Experts , however , have commented that there may be under-utilization of OAT due to patient factors , physician factors , or regional practice variations and that sub-optimal patient management may also occur .
These models are not suitable for all patients and require the identification and education of suitable c and i date s. Potential advantages of POC devices include improved convenience to patients , better treatment compliance and satisfaction , more frequent monitoring and fewer thromboembolic and hemorrhagic complications .
Potential disadvantages of the device include the tendency to underestimate high INR values and overestimate low INR values , low thromboplastin sensitivity , inability to calculate a mean normal PT , and errors in INR determination in patients with antiphospholipid antibodies with certain instruments .
Although treatment satisfaction and quality of life ( QoL ) may improve with POC INR monitoring , some patients may experience increased anxiety or preoccupation with their disease with these strategies .
The majority of studies reported favourable impacts of POC INR monitoring on QoL and found better treatment satisfaction with POC monitoring .
UNLABELLED SUBJECT OF THE EVIDENCE -BASED ANALYSIS : The purpose of this evidence based analysis report is to examine the safety and effectiveness of point-of-care ( POC ) international normalized ratio ( INR ) monitoring devices for patients on long-term oral anticoagulation therapy ( OAT ) .
CLINICAL NEED TARGET POPULATION AND CONDITION Long-term OAT is typically required by patients with mechanical heart valves , chronic atrial fibrillation , venous thromboembolism , myocardial infa rct ion , stroke , and /or peripheral arterial occlusion .
It is estimated that approximately 1 % of the population receives anticoagulation treatment and , by applying this value to Ontario , there are an estimated 132,000 patients on OAT in the province , a figure that is expected to increase with the aging population .
Patients on OAT  are regularly monitored and their medications adjusted to ensure that their INR scores remain in the therapeutic range .
This can be challenging due to the narrow therapeutic window of warfarin and variation in individual responses .
Optimal INR scores depend on the underlying indication for treatment and patient level characteristics , but for most patients the therapeutic range is an INR score of between 2.0 and 3.0 .
There is currently no population -based Ontario data to permit the assessment of patient care , but recent systematic review s have estimated that less that 50 % of patients receive OAT on a routine basis and that patients are in the therapeutic range only 64 % of the time .
OVERVIEW OF POC INR DEVICES POC INR devices offer an alternative to laboratory-based testing and venipuncture , enabling INR determination from a fingerstick sample of whole blood .
Independent evaluations have shown POC devices to have an acceptable level of precision .
They permit INR results to be determined immediately , allowing for more rapid medication adjustments .
POC devices can be used in a variety of setting s including physician offices , ACCs , long-term care facilities , pharmacies , or by the patients themselves through self-testing ( PST ) or self-management ( PSM ) techniques .
With PST , patients measure their INR values and then contact their physician for instructions on dose adjustment , whereas with PSM , patients adjust the medication themselves based on pre-set algorithms .
EffectivenessDoes POC INR monitoring improve clinical outcomes in various setting s compared to st and ard laboratory-based testing?Does POC INR monitoring impact patient satisfaction , QoL , compliance , acceptability , convenience compared to st and ard laboratory-based INR determination ?
Cost-effectivenessWhat is the cost-effectiveness of POC INR monitoring devices in various setting s compared to st and ard laboratory-based INR determination ?
ANTICOAGULATION CONTROL Anticoagulation control is measured by the percentage of time INR is within the therapeutic range or by the percentage of INR values in the therapeutic range .
The value of education and self‐monitoring in the management of warfarin therapy in older patients with unstable control of anticoagulation
Of 125 patients aged 65 years or over , with atrial fibrillation taking warfarin for at least 12 months , with a st and ard deviation ( SD ) of prothrombin time , expressed as the International Normalized Ratio ( INR ) > 0·5 over the previous 6 months , 40 were r and omized to continue with usual clinic care and 85 to receive education about warfarin .  Of these , 44 were r and omized to self‐monitor their INR and 41 returned to clinic . Compared with the previous 6 months there was a significant increase in percentage time within the therapeutic range for the 6 months following education [ 61·1 vs. 70·4 ; mean difference 8·8 ; 95 % confidence interval ( CI ) : −0·2–17·8 ; P = 0·054 ] and following education and self‐monitoring ( 57 vs. 71·1 ; mean difference 14·1 ; 95 % CI : 6·7–21·5 ; P < 0·001 ) , compared with those patients following usual clinic care ( 60·0 vs. 63·2 ; mean difference 3·2 ; 95 % CI : −7·3–13·7 ) . Using the same comparative periods , the INR SD fell by 0·24 ( P < 0·0001 ) in the group allocated to education and self‐monitoring , 0·26 ( P < 0·0001 ) in the group receiving education alone and 0·16 ( P = 0·003 ) in the control group . Inter‐group differences were not statistically significant ( intervention groups 0·26 ± 0·30 vs. control 0·16 ± 0·3 , P = 0·10 ) . Quality ‐of‐life measurements and health beliefs about warfarin were unchanged ( apart from emotional role limitation ) with education or education and self‐monitoring .  Patient  education regarding anticoagulation therapy could be a cost‐effective initiative and is worthy of further study
Experts , however , have commented that there may be under-utilization of OAT due to patient factors , physician factors , or regional practice variations and that sub-optimal patient management may also occur .
These models are not suitable for all patients and require the identification and education of suitable c and i date s. Potential advantages of POC devices include improved convenience to patients , better treatment compliance and satisfaction , more frequent monitoring and fewer thromboembolic and hemorrhagic complications .
Potential disadvantages of the device include the tendency to underestimate high INR values and overestimate low INR values , low thromboplastin sensitivity , inability to calculate a mean normal PT , and errors in INR determination in patients with antiphospholipid antibodies with certain instruments .
Although treatment satisfaction and quality of life ( QoL ) may improve with POC INR monitoring , some patients may experience increased anxiety or preoccupation with their disease with these strategies .
The majority of studies reported favourable impacts of POC INR monitoring on QoL and found better treatment satisfaction with POC monitoring .
UNLABELLED SUBJECT OF THE EVIDENCE -BASED ANALYSIS : The purpose of this evidence based analysis report is to examine the safety and effectiveness of point-of-care ( POC ) international normalized ratio ( INR ) monitoring devices for patients on long-term oral anticoagulation therapy ( OAT ) .
CLINICAL NEED TARGET POPULATION AND CONDITION Long-term OAT is typically required by patients with mechanical heart valves , chronic atrial fibrillation , venous thromboembolism , myocardial infa rct ion , stroke , and /or peripheral arterial occlusion .
It is estimated that approximately 1 % of the population receives anticoagulation treatment and , by applying this value to Ontario , there are an estimated 132,000 patients on OAT in the province , a figure that is expected to increase with the aging population .
Patients on OAT  are regularly monitored and their medications adjusted to ensure that their INR scores remain in the therapeutic range .
This can be challenging due to the narrow therapeutic window of warfarin and variation in individual responses .
Optimal INR scores depend on the underlying indication for treatment and patient level characteristics , but for most patients the therapeutic range is an INR score of between 2.0 and 3.0 .
There is currently no population -based Ontario data to permit the assessment of patient care , but recent systematic review s have estimated that less that 50 % of patients receive OAT on a routine basis and that patients are in the therapeutic range only 64 % of the time .
OVERVIEW OF POC INR DEVICES POC INR devices offer an alternative to laboratory-based testing and venipuncture , enabling INR determination from a fingerstick sample of whole blood .
Independent evaluations have shown POC devices to have an acceptable level of precision .
They permit INR results to be determined immediately , allowing for more rapid medication adjustments .
POC devices can be used in a variety of setting s including physician offices , ACCs , long-term care facilities , pharmacies , or by the patients themselves through self-testing ( PST ) or self-management ( PSM ) techniques .
With PST , patients measure their INR values and then contact their physician for instructions on dose adjustment , whereas with PSM , patients adjust the medication themselves based on pre-set algorithms .
EffectivenessDoes POC INR monitoring improve clinical outcomes in various setting s compared to st and ard laboratory-based testing?Does POC INR monitoring impact patient satisfaction , QoL , compliance , acceptability , convenience compared to st and ard laboratory-based INR determination ?
Cost-effectivenessWhat is the cost-effectiveness of POC INR monitoring devices in various setting s compared to st and ard laboratory-based INR determination ?
ANTICOAGULATION CONTROL Anticoagulation control is measured by the percentage of time INR is within the therapeutic range or by the percentage of INR values in the therapeutic range .