemorrhagic complications .
Potential disadvantages of the device include the tendency to underestimate high INR values and overestimate low INR values , low thromboplastin sensitivity , inability to calculate a mean normal PT , and errors in INR determination in patients with antiphospholipid antibodies with certain instruments .
Although treatment satisfaction and quality of life ( QoL ) may improve with POC INR monitoring , some patients may experience increased anxiety or preoccupation with their disease with these strategies .
The majority of studies reported favourable impacts of POC INR monitoring on QoL and found better treatment satisfaction with POC monitoring .
UNLABELLED SUBJECT OF THE EVIDENCE -BASED ANALYSIS : The purpose of this evidence based analysis report is to examine the safety and effectiveness of point-of-care ( POC ) international normalized ratio ( INR ) monitoring devices for patients on long-term oral anticoagulation therapy ( OAT ) .
CLINICAL NEED TARGET POPULATION AND CONDITION Long-term OAT is typically required by patients with mechanical heart valves , chronic atrial fibrillation , venous thromboembolism , myocardial infa rct ion , stroke , and /or peripheral arterial occlusion .
It is estimated that approximately 1 % of the population receives anticoagulation treatment and , by applying this value to Ontario , there are an estimated 132,000 patients on OAT in the province , a figure that is expected to increase with the aging population .
Patients on OAT  are regularly monitored and their medications adjusted to ensure that their INR scores remain in the therapeutic range .
This can be challenging due to the narrow therapeutic window of warfarin and variation in individual responses .
Optimal INR scores depend on the underlying indication for treatment and patient level characteristics , but for most patients the therapeutic range is an INR score of between 2.0 and 3.0 .
There is currently no population -based Ontario data to permit the assessment of patient care , but recent systematic review s have estimated that less that 50 % of patients receive OAT on a routine basis and that patients are in the therapeutic range only 64 % of the time .
OVERVIEW OF POC INR DEVICES POC INR devices offer an alternative to laboratory-based testing and venipuncture , enabling INR determination from a fingerstick sample of whole blood .
Independent evaluations have shown POC devices to have an acceptable level of precision .
They permit INR results to be determined immediately , allowing for more rapid medication adjustments .
POC devices can be used in a variety of setting s including physician offices , ACCs , long-term care facilities , pharmacies , or by the patients themselves through self-testing ( PST ) or self-management ( PSM ) techniques .
With PST , patients measure their INR values and then contact their physician for instructions on dose adjustment , whereas with PSM , patients adjust the medication themselves based on pre-set algorithms .
EffectivenessDoes POC INR monitoring improve clinical outcomes in various setting s compared to st and ard laboratory-based testing?Does POC INR monitoring impact patient satisfaction , QoL , compliance , acceptability , convenience compared to st and ard laboratory-based INR determination ?
Cost-effectivenessWhat is the cost-effectiveness of POC INR monitoring devices in various setting s compared to st and ard laboratory-based INR determination ?
ANTICOAGULATION CONTROL Anticoagulation control is measured by the percentage of time INR is within the therapeutic range or by the percentage of INR values in the therapeutic range .